Copyright: ©1999-2002 Trimble Navigation Limited. All rights reserved.
Citation: ©1999-2002 Trimble Navigation Limited. All rights reserved.


Trimble Geomatics Office - Vue d'ensemble

Trimble Geomatics Office vous permet d’intégrer des données provenant de diverses techniques de levé topographique, y compris GPS (RTK et post-traitées), instruments conventionnels/optiques (y compris les servomécanismes et la robotique), niveaux et lasers. Les nombreux outils de contrôle de qualité vous permettent de réaliser les tâches plus rapidement et plus efficacement pour un transfert transparent des données du terrain au bureau et inversement.

Utilisez Trimble Geomatics Office pour les opérations suivantes :

  Traitement de lignes de base GPS (lorsque le module Trimble WAVE Baseline Processing (Traitement de lignes de base WAVE de Trimble) est installé sur votre ordinateur)

  Ajustement de réseau de levé (lorsque le module Trimble Network Adjustment (Ajustement de réseau de Trimble) est installé sur votre ordinateur)

  Traitement de données topographiques conventionnelles ou GPS

  Assurance de la qualité et contrôle de la qualité des données (QA/QC)

  Importation et exportation de données de tracé de routes

  Importation et exportation de données de levé

  Réalisation de modèles de terrain et de courbes de niveau numériques

  Transformation de datum et projections

  Saisie de données SIG et exportation de données

  Codification

  Rapport de projet

  Gestion de projet de levé

Caractéristiques principales

La liste suivante présente les caractéristiques principales de Trimble Geomatics Office (version 1.5) :

  La Vue de levé et la Vue en plan vous permettent de visualiser les données du projet dans la fenêtre graphique

  La Barre du projet organise les tâches habituellement utilisées en raccourcis dans la fenêtre graphique principale

  Les Filtres d’affichage permettent d’afficher ou de masquer des types d’observations

  La fonction « glisser-déposer » vous permet de faire glisser tout fichier d’étude du bureau et de le déposer dans la zone du projet

  Importation de données collectées à partir de récepteurs GPS et/ou de stations totales conventionnelles

  Grande variété de formats d’importation et d’exportation

  Le dialogue Entrée d’un fichier DAT vous permet de vérifier et d’éditer les données brutes GPS avant de les importer

  La fenêtre Propriétés vous permet de visualiser et d’afficher toutes les informations relatives aux points ainsi que tous les types d’observations

  Le dialogue Edition multiple permet l’édition rapide d’entités multiples

  La prise en charge de collecte de données SIG

  La prise en charge de couches pour gérer les données CAD

  Rapports HTML professionnels

  Formats de rapport standard et personnalisés

  Modules RoadLink et DTMLink pour la visualisation et l’édition de fichiers de conception de routes tiers ainsi que pour la création de modèles de terrain numériques

NOTE : Trimble Geomatics Office ne prend pas en charge les systèmes de noms de fichier 8.3.

Modules

Cette section décrit les modules optionnels disponibles pour Trimble Geomatics Office :

  Le module de traitement de lignes de base WAVE calcule les lignes de base à partir des observations de terrain GPS réalisées selon des procédures de collecte des données cinématiques, FastStatic ou statiques. Le processeur de lignes de base WAVE assure un traitement par défaut et procure des commandes flexibles pour un traitement avancé. Timeline affiche les données GPS trouvées dans les fichiers d’observations brutes dans un format graphique basé sur l’heure. Utilisez-le pour visualiser et éditer les mesures GPS brutes et les informations sur le levé.

  Le module Ajustement de réseau ajuste vos données de levé GPS et terrestre pour évaluer et réduire les erreurs inhérentes.

Qu’est-ce que la géomatique ?

La géomatique est la conception, la collecte, le stockage, l’analyse, l’affichage et l’extraction d’informations spatiales. La collecte d’informations spatiales peut se faire à partir de nombreuses sources, y compris à l’aide des méthodes terrestres et GPS. La géomatique intègre les méthodes traditionnelles de levé topographique à des approches basées sur les nouvelles technologies, s’avérant ainsi utile pour un grand nombre d’applications.


Suppression du logiciel

ATTENTION : La procédure décrite sous cette rubrique permet de supprimer tous les fichiers à l’exception des fichiers communs, des modèles et des fichiers de données. Pour supprimer des fichiers de données, utilisez Microsoft Windows Explorer.

Pour supprimer Trimble Geomatics Office de votre ordinateur :

1.	Sélectionnez Démarrer / Paramètres / Panneau de configuration.

2.	Dans le Panneau de configuration, sélectionnez Ajout/Suppression de programmes. Le dialogue Propriétés d’ajout/suppression de programmes s’affiche.

3.	Assurez-vous que la tabulation Installer/Désinstaller est active.

4.	Sélectionnez Trimble Geomatics Office.

5.	Cliquez sur Ajout/Suppression. Le message Confirmer la suppression du fichier s’affiche.

6.	Cliquez sur Oui.

Les fichiers sont supprimés.

ATTENTION : Lorsque vous désinstallez Trimble Geomatics Office, vous perdez tous les formats personnalisés. Par exemple, si vous avez ajouté des types de symbole à la bibliothèque de symboles créée par le système (Symbols.sym) ou ajouté des formats d’importation/exportation personnalisés, ceux-ci seront perdus.

Les fichiers suivants servent au stockage des informations personnalisées :

  Fichiers de définition de modèle de plan (*.frm) — personnalisation en éditant ces fichiers à l’aide d’un éditeur de texte

  Fichiers de formats de rapport (Asciirpt.dat) — personnalisation lorsque vous créez des fichiers personnalisés

  Fichiers de définition d’importation ASCII (Asciiimp.dat) — personnalisation lorsque vous créez des formats d’importation personnalisés

  Fichiers de définition d’exportation ASCII (Asciiexp.dat) — personnalisation lorsque vous créez des formats d’exportation personnalisés

  Bibliothèque de symboles créée par le système (Symbols.sym) — personnalisation lorsque vous sauvegardez de nouveaux types de symbole dans ce fichier

  Bibliothèque de types de ligne créée par le système (Linetype.ltp) — personnalisation lorsque vous sauvegardez de nouveaux types de ligne dans ce fichier


Structure des dossiers

Lors de l’installation, tous les fichiers requis par Trimble Geomatics Office sont copiés dans les dossiers listés ci-dessous :

Dossier	Ce qui est enregistré
\Program Files\Trimble\ Trimble Geomatics Office	Programme et fichiers d’aide de Trimble Geomatics Office
...\System	Fichiers de système ; y compris les fichiers logiciels de bibliothèque de caractéristiques et attributs, les utilitaires Symbol Editor™ et Line Type Editor™ et les modèles de rapport HTML
...\Template	Fichiers de modèles de projet
...\Utility	Liens pour les utilitaires A1 Viewer, Convert to RINEX™, Coordinate System Manager™, Data Transfer, DC File Editor™, Feature and Attribute Editor™, Line Type Editor, Planning et Symbol Editor
...\RoadLink - DTMLink	Fichiers programmes et fichiers associés pour RoadLink et DTMLink.
\Program Files\Common Files\ Trimble	Fichiers utilisés à la fois par Trimble Geomatics Office et les autres produits Trimble
...\GeoData	Fichiers d’informations sur le système de coordonnées et de géoïdes
...\GeoDB	Fichiers programmes de Coordinate System Manager (gestionnaire de système de coordonnées)
...\Data Transfer	Fichiers programmes de Data Transfer (transfert de données)
...\DatToRinex	Conversion en fichiers programmes RINEX
...\DCeditor	Fichiers programmes de DC File Editor (éditeur de fichiers DC)
...\Plan	Fichiers programmes de Planning


A propos de Trimble

Trimble fait partie des leaders mondiaux dans le domaine de la conception et du développement de produits innovateurs basés sur la technologie GPS. En sus de fournir des composants GPS avancés, Trimble augmente le GPS avec d’autres technologies de positionnement ainsi que des communications sans fil et des logiciels pour créer des solutions client complètes. Les applications Trimble apparaissantes comprennent la topographie, la navigation de véhicules, le guidage des engins, les plates-formes sans fil et l’infrastructure de télécommunications.

Le siège de Trimble se trouve à 645 N. Mary Avenue à Sunnyvale, CA. Nous possédons également des agences dans le monde entier afin d’être en mesure de vous servir où que vous soyez.

Pour une vue interactive de Trimble, consultez notre site sur le World Wide Web
www.trimble.com


Commande Inverse

Utilisez cette commande pour déterminer l'inverse entre deux points du projet.

Pour y accéder, sélectionnez Levé / Inverse.

Le dialogue Inverse s'affiche.


Dialogue Inverse

Utilisez ce dialogue pour déterminer l'inverse entre deux points du projet. Vous pouvez utiliser le dialogue pour afficher des informations détaillées ou un résume de l'inverse ainsi que générer un rapport.

Pour y accéder effectuez l'une des choses suivantes:

  Sélectionnez Levé / Inverse.

  Appuyez sur F7.

NOTE: L'outil Inverse ne fonctionne qu'avec des points existants. Pour afficher les azimuts et les distances entre toute autre position de coordonnées, utilisez l'outil Mesurer. Pour y accéder, sélectionnez Levé / Mesurer.

De

Entrez un nom de point à l'aide du clavier, ou utilisez remplissage du champ pour cliquer sur un point.

Vers

Entrez un nom de point à l'aide du clavier, ou utilisez remplissage du champ pour cliquer sur un point.

Lorsque vous déplacez le pointeur près d'un autre point, une ligne pointillée apparaît. Elle indique l'inverse à calculer. Lorsque vous cliquez sur le deuxième point, la ligne pointillée change à une ligne solide. Celle-ci indique la ligne à laquelle les détails d'inverse font référence.

Si le paramètre Afficher distance dans la tabulation Unités et format du dialogue Propriétés du projet est configuré sur les distances au niveau de la mer ou de grille, et les deux points n'ont pas des élévations, la distance de grille et de l'azimut est affichée. Cependant, si le champ Afficher distance est configuré sur Distance au sol, et les deux points d'inverse ont des élévations nulles, la distance s'affiche comme nulle (lorsque l'élévation par défaut est configurée sur nulle). Dans l'absence des informations d'élévation, il n'est pas possible de calculer la distance au sol.

Azimut

Ce champ indique l'azimut de grille entre les points.

Distance

Ce champ indique la distance de grille entre les points.

Delta élévation

Lorsque les deux points ont des élévations, la delta élévation s'affiche. Lorsque l'une des deux élévations n'est pas connue, le champ indique nulle.

Méthode

Il y a deux méthodes de calcul de l'inverse disponibles:

Séquentielle	Pour calculer une séquence d'inverses l'un après l'autre.

Radial	Pour calculer une séquence d'inverses partant d'un point de début commun.

Calculer

Cliquez sur Calculer pour calculer l'inverse lorsque vous avez utilisé le clavier pour la saisie des points sélectionnés.

Envoyer vers rapport

Cliquez sur Envoyer vers rapport pour créer un rapport du calcul de l'inverse. Un résumé ou un rapport détaillé de l'inverse peut être généré selon l'option d'affichage utilisée pour l'inverse.

NOTE: Lorsque le rapport a été généré d'une option d'affichage et puis régénéré dans un autre mode d'affichage, les détails de l'inverse du rapport sont ajoutés au rapport existant.

Afficher rapport

Cliquez sur Afficher rapport pour visualiser le Rapport d'inverse créé.

Le résumé affiche les détails d'inverse comme affichés dans le dialogue Inverse en mode résumé.

Le rapport détaillé affiche les positions des points "de" et "vers" dans les systèmes de coordonnées de grille, local (géodésique) et cartésien (WGS-84), ainsi que les valeurs d'inverse calculées et les paramètres utilisés pour réduire l'inverse dans ces systèmes de coordonnées.

Détaillé / Résumé

Cliquez sur Détaillé pour afficher le groupe Détails. Lorsque le groupe Détails s'affiche le bouton change à Résumé. Cliquez sur Résumé pour masquer le groupe Détails.

Détails

Le groupe Détails affiche des informations supplémentaires concernant l'inverse.

Azimut avant SN

Ce champ affiche l'azimut (géodésique) avant de section normale du premier point au deuxième point.

Azimut arrière SN

Ce champ affiche l'azimut (géodésique) arrière de section normale du deuxième point au premier point.

Distance d'ellipsoïde

Ce champ affiche la distance d'ellipsoïde entre les points.

Distance au sol

Ce champ affiche la distance au sol entre les points.

Delta hauteur

Ce champ affiche la delta hauteur entre les points.


Commande Mesurer

Utilisez cette commande pour mesurer les azimuts, les distances et les zones de toute fenêtre de graphiques.

Pour y accéder, sélectionnez Levé / Mesurer.

Le dialogue Mesurer s'affiche.


Dialogue Mesurer

Utilisez ce dialogue pour mesurer les azimuts, les distances et les zones de toute fenêtre de graphiques.

Pour y accéder, sélectionnez Levé / Mesurer.

Vous pouvez mesurer entre les points ou les positions du projet sur l'écran. (Cliquez sur la position appropriée).

Une fois que vous avez sélectionné deux ou plus positions, la distance cumulative entre la première et la dernière position s'affiche dans le champ Distance. Le champ Azimut affiche l'azimut entre les deux dernières positions sélectionnées.

ASTUCE: Lorsque vous cliquez et déplacez la souris, les champs affichent l'azimut courant, la distance cumulative du premier point sélectionné et la surface entourée.

Azimut

Lorsque vous sélectionnez deux positions, l'azimut entre les points s'affiche. Avec trois ou plus points sélectionnés, ce champ affiche l'azimut entre les deux derniers points. Le type d'azimut affiché (grille ou géodésique) dépend du champ Afficher azimut sous dans le dialogue Propriétés – tabulation Unités et format.

Distance

Lorsque vous sélectionnez deux positions, la distance entre les deux points s'affiche. Avec trois ou plus points sélectionnés, ce champ affiche le total cumulatif (la somme) des distances mesurées.

Les distances totales sont rapportées selon vos paramètres d'affichage.

NOTE: Si le champ Afficher azimut sous dans le dialogue Propriétés du projet est configuré sur Distances au sol, et toutes les deux positions mesurées ont des élévations nulles, la distance s'affiche comme nulle (lorsque l'élévation par défaut est configurée sur nulle). Dans l'absence des informations d'élévation, il n'est pas possible de calculer la distance au sol.

Surface

Avec trois ou plus points sélectionnés, la surface de la figure formée par la fermeture au premier point s'affiche.

La surface calculée varie selon les paramètres d'affichage de distance:

  Un paramètre de distance au sol donne une surface calculée à l'élévation moyenne au sol.

  Une distance au niveau de la mer donne une surface calculée sur la surface d'ellipsoïde.

  Un paramètre de distance de grille donne une surface calculée des coordonnées de grille.

Remettre

Cliquez sur Remettre pour annuler tous les champs. Vous pouvez également remettre en double-cliquant sur le point de début.

ASTUCE: Vous pouvez rapidement déterminer l'étendue du projet représenté dans la fenêtre de graphiques en cliquant sur l'indicateur de dimensions [image] de la fenêtre graphique. Pour afficher cet indicateur, tenez le curseur hors de la fenêtre de graphiques.

Lors de l'utilisation du pointeur [image] appuyez sur Ctrl. Ainsi il ne passe pas instantanément à un point voisin.


Transformation de coordonnées - Vue d'ensemble

Plus amples informations

Comment...	Dialogue	Voir aussi
Vue d'ensemble

Une transformation de coordonnées déplace les points dans le projet. Il peut impliquer des décalages, les rotations et la mise à l'échelle en deux dimensions. Les points ayant une qualité de Contrôle ou Ajusté ne sont pas déplacés. Un message s'affiche si un point que vous avez sélectionné possède une qualité ajusté ou de contrôle.

Il faut utiliser une transformation de coordonnées lorsque, par exemple, vous avez des points dans un projet d'un progiciel tiers qui sont dans un système de coordonnées supposé et qu'il faut transformer les points afin d'obtenir les coordonnées correctes.

Pour définir la transformation de coordonnées, vous pouvez effectuer l'une des choses suivantes:

  Utilisez le remplissage du champ, cliquez sur les points, puis entrez leurs valeurs de coordonnées connues. Lorsque plus de trois points sont utilisés, le logiciel utilise la méthode par les moindres carrés pour calculer les paramètres de transformation pour les points sélectionnés.

  Spécifiez les paramètres de transformation

Lorsque vous effectuez une transformation de coordonnées, toutes les autres observations ou positions de grille sont désactivées. Les coordonnées WGS-84, de grille et locales pour les points transformés sont modifiées, mais le système de coordonnées n'est pas changé.

Une fois que vous avez effectué une transformation, vous pouvez créer un rapport affichant les valeurs de coordonnées transformées et d'origine.


Commande Transformation de coordonnées

Utilisez cette commande pour déterminer et appliquer une transformation horizontale.

Pour y accéder, sélectionnez Levé / Transformation de coordonnées.

Le dialogue Transformation de coordonnées s'affiche.


Dialogue Transformation de coordonnées

 Vue d'ensemble et plus amples informations

Utilisez ce dialogue pour déterminer et appliquer une Transformation horizontale.

Pour y accéder, sélectionnez Levé / Transformation de coordonnées.

Pour saisir des valeurs dans ce dialogue, effectuez l'une des choses suivantes:

  Cliquez sur Liste points pour sélectionner les points du projet dont vous avez des coordonnées connues et entrez les valeurs connues pour chaque point. Le logiciel calcule les paramètres de transformation. Lorsqu'il y a plus de trois valeurs de point, il utilise la méthode par les moindres carrés. Le dialogue Transformation de coordonnées affiche les valeurs de transformation.

  Entrez les valeurs à l'aide du clavier.

Lorsque vous effectuez une transformation de coordonnées, toutes les autres observations aux positions de point ou de grille du point sont désactivées. Les coordonnées de grille, locales et WGS-84 des points transformés sont modifiées. Le système de coordonnées n'est pas changé.

Translation est

Indique le degré de translation est (positive) ou ouest (négative).

Translation nord

Indique le degré de translation nord (positive) ou sud (négative).

Rotation

Indique le degré de rotation angulaire à appliquer dans la transformation.

Echelle

Indique le changement d'échelle de la transformation.

Origine de rotation

Les champs suivants affichent le centre de rotation.

Point	Indique le nom du point à utiliser comme le point d'origine de la rotation. Lorsque vous entrez un nom de point, les coordonnées s'affichent dans les champs Nord et Est. Lorsque Liste points est utilisé pour calculer les paramètres, ce champ affiche la moyenne des coordonnées de la liste de points.
Nord	Indique la coordonnée Nord du point d'origine.
Est	Indique la coordonnée Est du point d'origine.

Transformer

Pour transformer les points, sélectionnez l'une des options suivantes:

Sélection	Applique la transformation aux points sélectionnés.
BD entière	Applique la transformation à tous les points du projet.
Lorsqu'aucun point n'est sélectionné, ces options ne sont pas disponibles et tous les points sont transformés.

Générer le rapport de transformation de coordonnées

Sélectionnez cette case à cocher pour créer un rapport des coordonnées d'origine et transformées. Le rapport s'affiche dans l'afficheur HTML par défaut.

Liste points

Cliquez sur Liste points pour sélectionner les points de contrôle. Entrez les valeurs connues des points.

Le dialogue Liste points s'affiche.


Transformation de coordonnées - dialogue Liste points

 Vue d'ensemble et plus amples informations

Utilisez le dialogue Transformation de coordonnées – Liste points pour sélectionner les points utilisés pour calculer les paramètres de transformation de coordonnées. Vous sélectionnez les points, le logiciel Trimble Geomatics Office affiche les coordonnées courantes et vous entrez les coordonnés connues. Le logiciel Trimble Geomatics Office calcule les paramètres de transformation, puis applique la transformation aux points que vous avez sélectionné pour la transformation.

Pour y accéder:

1.	Sélectionnez Levé / Transformation de coordonnées. Le dialogue Transformation de coordonnées s'affiche.

2.	Cliquez sur Liste points.

Ce dialogue présente une liste des points pouvant être défilée. Chaque point a des champs indiquant les coordonnées courantes et des champs indiquant les coordonnées connues. Lorsque vous entrez le nom du point, la Coordonnée nord actuelle et la Coordonnée est actuelle s'affichent. Puis vous entrez les coordonnées connues dans les champs Nord connu et Est connu.

ASTUCE: Vous pouvez laisser le nom du point vide et utiliser le remplissage du champ pour entrer les coordonnées courantes.

Il n'y a aucune limite au nombre de définitions de position pouvant être définies pour calculer une transformation de coordonnées. Lorsque vous ajoutez beaucoup de définitions de position les résultats de transformation calculés ne sont pas toujours améliorés, mais il fournit des vérifications supplémentaires concernant la validité des paramètres calculés.

Insérer

Cliquez sur Insérer pour insérer un enregistrement de définition de position vide. En général, le logiciel Trimble Geomatics Office ajoute automatiquement les champs de définition de position, donc il ne faut pas le faire très souvent. Cependant, à cause du fait que le premier point dans la Liste points est utilisé comme l'origine de rotation dans les paramètres de transformation calculés, il se peut que vous voulez insérer une définition de position au début de la liste.

Supprimer

Cliquez sur Supprimer pour enlever une définition de position de la liste.

Fermer

Lorsque vous cliquez sur Fermer, Trimble Geomatics Office calcule les paramètres de transformation de coordonnées. En outre il fait un rapport des erreurs moyennes quadratiques et résiduelles maximum des positions transformées. Si le facteur d'échelle calculé n'est pas exactement 1.0000000 un message s'affiche.


Ajustement d'élévation - Vue d'ensemble

Plus amples informations

Comment ...	Dialogue	Voir aussi
Un ajustement d'élévation ajuste les élévations des points.

Une fois que l'ajustement est réalisé, Trimble Geomatics Office applique l'ajustement d'élévation en utilisant l'une des méthodes suivantes:

  Lorsque vous entrez les élévations connues des points existants, le logiciel Trimble Geomatics Office calcule les paramètres de l'ajustement d'élévation.

  Lorsque vous entrez une constante (positive ou négative), Trimble Geomatics Office l'applique à tous les points sélectionnés.

NOTE: L'ajustement d'élévation ne sera pas appliqué lorsque les qualités d'élévation ou de hauteur du point sont celles de contrôle ou ajustées.

Résultats de l'ajustement d'élévation

Lorsque vous effectuez un ajustement d'élévation, toutes les autres observations aux positions de grille ou de point d'un point sont désactivées. Les coordonnées de grille, locale et WGS-84 pour les points transformés sont modifiées. le système de coordonnées n'est pas changé.

La table suivante montre les paramètres calculés lorsqu'un nombre d'élévations sont comprises dans la liste de points.

NOTE: L'ajustement de l'élévation calculé n'est pas compris dans la définition du système de coordonnées, contrairement à l'ajustement vertical pouvant être effectué dans la commande Levé /  Calibration du site GPS.

Vous pouvez créer un rapport indiquant l'élévation d'origine, l'élévation ajustée et la différence entre les deux.


Commande Ajustement d'élévation

Utilisez cette commande pour déterminer et appliquer un ajustement d'élévation.

Pour y accéder, sélectionnez Levé / Ajustement d'élévation.

Le dialogue Ajustement d'élévation s'affiche.


Dialogue Ajustement d'élévation

 Vue d'ensemble et plus amples informations

Utilisez ce dialogue pour déterminer et appliquer un ajustement d'élévation.

Pour y accéde :

Sélectionnez Levé / Ajustement d'élévation.

Pour ajuster les élévations des points du projet sélectionnés, utilisez l'une des méthodes suivantes:

  la méthode Constante

  la méthode Paramètres de plan (entrez les valeurs ou sélectionnez les points)

Constante

Sélectionnez cette option pour ajouter une valeur constante aux élévations des points. Appliquez la constant aux points sélectionnés ou au projet entier..

Ajouter à l'élévation

Le champ Ajouter à l'élévation devient disponible lorsque vous sélectionnez l'option Constante. Entrez une valeur d'élévation positive ou négative à ajouter aux points.

Paramètres

Sélectionnez cette option pour spécifier ou calculer les paramètres pour un ajustement d'élévation de plan incliné. Pour calculer les paramètres, cliquez sur Liste points.

Point d'origine

Indique le nom du point à utiliser comme le point d'origine pour le plan incliné.

NOTE: Si Liste points est utilisé pour sélectionner les points pour le calcul des paramètres, ce champ affiche le premier point de contrôle sélectionné.

Origine est

Indique la coordonnée Est du point d'origine. Si vous sélectionnez un point dans le champ Point d'origine, ce champ est rempli avec la coordonnée du point sélectionné.

Origine nord

Indique la coordonnée Nord du point d'origine. Si vous sélectionnez un point dans le champ Point d'origine, ce champ est rempli avec la coordonnée du point sélectionné.

Correction constante

Indique la correction d'élévation appliquée à la position d'origine.

Pente nord

Cette valeur définit la pente du plan dans la direction nord-sud. Un ajustement d'élévation est calculé de cette valeur en parties par million (ppm) et la distance (direction nord-sud) de l'origine. Elle est également appliquée à la correction constante.

Pente est

Cette valeur définit la pente du plan dans la direction est-ouest. Un ajustement d'élévation est calculé de cette valeur ppm et la distance (direction est-ouest) de l'origine. Elle est également appliquée à la correction constante.

Liste points

Cliquez sur Liste points pour sélectionner les points de contrôle à utiliser pour calculer les paramètres d'ajustement d'élévation. Le dialogue Liste points s'affiche.

Ajuster

Sélectionnez cette option pour appliquer l'ajustement. Deux options sont disponibles:

Sélection	Sélectionnez cette option pour appliquer l'ajustement aux points sélectionnés.
BD entière	Sélectionnez cette option pour appliquer l'ajustement à tous les points du projet.
Lorsqu'aucun point n'est sélectionné, les boutons de sélection ne sont pas disponibles et l'ajustement est appliqué à tous les points du projet.

Générer le rapport d'ajustement d'élévation

Sélectionnez cette case à cocher pour créer un rapport des élévations ajustées et d'origine. Le rapport s'affiche dans l'afficheur HTML par défaut.


Ajustement d'élévation - dialogue Liste points

 Vue d'ensemble et plus amples informations

Trimble Geomatics Office utilise les points sélectionnés dans ce dialogue pour calculer les paramètres d'ajustement d'élévation. Vous sélectionnez des points du projet et entrez des élévations connues pour ces points.

Pour y accéder:

1.	Sélectionnez Levé / Ajustement d'élévation. Le dialogue Ajustement d'élévation s'affiche.

2.	Cliquez sur Liste points.

Sélectionnez chaque point de contrôle utilisant le remplissage du champ ou l'entrée à l'aide du clavier et entrez l'élévation connue.

ASTUCE: Les champs d'élévation et de coordonnées peuvent être remplis en utilisant le remplissage du champ pour choisir des points appropriés sur l'écran.

En général, trois ou plus points de contrôle sont saisis. Ces points devraient être répartis uniformément autour du périmètre des points en train d'être ajustés.

Il n'y a aucune limite au nombre de points pouvant être définies pour calculer un ajustement d'élévation. Lorsque vous ajoutez beaucoup de définitions de points les résultats de transformation calculés ne sont pas toujours améliorés, mais il fournit des vérifications supplémentaires concernant les paramètres calculés.

Statistiques

Le groupe Statistiques affiche l'inclinaison d'ajustement maximum et le résidu vertical maximum. Utilisez ces statistiques pour vérifier si des valeurs incorrectes ont été saisies.

Insérer

Cliquez sur Insérer pour insérer un enregistrement de définition de point. En général Trimble Geomatics Office ajoute automatiquement des nouveaux champs de définition de point donc il ne faut pas utiliser ce bouton très souvent. Cependant, à cause du fait que le premier point dans la liste des points est utilisé comme le point d'origine de l'Ajustement d'élévation, vous pouvez utilisez cette fonction pour placer un point au début de la liste des points voulue.  Cependant, la position utilisée pour l'origine de l'ajustement n'a pas d'importance.

Supprimer

Cliquez sur Supprimer pour supprimer une définition de point de la liste.

Fermer

Lorsque vous cliquez sur Fermer le logiciel Trimble Geomatics Office calcule les paramètres d'ajustement d'élévation et il fait un rapport des erreurs moyennes quadratiques et résiduelles maximum des élévations ajustées. Les erreurs moyennes quadratiques et résiduelles maximum ne sont calculées que lorsque quatre ou plus points sont spécifiés.


 Commande Annuler

Utilisez cette commande pour renverser des actions.

Pour l'accéder, effectuez l'une des choses suivantes:

  Sélectionnez Editer / Annuler.

  Appuyez sur Ctrl+Z.

Dans le menu Editer, le texte s'affichant à côté de la commande Annuler indique l'action à annuler. Les actions peuvent être annulées une à la fois. L'action effectuée le plus récemment est annulée premièrement. Lorsque vous annulez une action par erreur, utilisez la commande Rétablir.

Lorsque vous fermez le projet, ou Trimble Geomatics Office, vous ne pouvez annuler aucune action effectué pendant qu'il était ouvert.

ASTUCE: Pour inverser une opération d'importation après avoir fermer le projet, utilisez l'option De source dans le dialogue Sélectionner points / tabulation Général et supprimez la sélection pour annuler l'opération d'importation.


Commande Rétablir

Utilisez cette commande pour rétablir des actions annulées en utilisant la commande Editer / Annuler.

Pour l'accéder, effectuez l'une des choses suivantes:

  Sélectionnez Editer / Rétablir

  Appuyez sur Ctrl+Y.

Dans le menu Editer, le texte s'affichant à côté de la commande Rétablir indique l'action à rétablir.

Les actions peuvent être rétablies une à la fois.

Lorsque vous fermez le projet, vous ne pouvez rétablir aucune action effectué pendant qu'il était ouvert.


Commande Couper / Copier / Coller

Les commandes Couper, Copier et Coller sont une façon commode de déplacer ou de copier la sélection courante de données.

Du menu Editer, sélectionnez Couper, Copier ou Coller:

  Couper enlève la sélection du projet courant.

  Copier fait un double de la sélection.

  Coller insère la sélection dans un autre projet ou une autre application informatique.

Lorsque vous coupez ou copiez une sélection, elle reste dans le presse-papiers. Chaque nouveau coupage ou copie remplace toute sélection précédente.

La copie de la sélection reste dans le presse-papiers après que vous l'avez collé dans un projet ou un autre application informatique. Vous pouvez coller la sélection à nouveau ou autant de fois que nécessaire.

Utilisez ces commandes pour:

 transférer les données entre vos projets (par exemple, vous pouvez copier les points de contrôle entre projets)

  copier et coller des points ayant des attributs d'un projet vers un autre

  copier et coller le dessin CAD vers un rapport

  copier les données dans un projet avec un système de coordonnées différent (par exemple, les points d'un projet avec un système de coordonnées NAD-27 peuvent être copiés et collés dans un projet avec un système de coordonnées NAD-83)

Toutes les entités sont collées dans un projet en utilisant les valeurs de coordonnées associées aux entités. Ainsi les entités sont positionnées correctement parce que si le projet dans lequel vous collez les entités a une définition de système de coordonnées qui diffère du projet de source des entités, les valeurs des coordonnées sont converties automatiquement au nouveau système de coordonnées.

NOTE: Pour copier et coller des observations d'un projet vers un autre, vous devez utiliser le format Trimble Data Exchange. Vous ne pouvez pas copier des observations entre les projets dans Trimble Geomatics Office.


Commande Supprimer

Utilisez cette commande pour supprimer les entités sélectionnées actuellement.

Pour l'accéder, effectuez l'une de choses suivantes:

  Sélectionnez Editer / Supprimer.

  Appuyez sur Supprimer.

Pour rétablir une commande Supprimer:

  Sélectionnez Editer / Rétablir.


 Commande Renommer points

Utilisez cette commande pour renommer les points.

Pour l'accéder, sélectionnez Editer / Renommer points.

Le dialogue Renommer s'affiche.


Dialogue Renommer

 Vue d'ensemble et plus amples informations

Lorsque vous avez beaucoup de points dans un projet qui ne sont pas le même point physique, vous pouvez résoudre les points en double en le renommant à l'aide du dialogue Renommer.

Exemple

Vous importez deux fichiers Survey Controller collectés par des différentes équipes:

  Une équipe utilise des noms de point de 1000 à 2000.

  L'autre équipe utilise des noms de point de 1000 à 1050.

Quand les fichiers sont importés pendant que l'option Fusionner points en double dans la tolérance est configurée dans le dialogue Options de points en double, 51 points en double (1000 à 1050) sont créés dans la base de données.

Lorsque les fichiers sont importés avec l'option Toujours fusionner les points en double configurée dans le dialogue Options de points en double,  vous indique quels points en double sont fusionnés. Si vous voulez renommer les points en double, annuler l'option d'importation et puis les renommer.

Pour ce faire:

  Sélectionner Editer / Renommer points. Le dialogue Renommer s'affiche.

Renommer

Pour renommer les points, sélectionnez l'une des options suivantes:

Sélection	Cette option renomme les points sélectionnés.

BD entière	Cette option renomme tous les points.

Lorsqu'aucun point n'est sélectionné, tous les points dans la base de données sont renommés par défaut.

ASTUCE: Pour renommer les coordonnées d'un fichier .dc, utilisez le dialogue Sélectionner points pour sélectionner les points de la source.

Méthode

Le groupe Méthode a les options suivantes:

Renommer de

Cliquez sur Renommer de et saisissez le premier des noms que vous voulez remplacer.

Note: Lorsque vous saisissez un nom alphabétique, sous les points dans la sélection seront donnés le même nom. Lorsqu'une sélection est réalisée, saisissez une valeur numérique pour le dernier caractère du nom afin d'incrémenter les noms de point.

Ajouter préfixe

Cliquez sur Ajouter préfixe et saisissez un préfixe. Par exemple, pour indiquer les points Topo, saisissez le préfixe TOPO au nom de point existant sélectionné.

ASTUCE: Mettez un espace pour le dernier caractère du préfixe. Ainsi le point renommé est distingué du point originel.

Ajouter suffixe

Cliquez sur Ajouter suffixe et saisissez un suffixe. Par exemple, ajoutez un suffixe de date aux points pour indiquer la date du levé (par exemple, 08/10/99).

ASTUCE: Mettez un espace pour le premier caractère du suffixe. Ainsi le point renommé est distingué du point original.

Ajouter constante

Utilisez cette option pour ajouter une constante aux points sélectionnés. La valeur constate peut être positive ou négative. La table suivante fournit des exemples:

NOTE: P1023 - 1000 mène à P23. Si le résultat requis est P0023 vous pouvez effectuer une deuxième opération de renommer, P23 + 0000.

Rapport

Sélectionnez cette case à cocher pour créer un rapport des noms originaux et nouveaux. Lorsque vous cliquez sur OK, le rapport s'affiche dans l'afficheur HTML par défaut.


Commande Propriétés

Utilisez cette commande pour  ouvrir la fenêtre Propriétés.

Pour l'accéder, sélectionnez Editer / Propriétés.

La fenêtre Propriétés ouvre.


    Style CAD - Vue d'ensemble

Plus amples informations

Voir aussi	Dialogue
Vue d'ensemble

Dans la Vue en plan, les styles CAD configure comment les entités (les points, les lignes et le texte) sont affichées dans la fenêtre graphique de Trimble Geomatics Office.

Les styles CAD comprennent:

  Styles de point

  Styles de ligne

  Styles de point

Styles de point et styles de ligne

Les styles de point et les styles de ligne configure comment les points et les lignes s'affichent.

Un style de point consiste en un symbole de point et les paramètres qui configurent la taille, la couleur et les dimensions du symbole. Pour créer et éditer les symboles de point, utilisez l'utilitaire Editeur de symboles.

Un style de ligne consiste en un type de ligne et les paramètres qui configurent la taille, la couleur et les dimensions du type. Pour créer et éditer les types de ligne, utilisez l'utilitaire Editeur de symboles.

Pour configurer les styles d'un point ou d'une ligne, utilisez l'une des options suivantes:

  La fenêtre Propriétés.

  Le traitement de codes de caractéristiques.

Pour créer et éditer les styles de point, utilisez le dialogue Styles CAD dans Trimble Geomatics Office, ou l'utilitaire Editeur de caractéristiques et attributs.

Les modèles par défaut de Trimble ont un style de point et un style de ligne pour chaque symbole et type de ligne dans le fichier Symbols (*.sym) et le fichier Linetype (*.ltp). Les styles ont le même nom que le symbole ou le type de ligne.

Si un style de point ou de ligne a son champ de style de point ou son champ de style de ligne configuré sur <Par couche>, le style de point ou le style de ligne assigné aux points ou aux lignes est pris de la couche. Les styles par défaut sont configurés pour la couche dans laquelle les points et les lignes se trouvent.

Pour configurer les styles par défaut des différentes couches, utilisez le dialogue Couches.

Styles de texte

Les styles de texte configurent comment les entités de texte s'affichent dans Trimble Geomatics Office. Utilisez la fenêtre Propriétés pour configurer les style de texte des entités texte. Un style de texte consiste en une police, d'informations de taille et du format de police. Trimble fournit un style de texte, appelé Standard, mais vous pouvez en créer vos propres styles. Pour ce faire, utilisez le dialogue Styles CAD dans Trimble Geomatics Office.


    Commande Styles

Utilisez cette commande pour créer et gérer des styles de point, des styles de ligne et des styles de texte.

Pour l'accéder, effectuez l'une de choses suivantes:

  Sélectionnez Editer / Styles.

  Appuyez sur Ctl+S.

Le dialogue Styles CAD s'affiche.


    Dialogue Styles CAD

 Vue d'ensemble et plus amples informations

Utilisez le dialogue Styles CAD pour créer et gérer des styles de point, de ligne et de texte.

Pour y accéder:

  Sélectionnez Editer / Styles.

Nouveau

Ouvre le dialogue Créer style et crée un nouveau style basé sur la tabulation courante.

Editer

Cliquez sur Editer pour éditer un style existant. Lorsque vous éditez un style CAD utilisé  par une entité dans le projet, l'entité est mise à jour avec les nouveaux paramètres de style CAD.

Supprimer

Cliquez sur Supprimer pour supprimer un style. Vous pouvez supprimer tout style CAD qui n'est pas utilisé par une entité dans le projet.


     Dialogue Créer/Editer le style de point

Utilisez ce dialogue pour créer ou éditer un style de point.

Pour y accéder:

1.	De la Vue en plan, sélectionnez  Editer / Styles. Le dialogue Styles CAD s'affiche.

2.	Dans la tabulation Styles de point, effectuez l'une des chose suivantes:

  Cliquez sur Nouveau. Le dialogue Créer style de point s'affiche.

  Cliquez sur Editer. Le dialogue Editer le style de point s'affiche.

Une fois que vous avez sélectionné un style, le bouton Editer devient disponible.

Nom

Utilisez le champ Nom pour entrer le nom du nouveau Style de point ou renommer le nom courant du style de point . Tous les noms de styles de point doivent être uniques, mais vous pouvez utiliser le même nom pour un style de point, un style de ligne ou un style de texte.

Symbole

Sélectionnez un symbole de la liste Symbole. Pour afficher et éditer un symbole utilisez l'utilitaire Symbol Editor™.

Couleur

Sélectionnez une couleur pour le symbole. Lorsque vous sélectionnez Texte de fenêtre, le style sera affiché utilisant les paramètres Texte de la fenêtre dans Microsoft Windows. Pour éditer les paramètres de Texte de la fenêtre, utilisez le Panneau de configuration. Si Texte de la fenêtre est utilisé, le style par défaut est noir pendant le traçage lorsque vous avez le Style de légende par défaut sélectionné. Pour sélectionner le Style de légende par défaut, choisissez Affichage / Options / tabulation Légende de levé. Si vous avez le Style de légende noir sélectionné, le style par défaut est blanc pendant le traçage.

Certaines définitions de symbole ont des couleurs définies. Celles-ci sont configurées lors de la création du symbole dans l'utilitaire Editeur de symboles de Trimble. Par exemple, en utilisant l'utilitaire Editeur de symboles de Trimble, la couleur pour le symbole Haie peut être définie comme vert. Si vous sélectionnez des symboles ayant des définitions de couleur, le champ Couleur n'est pas disponible, et le champ Couleur affichera 'de symbole'.

Diamètre

Saisissez le diamètre du symbole. Les unités du champ Diamètre changent selon si l'option Dimensions est configuré sur Papier ou Au sol. Pour certains symboles pour lesquels le diamètre n'est pas applicable (tel qu'un point pixel unique) ce champ n'est pas disponible.

Papier	Quand l'option Papier est sélectionnée, la valeur du diamètre du symbole contrôle la dimension du symbole sur papier.

Par exemple, lorsque Dimensions est configurée à Papier, et vous entrez 5 dans le champ Diamètre, Trimble Geomatics Office tracera un symbole de cinq millimètres dans un report métrique.
Les unités utilisées par Trimble Geomatics Office sont les Unités de coordonnées. Pour configurer les Unités de coordonnées, sélectionnez Fichier / Propriétés du projet /  tabulation Unités et format.

La représentation des unités papier sur l'écran est calculée utilisant l'échelle du report. Pour configurer l'échelle du report, sélectionnez le dialogue Fichier / Mise en page.

Au sol	Quand l'option Au sol est sélectionnée, la valeur du diamètre du symbole est la taille au sol.

Par exemple, si Dimensions est configurée sur Au sol, et si vous entrez 5 dans le champ Diamètre, Trimble Geomatics Office produira un symbole d'une taille de cinq mètres (ou un symbole d'une taille de cinq pieds, selon les paramètres des Unités de coordonnées). Ce résultat se produit sur un rapport et sur l'écran.

Pour configurer les Unités de coordonnées, sélectionnez Fichier / Propriétés du projet / tabulation Unités et format.

Rotation

Utilisez le champ Rotation pour contrôler l'orientation du symbole. L'interprétation du champ dépend de la configuration des dimensions.

Papier	Quand l'option Papier est sélectionné, le symbole est toujours tracé de façon qu'il est orienté vers la tête de la page sur laquelle il est tracé.

Au sol	Quand l'option Au sol est sélectionnée, le symbole est tracé par rapport au nord.

Dimensions

Cliquez sur Papier ou Au sol pour contrôler l'opération des champs de Diamètre et de Rotation comme décrit ci-dessus.


    Dialogue Créer/Editer le style de ligne

Utilisez ce dialogue pour créer ou éditer un style de ligne.

Pour y accéder:

1.	De la Vue en plan, sélectionnez  Editer / Styles. Le dialogue Styles CAD s'affiche.

2.	Dans la tabulation Styles de ligne, effectuez l'une des chose suivantes:

  Cliquez sur Nouveau. Le dialogue Créer style de ligne s'affiche.

  Cliquez sur Editer. Le dialogue Editer le style de ligne s'affiche.

(Le bouton Editer devient disponible lorsque vous avez sélectionné un style.)

Nom

Saisissez le nom d'un nouveau style de ligne ou renommez un style de ligne existant. Tous les noms de styles de ligne doivent être uniques, mais vous pouvez utiliser le même nom pour un style de point, un style de ligne ou un style de texte.

Paramètres

Utilisez le groupe Paramètres pour définir l'apparence du style.

Types de ligne

Sélectionnez le type de ligne pour la ligne. Pour afficher et éditer un type de ligne, utilisez l'utilitaire Trimble Line Type Editor™.

Couleur

Sélectionnez une couleur pour le style de ligne. Lorsque vous sélectionnez Texte de fenêtre, le style sera affiché utilisant les paramètres Texte de fenêtre de Microsoft Windows. Pour éditer les paramètres Windows, utilisez le Panneau de configuration. Si Texte de fenêtre est utilisé, le style par défaut est noir pendant le traçage lorsque vous avez le Style de légende par défaut sélectionné. Pour sélectionner le Style de légende par défaut, choisissez Affichage / Options / tabulation Légende de levé. Si vous avez le Style de légende noir sélectionné, le style par défaut est blanc pendant le traçage.

Certaines définitions de type de ligne ont des couleurs définies. Celles-ci sont configurées lors de la création du type de ligne dans l'utilitaire Editeur de types de ligne de Trimble. Par exemple, en utilisant l'utilitaire Editeur de types de ligne de Trimble, la couleur pour le type de ligne Haie peut être définie comme vert. Si vous sélectionnez des types de ligne ayant des définitions de couleur, le champ Couleur n'est pas disponible, et le champ Couleur affichera 'de type de ligne'

Longueur

Saisissez la longueur de la répétition du modèle. Ce champ est disponible si le type de ligne est un modèle qui se répète  lorsque le style de ligne est sélectionné pour une ligne. Par exemple, vous pouvez sélectionner une 'ligne de barrière' comme un type à répéter. La figure suivante en est une illustration

Certains types de lignes ont une longueur inhérente (par exemple, Porte) et ce champ n'est pas disponible.

L'interprétation de ce champ dépend de la configuration du champ Dimensions. Les unités du champ Longueur changent selon que le champ Dimensions est configuré sur Papier ou Au sol.

Papier	Quand l'option Papier est sélectionnée, la valeur de la longueur est configurée en millimètres (mm) ou pouces. (Les unités dépendront des paramètres des Unités de coordonnées. Pour configurer ces unités, sélectionnez Fichier / Propriétés du projet / tabulation Unités et format.)

Par exemple, si Dimensions est configurée sur Papier, l'entrée de 5 dans le champ Longueur tracera une ligne en utilisant une répétition de modèle de cinq mm (ou pouces).

La représentation des unités de papier à l'écran est calculée en utilisant l'échelle de report. Pour configurer l'échelle de report, sélectionnez Fichier / Mise en page.

Au sol	Quand l'option Au sol est sélectionnée, la valeur de la longueur sera la valeur au sol configurée pour les unités de coordonnées. Pour configurer ces unités, sélectionnez Fichier / Propriétés du projet / tabulation Unités et format.

Par exemple, si Dimensions est configurée sur Au sol, l'entrée de 5 dans le champ Longueur produira (sur un report et à l'écran) une ligne avec une répétition de modèle de cinq mètres (ou pieds).

Largeur

Saisissez la largeur de la ligne. Pour certains types de ligne (par exemple, solide) la largeur n'est pas applicable et le champ n'est pas disponible. La figure suivante illustre comment le champ Largeur est appliqué:

L'interprétation de ce champ dépend de la configuration du champ Dimensions.

Papier	Lorsque l'option Dimensions est configurée sur Papier, le type de ligne est configuré en millimètres (mm) ou pouces. (Les unités dépendront des paramètres des Unités de coordonnées. Pour configurer ces unités, sélectionnez Fichier / Propriétés du projet / tabulation Unités et format.)

Au sol	Lorsque l'option Dimensions est configurée sur Au sol, la valeur du type de ligne sera les valeurs au sol configurées pour les Unités de coordonnées dans Fichier / Propriétés du projet / tabulation Unités et format.

Dimensions

Cliquez sur Papier ou Au sol pour contrôler l'opération des champs Longueur et Largeur comme décrit ci-dessus.


     Dialogue Créer/Editer le style de texte

Utilisez le dialogue Créer/Editer le style de texte pour créer ou éditer un style de texte.

Pour y accéder:

1.	De la Vue en plan, sélectionnez  Editer / Styles.

Le dialogue Styles CAD s'affiche.

2.	Sélectionnez la tabulation Styles de texte

3.	Effectuez l'une des chose suivantes:

{ [image]  Cliquez sur Nouveau. Le dialogue Créer style de texte s'affiche.

  Cliquez sur Editer. Le dialogue Editer le style de texte s'affiche.

Une fois que vous avez sélectionné un style, le bouton Editer devient disponible.

Nom

Saisissez le nom du nouveau style de texte ou renommez le nom courant du style de texte. Tous les noms de styles de texte doivent être uniques, mais vous pouvez utiliser le même nom pour un style de point, un style de ligne ou un style de texte.

Police

Sélectionnez la police pour le style de texte.

Les noms de police affichés dans la liste sont les polices disponibles dans votre système Microsoft Windows.

Couleur

Sélectionnez la couleur du texte. Lorsque vous ne sélectionnez pas une couleur, la couleur par défaut est le paramètre Texte de la fenêtre de Microsoft Windows. Pour éditer le paramètre Texte de la fenêtre, utilisez le Panneau de configuration. Si Texte de la fenêtre est utilisé, le style par défaut est noir pendant le traçage lorsque vous avez le Style de légende par défaut sélectionné. Pour sélectionner le Style de légende par défaut, choisissez Affichage / Options / tabulation Légende de levé. Si vous avez le Style de légende noir sélectionné, le style par défaut est blanc sur l'écran.

Hauteur

Saisissez la hauteur du style de texte.

L'interprétation de ce champ dépend du paramètre des dimensions.

Papier	Lorsque l'option Papier est sélectionnée, la valeur de hauteur est configurée en millimètres (mm) ou pouces. (Les unités dépendent des unités de coordonnées configurées dans Fichier / Propriétés du projet / tabulation Unités et format.)

Par exemple, si l'option est configurée sur Papier, l'entrée de 5 dans le champ Hauteur crée une hauteur de texte de 5 mm sur un report métrique. La représentation des unités de papier à l'écran est calculée en utilisant l'échelle de report spécifiée à partir de  Fichier / Mise en page.

Au sol	Avec l'option Au sol sélectionnée, la valeur de hauteur est les valeurs au sol configurées pour les unités de coordonnées dans Fichier / Propriétés du projet / tabulation Unités et format.

Par exemple, si l'option est configurée sur Au sol, l'entrée de 5 dans le champ Hauteur affichera (sur un report et à l'écran) sous la forme d'un texte de hauteur de cinq mètres sur un report métrique.

Style de police

Sélectionnez les cases à cocher gras, italique ou souligné pour créer l'effet de texte requis.

Dimensions

Cliquez sur Papier ou Au sol pour contrôler l'opération du champ Hauteur comme décrit ci-dessus.


    Commande Couches

Utilisez cette commande pour gérer les couches du projet courant.

Pour  y accéder:

  Sélectionnez Editer / Couches.

Le dialogue Couches s'affiche.


    Dialogue Couches

 Vue d'ensemble et plus amples informations

Utilisez ce dialogue pour gérer les couches du projet courant.

Pour l'accéder, effectuez l'une de choses suivantes:

  Sélectionnez Editer / Couches.

  Cliquez sur l'outil Couches [image].

  Appuyez sur Ctrl+L.

La couche courante s'affiche en haut du dialogue.

Nom

Saisissez un nom pour une couche. Le nom doit être unique. Si vous changes le nom d'une couche contenant des entités, les entités sont déplacées à la nouvelle couche.

Style de point

Sélectionnez un style de point de la liste. Les entrées dans la liste sont les styles prédéfinis de Trimble et les nouveaux styles de point que vous avez créé dans Trimble Geomatics Office (utilisant Editer / Styles), ou pendant le traitement des codes de caractéristiques. (Pour plus amples informations, référez-vous à l'aide de l'utilitaire Feature and Attribut Editor .)

Ce style de point est le style utilisé pour tous les points dans cette couche avec l'option <Par couche> sélectionnée pour leur style de point.

Style de ligne

Sélectionnez un style de ligne de la liste. Les entrées dans la liste sont les styles prédéfinis de Trimble et les nouveaux styles de ligne que vous avez créé dans  Trimble Geomatics Office (utilisant Editer / Styles), ou pendant le traitement des codes de caractéristiques. (Pour plus amples informations, référez-vous à l'aide l'utilitaire Feature and Attribut Editor .)

Ce style de ligne est le style utilisé pour tous les points dans cette couche avec l'option <Par couche> sélectionnée pour leur style de ligne.

Style de texte

Sélectionnez un style de texte de la liste. Les entrées dans la liste sont les styles prédéfinis de Trimble et les nouveaux styles de texte que vous avez créé dans  Trimble Geomatics Office (utilisant Editer / Styles).

Ce style de texte est le style utilisé pour tout le texte et les annotations dans cette couche avec l'option <Par couche> sélectionnée pour leur style de texte.

Verrouillée

Afin de permettre la sélection des entités dans la couche de la fenêtre Trimble Geomatics Office, effacez cette case à cocher. Lorsque vous ne voulez pas la sélection des entités dans cette couche, sélectionnez cette case à cocher. La valeur de ce champ s'affiche dans l'outil Sélection de couche dans la barre d'outils Couches.

Si la couche est verrouillée, l'icône [image] s'affiche dans la barre d'outils Couches.

Si la couche n'est pas verrouillée, l'icône [image] s'affiche dans la barre d'outils Couches.

Si une ou plusieurs couches ont leur champ Verrouillée sélectionné, l'icône [image] s'affiche dans la barre d'état.

NOTE: Lorsque vous sélectionnez cette case à cocher, vous ne pouvez afficher aucune entité dans la couche à l'aide des méthodes suivantes:

  utilisant la fenêtre Propriétés

  les sélectionnant et en cliquant

  utilisant les commandes Sélectionner dans le menu Sélectionner

Pour visualiser des entités dans la couche de la fenêtre graphique, sélectionnez cette case à cocher. La valeur de ce champ s'affiche dans l'outil Sélection de couche dans la barre d'outils Couches.

Si la couche est visible, l'icône [image] s'affiche dans la barre d'outils Couches.

Si la couche n'est pas visible, l'icône [image] s'affiche dans la barre d'outils Couches.

Pour masquer les entités dans la couche, effacez cette case à cocher.

NOTE: Lorsque vous effacez cette case à cocher, vous ne pouvez visualiser aucune entité dans la couche de la fenêtre graphique Trimble Geomatics Office.

L'icône Couches masquées [image] s'affiche dans la barre d'état lorsqu'une ou plusieurs couches sont masquées. C'est à dire, cette icône s'affiche dans la barre d'état lorsqu'une ou plusieurs couches ont la propriété Visible effacée.

ATTENTION: Lorsqu'une couche n'est pas visible vous ne pouvez pas afficher ou sélectionner des entités de cette couche. C'est applicable lorsque la couche est verrouillée ou non. C'est à dire, l'option Visible annule l'option Verrouillée.

Nouveau

Cliquez sur Nouveau pour ajouter une nouvelle couche au projet. Une nouvelle ligne est ajoutée dans le dialogue Couche. Saisissez des valeurs pour chaque colonne dans la nouvelle couche.

Supprimer

Cliquez sur Supprimer pour supprimer la couche soulignée actuellement. Lorsque vous supprimez une couche contenant des entités, un dialogue d'avertissement s'affiche. Le dialogue vous avertit que si vous supprimez la couche, les entités dans cette couche seront supprimées.

Courant(e)

Cliquez sur Courant(e) pour rendre la couche sélectionnée la couche courante. La couche courante est soulignée. La couche courante est la couche à laquelle les nouvelles entités sont ajoutées. Toute entité (points, lignes, arcs, courbes, texte et annotations) ajoutée à la base de données sans un nom de couche compris est ajoutée à la couche courante.

ASTUCE: Vous pouvez éditer les entrées multiples dans le dialogue Couches:

1.	Utilisez la touche Ctrl pour sélectionne les entrées multiples.

2.	Editez le dernier champ souligné. Toutes les entrées soulignées seront mises à jour.


    Couches - Vue d'ensemble

Plus amples informations

Comment ...	Dialogue	Voir aussi
Vue d'ensemble

Dans le Vue en plan, chaque entité appartient à une couche. Une couche consiste en un nom, un style de point, un style de ligne et un style de texte. L'organisation des données en couches Il est plus facile de gérer les données lorsqu'elles sont organisées en couches. Ce sert lors de la gestion des groupes de données grandes.  Par exemple, tous les points avec le code de caractéristiques 'arbre' peuvent être groupés dans une couche appelée "Arbres". A moins que vous n'avez créé des nouvelles couches dans le projet, la couche Par défaut est utilisée pour toutes les entités.

Toutes les entités (points, lignes, arcs, courbes, texte et annotations) ont un champ Couches. Utilisez la fenêtre Propriétés pour afficher les entités. Pour sélectionner une couche pour une entité, utilisez l'une des méthodes suivantes:

  Lors de l'ajout d'une entité au projet, sélectionnez la couche du champ Couche dans le dialogue approprié.

  Importez un fichier avec la couche spécifiée.

  Utilisez le traite des codes de caractéristiques.

  Pour sélectionner la couche pour les entités existantes dans le projet. utilisez la fenêtre Propriétés.

Pour gérer les couches du projet, utilisez le dialogue Couches.

Utilisez la barre d'outils Sélection pour spécifier quelles couches vous pouvez visualiser et sélectionner. Pour sélectionner les entités dans une couche choisie:

  Choisissez Sélectionner / Par couche.

Si un point dans la base de données a <Par couche> sélectionné dans le champ Style de point, le style de point spécifié pour la couche est assigné au point. Si une ligne, une courbe ou un arc a <Par couche> sélectionné dans le champ Style de ligne, le style de ligne spécifié pour la couche est assigné à la ligne, la courbe ou l'arc. De la même façon, si un texte ou une annotation a <Par couche> sélectionné dans le champ Style de texte, le style de texte spécifié pour la couche est assigné au texte ou à l'annotation.


Points en double - Vue d'ensemble

Plus amples informations

Comment ...	Dialogue
Vue d'ensemble

 Trimble Geomatics Office fournit deux options pour l'importation des points avec le même nom:

1.	Toujours fusionner les points en double

2.	Fusionner points en double dans la tolérance. Vous pouvez spécifier dans quelle tolérance verticale et horizontale  fusionnera les points, ou que  n'effectuera pas la résolution des points en double lors de l'importation.

Par défaut,  Trimble Geomatics Office fusionne les points avec le même nom, donc les points en double ne sont pas créés. (C'est la façon la plus rapide pour l'importation des données.)

Si vous décidez de ne pas fusionner des points en double, les points en double sont créés lorsque vous importer un ou plusieurs fichiers qui ont des observations ou des coordonnées pour le même point qui sont hors de tolérance. Par exemple, lorsque vous importez:

  un fichier contenant un point avec le même nom qu'un point existant dans la base de données et ils ne sont pas dans la tolérance.

  les points avec le même nom à partir de fichiers Survey Controller (*.dc) différents

Si nécessaire, vous pouvez également fusionner les points à partir du logiciel, à l'aide de la commande Fusionner points en double.

Comment  Trimble Geomatics Office fusionne les points en double

Un point ne peut avoir qu'une coordonnée WGS-84 saisie de l'utilisateur et une coordonnée de grille/locale saisie de l'utilisateur.  Trimble Geomatics Office choisit quelles coordonnées WGS-84 ou de grille/locale existantes à écraser selon ces règles:

1.	Les coordonnées avec la qualité la plus haute sont prises.

2.	Le dernier composant de coordonnée saisi dans la base de données est utilisé.

Le Rapport de dérivation du point fournit des détails concernant les données de levé utilisées pour coordonner le points, pendant que le Rapport d'importation détaille tout composant de coordonnée écrasé ou écarté.

NOTE: Si  écrase une coordonnée que vous voulez utiliser, vous devez annuler l'opération d'importation, éditer le fichier d'importation (ou le nom du point dans le projet Trimble Geomatics Office), et réimporter les données. Sinon, annuler l'opération d'importation dans le dialogue Options de points en double. Vous pouvez alors renommer les points en double dans  Trimble Geomatics Office.


Dialogue Options de points en double

Plus amples informations

Comment ...	Voir aussi

Utilisez le dialogue Options de points en double pour spécifier comment  effectue la gestion des points avec le même nom lors de leur importation dans un projet.

Les paramètres dans ce dialogue s'appliquent à toutes les options d'importation; si vous importez les fichiers à l'aide de la barre du projet, ou si vous faites glisser déposer des fichiers dans un projet Trimble Geomatics Office,  utilise les derniers paramètres d'Options de points en double que vous avez spécifié. Vous pouvez créer un modèle pour configurer les Options de points en double.

Pour accéder au dialogue Options de points en double:

1.	Sélectionnez Fichier / Importer. Le dialogue Importer s'affiche.

2.	Cliquez sur Options.

Les sections suivantes décrivent les options disponibles.

Toujours fusionner les points en double

Par défaut,  Trimble Geomatics Office fusionne les points avec le même nom, donc les points en double ne sont pas créés. Après l'importation, le Rapport d'importation détaille tout composant de coordonnée écrasé ou écarté.

NOTE: Si  écrase une coordonnée que vous voulez utiliser, vous devez annuler l'opération d'importation, éditer le fichier d'importation (ou le nom du point dans le projet Trimble Geomatics Office), et réimporter les données. Sinon, annuler l'opération d'importation dans le dialogue Options de points en double. Vous pouvez alors renommer les points en double dans  Trimble Geomatics Office.

Fusionner points en double dans la tolérance

Si vous sélectionnez cette option, toutes les données de levé d'un point sont mises sous le même point si c'est dans la tolérance verticale et/ou horizontale que vous spécifiez. La tolérance par défaut est déterminée à partir de la tolérance horizontale et verticale inconnue spécifiée dans la tabulation Recalculer du dialogue Propriétés du projet. Après l'importation, le Rapport d'importation détaille toute coordonnée écrasée ou écartée.

Vous pouvez également utiliser cette option pour spécifier que le logiciel ne fusionne pas les points en double lors de l'importation. Pour ce faire, configurez la tolérance pour l'importation des données sur 0,02 (c'est à dire, la tolérance de qualité de levé horizontale) et 0,050 (c'est à dire, la tolérance de qualité de levé verticale). Vous pouvez alors renommer les points en double dans  Trimble Geomatics Office. Si la tolérance horizontale et la tolérance verticale sont spécifiées, afin que les points en double soient fusionnés ils doivent être dans la tolérance verticale ainsi que horizontale. Si une tolérance n'est pas spécifiée (c'est à dire, dans le groupe Tolérances, les cases à cocher Horizontal et Vertical ne sont pas sélectionnées) les points en double sont fusionnés.

NOTE: Lors de l'importation de données de niveaux numériques, seule la tolérance verticale est utilisée.

NOTE: Si la tolérance est configurée sur 0,000, les coordonnées des points en double doivent être exactement les mêmes pour les points à fusionner.


Commande Exploser points

Utilisez cette commande pour créer plusieurs points d'un point, en séparant les données:

  selon la source de données

  selon toutes les observations

  selon le type de données (GPS/Terrestre)

Pour l'accéder, sélectionnez Editer / Exploser points.

Le dialogue Exploser points s'affiche.

Utilisez l'option Exploser points pour:

  résoudre les points en double

  calculer les positions pour chaque observation

  effectuer une calibration dans  Trimble Geomatics Office lorsque les données GPS et les données de grille/locales sont sous le même point

  séparer les observations GPS et terrestres


Dialogue Exploser points

Un point peut avoir des données de levé de plus d'une source de données. La source de données peut également contenir plus d'une observation. Cependant seulement une observation est utilisée afin de dériver la position calculée du point.

Vous pouvez utiliser l'option Exploser points pour créer des points en séparant les données selon la source de données, toutes les observations, ou le type de données (GPS/Terrestre). Exploser les points pour:

  résoudre les points en double

  calculer les positions pour chaque observation

  effectuer une calibration dans  Trimble Geomatics Office lorsque les données GPS et les données de grille/locales sont sous le même point

  séparer les observations GPS et terrestres

Lorsque vous explosez des points, l'opération crée des points avec le même nom.

Par exemple, vous importez deux fichiers Survey Controller (*.dc) au même projet:

  Le premier fichier .dc file contient une observation conventionnelle au point ‘101’.

  Le deuxième fichier .dc contient une coordonnée saisie du clavier et une observation GPS au même point ‘101’.

Lorsque le premier fichier est importé, un point unique ‘101’ est créé dans le projet et l'observation est utilisée pour dériver la position du point/ Lorsque le deuxième fichier est importé, Trimble Geomatics Office fait une comparaison entre la coordonnée saisie du clavier du fichier .dc et la position du point dans le projet. Etant donné que les données de levé du fichier .dc sont pour le même point ‘101’, le point du projet et le fichier .dc sont fusionnés automatiquement. Les données de levé seront de deux sources de données: le premier et le deuxième fichiers Survey Controller (*.dc). Il y aura trois groupes de sources de coordonnées pour le point:

  Observation conventionnelle

  Coordonnée saisie du clavier

  Occupation GPS

ASTUCE: Vous pouvez afficher la source de données et le contenu d'observation et coordonnées d'un point dans le menu de fenêtre Propriétés .

Vous ne pouvez exploser les points que s'il y a un point dans le groupe de sélection courant qui contient des observations ou des coordonnées multiples.

Pour accéder au dialogue Exploser points, sélectionnez Editer / Exploser points.

Exploser

Pour exploser les points afin de créer des points séparés,  effectue la séparation des données en utilisant l'une des méthodes suivantes:

Par source de données	Par exemple, un fichier .dc.
Par toutes les observations	Des points séparés sont créés à partir de toutes les observations et les coordonnées du point.
Par type de données	Toutes les observations GOS et les coordonnées WGS-84 sont mises sous un point. Toutes les coordonnées de grille et les observations terrestres sont mises sous un autre point.

NOTE: Si vous devez effectuer une calibration du site GPS dans  Trimble Geomatics Office, et les observations GPS et les coordonnées de grille sont sous le même point, utilisez l'option Exploser points par type de données. Cette opération crée deux points: un point de grille et un point GPS. Vous pouvez ainsi effectuer une calibration du site utilisant ces points.

Renommer les points GPS avec suffixe (_GPS)

Si vous sélectionnez l'option Par types de données, l'option Renommer les points GPS avec suffixe devient disponible. Vous pouvez rendre les noms de point uniques en choisissant le suffixe à ajouter aux points GPS. Si vous ne sélectionnez pas cette option, les points sont explosés par type de données et chaque point a le même nom.

Afficher statistiques d'explosion

Sélectionnez cette case à cocher pour afficher le nombre de points pouvant être créés des points sélectionnés en utilisant la méthode d'explosion sélectionnée.


Commande Edition multiple

 Voir aussi

Utilisez cette commande pour éditer les propriétés ayant rapport au levé ou ayant rapport à CAD des entités dans la sélection. Cette commande n'est disponible que si vous avez un groupe de sélection courant.

Pour y accéder, effectuez l'une de choses suivantes:

  Sélectionnez Editer / Edition multiple.

  Choisissez Edition multiple du menu de raccourcis.

Le dialogue Edition multiple (tabulation Levé ou tabulation CAD) s'affiche.


Dialogue Edition multiple - tabulation Levé

Utilisez ce dialogue pour éditer les propriétés ayant rapport au levé des entités multiples. Par exemple, vous pouvez corriger les hauteurs d'antenne pour les occupations d'un groupe de points GPS lorsque vous savez qu'elles ont été saisies inexactement sur le terrain..

Pour l'accéder, effectuez l'une de choses suivantes:

  Sélectionnez Editer / Edition multiple.

  Choisissez Edition multiple du menu de raccourcis.

NOTE: La sélection est validée lorsqu'elle est sélectionnée. Elle peut devenir non disponible si elle n'est pas une option valide pour la sélection courante.

Effectuer ces éditions aux points sélectionnés

Ce groupe contient les propriété ayant rapport au levé que vous pouvez éditer pour les points sélectionnés.

Configurer la hauteur d'antenne sur

Sélectionnez cette case à cocher pour activer les champs suivants. Utilisez ce champs pour changer les hauteurs d'antenne des occupations du point.

Changer les hauteurs de taille	Sélectionnez la valeur de hauteur d'antenne des points à changer. La liste contient les hauteurs saisies des points sélectionnés. Seules les hauteurs d'antenne avec la valeur sélectionnée changeront.
Nouvelle hauteur d'antenne	Saisissez la valeur de hauteur d'antenne correcte que vous voulez appliqué.

Configurer les détails d'antenne GPS sur

Sélectionnez cette case à cocher pour activer les champs suivants. Utilisez ce champs pour changer le type d'antenne et/ou la méthode de mesure des points sélectionnés.

Type d'antenne	Sélectionnez le type d'antenne pour les points à changer.
Mesurer à	Sélectionnez la méthode de mesure pour les points à changer. La liste dans ce champ dépend du type d'antenne sélectionné. Lorsqu'un nouveau type d'antenne est sélectionné, ce champ est effacé.

Configurer les détails de destination conventionnelle sur

Sélectionnez cette case à cocher pour activer les champs suivants. Utilisez ces champs pour changer les hauteurs de destination et les constantes prisme des points sélectionnés.

Hauteur de destination	Saisissez la hauteur de destination correcte à appliquer.
Constante prisme	Saisissez la constante prisme correcte à appliquer.

Configurer la hauteur de destination laser

Sélectionnez cette case à cocher pour activer ce champ. Saisissez la hauteur de destination correcte à appliquer.

Effectuer ces éditions aux observation sélectionnées:

Ce groupe contient les propriété ayant rapport au levé que vous pouvez éditer pour les observations sélectionnées (GPS, Conventionnelles, Laser).

Utiliser dans Ajustement de réseau

Sélectionnez cette case à cocher pour activer ce champ. Utilisez ce champ pour utiliser (ou exclure) les observations sélectionnées dans un Ajustement de réseau (si vous avez le module Ajustement de réseau installé).

Cette configuration ne s'applique qu'aux observations GPS ayant d'informations de covariance (QC2). Lorsque c'est configuré sur Non, les observations seront désactivées pour un ajustement de réseau. Cependant, elles sont toujours utilisées pendant le recalcul pour déterminer les positions des points.

Configurer état d'observation sur

Sélectionnez cette case à cocher pour activer le champ. Utilisez le champ pour activer ou désactiver une observation pendant le recalcul ou pour activer les observations comme une vérification.

Lorsque les observations sont désactivées, elles ne seront pas utilisées pendant le recalcul et toute erreur de fermeture sera rapportée pour ces observations. Lorsque vous activez les observations comme une vérification, les observations seront utilisées pour coordonner un point si aucune autre observation ou coordonnée activé n'est disponible.

Supprimer les indicateurs du type

Sélectionnez cette case à cocher pour activer le champ. Sélectionnez les types d'indicateur d'avertissement que vous voulez supprimer dans Affichage du levé. Les indicateurs d'avertissement s'afficheront à nouveau lorsqu'un recalcul est effectué.

NOTE: Lorsque vous sélectionnez une case à cocher afin d'éditer un champ spécifique,  Trimble Geomatics Office effectue une vérification des entités sélectionnées. S'il trouve que les champs ne sont pas utilisés dans les entités sélectionnées, la (les option(s) de case à cocher seront désactivées.


Dialogue Edition multiple - tabulation CAD

Utilisez ce dialogue pour éditer les propriétés ayant rapport à CAD des entités dans la sélection.

Pour l'accéder, effectuez l'une de choses suivantes:

  Sélectionnez Editer / Edition multiple.

  Du menu de raccourcis, choisissez Edition multiple.

Effectuer ces éditions aux points sélectionnés

Ce groupe contient les propriétés ayant rapport à CAD que vous pouvez éditer pour les points sélectionnés.

Configurer le code de caractéristiques sur

Sélectionnez cette case à cocher pour activer ce champ. Saisissez la valeur du code de caractéristiques que vous voulez appliquer aux points sélectionnés.

Configurer couche sur

Sélectionnez cette case à cocher pour activer ce champ. Sélectionnez la couche que vous voulez assigner aux points sélectionnés. Si vous saisissez une couche que n'existe pas, le dialogue Nouvelle couche s'affiche.

Configurer style de point sur

Sélectionnez cette case à cocher pour activer le champ. Sélectionnez le style de point que vous voulez appliquer aux points sélectionnés. Si vous saisissez un style de point que n'existe pas, le dialogue Créer style de point s'affiche.

Supprimer toutes les annotations

Sélectionnez cette case à cocher pour supprimer toute annotation assignée aux points sélectionnées.

Effectuer ces éditions aux lignes sélectionnées

Ce groupe contient les propriétés ayant rapport à CAD que vous pouvez éditer pour les lignes sélectionnées (lignes, courbes, arcs).

Configurer couche sur

Sélectionnez cette case à cocher pour activer le champ. Sélectionnez la couche que vous voulez assigner aux lignes sélectionnées. Si vous saisissez une couche que n'existe pas, le dialogue Nouvelle couche s'affiche.

Configurer style de ligne sur

Sélectionnez cette case à cocher pour activer le champ. Sélectionnez le style de ligne que vous voulez appliquer aux lignes sélectionnées. Si vous saisissez une style de ligne que n'existe pas, le dialogue Créer style de ligne s'affiche.

Supprimer toutes les annotations

Sélectionnez cette case à cocher pour supprimer toute annotation assignée aux lignes sélectionnées.

Effectuer ces éditions aux texte sélectionné

Ce groupe contient les champs que vous pouvez éditer pour le texte sélectionné.

Configurer couche sur

Sélectionnez cette case à cocher pour activer le champ. Sélectionnez la couche que vous voulez assigner au texte sélectionné. Si vous saisissez une couche que n'existe pas, le dialogue Nouvelle couche s'affiche.

Configurer style de texte sur

Sélectionnez cette case à cocher pour activer le champ. Sélectionnez le style de texte que vous voulez appliquer au texte sélectionné. Si vous saisissez un style de texte que n'existe pas, le dialogue Créer style de texte s'affiche.


    Commande Activer/Désactiver observation

Utilisez cette commande pour activer ou désactiver une observation. La désactivation d'une observation la supprimera de toute utilisation dans un calcul de levé.

Pour l'accéder, effectuez l'une de choses suivantes:

  Sélectionnez Editer / Activer/Désactiver observation.

 Dans la fenêtre Propriétés, utilisez le champ Etat d'un observation

 Dans le dialogue Edition multiple / tabulation Levé, utilisez le champ Configurer état d'observation sur

NOTE: Un recalcul prend en considération l'état des observations pendant la détermination de la position calculée d'un point. Vous pouvez utiliser la fonction Activer/Désactiver observations pour désactiver l'état d'une observation afin de la supprimer d'un recalcul.

Vous ne pouvez changer l'état des observation que lorsque:

  Il y a un groupe de sélection courant, et

  Toutes les observations dans le groupe de sélection ont le même état d'observation.

NOTE: Vous pouvez configurer l'état des observations multiples lorsque vous en avez plus d'une dans votre groupe de sélection courant.

Lorsque vous désactivez une observation, la couleur des lignes d'observation se change à rouge foncé.

NOTE: S'il y a des observations multiples le long de la même ligne, l'observation ne s'affiche en rouge que si toutes les observations sont désactivées.


    Commande Inverser flux d'observation

Les vecteurs sont appliqués dans la direction dans laquelle ils sont observés. Utilisez cette commande pour renverser la direction d'une observation et le forcer d'être appliqué dans la direction opposée.

Pour y accéder, sélectionnez Editer/ Inverser flux d'observation

Lorsque vous renversez la direction de l'observation les coordonnées calculées et les qualités d'une point peuvent changer. Le procédé de recalcul doit être utilisé pour appliquer l'observation et calculer les nouvelles positions de point.

CAUTION: Lorsque le point à partir duquel vous voulez faire sortir l'observation n'a pas une position connue (tel qu'un point observé avec RTK), l'observation n'est pas appliquée.


Dialogue Nouvelle couche

Utilisez ce dialogue pour créer une nouvelle couche.  Le dialogue s'affiche lorsque vous saisissez un nom de couche que n'existe pas lors de l'insertion ou l'édition des entités (points, lignes, courbes, arcs, texte).

Nom

Saisissez un nom pour la couche à laquelle la (les) entité(s) sélectionnées seront déplacées.

Point style

Sélectionnez un style de point de la liste. Les entrées dans la liste sont les styles prédéfinis de Trimble et les nouveaux styles de point que vous avez créé dans  Trimble Geomatics Office (utilisant Editer / Styles), ou pendant le traitement des codes de caractéristiques.

Pour plus amples informations, référez-vous à l'Aide de l'utilitaire Editeur de caractéristiques et attributs de Trimble .

Ce style de point est le style utilisé pour tous les points dans la couche avec <Par couche> sélectionné pour leur style de point.

Style de ligne

Sélectionnez un style de ligne de la liste. Les entrées dans la liste sont les styles prédéfinis de Trimble et les nouveaux styles de ligne que vous avez créé dans  Trimble Geomatics Office (utilisant Editer / Styles) ou pendant le traitement des codes de caractéristiques.

Ce style de ligne est le style utilisé pour tous les lignes dans la couche avec <Par couche> sélectionné pour leur style de ligne.

Style de texte

Sélectionnez un style de texte de la liste. Les entrées dans la liste sont les styles prédéfinis de Trimble et les nouveaux styles de texte que vous avez créé dans  Trimble Geomatics Office (utilisant Editer / Styles).

Ce style de texte est le style utilisé pour tout le texte et les annotations dans la couche avec <Par couche> sélectionné pour leur style de texte.

Verrouillée

Afin de permettre la sélection des entités dans la couche de la fenêtre Trimble Geomatics Office, effacez cette case à cocher. Lorsque vous ne voulez pas la sélection des entités dans cette couche, sélectionnez cette case à cocher. La valeur de ce champ s'affiche dans l'outil Sélection de couche dans la barre d'outils Couches.

Si la couche est verrouillée, l'icône [image] s'affiche dans la barre d'outils Couches.

Si la couche n'est pas verrouillée, l'icône [image] s'affiche dans la barre d'outils Couches.

Si une ou plusieurs couches ont leur champ Verrouillée sélectionné, l'icône [image] s'affiche dans la barre d'état.

NOTE: Lorsque vous sélectionnez cette case à cocher, vous ne pouvez afficher aucune entité dans la couche à l'aide des méthodes suivantes:

  utilisant la fenêtre Propriétés

  les sélectionnant et en cliquant

  utilisant les commandes Sélectionner dans le menu Sélectionner

Visible

Pour visualiser des entités dans la couche de la fenêtre graphique, sélectionnez cette case à cocher. La valeur de ce champ s'affiche dans l'outil Sélection de couche dans la barre d'outils Couches.

Si la couche est visible, l'icône [image] s'affiche dans la barre d'outils Couches.

Si la couche n'est pas visible, l'icône [image] s'affiche dans la barre d'outils Couches.

Pour masquer les entités dans la couche, effacez cette case à cocher.

NOTE: Lorsque vous effacez cette case à cocher, vous ne pouvez visualiser aucune entité dans la couche de la fenêtre graphique Trimble Geomatics Office.

L'icône Visible [image] s'affiche dans la barre d'état lorsqu'une ou plusieurs entités dans la couche sont masquées. C'est à dire, cette icône s'affiche dans la barre d'état lorsqu'une ou plusieurs couches ont la propriété Visible effacée.

ATTENTION: Lorsqu'une couche n'est pas visible vous ne pouvez pas afficher ou sélectionner des entités de cette couche. C'est applicable lorsque la couche est verrouillée ou non. C'est à dire, l'option Visible annule l'option Verrouillée.


Commande Déplacer points et texte

Utilisez cette commande pour accéder au dialogue Déplacer points et texte. Cette commande n'est disponible que lorsque vous avez un groupe de sélection courant. Pour l'accéder, sélectionnez Editer / Déplacer points et texte.

Le dialogue Déplacer points et texte s'affiche.


Dialogue Déplacer points et texte

Utilisez ce dialogue pour déplacer des points ou du texte.

Pour l'accéder, sélectionnez Editer / Déplacer points et texte.

Les points avec la qualité de Contrôle, Contrôle fixe, ou Ajusté ne sont pas déplacés. Lorsqu'un point que vous avez sélectionné à déplacer possède une qualité de Contrôle, Contrôle fixe, ou Ajusté, un message s'affiche.

AVERTISSEMENT: Le déplacement d'un point rendra désactivée toute observation au point ou toute coordonnée pour le point. Vous pouvez les réactiver en utilisant la fenêtre Propriétés. La réactivation des observations et des coordonnées peut résulter en le transfert du point déplacé à sa position d'origine.

Sélectionnez le texte ou les points et saisissez le nouvel emplacement dans le dialogue. Le déplacement se produit lorsque vous cliquez sur OK dans le dialogue.

Déplacer par

Sélectionnez l'option Déplacer par pour déplacer le texte ou les points sélectionné au nouvel emplacement défini par les décalages nord et est (deltas).

Deux options sont disponibles:

Delta est	Cette option ajoute la valeur delta est affichée aux Ests de toutes les entités sélectionnées.

Delta nord	Cette option ajoute la valeur delta nord affichée aux Nords de toutes les entités sélectionnées.

Déplacer au point

Sélectionnez la case à cocher Déplacer au point pour déplacer les points sélectionnés à la position d'un autre point.

Nom

Utilisez ce champ pour nommer ou sélectionner le point au nouvel emplacement.

Déplacer aux coord.

Sélectionnez la case à cocher Déplacer aux coord. pour déplacer le texte ou le point sélectionné à la nouvelle position. Deux options sont disponibles:

Est	Utilisez ce champ pour saisir ou sélectionner la nouvelle valeur Est pour le point sélectionné.

Nord	Utilisez ce champ pour saisir ou sélectionner la nouvelle valeur Nord pour le point sélectionné.

Configurer élévations sur nulle

Sélectionnez la case à cocher Configurer élévations sur nulle pour configurer les élévations de point pour les points déplacés à nulle. Si vous ne sélectionnez pas cette option, l'élévation existante du point est conservée.

NOTE: Toute observation à un point qui a été déplacé sera désactivée.


Dialogue Points en double

Plus amples informations

 Voir aussi

Lorsque vous importez un fichier contenant un point avec le même nom qu'un autre point dans le projet, il y aura des points avec le même nom créés dans le projet.  Le dialogue Points en double s'affiche avec le message suivant:

Cette opération d'importation a créé des points ayant le même nom dans la base de données, (points en double).

Vous pouvez résoudre ces points en double en:

   fusionnant les points avec des noms en double lorsque vous êtes sûr qu'ils sont au même point physique

  renommant les points avec des noms en double lorsqu'ils ne sont pas au même point physique.


Feedback

Vos commentaires concernant la documentation nous assistent d'améliorer la documentation avec chaque version.

Pour nous envoyer vos commentaires, effectuez l'une des choses suivantes:

  Envoyez un email à ReaderFeedback@trimble.com

  Remplissez le Formulaire de commentaires de lecture à la fin du manuel et le retourner par courrier ou par télécopie selon les instructions au bas du formulaire. Lorsque le formulaire de commentaires n'est pas disponible, vous pouvez nous envoyer vos commentaires et suggestions à l'adresse suivante:

Land Survey Marketing
Trimble Navigation
7403 Church Ranch Boulevard, Suite 100
Westminster, CO 80021
Etats-Unis

Attention: Publications techniques

Télécopie: +1 720 887-6101

www.trimble.com/

Tout commentaire et suggestion deviennent la propriété de Trimble Navigation Limited.

Nous vous remercions de votre assistance.


Commande Nouveau projet

Utilisez cette commande pour créer un nouveau projet ou modèle.

Pour l'accéder, effectuez l'une des choses suivantes:

  Sélectionnez Fichier / Nouveau projet.

  Cliquez sur le bouton Nouveau projet [image] sur la barre d'outils standard.

  Appuyez sur Ctrl+N.

Le dialogue Nouveau projet s'affiche.


Dialogue Nouveau projet

Utilisez cette commande pour créer un nouveau projet ou modèle.

Pour l'accéder, effectuez l'une des choses suivantes:

 Sélectionnez Fichier / Nouveau projet.

 Cliquez sur le bouton Nouveau projet [image] sur la barre d'outils standard.

 Appuyez sur Ctrl+N.

  Cliquez sur le raccourci Nouveau projet sur la barre du Projet.

Nom

Saisissez le nom du projet que vous voulez créer. Un nouveau dossier avec le nom du projet est créé sur le dossier du projet courant comme sélectionné utilisant le bouton Dossier.

Modèle

La liste Modèle affiche les modèles de projet disponibles dans votre système. L'utilisation des modèles de projet dans Trimble Geomatics Office est semblable à l'utilisation des documents modèles dans Microsoft Word ou des dessins prototypes dans AutoCAD. Vous créez chaque nouveau projet utilisant un modèle pour former la base de votre nouveau projet. Le modèle utilisé reste inchangé.

ASTUCE: Les nouveaux styles CAD ajoutés à un modèle sont disponibles dans les projets basés sur ce modèle. Lorsqu'il est probable que le travail encours dans une zone spécifique nécessitera une série de projets individuels, il se peut que vous voudriez créer un modèle qui comprend le système de coordonnées spécifique de la zone, les styles spéciaux, et les points de contrôle, par exemple, afin que ceux-ci soient disponibles immédiatement dans tout nouveau projet.

Une fois que les modèles du projet sont définis, vous pouvez utiliser les modèles beaucoup de fois pour des projets divers.

Sélectionnez le modèle le plus approprié des noms de modèle affichés dans la liste. Le modèle que vous sélectionnez forme la base du nouveau projet que vous créez.

Détails

La liste Détails affiche les détails du système de coordonnées et les En-têtes du modèle sélectionné. Ce vous assiste de sélectionner le modèle le plus approprié.

En plus, le dossier dans lequel le projet est créé et maintenu s'affiche. Ce dossier reflète ce qui est défini directement dans le champ Nom ou en utilisant Dossier.

Les options de projection sont prédéfinies dans le modèle de projet, et la projection peut être changée.

Dossier

Cliquez sur Dossier pour sélectionner le dossier dans lequel vous voulez créer un nouveau projet.

La sélection d'un dossier en utilisant Dossier change l'emplacement par défaut utilisé par Trimble Geomatics Office la prochaine fois que vous ouvrez ou créez un projet.

Utilisez l'arborescence du dossier affichée dans le dialogue Recherche d'un dossier pour effectuer l'une des choses suivantes:

  Sélectionner un dossier du projet directement.

  Sélectionner un dossier contenant un nombre de projets.

Nouveau

Cliquez sur Nouveau pour créer un nouveau projet ou un nouveau modèle.

Lorsque configuré sur l'option modèle, vous pouvez créer un nouveau modèle dans le dossier approprié basé sur le modèle sélectionné actuellement. Vous pouvez l'utiliser pour générer des nouveaux modèles pour les besoins particuliers.


Commande  Ouvrir projet

Utilisez cette commande pour ouvrir un projet existant.

Pour l'accéder, effectuez l'une des choses suivantes:

  Sélectionnez Fichier / Ouvrir projet.

  Cliquez sur le bouton Ouvrir projet [image] sur la barre d'outils standard.

  Appuyez sur Ctrl+O.

Le dialogue Ouvrir projet s'affiche.


Dialogue Ouvrir projet

Utilisez ce dialogue pour ouvrir un projet existant.

Pour l'accéder, effectuez l'une des choses suivantes:

 Sélectionnez Fichier / Ouvrir projet.

 Cliquez sur le bouton Ouvrir projet [image] sur la barre d'outils standard.

  Appuyez sur Ctrl+O.

NOTE: Lorsqu'il y a un projet ouvert déjà, vous êtes averti que cette action fermera le projet courant.

NOTE: Un aperçu du projet est affiché dans le dialogue Ouvrir projet

Nom

Saisissez le nom du projet que vous voulez ouvrir.

Une liste des projets dans le dossier du projet courant s'affiche sous le champ Nom. Sélectionnez l'un de ces noms de projet pour saisir le nom automatiquement dans le champ Nom.

Vous pouvez également saisir un chemin d'un projet que vous voulez ouvrir dans le champ Nom.

Par exemple, vous pourriez avoir plusieurs projets dans le dossier MONPROJ du lecteur C:\.

Saisissez C:\monproj\. Trimble Geomatics Office affiche les projets dans ce dossier disponibles à la sélection. L'ouverture d'un projet de cette façon ne change pas le dossier du projet par défaut comme lorsque vous cliquez sur Dossier. Référez-vous à la description suivante à propos le bouton Dossier.

Dossier

Cliquez sur Dossier pour sélectionner le dossier à partir duquel un projet sera ouvert.

Utilisez l'arborescence du dossier affichée dans le dialogue Recherche d'un dossier pour effectuer l'une des choses suivantes:

  Sélectionner un dossier du projet directement.

  Sélectionner un dossier contenant un nombre de projets.

Détails

La liste Détails affiche le système de coordonnées et les en-têtes du projet et contient le dossier dans lequel le projet est stocké. Utilisez ces détails pour vous assister d'ouvrir le projet approprié. La case Détails peut être modifiée en utilisant Fichier / Propriétés du projet / tabulation Unités et format.

NOTE: Si vous n'avez pas le module Traitement de lignes de base WAVE ou le module Ajustement de réseau installé, lorsque vous ouvrez un projet existant créé avec ces modules, les données de levé restent inchangées. Vous pouvez visualiser les données dérivées des modules désactivés (telles que les ellipses d'erreur et les lignes de bases traitées). D'autres tâches nécessitant l'utilisation des modules désactivés ne sont pas disponibles.


Commande Fermer projet

Utilisez cette commande pour fermer le projet courant.

Pour l'accéder, sélectionnez Fichier / Fermer projet.

Tous les modifications faites au projet (y compris Groupes de sélection enregistrés et Affichages) sont enregistrées pendant le travail dans le projet. Vous pouvez fermer un projet sans la nécessité d'en premier l'enregistrer.


Commande Supprimer projet

Utilisez cette commande pour supprimer un projet.

Pour l'accéder, sélectionnez Fichier / Supprimer projet.

Le dialogue Supprimer projet s'affiche.


Dialogue Supprimer projet

Utilisez ce dialogue pour supprimer un projet. Le projet doit être fermé.

Pour l'accéder, sélectionnez Fichier / Supprimer projet.

ASTUCE: Vous pouvez récupérer un projet supprimé de la Corbeille.

Nom

Saisissez le nom du projet que vous voulez supprimer.

Dans le dialogue Supprimer, une liste des projet dans le dossier du projet courant  s'affiche sous le champ Nom. Sélectionnez l'un de ces noms de projet pour saisir le nom automatiquement dans le champ Nom.

Vous pouvez également saisir un chemin de répertoire d'un projet que vous voulez supprimer.

Par exemple, vous pourriez avoir plusieurs projets dans votre dossier MONPROJ du lecteur C:\.

Saisissez C:\monproj\.  Trimble Geomatics Office affiche les projets dans ce dossier disponibles à la sélection. La suppression d'un projet de cette façon ne change pas le dossier du projet par défaut comme lorsque vous cliquez sur le bouton Dossier. Référez-vous à la description suivante de Supprimer groupe.

Supprimer groupe

Pour afficher les projets à supprimer, sélectionnez le option Projet.

Pour afficher les modèles à supprimer, sélectionnez l'option Modèle.

Dossier

Cliquez sur Dossier pour sélectionner le dossier duquel il faut supprimer un projet. La sélection d'un dossier en utilisant Dossier change le paramètre par défaut utilisé par Trimble Geomatics Office la prochaine fois que vous ouvrez ou créez un projet.

Utilisez l'arborescence du dossier affichée dans le dialogue Recherche d'un dossier pour effectuer l'une des choses suivantes:

  Sélectionner un dossier du projet directement.

  Sélectionner un dossier contenant un nombre de projets.

Supprimer

Cliquez sur Supprimer pour supprimer le projet sélectionné. Tous les dossiers et les fichiers compris dans le projet sont supprimés.

Détails

Le groupe Détails affiche le système de coordonnées et les en-têtes du projet et montre le dossier dans lequel le projet est stocké. Utilisez ces détails pour vous assister de confirmer que vous supprimez le projet correct.


Commande Importer

Utilisez cette commande pour ouvrir le dialogue Importer.

Pour l'accéder, effectuez l'une des choses suivantes:

  Sélectionnez Fichier / Importer.

  Cliquez sur le bouton Importer [image] sur la barre d'outils standard.

  Utilisez le groupe Importer sur la Barre du projet.

Le dialogue Importer s'affiche.


Commande Exporter

Utilisez cette commande pour exporter les données à un autre format ou à une périphérique Trimble.

Pour l'accéder, effectuez l'une des choses suivantes:

  Sélectionnez Fichier / Exporter.

  Cliquez sur le bouton Exporter [image] sur la barre d'outils standard.

Le dialogue Exporter s'affiche.


Commande Mise en page

Utilisez cette commande pour changer les paramètres de la mise en page avant de produire un report.

Vous pouvez:

  sélectionner le modèle de plan à utiliser pour votre report

  assigner l'échelle, l'origine et l'orientation du report

Pour l'accéder, sélectionnez Fichier / Mise en page.

Le dialogue Mise en page s'affiche.


Dialogue Mise en page

Utilisez ce dialogue pour changer les paramètres de la mise en page avant de produire un report. Vous pouvez sélectionner le modèle de plan à utiliser pour votre report et assigner l'échelle, l'origine et l'orientation du report.

Pour l'accéder, sélectionnez Fichier / Mise en page.

Modèle de plan

Un modèle de plan fournit un modèle dans lequel vous tracez votre dessin. Vous pouvez personnaliser ce modèle afin de fournir des composants du dessin (tels que une flèche Nord ou le logo de votre compagnie) ainsi que la région de découpage, le cadre, les lignes et l'annotation de texte. Le modèle de plan est appliqué à la taille de papier configurée pour l'imprimante / ou traceur sélectionné et se conformera à la configuration portrait / ou paysage de l'imprimante / traceur.

Vous pouvez modifier ces paramètres en utilisant Config.d'impression.

Dans le champ Fichier, saisissez le nom de fichier du modèle de plan requis.

Parcourir

Le dialogue Parcourir affiche une liste de fichiers de modèles de plan (*.frm) disponibles.

Sélectionnez le nom de fichier du modèle de plan requis.

Trimble Navigation Limited fournit plusieurs différents modèles de plan à utiliser. Vous pouvez créer vos propres modèles de plan. Pour plus amples informations, référez-vous à Modèles de plan – Vue d'ensemble.

Echelle, Rotation du report, Origine N (Nord), Origine E (Est)

Utilisez ces champs pour définir la surface du projet à tracer.

Une fois que vous avez configuré une échelle, une "case de report" représentant la surface du projet à tracer, s'affiche dans la fenêtre graphique. Elle est basée sur les coordonnées d'origine. Changez l'échelle pour augmenter ou réduire la couverture du report.

ASTUCE: Configurez le champ Rotation du report sur une valeur pour l'orientation du report requise. La valeur dans ce champ configure l'orientation du côté de la feuille de report, plutôt que la rotation du point nord sur le report. La case de report peut se déplacer. Si c'est le cas, déplacez-la de façon graphique autour des données afin de s'adapter aux données du projet..

Cliquez sur le champ Origine nord ou Origine est, puis cliquez et faites glisser déplacer la 'case de report' afin de couvrir la surface requise. Vous pouvez configurer le champ Rotation de report sur une valeur que vous donne l'orientation de report requise. Lors de l'utilisation de ce champ vous spécifiez l'orientation du côté de la feuille de report plutôt que la rotation du point nord sur le report.

OK

Cliquez sur OK pour confirmer votre configuration de feuille de report. Ces détails de configuration sont stockés et puis utilisés dans l'option Report/Imprimer.

Meilleure adaptation

Cliquez sur Meilleur adaptation pour que le système puisse calculer les valeurs optimales pour les champs Echelle et Origine afin de tracer toutes les données de projet affichées actuellement. La Rotation du report passe par défaut à 0° lorsque vous utilisez cette option.

Config. d'impression

Cliquez sur Config. d'impression pour ouvrir le dialogue Config. d'impression pour votre système. Utilisez ce dialogue pour sélectionner l'imprimante ou le traceur approprié, choisissez la taille de papier et la source, et sélectionnez l'orientation de la sortie du report sur papier (portrait ou paysage).


Modèles de plan - Vue d'ensemble

Un modèle de plan (*.frm) est stocké comme un fichier de texte. Il contient une série de formulations concernant le contenu du modèle de plan et son emplacement. Dans un modèle de plan vous pouvez comprendre des lignes, du texte et des symboles de définition (par exemple Point nord) et des métafichiers étendus (*wmf) de Microsoft Windows pour construite la mise en page requise.

Commentaires

Dans le modèles de plan fournis, il y a des commentaires pour expliquer la signification des différentes commandes disponibles dans la définition de modèle de plan. Utilisez un éditeur texte pour éditer une définition de modèle de plan. Les définitions de modèle plan fournies se trouvent en général dans le dossier Système sous le dossier programme principal (typiquement:C:\Fichiers programme\Trimble\Trimble Geomatics Office).

ASTUCE: Pendant l'installation du logiciel Trimble Geomatics Office, l'éditeur de texte Bloc-notes pour Microsoft est assigné à l'opération Ouvrir pour l'extension de fichier. Donc, lors de la sélection d'un fichier de modèle de plan, ou lorsque vous êtes dans le programme Explorateur de Microsoft Windows, si vous soulignez un fichier .frm, vous pouvez utiliser l'option Ouvrir dans le menu des raccourcis pour afficher les détails du modèle de plan dans Bloc-notes pour Microsoft.

Pour plus amples informations, référez-vous aux Détails de définition du modèle de plan.


Détails de définition du modèle de plan

 Vue d'ensemble

Les définitions de modèle de plan sont stockées sous un format similaire au format fichier *.ini standard de Windows. Ce format consiste en plusieurs en-têtes de section enfermées entre crochets ([ ]). Dans chaque section des éléments individuels sont définis, chaque élément ayant un nom unique et ayant une valeur appropriée assigné en utilisant un caractère '='. Les commentaires peuvent être placés dans un fichier mais ils doivent être sur leur propre ligne et la ligne doit commencer par un point-virgule (;).

Un système de coordonnées spécial dans les définitions de modèles de plan permet l'utilisation du même modèle de plan sur des tailles de feuilles de report différentes. Il est supposé que toutes les valeurs de coordonnées sont en millimètres et qu'elles ont toujours spécifiées en paires séparées par virgule entre parenthèses, avec la valeur x (ou horizontale) premier et la valeur y (ou verticale) deuxième.

Par exemple (20,40) est une position de coordonnées valide. Le coin traçable de la feuille en haut à gauche est défini avec les coordonnées +0.0, +0.0. Toutes les valeurs de coordonnées positives sont interprétées comme étant à droite et au-dessous de cette position. Le coin traçable de la feuille en bas à droite est défini avec les coordonnées –0.0, –0.0. Toutes les valeurs de coordonnées négatives sont interprétées comme étant à gauche et au-dessus de cette position.

NOTE: Le coins traçables d'une feuille sont les limites du traçage (ou d'impression) pouvant être effectué sur une feuille spécifique par le traceur ou l'imprimante. Ces limites seront diverses selon le traceur ou l'imprimante. Les limites sont influencées par des facteurs tels que la façon dans laquelle la feuille est tenue entre les rouleaux d'un traceur, par exemple.

Comme une illustration, la table ci-dessous montre les diverses positions de coordonnées et leurs interprétations.

Sections

Les sections dans un fichier de définition du modèle de plan sont les suivants:


Commande Report / Imprimer

Utilisez cette commande pour:

  Afficher une liste des traceurs et des imprimantes installés.

  Configurer le traceur ou l'imprimante par défaut.

  Accéder aux autres options de traçage ou d'impression pour le traceur ou l'imprimante sélectionné.

Pour l'accéder, effectuez l'une des choses suivantes:

  Sélectionnez Fichier / Report/Imprimer.

  Appuyez sur Ctrl+P.

Avant que vous pouvez imprimer dans  Trimble Geomatics Office, vous devez:

  Connecter le traceur ou l'imprimante à votre ordinateur ou un réseau.

  Installer un pilote de traceur ou d'imprimante en utilisant le programme d'installation de Microsoft Windows ou le Panneau de configuration de Microsoft Windows, lorsqu'il n'y en a pas un traceur/imprimante présent déjà. Pour plus amples informations, référez-vous à l'Aide en ligne Microsoft Windows.

  Dans le dialogue Report/Imprimer, sélectionnez le traceur ou l'imprimante que vous voulez utiliser.

NOTE: Si vous avez des lignes de grille affichées dans la fenêtre de graphiques, elles sont tracées sur votre report.


Commande Propriétés du projet

Utilisez cette commande pour ouvrir le dialogue Propriétés du projet . Pour l'accéder, sélectionnez Fichier / Propriétés du projet.


Commande Dernier projet

Le menu Fichier liste les derniers quatre projets que vous avez ouvert récemment.

Pour ouvrir un projet utilisé récemment, sélectionnez-en un du menu Fichier.


Commande Quitter

Utilisez cette commande pour fermer le projet courant et quitter  Trimble Geomatics Office.

 Trimble Geomatics Office enregistre vos modifications automatiquement.


Commande Copier projet

Utilisez cette commande pour faire une copie d'un projet Trimble Geomatics Office existant. Cette commande n'est disponible lorsqu'il n'y a aucun projet ouvert.

Le dialogue Projet à copier s'affiche.


Dialogue Projet à copier

Utilisez ce dialogue pour sélectionner le projet que vous voulez copier.

Pour accéder au dialogue:

  Sélectionnez Fichier / Copier projet.

Cette commande n'est disponible lorsqu'il n'y a aucun projet ouvert.

Sélectionnez un projet de la liste.

Les détails du projet sélectionné s'affichent au-dessous de la liste de projets.

Dossier

Pour sélectionner un projet d'un dossier différent, cliquez sur Dossier.

Lorsque vous avez sélectionné le projet à copier, cliquez sur OK.

Le dialogue Copier projet vers s'affiche.


Dialogue Copier projet vers

Utilisez ce dialogue pour saisir le nouveau nom du projet à copier.

Titre du projet

Saisissez le nom du projet.

Dossier

Effectuez l'une des choses suivantes:

  Cliquez sur le bouton d'ellipse pour copier le projet vers un dossier différent.

  Saisissez le chemin du dossier à l'aide du clavier.


Site FTP (protocole de transfert de fichier)

Utilisez le site FTP de Trimble pour envoyer des fichiers ou pour recevoir des fichiers tels que les corrections de logiciel, les utilitaires et les réponses du Forum aux questions (FAQ). Vous pouvez accéder au site FTP à partir du site World Wide Web de Trimble: www.trimble.com/support/support.htm.


Formats SIG - Vue d'ensemble

Plus amples informations

Comment ...	Voir aussi
Vue d'ensemble

Vous pouvez exporter des points et leurs attributs associés depuis  Trimble Geomatics Office vers une variété de formats du Système d'information géographique (SIG). Les lignes, les arcs, les courbes, le texte et les annotations ne sont pas exportés.

Référez-vous à la table suivante:

Pendant l'exportation, vous pouvez créer des nouveaux attributs à exporter (par exemple, l'heure ou la date à laquelle l'attribut a été collecté, ou l'élévation du point).

Vous pouvez également créer des formats de rapport ou d'exportation ASCII personnalisés pour exporter les points et leurs attributs.

Utilisez la tabulation SIG pour l'exportation des données à une variété de formats SIG.

Pour accéder à cette tabulation:

1.	Effectuez l'une des choses suivantes:

  Sélectionner Fichier / Exporter.

  Utilisez le groupe Exporter sur la Barre du projet.

Le dialogue Exporter d'affiche.

2.	Sélectionnez la tabulation SIG.

Vous pouvez déplacer des formats personnalisés à cette tabulation. Pour plus amples informations, référez-vous au fichier Asciiexp.dat.

Pour des informations à propos de l'exportation vers un format SIG, référez-vous aux Formats d'exportation SIG.

Cette tabulation a les formats suivants:


Option de fichiers ArcView (*.dbf,*.shp,*shx)

Lorsque vous exportez au format ArcView, trois fichiers sont créés pour chaque code de caractéristiques avec attributs.

La table suivante décrit ces fichiers:

Ces fichiers sont des fichiers binaires. Vous ne pouvez pas afficher ou éditer ces fichiers avec un éditeur de texte.

 Trimble Geomatics Office donne un nom automatiquement à ces fichiers.

Par exemple, si vous avez des points avec attributs pour le code de caractéristiques 'Arbre", les fichiers suivants sont créés:

  Arbre.shp

  Arbre.shx

  Arbre.dbf

Pour les informations à propos des paramètres ArcView, référez-vous à la rubrique dialogue Options d'exportation. ArcView n'a pas des paramètres supplémentaires autre que ceux de tous les formats.

NOTE: Seule la version ArcView 3.1 ou ultérieure prend en charge les points 3D.


Option de fichier AutoCAD avec attributs (*.dxf)

Lorsque vous exportez à l'option AutoCad avec attributs (*.dxf), un fichier .dxf est créé.

Utilisez les fichiers Drawing eXchange Format (.dxf) pour échanger et afficher les données tabulaires et graphiques.

Configurer

En outre des paramètres standard décrits sous la rubrique dialogue Options d'exportation, il y a également des paramètres AutoCAD spécifiques.

Cliquez sur Options. le dialogue Options d'exportation s'affiche.

Taille de texte

Il y a deux options pour la spécification de la taille de texte dans le fichier de sortie:

Automatique	La taille de texte est calculée automatiquement du fichier d'entrée.
Fixe	La taille de texte est fixe. Effectuez l'une des choses suivantes:   Dans le champ Taille de texte, saisissez la taille requise pour le texte.   Utilisez les boutons directionnels.
Options de bloc

Sélectionnez la case à cocher Exporter les caractéristiques en blocs pour exporter les informations d'attributs en blocs. Ce signifie que les attributs peuvent être interrogés et édités lors de la lecture dans un système SIG ou CAD que prend en charge le format .dxf. Le groupe Point d'insertion du bloc devient disponible. Il existe deux options pour mettre un point d'insertion dans un bloc:

  Origine	Mettez le point d'insertion à la position 0,0 dans le bloc.

  Centroïde	Mettez le point d'insertion dans le centroïde calculé du bloc.

Si la case à cocher Exporter les caractéristiques en blocs n'est pas sélectionnée ces options ne sont pas disponibles. Plutôt, les informations de position d'une caractéristique sont stockées sous forme d'une entité POLYLIGNE ou POINT et les informations d'attribut sont écrites sous forme de TEXTE. TEXTE est écrit à la couche appropriée pour chaque caractéristique. Le groupe Point d'insertion du bloc n'est pas disponible.

NOTE: Si vous n'exportez pas en blocs, les noms d'attribut ne sont pas exportés—seulement les valeurs des attributs. Si vous exportez à tout logiciel AutoCAD SIG, par exemple AutoCAD Map, toujours exportez en blocs afin que les noms d'attribut sont compris.

Options de couche

Les fichiers .dxf organisent des informations différentes en couches. Il est possible de manipuler ces couches indépendamment l'une des autres. Il y a deux options de couche:

  Une caractéristique par couche - Une couche différente est créée pour chaque caractéristique. Par exemple, les caractéristiques de poteau sont mises dans une couche différente à celle des caractéristiques d'arbre. Le nom de couche de chaque couche est le nom de la caractéristique.

  Toutes les caractéristiques dans une couche - Toutes les caractéristiques sont mises dans une seule couche. Le champ Nom de couche est disponible. Saisissez le nom de la couche dans le champ Nom de couche. Vous ne pouvez pas laisser le champ Nom de couche vide.

Changer les espaces aux soulignements dans les noms de caractéristiques

Certains programmes qui lisent les fichiers .dxf nécessitent que les noms de couche dans les fichiers .dxf n'ont pas des espaces.  Sélectionnez cette option pour convertir tout espace dans les noms de caractéristiques au caractère de soulignement (_).


Option de fichiers MapInfo  (*.mif,*.mid)

Lorsque vous exportez au format MapInfo, deux fichiers sont créés pour chaque code de caractéristiques ayant attributs. Ils sont discutés dans la table suivante:

Ces fichiers sont les fichiers texte. Vous pouvez les afficher ou les éditer avec un éditeur de texte. Un exemple de chacun de ces fichiers est illustré ci-dessous:

Fichier MIF

VERSION 2

  DELIMITEUR "	"

  SYSCOOR PROJECTION AU SOL 1, 104

COLONNES 3

  Hauteur_m  DECIMALES (5,0)

  Besoin_attention  CARACT(30)

  Remarques  CARACT(30)

DONNEES

POINT 700345.32 306565.45

POINT 700356.47 306125.38

POINT 700937.71 306505.41

Fichier MID

1	Oui	Branches entre les câbles

2	Oui	Morte

3	Non	Tout près du poteau

 Trimble Geomatics Office fournit un nom automatiquement à ces fichiers. Par exemple, si vous avez des points avec attributs pour le code de caractéristiques 'arbre'. les fichiers suivants sont créés:

  arbre.mif

  arbre.mid

Le programme MapInfo ne possède aucun paramètre supplémentaire à part ceux de tous les formats. Ils sont décrits dans la rubrique dialogue Options d'exportation .

NOTE: MapInfo ne prend en charge que les points 2D.


Option de fichier Microstation (*.dgn)

Lorsque vous exportez au format Microstation (*.dgn), un fichier est créé. Vous devez saisir le nom pour le fichier que vous voulez exporter.

Configurer

En plus des paramètres disponibles pour tous les formats dans le dialogue Options d'exportation, les options suivantes sont disponibles pour configurer la façon dont vous voulez exporter au format Microstation DGN.

Faire disponible pour sélection

Sélectionnez cette case à cocher pour assurer que le fichier .dgn créé a des caractéristiques 'disponible pour sélection'. Avec cette option dans MicroStation, lorsque vous cliquez près de la caractéristique, le curseur se déplace à sa position exacte.

Unité de résolution

Saisissez l'unité de résolution pour les positions GPS dans le fichier de sortie. Si vous voulez recouvrir vos données GPS avec des données existantes en format .dgn, il est important que les unités de résolution des deux fichiers sont les mêmes.

Les deux options sont:

  Automatique − Les unités de résolution sont calculées automatiquement du champ Décimales de coordonnées dans la tabulation Fichier / Propriétés du projet / Unités et format du projet Trimble Geomatics Office.

  Défini de l'utilisateur − Les unités de résolution peuvent être spécifiées manuellement dans ce champ.

Origine globale

Sélectionnez comment l'origine globale est calculée. Si vous voulez recouvrir vos données GPS avec des données existantes en format .dgn, il est important que l'origine globale des deux fichiers est la même.

Les deux options sont:

  Automatique − L'origine globale est calculée est calculée du fichier (ou des fichiers) d'entrée automatiquement.

  Défini de l'utilisateur − Les coordonnées X, Y et Z de l'origine globale peuvent être spécifiées manuellement dans ce champ.


Dialogue Options d'exportation (exportation SIG)

Utilisez ce dialogue pour éditer les options du format Système d'information géographique (SIG) auquel vous exportez.

Pour accéder à ce dialogue:

1.	Sélectionnez Fichier / Exporter. Le dialogue Exporter ouvre.

2.	Dans la tabulation SIG, cliquez sur Options.

NOTE: Les options disponibles dans ce dialogue dépendent du format d'exportation choisi. Ces paramètres sont décrits dans les rubriques de format appropriées:

  fichiers ArcView (*.dbf, *.shp, *.shx)

  fichier AutoCAD avec attributs (*.dxf)

  fichiers MapInfo (*.mif, *.mid)

  fichier Microstation (*.dgn)

A part des paramètres dépendant du format, les paramètres communs à tous les formats sont discutés ci-dessous.

Format du fichier

Il y a trois formats de fichier différents:

  le format DOS

Sélectionnez cette option si le système cible s'exécute sous MS-DOS ou contient du texte dans le jeu de caractères MS-DOS ASCII. Les fichiers texte dans ce format ont les terminateurs d'enregistrement <RetourChariot> <NouvelleLigne>.

  le format Windows

Sélectionnez cette option si le système cible s'exécute sous un système d'exploitation Microsoft Windows ou contient du texte dans le jeu de caractères ANSII. Les fichiers texte dans ce format ont les terminateurs d'enregistrement <RetourChariot> <NouvelleLigne>.

  le format UNIX

Sélectionnez cette option si le système cible s'exécute sous un système d'exploitation UNIX. Les fichiers texte dans ce format ont les terminateurs d'enregistrement <NouvelleLigne>.

NOTE: Seuls les formats de fichiers valides pour le format SIG sélectionné sont disponibles.

Dimensions

Si le format SIG prend en charge les points à trois dimensions (3D), vous pouvez exporter les points en deux dimensions (2D) (Nord, Est), ou les points à trois dimensions (Nord, Est, Elévation). Cette option n'est pas disponible lorsque le format SIG ne prend pas en charge les points 3D.

ASTUCE: Si le SIG ne prend pas en charge les points 3D, pour exporter des élévations cliquez sur Attributs et sélectionnez l'option Elévation de point.

Attributs

Cliquez sur Attributs pour sélectionner les attributs générés que vous voulez exporter avec les points. Le dialogue Attributs ouvre. Utilisez ce dialogue pour configurer comment vous voulez exporter les valeurs d'attribut de menu.

NOTE: Les options disponibles dans le dialogue Attributs sont les mêmes pour tous les formats SIG.


Dialogue Attributs  (Options d'exportation SIG)

Utilisez ce dialogue pour sélectionner toute autre valeur d'attribut que vous voulez avoir comprise avec l'exportation du fichier par  Trimble Geomatics Office. Les attributs sont créés utilisant des valeurs appropriées stockées avec les points dans la base de données. Dans ce dialogue, vous ne pouvez pas créer vos propres attributs à ajouter.

Les options disponibles dans le dialogue Attributs sont les mêmes pour tous les formats du Système d'information géographique (SIG).

Attributs générés

Sélectionnez toute combinaison de ces options à ajouter pour chaque point exporté. Ce sert lorsque vous exportez à un SIG 2D et vous voulez exporter les élévations. Dans  SIG auquel vous exportez, tout attribut généré apparaît comme un attribut normal.

NOTE: Les attributs de date et heure générés sont exportés dans le format utilisé le plus récemment dans la bibliothèque d'attributs et caractéristiques.

Exporter attributs de menu sous

Lorsqu'un fichier de bibliothèque d'attributs et caractéristiques (*.fcl) ou un fichier de dictionnaire d'attributs (*.ddf) est défini, vous pouvez spécifier jusqu'à deux valeurs de code pour chaque élément d'attribut de menu.

Par exemple, considérez un code de caractéristiques 'Arbre' avec un attribut de menu 'Type'. Chaque élément de menu peut avoir des valeurs de code. Le type 'Indigène' peut avoir des valeurs de code IV et IT. Les points ayant un code de caractéristiques Arbre avec des attributs collectés peuvent avoir leurs attributs de type exportés comme la valeur d'attribut actuelle (par exemple, Indigène), 'Code d'utilisateur 1' (par exemple, IV), ou 'Code d'utilisateur 2' (par exemple, IT). Ce sert lorsque votre SIG ne prend en charge que les valeurs d'attribut à deux caractères.

Pour plus amples informations concernant les valeurs de code, référez-vous à l'aide de l'utilitaire Trimble Feature and Attribute Editor.


Commande Rubriques d'aide

Utilisez la commande Rubriques d'aide pour ouvrir l'Aide en ligne.


Commande Astuce du jour

Le système Astuce du jour fournit des astuces pour vous assister avec l'utilisation de Trimble Geomatics Office.

Sélectionnez cette commande pour afficher l'astuce du jour suivant.

Pour afficher des astuces quand vous démarrez Trimble Geomatics Office, sélectionnez la case à cocher Afficher les astuces au démarrage.


Dialogue Astuce du jour

Le système Astuce du jour fournit des astuces pour vous assister avec l'utilisation Trimble Geomatics Office.

Pour afficher des astuces quand vous démarrez Trimble Geomatics Office, sélectionnez la case à cocher Afficher les astuces au démarrage.


Commande A propos de Trimble Geomatics Office

 Comment ...

Utilisez cette commande pour afficher les informations concernant le programme et le copyright ainsi que le numéro de version de Trimble Geomatics Office.


Dialogue A propos de Trimble Geomatics Office

Ce dialogue affiche les informations concernant le programme et le copyright ainsi que le numéro de version de Trimble Geomatics Office.


Utilisation de l'Aide

Il y a plusieurs méthodes pour obtenir de l'aide lorsque vous utilisez Trimble Geomatics Office.

Pour obtenir de l'Aide dans un dialogue	V érifiez que le dialogue soit actif et appuyez sur F1.
Pour obtenir de l'Aide sur une commande de menu	1. Placez votre curseur sur la commande de menu et enfoncez le bouton gauche de la souris. 2. Appuyez sur F1.
Pour rechercher une rubrique dans le système d'Aide en utilisant le Contenu	Sélectionnez la tabulation Contenu pour rechercher les rubriques par catégorie, comme un livre.
Pour rechercher une rubrique dans le système d'Aide en utilisant l'Index	Sélectionnez la tabulation Index pour afficher une liste d'entrées de l'index. Entrez le mot que vous recherchez ou défilez à travers la liste.
Pour rechercher un terme ou une phrase dans le système d'Aide	1. Sélectionnez la tabulation Rechercher. 2. L'Assistant création de recherche peut s'afficher. Si c'est le cas, sélectionnez l'option Minimise la taille de la base de données et cliquez sur Suivant >, puis cliquez sur Terminer. 3. Dans le champ Taper le mot(s) que vous voulez rechercher, saisissez le mot ou la phrase que vous voulez recherchez. 4. Si nécessaire, dans le champ Sélectionnez des mots correspondants pour approfondir votre recherche, sélectionnez un mot pour approfondir la recherche. 5. Dans le champ Choisir la rubrique à afficher, sélectionnez une rubrique. Elle sera affichée au côté droit de la fenêtre d'Aide. ASTUCE: Pour reconstruire l'Index, sélectionnez Rechercher / Options.
Pour imprimer une rubrique d'Aide	Pour imprimer la rubrique d'Aide courante, effectuez l'une des choses suivantes:   Cliquez sur Impression.  Du menu Fichier choisissez Imprimer la rubrique.  Utilisez le bouton droit de votre souris pour cliquer à l'intérieur de la fenêtre d'Aide, et puis sélectionnez Imprimer la rubrique.       NOTE: Vous pouvez imprimer des rubriques d'Aide individuelles mais pas des livres d'Aide.  Pour imprimer d'informations dans une fenêtre automatique, utilisez le bouton droit de la souris pour cliquez à l'intérieur, puis cliquez sur Imprimer la rubrique.

ASTUCE: Cliquez sur Précédent sur la barre d'outils Aide pour afficher les rubriques que vous avez déjà affiché.


Rapporter les enregistrements de notes

Vue d'ensemble et plus amples informations
Pour rapporter les enregistrements de notes de la base de données du projet Trimble Geomatics Office, créez un rapport de base de données personnalisé. Les rapports personnalisés travaille sur les types d'enregistrement primaires ou les Interrogations spécifiés dans la base de données.

Il y a deux étapes dans la création d'un rapport personnalisé:

  La définition d'un groupe de sélection

  La création d'un format du rapport


Définition du groupe de sélection

Pour sélectionner des types 'enregistrement non graphique (par exemple, les enregistrements de note):

1.	Effectuez l'une des choses suivantes:

  Utilisez le groupe de sélection de fichier .dc.

 	  Choisissez Sélectionner / Par interrogation.

2.	Sélectionnez les types d'enregistrement.

NOTE: Si vous sélectionnez l'option BD entière, le rapport comprend toutes les notes dans la base de données.


Création d'un format du rapport

Pour ce faire:

1.	Sélectionnez Rapports / Rapports supplémentaires. Le dialogue Rapports supplémentaires s'affiche. La liste Rapports affiche une liste des rapports actuellement disponibles.

2.	Cliquez sur Nouveau pour créer un nouveau format de rapport. Le dialogue Définir le format du rapport s'affiche.

3.	Dans le champ Nom, saisissez un nom pour le rapport.

4.	Dans le champ Rapport sur, sélectionnez Enregistrements de note.

5.	Dans le champ Corps de format, effectuez l'une des choses suivantes:

  Entrée [Note].

  Utilisez le menu de raccourcis et sélectionnez Note.

L'afficheur HTML par défaut ouvre et affiche le rapport. Vous pouvez imprimer ou enregistrer le rapport.


Rapports d'enregistrements de note (collectées sur le terrain) avec des informations de point

Les enregistrements de note ne sont pas liés directement aux points dans  Trimble Geomatics Office. Donc, vous ne pouvez pas exporter les enregistrements de note avec des informations de point directement (par exemple N, E, notes) utilisant le procédé décrit ci-dessus. Cependant, vous pouvez exporter des notes au moyen du champ de codes de caractéristiques.

Pour ce faire:

1.	Utilisant l'utilitaire DC Editor, dans le fichier .dc, recherchez les enregistrements de vecteur GPS (GPS VEC) qui sont suivis d'une note.

2.	Dans ces champs d'Enregistrement de vecteur GPS, allez au champ CODE et entrez un tiret après le code courant, suivi des informations de Note. Par exemple, ARBRE – Pin

Lors de l'importation de l'enregistrement, l'enregistrement de note est ajouté à la fin du champ de codes de caractéristiques. La note est maintenant associé avec un point et vous pouvez rapporter:

  Point, Nord, Est, Code de caractéristiques

NOTE: Quelques fois le champ Enregistrement de vecteur GPS est suivi d'un cachet de date et puis l'enregistrement de note. En ce cas le cachet de date est associé au code de caractéristiques au lieu du point. Vous devrez supprimer l'enregistrement de cachet de date.


Créer un rapport de points personnalisé

Vue d'ensemble et plus amples informations
Pour rapporter les enregistrements de point de la base de données Trimble Geomatics Office, créez un rapport de base de données personnalisé. Les rapports personnalisés travaille sur les types d'enregistrement primaires ou les Interrogations spécifiés dans la base de données.


Création d'un rapport de base de données personnalisé

1.	Sélectionnez Rapports / Rapports supplémentaires. Le dialogue Rapports supplémentaires s'affiche. La liste Rapports affiche une liste des rapports disponibles actuellement.

2.	Cliquez sur Nouveau pour créer un nouveau format de rapport. Le dialogue Définir le format du rapport s'affiche.

3.	Dans le champ Nom, saisissez un nom pour le rapport. Par défaut, l'élément Enregistrements de point s'affiche dans le champ Rapport sur.

4.	Cliquez sur Spécifier format automatiquement et  Trimble Geomatics Office complétera les champs En-tête de format et Corps de format. Vous pouvez éditer ces formats.

ASTUCE: Cliquez avec le bouton droit de la souris dans les champs En-tête de format et Corps de format, sélectionnez Champs, et sélectionnez les codes de champ disponibles de la liste.

6.	Cliquez sur OK.

L'afficheur HTML par défaut ouvre et affiche le rapport. Vous pouvez imprimer ou enregistrer le rapport.


Changer la zone dans la projection du projet

 Vue d'ensemble et plus amples informations

NOTE: La zone doit se trouver dans la base de données du système de coordonnées.

1.	Sélectionnez Fichier / Propriétés du projet. Le dialogue Propriétés du projet s'affiche.

2.	Sélectionnez la tabulation Système de coordonnées.

3.	Dans le groupe Paramètres du système de coordonnées, cliquez sur Changer. Le dialogue Sélectionner système de coordonnées s'affiche.

4.	Sélectionnez l'option Nouveau système et puis cliquez sur Suivant. Le dialogue Sélectionner le type de système de coordonnées s'affiche.

5.	Sélectionnez l'option Système de coordonnées et zone et puis cliquez sur Suivant.

6.	Dans le groupe Zone du système de coordonnées, soulignez le système de coordonnées contenant la zone requise.

7.	Dans le groupe de zone, soulignez la zone requise et puis cliquez sur Suivant. Le dialogue Sélectionner le modèle du géoïde s'affiche.

8.	Effectuez l'une des choses suivantes:

  Sélectionnez le modèle du géoïde requis pour le projet.

  Cliquez sur Pas de modèle géoïde.

9.	Cliquez sur Finir.

Le système de coordonnées est appliqué à la base de données. Tout point existant dans la base de données est converti au nouveau système de coordonnées.

Lorsque vous avez des positions de points ou que vous saisissez des coordonnées avec des élévations nulles et une élévation de projet n'est pas définie, la sélection du système de coordonnées échoue. C'est à cause du fait que  Trimble Geomatics Office ne peut pas convertir les points existants d'un système à l'autre sans une élévation approximative.


Exporter une carte d'arrière-plan au logiciel Trimble Survey Controller

 Vue d'ensemble et plus amples informations

1.	Effectuez l'une des choses suivantes:

  Sélectionnez Fichier / Exporter.

  Cliquez sur l'outil Exporter.

Le dialogue Exporter s'affiche.

2.	Dans la tabulation Levé, sélectionnez l'option Bibliothèque de caractéristiques et attributs au Survey Controller (*.fcl). Le dialogue Sélectionner bibliothèque d'attributs et caractéristiques s'affiche.

3.	Cliquez sur Parcourir pour trouver le fichier .fcl que vous voulez exporter.

4.	Sélectionnez la case à cocher Exporter attributs pour exporter les définitions d'attribut contenues dans la bibliothèque d'attributs et caractéristiques. Cette option fait les options disponibles dans  Trimble Survey Controller. Si la case à cocher est vide, tout attribut défini dans le fichier .fcl n'est pas exporté.

NOTE: Les codes de contrôles sont exportés aussi.

5.	Cliquez sur OK. Le dialogue Transfert de données enregistrer sous s'affiche.

6.	Vérifiez que votre carnet de terrain (exécutant  Trimble Survey Controller) est connecté à votre ordinateur.

7.	Sélectionnez un périphérique Survey Controller et cliquez sur Ouvrir. Le dialogue Connexion s'affiche.

Lorsque la connexion est effectuée, le dialogue Enregistrer s'affiche.

8.	Effectuez l'une des choses suivantes:

  Sélectionnez l'option Mémoire principale (C:) pour stocker la bibliothèque dans la mémoire principale du logiciel Trimble Survey Controller.

  Sélectionnez l'option Carte PC (A:) lorsque vous avez une carte PC insérée et vous voudriez stocker la bibliothèque sur la carte PC.

NOTE: Si votre périphérique n'a pas une carte PC, ce dialogue n'est pas disponible.

9.	Cliquez sur Ouvrir. L'emplacement que vous avez sélectionné est lu. Tous les fichiers de bibliothèque d'attributs and caractéristiques dans le répertoire sont affichés. Le nom interne de la bibliothèque d'attributs et caractéristiques s'affiche dans le champ Nom du fichier.

10.	Editez le champ Nom du fichier pour changer le nom du fichier dans Trimble Survey Controller si nécessaire. Le nom du fichier d'origine sur votre ordinateur n'est pas changé.

11.	Cliquez sur Enregistrer. Le dialogue Téléchargement s'affiche. La barre de progression affiche le progrès du transfert.

Lorsque le transfert est terminé, le dialogue Téléchargement disparaît et la connexion au carnet de terrain est interrompue.

La bibliothèque a été transférée.


Exporter à un format SIG

Vue d'ensemble et plus amples informations
1.	Effectuez l'une des choses suivantes:

  Sélectionnez Fichier / Exporter.

  Cliquez sur l'outil Exporter sur la barre d'outils standard.

Le dialogue Exporter s'affiche.

2.	Dans la tabulation SIG, sélectionnez le format requis.

3.	Pour changer les options de configuration des formats d'exportation, cliquez sur Options. Ce vous permet de configurer des options appropriées au format SIG auquel vous exportez.

4.	Cliquez sur OK. Le dialogue Nom du fichier s'affiche.

5.	Spécifiez un nom pour le nouveau fichier et cliquez sur Enregistrer pour créer le nouveau fichier (ou fichiers).


Créer un Fichier de grille de datum combiné (.cdg) pour transférer au Survey Controller

 Vue d'ensemble et plus amples informations

NOTE: Pour une nouvelle grille de datum combiné vous ne pouvez utiliser que le fichier de grille de datum utilisé actuellement dans le projet. Cependant, vous pouvez exporter toute grille de datum combiné. Vous ne pouvez transférer un fichier de grille de datum combiné qu'au logiciel Trimble Survey Controller de version 7.00 ou ultérieure.

1.	Sélectionnez Fichier / Exporter. Le dialogue Exporter s'affiche.

2.	Dans la tabulation Levé, sélectionnez l'option Grille de datum vers Survey Controller (*.cdg) et cliquez sur OK. Le dialogue Exporter une grille de datum s'affiche.

3.	Sélectionnez l'option Créer un fichier de grille de datum combiné pour l'exportation vers le Survey Controller. Le dialogue Créer fichier de grille de datum combiné s'affiche.

La boîte jaune sur le globe définit la zone couverte par les fichiers de grille de latitude et de longitude. Ils sont spécifiés au côté droit du dialogue Créer grille de datum combiné.

4.	Pour réduire la taille du fichier, utilisez l'outil Sélectionner au côté gauche du dialogue Créer grille de datum combiné.  Faites glisser déplacer la boîte de sélection noire sur la surface à l'intérieur de la boîte jaune pour laquelle vous voulez créer une grille de datum combiné.

La taille du fichier indique la taille et les limites de latitude et longitude indiquent l'étendue du fichier de grille de datum combiné à créer.

5.	Le champ Nom du Survey Controller affiche le nom de la grille de datum combiné à sous-grilles utilisé dans l'étude Trimble Survey Controller. Ajoutez un suffixe ou un préfixe au nom si le Trimble Survey Controller déjà utilise une grille de datum du même nom. Le nom doit contenir le nom de la grille de datum d'origine.

NOTE: Lors de l'importation du fichier Survey Controller (*.dc),  Trimble Geomatics Office utilise le nom du Survey Controller pour vérifier si le nom du fichier .cdg utilisé dans le champ correspond à la grille de datum spécifiée pour le projet Trimble Geomatics Office. Si le nom ne comprend pas le nom de la grille de datum d'origine, la grille de datum ne sera pas reconnue.

6.	Cliquez sur OK.

Le dialogue Nom du fichier s'affiche. Le nom du fichier par défaut est les premiers six caractères du nom de la grille de datum. Ce n'est pas le nom utilisé dans le Trimble Survey Controller – c'est le nom du fichier .cdg à stocker dans le répertoire Program Files\Common Files\Trimble\GeoData.

7.	Pour spécifier un autre nom pour le fichier, utilisez le clavier pour éditer le champ Nom du fichier et cliquez sur Enregistrer. Le dialogue Transfert de données Enregistrer sous s'affiche.

8.	Sélectionnez un périphérique Survey Controller et cliquez sur Ouvrir. Le dialogue Connexion s'affiche. Lorsque la connexion est effectuée, le dialogue Enregistrer s'affiche.

9.	Effectuez l'une des choses suivantes:

  Sélectionnez l'option Mémoire principale (C:) pour stocker la bibliothèque dans la mémoire principale du logiciel Trimble Survey Controller.

  Sélectionnez l'option Carte PC (A:) lorsque vous avez une carte PC insérée et vous voudriez stocker la bibliothèque sur la carte PC.

NOTE: Si votre périphérique n'a pas une carte PC, ce dialogue n'est pas disponible.

10.	Cliquez sur Ouvrir.

le répertoire sélectionné est lu. Tous les fichiers de grille de datum combiné dans le répertoire sont affichés. Le champ Nom du fichier n'est pas disponible. Il affiche le nom de la grille de datum combiné qui apparaît dans Trimble Survey Controller.

11.	Cliquez sur Enregistrer. Le dialogue Téléchargement s'affiche. La barre de progression affiche le progrès du transfert. Lorsque le transfert est terminé, le dialogue Téléchargement disparaît et la connexion au périphérique est interrompue.

Maintenant vous pouvez utiliser le fichier dans Trimble Survey Controller.


Faire une sous-grille et transférer un Fichier de grille de géoïde (.ggf) au Trimble Survey Controller

 Vue d'ensemble et plus amples informations

NOTE: Vous ne pouvez transférer un Fichier de grille de géoïde qu'au Trimble Survey Controller de version 6.00 ou ultérieur.

1.	Sélectionnez Fichier / Exporter. Le dialogue Exporter s'affiche.

2.	Dans la tabulation Levé, sélectionnez l'option Fichier de géoïde vers Survey Controller (*.ggf) et cliquez sur OK. 	Le dialogue Exporter un modèle de géoïde s'affiche.

3.	Pour créer un nouveau Fichier de grille de géoïde, sélectionnez l'option Créer une sous-grille d'un modèle de géoïde pour l'exportation au Survey Controller et cliquez sur OK. Le dialogue Créer fichier de grille de géoïde s'affiche.

NOTE: Choisissez l'option Exporter le modèle de géoïde existant au Survey Controller pour exporter un fichier de ce type.

La boîte jaune sur le globe définit la zone couverte par le modèle de géoïde. Le modèle de géoïde et le fichier de grille de géoïde sont spécifiés au côté droit du dialogue Créer grille de géoïde. Si le modèle de géoïde couvre le monde la boîte jaune n'est pas visible.

4.	Pour réduire la taille du fichier, utilisez l'outil Sélectionner au côté gauche du dialogue Créer grille de géoïde.  Faites glisser-déplacer la boîte de sélection noire sur la surface à l'intérieur de la boîte jaune pour laquelle vous voulez créer une grille de géoïde.

La taille du fichier indique la taille et les limites de latitude et longitude indiquent l'étendue du fichier de grille de géoïde à créer

5.	Le champ Nom affiche le nom de la grille de géoïde à sous-grilles utilisé dans l'étude Trimble Survey Controller. Ajoutez un suffixe ou un préfixe au nom si le Survey Controller utilise déjà un modèle de géoïde du même nom. Le nom doit contenir le nom du modèle de géoïde.

NOTE: Lors de l'importation du fichier Survey Controller (*.dc),  Trimble Geomatics Office utilise le nom du Survey Controller pour vérifier si le nom du fichier .ggf utilisé dans le champ correspond au modèle de géoïde spécifié pour le projet Trimble Geomatics Office. Vérifiez que le nom du fichier comprend pas le nom du modèle de géoïde d'origine, ou le modèle de géoïde ne sera pas reconnu.

6.	Cliquez sur OK.

Le dialogue Nom du fichier s'affiche. Le nom du fichier par défaut est les premiers six caractères du nom du modèle de géoïde. Ce n'est pas le nom utilisé dans le Trimble Survey Controller – c'est le nom du fichier de grille de géoïde à stocker dans le répertoire Program Files\Common Files\Trimble\GeoData.

7.	Pour spécifier un autre nom pour le fichier, utilisez le clavier pour éditer le champ Nom du fichier et cliquez sur Enregistrer. Le dialogue Transfert de données Enregistrer sous s'affiche.

8.	Sélectionnez un périphérique Survey Controller et cliquez sur Ouvrir. Le dialogue Connexion s'affiche. Lorsque la connexion est effectuée, le dialogue Enregistrer s'affiche.

9.	Effectuez l'une des choses suivantes:

  Sélectionnez l'option Mémoire principale (C:) pour stocker la bibliothèque dans la mémoire principale du logiciel Trimble Survey Controller.

  Sélectionnez l'option Carte PC (A:) lorsque vous avez une carte PC insérée et vous voudriez stocker la bibliothèque sur la carte PC.

NOTE: Si votre périphérique n'a pas une carte PC, ce dialogue n'est pas disponible.

10.	Cliquez sur Ouvrir.

11.	Cliquez sur Enregistrer. Le dialogue Envoi s'affiche. La barre de progression affiche le progrès du transfert. Lorsque le transfert est terminé, le dialogue Envoi disparaît et la connexion au périphérique est interrompue.

Maintenant vous pouvez utiliser le fichier dans Trimble Survey Controller

NOTE: Vous pouvez également utiliser l'utilitaire Trimble Grid Factory™ pour faire des sous-grilles de fichiers de modèle de géoïde. L'utilitaire Grid Factory fait partie de l'option personnalisée de Trimble Geomatics Office lors de l'installation du logiciel.

NOTE: Si vous transférez les données vers ou depuis  Trimble Survey Controller de version 7.60, utilisez le format de fichier .dc de la version 7.50.


Faire une sous-grille et transférer un Fichier de grille de géoïde (.ggf) au logiciel TDS Survey Pro CE

 Vue d'ensemble et plus amples informations

NOTE: Vous ne pouvez transférer un Fichier de grille de géoïde qu'au logiciel TDS Survey Pro CE de version 4.00 ou ultérieur.

1.	Sélectionnez Fichier / Exporter. Le dialogue Exporter s'affiche.

2.	Dans la tabulation Levé, sélectionnez l'option Fichier de géoïde vers TDS Survey Pro CE (*.ggf) et cliquez sur OK. 	Le dialogue Exporter un modèle de géoïde s'affiche.

3.	Pour créer un nouveau Fichier de grille de géoïde, sélectionnez l'option Créer une sous-grille d'un modèle de géoïde pour l'exportation au Survey Controller et cliquez sur OK. Le dialogue Créer fichier de grille de géoïde s'affiche.

NOTE: Choisissez l'option Exporter le modèle de géoïde existant pour exporter un fichier existant de ce type.

La boîte jaune sur le globe définit la zone couverte par le modèle de géoïde. Le modèle de géoïde et le fichier de grille de géoïde sont spécifiés au côté droit du dialogue Créer grille de géoïde. Si le modèle de géoïde couvre le monde la boîte jaune n'est pas visible.

4.	Pour réduire la taille du fichier, utilisez l'outil Sélectionner au côté gauche du dialogue Créer grille de géoïde.  Faites glisser-déplacer la boîte de sélection noire sur la surface à l'intérieur de la boîte jaune pour laquelle vous voulez créer une grille de géoïde.

La taille du fichier indique la taille et les limites de latitude et longitude indiquent l'étendue du fichier de grille de géoïde à créer.

Le champ Nom affiche le nom du modèle de géoïde à sous-grille utilisé.

NOTE: Lors de l'importation du fichier, Trimble Geomatics Office utilise le champ Nom pour vérifier si le nom du fichier .ggf utilisé dans le champ correspond au modèle de géoïde défini pour le projet Trimble Geomatics Office. Vérifiez que le nom du fichier comprend pas le nom du modèle de géoïde d'origine, ou le modèle de géoïde ne sera pas reconnu

6.	Cliquez sur OK.

Le dialogue Enregistrer sous s’affiche. le nom du fichier par défaut est les premiers huit caractères du nom de modèle de géoïde. Il est le nom du fichier de grille de gIl est le nom du fichier de grille de géoïde qui sera stocké dans le répertoire Program Files\Common Files\Trimble\GeoData.

7.	Pour spécifier un nom alternatif pour le fichier, utilisez le clavier pour modifier le champ Nom du fichier et cliquez sur Enregistrer. Le dialogue Enregister  sous de Data Transfer  s’affiche.

8.   Sélectionnez le périphérique TDS Survey Pro CE et cliquez sur Ouvrir. Le dialogue Connexion s’affiche. Lorsque la connexion est effectuée, le dialogue Enregistrer s’affiche.

9.	Transférez le fichier au périphérique TDS Survey Pro CE.

Maintenant vous pouvez utiliser le fichier dans le logiciel TDS Survey Pro CE.

NOTE: Vous pouvez également utiliser l'utilitaire Grid Factory pour faire des sous-grilles de fichiers de modèle de géoïde. L'utilitaire Grid Factory fait partie de l'option personnalisée de Trimble Geomatics Office lors de l'installation du logiciel.


Utiliser les données du logiciel TRIMMAP

Vous pouvez importer les données des études créées avec  TRIMMAP au logiciel Trimble Geomatics Office.

1.	Dans  TRIMMAP, ouvrez l'étude que vous voulez traiter avec  Trimble Geomatics Office.

2.	Sélectionnez Outils / Importer/Exporter / Interfaces graphiques. Le menu Interfaces graphiques s'affiche.

3.	Sélectionnez Interfaces graphiques / Créer fichier.

4.	Saisissez un nom d'étude et un Nom du fichier de sortie.

NOTE: Dans le champ Nom du fichier de sortie, assurez-vous de spécifier le nom et l'emplacement du fichier à créer et fourni l'extension .tgf. Par exemple C:\Temp\Topo_pts.tgf

5.	Dans  Trimble Geomatics Office, sélectionnez Fichier / Importer. Le dialogue Importer s'affiche.

6.	Dans la tabulation CAD / ASCII, sélectionnez l'option Fichier de graphiques TRIMMAP et cliquez sur OK. Le dialogue Sélectionner fichier(s) s'affiche.

7.	Naviguez au dossier contenant le fichier .tgf et cliquez sur Ouvrir.

Le fichier est importé au logiciel Trimble Geomatics Office.


Créer une nouvelle interrogation

AVERTISSEMENT: Trimble Navigation Limited soutient l'utilisation ou la modification de cette base de données seulement en utilisant  Trimble Geomatics Office de marque déposée de Trimble. Trimble ne soutient ni autorise l'utilisation ou la modification, ni garantit le fonctionnement, de cette base de données en utilisant d'autre applications informatiques. Trimble se réserve le droit de modifier le format de cette base de données à tout moment le faisant incompatible avec d'autres applications informatiques.

 Vue d'ensemble et plus amples informations

NOTE: Pour définir des interrogations personnalisées, il vous faut  Microsoft Access 2000 (ou plus récent).

Cet exemple crée une interrogation que sélectionne tous les points de la base de données avec des valeurs d'élévation de zéro.

1.	Dans Microsoft Access, ouvrez la base de données du modèle ( ou du projet)  du logiciel Trimble Geomatics Office à laquelle il faut ajouter l'interrogation. Elle se trouve dans le dossier du projet ou du modèle (.mdb).

2.	Sélectionnez la tabulation Interrogations dans la fenêtre de graphiques principale puis cliquez sur Nouveau pour créer une nouvelle interrogation.

ASTUCE: Choisissez l'option Affichage du projet au lieu de l'option Assistant Interrogation.

3.	Dans le dialogue Afficher la table, sélectionnez la table de Points car vous voulez sélectionner les entités de point et elles sont stockées dans cette table.

4.	Cliquez sur Ajouter avant de fermer le dialogue.

Un dialogue d'interrogation s'affiche.

5.	Configurez la première colonne dans la définition d'interrogation sur la table de Points entière. Vous pouvez le faire soit en faisant glisser déplacer l'identificateur * de la table de Points dans la partie supérieure du dialogue d'interrogation, soit en sélectionnant Point.*  de la case Champ.

NOTE: La case à cocher Afficher est sélectionnée par défaut. A moins que des entités multiples ne sont à sélectionner par interrogation, en général c'est meilleur de retourner la table entière. De cette façon, il est possible d'utiliser l'interrogation pour d'autres buts (tel qu'un rapport).

6.	Configurez le paramètre Champ dans la deuxième colonne sur Elévation (l'identificateur de champ d'élévation).

Saisissez =0 dans le champ Critères dans cette colonne. Ainsi la vérification pour les valeurs d'élévation zéro est configurée.

7.	Pour tester la fonctionnalité de cette interrogation, vérifiez l'affichage Feuille de données. Tout point ayant une hauteur de zéro est listée dans la table.

8.	Pour enregistrer la nouvelle interrogation, fermez-la et nommez-la avec un préfixe "Sélectionner_". Par exemple, donnez-la le nom "Sélectionner_points avec des élévations zéro".

NOTE: Pour ajouter une interrogation aux outils de sélection de la fenêtre Propriétés, utilisez le préfixe “TrierSS_” pour l'outil Trier, “SvyFiltreSS_” pour l'outil Filtrer (pour les données de levé) et “FiltresPlanSS_” pour l'outils Filtrer (pour les données CAD). Pour utiliser l'interrogation pour un rapport personnalisé, utilisez le préfixe “Rapport_”.

9.	Avant que la nouvelle interrogation devient disponible dans Sélectionner / Par interrogation (et ailleurs dans  Trimble Geomatics Office) le fichier schéma de base de données doit être mis à jour. Utilisez l'Explorateur de Microsoft Windows pour supprimer le fichier de schéma de base de données existant (.dbs) du projet (ou le modèle du projet).

NOTE: Lorsqu'un projet sans un fichier de schéma de base de données est ouvert,  Trimble Geomatics Office crée un nouveau fichier de schéma automatiquement basé sur la configuration du fichier de base de données. Ce nouveau fichier de schéma contient la référence à l'interrogation ajoutée récemment.  La nouvelle interrogation apparaît dans Sélectionner / Par interrogation. Vous pouvez l'utiliser pour personnaliser les définitions de rapport et d'exportation ASCII.

10.	Ouvrez  Trimble Geomatics Office, et ouvrez le modèle ou le projet auquel vous avez ajouté l'interrogation.

Maintenant, vous pouvez utiliser la nouvelle interrogation en choisissant Sélectionner / Par interrogation. Votre nouvelle interrogation apparaît dans la liste.


Importer un fichier Survey Controller (*.dc) du Trimble Survey Controller

 Vue d'ensemble et plus amples informations

1.	Sélectionnez Fichier / Importer. Le dialogue Importer s'affiche.

2.	Dans la tabulation Levé, sélectionnez l'option Périphériques de levé et cliquez sur OK. Le Transfert de données Ouvrir s'affiche.

3.	Sélectionnez un périphérique Survey Controller et cliquez sur Ouvrir. Le dialogue Connexion s'affiche.

Lorsque la connexion est effectuée, le champ Rechercher dans du dialogue Ouvrir change au périphérique Survey Controller.

4.	Effectuez l'une des choses suivantes:

  Sélectionnez l'option Mémoire principale (C:) si le fichier que vous importez se trouve dans la mémoire principale du logiciel Trimble Survey Controller.

  Sélectionnez l'option Carte PC (A:) si vous avez une carte PC insérée et vous voulez importer le fichier que trouve sur la carte PC.

NOTE: Si votre périphérique Trimble n'a pas une carte PC, ce dialogue n'est pas disponible.

5.	Cliquez sur Ouvrir. Le répertoire sélectionné est lu. Toutes les études Trimble Survey Controller dans le répertoire sont affichées.

6.	Dans le champ Format du fichier, sélectionnez la version du fichier de carnet de terrain que vous utilisez.

7.	Sélectionnez le(s) fichier(s) que vous voulez importer et cliquez sur Ouvrir. Le dialogue Téléchargement s'affiche et la barre de progression indique le progrès du transfert du fichier. Lorsque le transfert est terminé, le dialogue Téléchargement disparaît et la connexion au périphérique est interrompue.

Le fichier d'étude a été converti à un fichier .dc et il est importé dans la base de données du projet. Le fichier .dc se trouve dans le dossier Fichiers de données\Fichiers Trimble du projet.

NOTE: Lorsque vous transférez un fichier .dc au logiciel  Trimble Geomatics Office d'un périphérique exécutant  Trimble Survey Controller, il est importé automatiquement dans le projet.

NOTE: Lorsque vous transférez les données vers ou depuis un Survey Controller de version 7.60, utilisez le format de fichier .dc de version 7.50.


Importer un fichier de données GPS Trimble (.dat) sur votre ordinateur

1.	Sélectionnez Fichier / Importer. Le dialogue Importer s'affiche.

2.	Dans la tabulation Levé, sélectionnez l'option Fichiers de données GPS (*.dat) et cliquez sur OK. Le dialogue Ouvrir s'affiche.

3.	Trouvez le fichier .dat que vous voulez importer et cliquez sur Ouvrir. Le dialogue Entrée DAT s'affiche.

4.	Effectuez toute modification nécessaire.

En particulière, vérifiez que la (les) valeur(s) de hauteur d'antenne sont correctes. La modification des valeurs ne modifiera les valeurs actuelles dans le fichier .dat. Seules les valeurs utilisées pour l'importation du fichier dans  Trimble Geomatics Office seront modifiées. Ainsi le fichier .dat est conservé dans son état d'origine.

5.	Cliquez sur OK.

Le fichier importé se trouve dans le dossier Fichiers de données\Fichiers Trimble du projet.

ASTUCE: Vous pouvez faire glisser-déplacer le fichier.


Importer un fichier de données GPS Trimble (.dat) d'un récepteur GPS Trimble

Vue d'ensemble et plus amples informations
1.	Sélectionnez Fichier / Importer. Le dialogue Importer s'affiche.

2.	Dans la tabulation Levé, sélectionnez l'option Périphériques de levé et cliquez sur OK. Le Transfert de données Ouvrir s'affiche.

3.	Sélectionnez un périphérique récepteur GPS et cliquez sur Ouvrir. Le dialogue Connexion s'affiche.

Lorsque la connexion est effectuée, le champ Rechercher dans du dialogue Ouvrir change au périphérique récepteur GPS.

4.	Sélectionnez le(s) fichier(s) que vous voulez importer et cliquez sur Ouvrir. Le dialogue Téléchargement s'affiche et la barre de progression indique le progrès du transfert du fichier.

Lorsque le transfert est terminé, le dialogue Téléchargement disparaît et la connexion au récepteur GPS est interrompue.

Le dialogue Entrée DAT s'affiche.

5.	Effectuez toute modification nécessaire.

En particulière, vérifiez que la (les) valeur(s) de hauteur d'antenne sont correctes. La modification des valeurs ne modifiera les valeurs actuelles dans le fichier .dat. Seules les valeurs utilisées pour l'importation du fichier dans  Trimble Geomatics Office seront modifiées. Ainsi le fichier .dat est conservé dans son état d'origine.

6.	Cliquez sur OK.

Le fichier importé se trouve dans le dossier Fichiers de données\Fichiers Trimble du projet.


Importer un fichier de données GPS RINEX

Vue d'ensemble et plus amples informations
1.	Sélectionnez Fichier / Importer. Le dialogue Importer s'affiche.

2.	Dans la tabulation Levé, sélectionnez l'option Fichiers RINEX et cliquez sur OK. Le dialogue Ouvrir s'affiche.

3.	Sélectionnez le(s) fichier(s) d'observation que vous voulez importer. Changez l'option Types de fichier si requis. Cliquez sur Ouvrir.

S'il y a un fichier .nav correspondant avec le même nom, il est lié automatiquement au fichier d'observation.

Si le fichier .nav correspondant n'a pas le même nom, le dialogue Correspondance de fichiers de navigation s'affiche. Les fichiers d'observation sont listés sous les fichier d'observation. Sélectionnez le fichier .nav approprié et cliquez sur OK.

NOTE: Chaque fichier d'observation doit avoir un fichier .nav correspondant. Si le fichier .nav correct n'est pas sélectionné, un message s'affichera lors de l'importation et il ne sera pas possible de traiter les lignes de base.

4.	Le(s) fichier(s) sont convertis au format de fichier de données GPS Trimble. Le dialogue Entrée DAT s'affiche.

5.	Effectuez toute modification nécessaire.

En particulière, vérifiez que la (les) valeur(s) de hauteur d'antenne et le type de récepteur sont corrects. La modification des valeurs ne modifiera les valeurs actuelles dans le fichier de données GPS, Seules les valeurs utilisées pour l'importation du fichier dans  Trimble Geomatics Office seront modifiées. Ainsi le fichier de données GPS est conservé dans son état d'origine

ASTUCE: Assurez-vous qu'à l'entrée  .dat les types de récepteur et d'antenne ne sont pas inconnus.

6.	Cliquez sur OK.

Le fichier importé se trouve dans le dossier Fichiers de données\Fichiers Trimble du projet.


Créer un format d'importation/exportation personnalisé

 Vue d'ensemble et plus amples informations

Cet exemple définit un format d'exportation personnalisé. Le format est un fichier texte délimité par des virgules contenant le nom du point (en majuscules), le nord (avec des unités), l'est (avec des unités) et l'élévation (avec des unités). Les codes de champ sont utilisés pour sélectionner les champs requis. Les formats d'importation personnalisés sont définis dans la même façon.

1.	Sélectionnez Fichier / Exporter. Le dialogue Exporter s'affiche.

2.	Dans la tabulation Personnalisé, cliquez sur Nouveau. Le dialogue Définir le format d'exportation ASCII s'affiche.

3.	Dans le champs Nom, saisissez un nom.

4.	Cliquez avec le bouton droit de la souris dans le champ Corps du format et sélectionnez Champs / Nom du menu de raccourcis.

[Nom] s'affiche dans la boîte Corps du format.

5.	Cliquez avec le bouton droit de la souris et sélectionnez Champs / Expert / Modificateurs / Majuscule dans le menu de raccourcis.

[Nom:Majuscule] s'affiche dans la boîte Corps du format. Ce signifie que tous les noms de point sont exporté en majuscules.

6.	Saisissez une virgule suivie d'un espace.

7.	Cliquez avec le bouton droit de la souris et sélectionnez Champs / Nord du menu de raccourcis.

[Nord] s'affiche à côté de la virgule.

8.	Cliquez avec le bouton droit de la souris et sélectionnez Champs / Expert / Modificateurs / Unités dans le menu des raccourcis.

[Nord] change à [Nord:unités]. Ce signifie que l'unité de coordonnées courantes est comprise dans le fichier.

9.	Saisissez une virgule suivie d'un espace.

10.	Cliquez avec le bouton droit de la souris et sélectionnez Champs / Est du menu de raccourcis.

[Est] s'affiche à côté de la virgule.

11.	Cliquez avec le bouton droit de la souris et sélectionnez Champs / Expert / Modificateurs / Unités dans le menu de raccourcis.

[Est] change à [Est:unités]. Ce signifie que l'unité de coordonnées courantes est comprise dans le fichier.

12.	Saisissez une virgule suivie d'un espace.

13.	Cliquez avec le bouton droit de la souris et sélectionnez Champs / Elévation du menu de raccourcis.

[Elévation] s'affiche à côté de la virgule.

14.	Cliquez avec le bouton droit de la souris et sélectionnez Champs / Expert / Modificateurs / Unités dans le menu des raccourcis.

[Elévation] change à [Elévation:unités]. Ce signifie que l'unité de coordonnées courantes est comprise dans le fichier.

15.	Cliquez sur OK pour enregistrer le format d'exportation. Le dialogue Exporter s'affiche. Le format que vous avez créé s'affiche dans la liste dans la tabulation Personnalisé.

16.	Pour exporter les données dans ce format, sélectionnez le format et cliquez sur OK. Le dialogue Enregistrer sous s'affiche.

17.	Dans le champ Nom du fichier, saisissez un nom pour le fichier d'exportation et cliquez sur Enregistrer pour créer le fichier.


Exporter un fichier Survey Controller (*.dc) au Trimble Survey Controller

Vue d'ensemble et plus amples informations
1.	Sélectionnez Fichier / Exporter. Le dialogue Exporter s'affiche.

2.	Dans la tabulation Levé, sélectionnez l'option Périphériques de levé.

3.   Cliquez sur Options pour spécifier les paramètres de l'étude du fichier.

4.   Dans le dialogue qui s’affiche, cliquez sur OK pour accepter toute modification et retourner au dialogue Exporter.

5.	Cliquez sur OK. Le dialogue Transfert de données Enregistrer sous s'affiche.

6.	Sélectionner un périphérique Trimble Survey Controller et cliquez sur Ouvrir. Le dialogue Connexion s'affiche.

7.	Lorsqu'une connexion est effectuée, le dialogue Enregistrer s'affiche.

8.	Effectuez l'une des choses suivantes:

  Sélectionnez l'option Mémoire principale (C:) pour stocker le fichier dans la mémoire principale du logiciel Trimble Survey Controller.

  Sélectionnez l'option Carte PC (A:) si vous avez une carte PC insérée et vous voulez stocker le fichier sur la carte PC.

NOTE: Si votre périphérique Trimble n'a pas une carte PC, ce dialogue n'est pas disponible.

9.	Cliquez sur Ouvrir.

Le dossier sélectionné est lu. Tous les fichiers Survey Controller dans le dossier sont affichés. Le nom du projet Trimble Geomatics Office s'affiche dans le champ Nom du fichier. C'est le nom de l'étude du logiciel Trimble Survey Controller que sera exportée au logiciel Trimble Survey Controller. Vous pouvez éditer ce nom si nécessaire.

10.	Cliquez sur Enregistrer. Le dialogue Téléchargement s'affiche. La barre de progression affiche le progrès du transfert. Lorsque le transfert est terminé, le dialogue Téléchargement disparaît et la connexion au périphérique est interrompue.

Maintenant vous pouvez utiliser le fichier dans Trimble Survey Controller.

NOTE: Lorsque vous transférez les données vers ou depuis un Survey Controller de version 7.60, utilisez le format de fichier .dc de version 7.50.


Exporter les fichiers AutoCAD (*.dxf)

 Vue d'ensemble et plus amples informations

1.	Sélectionnez Fichier / Exporter. Le dialogue Exporter s'affiche.

2.	Dans la tabulation CAD / ASCII, sélectionnez l'option AutoCAD (*.dxf).

3.	Cliquez sur Options pour configurer la version .dxf et le type auxquels vous voulez exporter.

4.	Cliquez sur OK. Le dialogue Enregistrer sous s'affiche.

5.	Naviguez au dossier dans lequel vous voulez enregistrer le fichier.

6.	Saisissez le nom du fichier et cliquez sur Enregistrer.

Le fichier est créé.

Par défaut,  Trimble Geomatics Office exporte le nom du point sous forme d'un attribut du point. Pour exporter d'autres informations concernant le point, éditez les définitions de bloc d'attribut DXF/DWG dans le fichier Options.ini.

NOTE: Vous pouvez utiliser ces étapes pour exporter un fichier de dessin (*.dwg). A l'étape 4, cliquez sur Options pour configurer le format et la version .dwg.


Changer les unités d'un projet

 Vue d'ensemble et plus amples informations

1.	Sélectionnez Fichier / Propriétés du projet. Le dialogue Propriétés du projet s'affiche.

2.	Dans la tabulation Unités et Format, sélectionnez le champ pour lequel vous voulez changer les unités et le format.

3.	Sélectionnez l'option requise de la liste et cliquez sur OK pour appliquer les modifications.


Configurer un projet pour des attributs

 Vue d'ensemble et plus amples informations

1.	Sélectionnez Fichier / Propriétés du projet. Le dialogue Propriétés du projet s'affiche.

2.	Dans la tabulation Caractéristiques, sélectionnez la case à cocher Utiliser attributs. Le champ Bibliothèque devient disponible.

3.	Cliquez sur Parcourir pour trouver le fichier de Bibliothèque de caractéristiques et attributs (*.fcl) ou le fichier de Dictionnaire d'attributs (*.ddf) requis.

4.	Cliquez sur OK. Le fichier sélectionné est utilisé pour créer des tables de base de données pour chaque code de caractéristiques avec des attributs. Vous pouvez importer les fichiers .dc avec des attributs maintenant.


Sélectionner un site enregistré comme un système de coordonnées du projet

Vue d'ensemble et plus amples informations
1.	Sélectionnez Fichier / Propriétés du projet. Le dialogue Propriétés du projet s'affiche.

2.	Sélectionnez la tabulation Système de coordonnés.

3.	Dans le groupe Paramètres du système de coordonnées, cliquez sur Changer. Le dialogue Sélectionner système de coordonnées s'affiche.

4.	Sélectionnez Nouveau système et cliquez sur Suivant. Le dialogue Sélectionner le type de système de coordonnées s'affiche.

5.	Sélectionnez Site calibré et cliquez sur Suivant. Le dialogue Sélectionner site calibré s'affiche.

6.	Sélectionnez le site requis et cliquez sur Finir.

Le système de coordonnées est appliqué à la base de données. Tout point existant dans la base de données est converti au nouveau système de coordonnées. Pour plus amples informations concernant le procédé de conversion, référez-vous à la rubrique Systèmes de coordonnées – Vue d'ensemble.

Si vous avez des positions de point ou vous saisissez des coordonnées avec des élévations nulles dans votre projet et une élévation par défaut n'est pas définie, la sélection du système de coordonnées échouera parce que  Trimble Geomatics Office ne peut pas convertir les points existants d'un système à l'autre.


Exporter des attributs

 Vue d'ensemble et plus amples informations

Vous pouvez exporter un fichier contenant:

  tous les points avec une caractéristique spécifique

  tous les attributs

Pour exporter des fichiers séparés pour chaque caractéristique:

1. 	Sélectionnez les points contenant la caractéristique d'intérêt.

2. 	Configurez un format d'exportation personnalisé.


Sélections des points contenant la caractéristique d'intérêt

Pour ce faire:

1.	Dans la Vue en plan, choisissez Sélectionnez / Par caract. de point. Par exemple, si vous voulez rapporter sur tous les attributs arbre, sélectionnez "arbre" dans le dialogue Sélectionner par caract. de point.

2.	Cliquez sur OK. Tous les points avec le code de caractéristiques spécifié sont sélectionnés maintenant.


Configuration du format de rapport personnalisé

Pour ce faire:

1.	Sélectionnez Fichier / Exporter. Le dialogue Exporter s'affiche.

2.	Dans la tabulation Personnalisé, cliquez sur Nouveau. Le dialogue Définir le format d'exportation ASCII  s'affiche.

3.	Dans le champ Nom, saisissez un nom pour le format d'exportation.

4.	Dans le champ Description, saisissez une description du format d'exportation. Une description n'est pas obligatoire pour le format.

5.	Assurez-vous que l'option Informations de points est sélectionnée dans le champ Exporter de. C'est à cause du fait que vous faites un rapport sur des points.

6.	Cliquez sur Spécifier format automatiquement pour remplir tous les champs de la table Points dans l'en-tête du format et le corps du format.

7.	Dans le dialogue Insérer format que s'affiche, sélectionnez les délimiteurs appropriés requis pour le format.

8.	Supprimez toute en-tête et tout champ que vous ne voulez pas inclure dans le format.

9.	Dans la liste En-tête du format, ajoutez toute en-tête d'attribut que vous voulez inclure dans le format.

10.	Les enregistrements d'attribut ne sont pas stockés dans la table Points. Ajoutez les champs à inclure dans le format de la table de caractéristiques appropriée. Pour accéder aux champs, cliquez avec la bouton droit de la souris dans la liste Corps du format et sélectionnez du menu de raccourci.

11.	Sélectionnez Champs / Expert / Attributs. Tous les codes de caractéristiques avec attributs (dans la bibliothèque d'attributs et caractéristiques utilisée pour configurer le projet pour des attributs) s'affichent.

12.	Choisissez le même code de caractéristiques que vous avez sélectionné (par exemple "arbre"). Les attributs s'affiche dans un sous-menu supplémentaire.

13.	Cliquez l'attribut dont vous voulez rapporter. L'attribut sélectionné s'affiche dans la boîte Corps du format.

NOTE: Si vous voulez rapportes sur tous les attributs d'une caractéristique spécifique, dans le menu de raccourcis, sélectionnez l'option d'astérisque (*). Tous les attributs sont ajouté au corps du format.

14.	Répétez les étapes ci-dessus pour chaque attribut que vous voulez exporter à un fichier.

15.	Complétez le format d'exportation et cliquez sur OK. Le format est ajouté à la liste Personnalisé.

16.	Pour exporter un format de ce format, soulignez le format et cliquez sur OK.


Pour exporter un seul fichier avec tous les attributs

Pour ce faire:

1.	Sélectionnez Fichier / Exporter. Le dialogue Exporter s'affiche.

2.	Dans la tabulation SIG, sélectionnez l'option Nom, Latitude, Longitude, Hauteur, Tous les attributs (Local) ou l'option Nom, Nord, Est, Hauteur, Tous les attributs.

3.	Cliquez sur OK.

4.	Le dialogue Enregistrer s'affiche. Saisissez un nom du fichier et un emplacement et cliquez sur OK.

Un fichier contenant des attributs est créé.


Importer un fichier Survey Controller (*.dc) avec attributs

Vue d'ensemble et plus amples informations
NOTE: Pour utiliser les attributs dans  Trimble Geomatics Office, d'abord vous devez configurer un projet pour des attributs.

1.	Sélectionnez Fichier / Importer. Le dialogue Importer s'affiche.

2.	Dans la tabulation Levé, sélectionnez l'option Fichier Survey Controller (*.dc) et cliquez sur OK.

 Trimble Geomatics Office vérifie que le fichier Bibliothèque d'attributs et caractéristiques (*.fcl) ou Dictionnaire d'attributs (*.ddf) spécifié dans le projet Trimble Geomatics Office courant est le même que celui du fichier .dc.

S'ils diffèrent vous avez les options suivantes:

  Annuler l'importation.

  Importer tous les points et les informations d'attribut reconnus par le projet Trimble Geomatics Office courant. Vous êtes averti que vous pourriez perdre des données. Le Rapport d'importation décrit quelles informations d'attribut n'ont pas été importées.

  Importer seulement des points (aucune information d'attribut n'est importée).

Si la bibliothèque d'attributs et caractéristiques n'est pas définie dans le projet Trimble Geomatics Office courant, vous pouvez importer le fichier .dc dans les informations d'attribut ou annuler l'importation.

NOTE: Si vous transférez des données vers ou depuis  Survey Controller de version 7.60, utilisez le format du fichier .dc 7.50.


Ajouter une couche

 Vue d'ensemble et plus amples informations

1.	De la Vue en plan, effectuez l'une des choses suivantes:

  Sélectionnez Editer / Couches.

  Cliquez sur l'outil Editer couches.

Le dialogue Couches s'affiche.

2.	Cliquez sur Nouveau. Une nouvelle ligne s'affiche dans le dialogue Couches.

3.	Dans le champ Nom saisissez un nom unique pour la nouvelle couche.

4.	Sélectionnez un style de point par défaut de la liste. Ce style de point est utilisé pour tous les points qui ont <Par couche> sélectionné dans leur champ de Style de point.

5.	Sélectionnez un style de ligne par défaut de la liste. Ce style de ligne est utilisé pour toutes les lignes qui ont <Par couche> sélectionné dans leur champ de Style de ligne.

6.	Sélectionnez un style de texte par défaut de la liste. Ce style de texte est utilisé pour tout texte avec  <Par couche> sélectionné dans son champ de Style de texte.

7.	Si vous voulez afficher tours les entités dans la couche, sélectionnez la case à cocher Visible.

8.	Effacez la case à cocher Verrouillé pour permettre la sélection de toutes les entités dans la couche.

9.	Cliquez sur OK.


Fusionner les points avec noms en double

 Vue d'ensemble et plus amples informations

Vous pouvez fusionner les points en double dans  Trimble Geomatics Office si vous savez qu'ils sont le même point physique, et vous n'avez pas besoin qu'ils sont des points séparés.

ATTENTION: Deux points avec le même nom mais des données qui diffèrent de façon significative peut être le résultat d'une erreur. Leur fusion peut produire des résultats inattendus.

1.	Assurez-vous que les points sont le même point physique. Pour ce faire, choisissez Sélectionner / Points en double / Par nom et distance

2.	Saisissez une Distance de séparation et cliquez sur OK.

3.	Sélectionnez Editer / Fusionner points en double.

 Trimble Geomatics Office fusionne toutes les observations aux points et les positions entrées au clavier des points. La barre d'état affiche un résumé de l'opération de fusion.


Spécifier comment les points en double sont gérés lors de l'importation

 Vue d'ensemble et plus amples informations

Par défaut,  Trimble Geomatics Office fusionne les points avec le même nom lors de l'importation, afin d'éviter la création des points en double. Vous pouvez spécifier dans quelle tolérance horizontale et/ou verticale  fusionnera les points, ou que  ne fera pas  la résolution des points en double dors de l'importation, dans le dialogue Options de points en double.

1.	Sélectionnez Fichier / Importer. Le dialogue Importer s'affiche.

2.	Cliquez sur Options. Le dialogue Options de points en double s'affiche.

Effectuez l'une des choses suivantes:

  Assurez-vous que l'option Toujours fusionner les points en double est sélectionnée.

  Sélectionnez l'option Fusionner points en double dans la tolérance et saisissez les tolérances appropriées.

  Pour spécifier que  ne fait pas la résolution des points en double lors de l'importation, sélectionnez l'option Fusionner points en double et saisissez une valeur de tolérance horizontale de 0, 02 (c'est à dire, la tolérance de qualité de levé horizontale) et une tolérance verticale de 0,050 (c'est à dire, la tolérance de levé verticale ).

NOTE: Si vous ne spécifiez pas une tolérance (c'est à dire, dans le groupe de tolérances, les cases à cocher Horizontal et Vertical ne sont pas sélectionnées) les points en double seront fusionnés.

NOTE: Si vous configurez la tolérance sur 0,000, les coordonnées des points en double doivent être exactement les mêmes pour les points à fusionner.

Ces paramètre s'appliquent à toutes les options d'importation; si vous importez les fichier à l'aide de la barre du projet, ou faites glisser déplacer les fichiers dans un projet Trimble Geomatics Office,  utilise le dernier paramètre de points en double spécifié.


Renommer les points en double

 Vue d'ensemble et plus amples informations

Utilisez cette procédure pour renommer les points en double dans le cas où, par exemple, deux équipes sur le terrain ont utilisé les mêmes noms pour des points différents dans des fichiers Survey Controller (*.dc) séparés:

1.	Importez les fichiers avec l'option Fusionner points en double dans la tolérance sélectionnée dans le dialogue Options de points en double. Configurez la tolérance horizontale sur 0,02 et la tolérance verticale sur 0,050.

NOTE: Si vous ne vous rendez pas compte que vous avez des points en double que jusqu'à après l'importation des données, annulez l'opération d'importation, configurez l'option Fusionner points en double dans la tolérance sur 0,02 pour la tolérance horizontale (c'est à dire, la tolérance de qualité de levé horizontale) et 0,05 pour la tolérance verticale (c'est à dire, la tolérance de qualité de levé verticale)  dans le dialogue Options de points en double, et importez les données à nouveau.

2.	Sélectionnez le groupe de sélection de fichiers Survey Controller (*.dc) qui a ajouté les points en double à la base de données. Sinon, sélectionnez Sélectionner / Points en double / Par nom.

3.	Sélectionnez Editer / Renommer. Le dialogue Renommer s'affiche.

4.	Saisissez le nouveau nom du premier point dans la sélection.

 Trimble Geomatics Office renomme tous les points dans le groupe de sélection. Les points sont renommés selon la valeur de début assignée au groupe.

Par exemple, supposez que vous avez cinq points sélectionnés. Si vous remplacez le nom du premier point avec le nom "2000", les quatre points restants sont renommés 2001, 2002, 2003, et 2004.


Etiqueter les points utilisants des étiquettes personnalisées

 Vue d'ensemble et plus amples informations

Cet exemple étiquete les points en utilisant l'étiquette personnalisée d'une description de point.

1.	Utilisez l'Explorateur de Microsoft Windows pour trouver le fichier Options.ini du modèle du projet ou du projet auquel vous voulez ajouter l'option d'étiquette personnalisée.

2.	Utilisez l'éditeur de texte pour ouvrir le fichier Options.ini.

3.	Recherchez l'en-tête:

[Etiquettes]

4.	Directement après la dernière option d'étiquette, utilisez la clavier pour saisir le texte suivant:

|X|[Y]

Où ‘X’ définit le nom de l'étiquette, et ‘Y’ définit le code de champ avec lequel vous voulez étiqueter le point.

5.	Ouvrez le modèle du projet ou le projet dans  Trimble Geomatics Office.

6.	Sélectionnez Affichage / Etiquettes de point. Le dialogue Etiqueter point avec s'affiche avec la nouvelle option ajoutée à la liste.

8.	Sélectionnez les options requises et cliquez sur OK.


Annoter les entités

 Vue d'ensemble et plus amples informations

1.	Sélectionner le points, les lignes, ou les arcs que vous voulez annoter. Pour ce faire, utilisez l'un des suivants:

  La souris

  Le menu Sélectionner

2.	Sélectionnez Insérer / Annotations. Le dialogue Annoter s'affiche.

3.	Sélectionnez le modèle requis de la liste.

4.	Pour éditer les modèles d'annotation, cliquez sur Modèles.

  Pour conserver les annotations existantes, sélectionnez la case à cocher Conserver annotations existantes.

  Pour supprimer des annotations existantes avant l'application des nouvelles annotations, effacez la case a cocher Conserver annotations existantes.

5.	Si vous avez sélectionné un groupe d'entités à annoter, vérifiez que le groupe Annoter est configuré sur Sélection.

Si vous voulez annoter la base de données entière, sélectionnez l'option BD entière.

S'il n'y a aucune entité sélectionnée, le groupe Annoter n'est pas disponible et la base de données entière est annotée.

6.	Cliquez sur OK pour appliquer les annotations.

Les annotations sont ajoutées à la base de données et s'affichent dans la fenêtre de graphiques.


Créer un groupe de sélection

 Vue d'ensemble et plus amples informations

1.	Sélectionnez les entités (points, lignes, observations, courbes, arcs, texte, annotations) que vous voulez inclure dans le groupe de sélection. Pour ce faire, utilisez l'un des suivants:

  La souris

  Le menu Sélectionner

2.	Choisissez Sélectionner / Groupes de sélection / Enregistrer. Le dialogue Enregistrer s'affiche.

3.	Saisissez un nom pour le groupe de sélection et cliquez sur OK.

Maintenant le groupe de sélection s'affiche dans la liste des groupes de sélection enregistrés lorsque vous choisissez Sélectionner / Groupes de sélection / Récupérer. En plus il s'affiche dans la liste de groupes de sélection utilisé récemment lorsque vous choisissez Sélectionner / Groupes de sélection.

ASTUCE: Utilisez le dialogue Filtres d'affichage pour afficher les groupes de sélection.


Effectuer une calibration du site GPS

 Vue d'ensemble et plus amples informations

Il peut être nécessaire d'effectuer une calibration du site GPS dans  Trimble Geomatics Office à cause des raisons suivantes:

  pour transférer une calibration au logiciel Trimble Survey Controller

  vous n'avez pas calibré sur le terrain

  pour créer un rapport des enregistrements de contrôle de qualité

  pour ajouter des points supplémentaires à une calibration effectuée dans  Trimble Survey Controller


Préparation pour effectuer une calibration du site GPS

Avant d'effectuer une calibration du site GPS, vous devez vous assurer que vous avez des coordonnées dérivées des données GPS (WGS-84) ou terrestres (de grille) pour chaque point de calibration.

Si vous avez effectué une calibration dans  Trimble Survey Controller, et les points de grille et les points GPS ont le même nom, il se peut que ces points ont été fusionnés lors de l'importation. Les points fusionnés seront séparés automatiquement si vous décidez d'effectuer une calibration du site GPS dans  Trimble Geomatics Office. A ce temps-là, un message vous informe que le même nom a été utilisé pour les points GPS et de grille dans .

Effectuez l'une des choses suivantes:

  Cliquez sur Oui pour créer un point GPS et un point de grille séparés avec le même nom dans la base de données.

  Cliquez sur Annuler si vous ne voulez pas effectuer une calibration du site GPS ou si vous voulez exploser les points manuellement. Si vous explosez les points manuellement,  vous aurez l'option de renommer les points GPS.


Réalisation de la calibration du site GPS

1.	Sélectionnez Levé / Calibration du site GPS. Le dialogue Calibration du site GPS s'affiche.

2.	Dans le groupe Composants de calibration, sélectionnez les paramètres de calibration que vous voulez calculer.

3.	Cliquez Liste points. Le Liste points s'affiche.

4.	Utilisez le dialogue Liste point pour sélectionner votre paire de points (point GPS et point de grille) pour la calibration. Il peut être le même point ou un point différent avec le même nom, à condition que des valeurs GPS et de grille séparées peuvent être trouvées.

5.	Avec le premier point requis dans la calibration, effectuez l'une des choses suivantes:

  Utilisez le remplissage du champ pour sélectionner le point.

  Saisissez le nom du point à l'aide du clavier.

Une boîte s'affiche sur le point GPS dans la fenêtre de graphiques.

NOTE: Si vous avez un point de grille correspondant avec le même nom, la Liste points est remplie automatiquement.

6.	Pour saisir le point de grille associé au point GPS sélectionné, effectuez l'une des choses suivantes:

 Utilisez le remplissage du champ pour sélectionner le point.

 Saisissez le nom du point à l'aide du clavier.

Une croix s'affiche sur le point de grille dans la fenêtre de graphiques et une nouvelle entrée est ajoutée au dialogue Liste points.

Le champ Type se met à jour pour indiquer si le point sélectionné a un composant de coordonnée horizontal ou vertical (ou tous les deux). S'il n'y a aucun composant de coordonnée commun, le Type est affiché comme non valide et les coordonnées ne sont pas utilisées.

7.	Répétez les deux étapes ci-dessus pour sélectionner toutes les paires de points requises dans la calibration.

8.	Lorsque vous avez sélectionné toutes les paires de points, cliquez sur Fermer et retournez au  dialogue Calibration du site GPS.

NOTE: Les statistiques de calibration s'affichent en bas du dialogue Points et sont calculés après l'entrée d'une paire de points valides. Utilisez ces statistiques pour identifier les paires de points incorrectes.

9.	Cliquez sur Calculer pour calculer les paramètres de calibration. Les résultats de la calibration s'affichent dans le groupe Résumé de calibration.

10.	Si les résultats sont hors des limites attendues, utilisez le rapport de calibration du site GPS pour vérifier tout résidu important entre les paires de points. Cliquez sur Rapport pour générer le rapport de calibration du site GPS. Le rapport s'affiche dans l'afficheur HTML par défaut.

11.	Vérifiez les résidus dans la section Différences résiduelles entre les coordonnées connues et GPS du rapport. Si nécessaire, supprimez toute paire de points suspecte et réappliquez la calibration.

12.	Cliquez sur OK pour appliquer la calibration au projet.

Le système de coordonnées est mis à jour avec les paramètres de calibration.


Traiter les codes de caractéristiques

 Vue d'ensemble et plus amples informations

1.	Choisissez Sélectionner / Groupes de sélection / Récupérer. Le dialogue Récupérer s'affiche.

2.	Sélectionnez le groupe de sélection avec le même nom que le fichier  à traiter (le fichier dans lequel les codes de caractéristique ont été collectés). Ce groupe de sélection était créé automatiquement lors de l'importation du fichier Survey Controller (*.dc), et a un enregistrement de l'ordre dont les points ont été collectés sur le terrain. En utilisant ce groupe de sélection, vous êtes assuré que les codes de caractéristiques sont traités dans l'ordre correct.

3.	Sélectionnez Outils / Traiter codes de caract. Le dialogue Traiter codes de caract. s'affiche.

4.	Sélectionnez la bibliothèque requise pour le traitement de codes de caractéristiques et cliquez sur OK.

 Trimble Geomatics Office traite les codes de caractéristiques. Les codes de caractéristiques s'affichent dans la fenêtre de graphiques.


Afficher un inverse

 Vue d'ensemble et plus amples informations

Pour déterminer l'inverse entre deux points du projet:

1.	Sélectionnez Levé / Inverse. Le dialogue Inverse s'affiche et le pointeur devient un pointeur de Remplissage de champ.

2.	Cliquez sur le premier point requis pour le calcul de l'inverse. Le nom du point s'affiche dans le champ De dans le dialogue Inverse.

3.	Cliquez sur le deuxième point requis pour le calcul de l'inverse. Lorsque la souris est positionnée sur un point, une ligne pointillée s'affiche.

Le nom du point s'affiche dans le champ Vers dans le dialogue Inverse et les valeurs d'inverse sont affichées automatiquement dans le groupe Inverse.

NOTE: Si vous saisissez des points dans les champs à l'aide du clavier, cliquez sur Calculer pour calculer l'inverse.

4.	Cliquez sur Détaillé pour afficher des informations supplémentaires concernant ce calcul d'inverse. Le groupe Détails est ajouté en bas du dialogue et affiche les informations supplémentaires concernant l'inverse.

5.	Cliquez sur Envoyer au rapport pour générer un rapport sur l'inverse. Un rapport de résumé ou détaillé sera créé selon l'option d'affichage (résumé ou détaillé) utilisée.

6.	Cliquez sur Afficher rapport pour afficher le rapport dans l'afficheur HTML.

7.	Pour afficher un autre inverse, cliquez dans les champs de points et résélectionnez les points.


Mesurer les surfaces

Utilisez l'outil Mesurer pour afficher l'azimut et la distance entre les derniers deux emplacements sélectionnés et la surface créée par les emplacements sélectionnés dans l'opération de mesure courante.

1.	Sélectionnez Levé / Mesurer. Le dialogue Mesurer s'affiche.

2.	Cliquez sur le premier emplacement (qui ne doit pas être un point) dans la fenêtre de graphiques.

3.	Continuez de cliquer sur les emplacements dans la fenêtre de graphiques. Chaque fois que vous sélectionnez un emplacement, les coordonnées sont utilisées pour calculer les champs du dialogue Mesurer. Une ligne s'affiche liant les emplacements sélectionnés.

Les champs dans le dialogue Mesurer sont mis à jour chaque fois que vous cliquez sur un nouvel emplacement. Les valeurs ne sont pas stockées dans la base de données.

4.	Cliquez sur Remettre pour effacer les valeurs.


Effectuer une transformation de coordonnées

 Vue d'ensemble et plus amples informations

Utilisez une transformation de coordonnées pour traduire et/ou pivoter les points sélectionnés. Cet exemple vous montre comment transformer tous les points dans la base de données en entrant des coordonnées connues pour un point existant.

1.	Sélectionnez les points que vous voulez transformer ou pivoter.

2.	Sélectionnez Levé / Transformation de coordonnées. Le dialogue Transformation de coordonnées s'affiche.

3.	Cliquez sur Liste points. Le dialogue Transformation de coordonnées – Liste points s'affiche.

4.	Effectuez l'une des choses suivantes:

  Dans le champ Point, saisissez le nom du point existant.

  Utilisez le remplissage du champ pour sélectionner le point.

Le champ Point, le champ Coordonnée nord actuelle, et le champ Coordonnée est actuelle sont remplis et le champ Nord connu est sélectionné.

5.	Dans les champs Nord connu et Est connu, effectuez l'une des choses suivantes:

  Saisissez le nord et l'est connus du point.

  Utilisez le remplissage du champ pour sélectionner le nord et l'est connus du point.

Une ligne s'affiche dans la fenêtre de graphiques pour indiquer la différence entre les coordonnées connues et courantes.

Le groupe Statistiques affiche le facteur d'échelle et le résidu horizontal maximum. Utilisez ces détails pour identifier les paires de points mauvaises.

6.	Cliquez sur OK. Un dialogue s'affiche indiquant les erreurs résiduelles.

7.	Le dialogue Transformation de coordonnées retourne et les champs Translation nord et Translation est sont remplis avec les valeurs calculées à partir des coordonnées connues et courantes.

8.	Cliquez sur OK pour appliquer la transformation de coordonnées à la base de données entière.

NOTE: Vous pouvez annuler une transformation de coordonnées.


Effectuer un ajustement d'élévation

 Vue d'ensemble et plus amples informations

Utilisez un ajustement d'élévation pour ajuster les élévations des points sélectionnés. Cet exemple vous montre comment ajuster les élévations de tous les points dans la base de données en entrant une élévation connue pour un point existant.

1.	Sélectionnez Levé / Ajustement d'élévation. Le dialogue Ajustement d'élévation s'affiche.

2.	Cliquez sur Liste points. Le dialogue Ajustement d'élévation – Liste points s'affiche

3.	Effectuez l'une des choses suivantes:

  Dans le champ Point, saisissez le nom du point existant.

  Utilisez le remplissage du champ pour sélectionner le point.

Une boîte s'affiche sur le point sélectionné.

Les champs Point, Coordonnée nord actuelle, le champ Coordonnée est actuelle et le champ Elévation actuelle sont remplis.

4.	Effectuez l'une des choses suivantes:

  Dans le champ Elévation connue, saisissez l'élévation connue du point.

  Utilisez le remplissage du champ pour sélectionner l'élévation d'un autre point.

Le groupe Statistiques affiche le facteur d'échelle et l'inclinaison d'ajustement maximum et le résidu vertical maximum. Utilisez ces détails pour identifier les paires de points mauvaises.

5.	Cliquez sur OK pour retourner au dialogue Ajustement d'élévation.

Les champs Origine nord et Origine est sont remplis avec les coordonnées du premier point utilisé. Les champ Correction constante est rempli avec l'ajustement d'élévation calculé. Les champs Correction par unité nord et and Correction par unité est ne sont remplis que si plus d'un point est utilisé pour définir l'ajustement d'élévation.

6.	Cliquez sur OK pour appliquer l'ajustement d'élévation à la base de données entière.

NOTE: Vous pouvez annuler un ajustement d'élévation.


Utiliser les codes de champ

 Vue d'ensemble et plus amples informations

Utilisez les codes de champ pour afficher les valeurs des:

  champs

  variables dans les annotations, les rapports, les formats d'exportation, le texte

Les codes de champ sont également utiles pour la définition des rapports et des formats d'exportation. Cet exemple vous montre comment utiliser les codes de champ pour définir un format d'exportation. Le format définit un fichier délimité par des virgules contenant le nom du point (en majuscules), le nord (avec unités), l'est (avec unités) et l'élévation (avec unités).

1.	Sélectionnez Fichier / Exporter. Le dialogue Exporter s'affiche.

2.	Dans la tabulation Personnalisé, cliquez sur Nouveau. Le dialogue Définir le format d'exportation ASCII s'affiche.

3.	Dans le champ Nom, saisissez un nom.

4.	Cliquez avec le bouton droit de la souris dans le champ Corps du format et à partir du menu de raccourcis, sélectionnez Champs / Nom.

[Nom] apparaît dans la boîte Corps du format.

5.	Cliquez avec le bouton droit de la souris pour accéder au menu de raccourcis et puis sélectionnez Champs / Expert / Modificateurs / Majuscules.

[Nom:Majuscules] apparaît dans la boîte de corps du format. Ce signifie que tous les noms de point seront exportés avec toute lettre comprise dans le nom de point en majuscules.

6.	Saisissez une virgule suivie d'un espace.

7.	Cliquez avec le bouton droit de la souris pour accéder au menu de raccourcis et sélectionnes Champs / Nord.

[Nord] apparaît à côté de la virgule.

8.	Cliquez avec les bouton droit de la souris pour accéder au menu de raccourcis et sélectionnez Champs / Expert / Modificateurs / Unités.

[Nord] change à [Nord:unités]. Ce signifie que l'unité de coordonnées courante est comprises dans le fichier.

9.	Saisissez un virgule suivie d'un espace.

10.	Cliquez avec le bouton droit de la souris et sélectionner Champs / Est du menu de raccourcis.

[Est] apparaît à côté de la virgule.

11.	Cliquez avec les bouton droit de la souris et sélectionnez Champs / Expert / Modificateurs / Unités du menu de raccourcis.

[Est] change à [Est:unités]. Ce signifie que l'unité de coordonnées courante est comprises dans le fichier.

12.	Saisissez un virgule suivie d'un espace.

13.	Cliquez avec le bouton droit de la souris et sélectionner Champs / Elévation du menu de raccourcis.

[Elévation] apparaît à côté de la virgule.

14.	Cliquez avec les bouton droit de la souris et sélectionnez Champs / Expert / Modificateurs / Unités du menu de raccourcis.

[Elévation] change à [Elévation:unités]. Ce signifie que l'unité de coordonnées courante est comprises dans le fichier.

15.	Cliquez sur OK pour enregistrer le format d'exportation. Le dialogue Exporter s'affiche. Le format que vous avez créé s'affiche dans la liste dans la tabulation Personnalisé.

16.	Pour exporter les données sous ce format, sélectionnez le format et cliquez sur OK. Le dialogue Enregistrer sous s'affiche.

17. 	Dans le champ Nom du fichier, saisissez un nom pour le fichier d'exportation.

18.	Cliquez sur Enregistrer pour créer le fichier.


Utiliser le remplissage du champ

 Vue d'ensemble et plus amples informations

Utilisez le remplissage du champ pour saisir les valeurs des champs à partir des entités existantes dans la base de données. Lorsque le remplissage du champ est disponible, le pointeur de Remplissage du champ s'affiche dans la fenêtre de graphiques. Cet exemple utilise le remplissage du champ pour sélectionner des données lors de l'ajout d'une entité de texte.

1.	Sélectionnez Insérer / Texte. Le dialogue Insérer texte s'affiche.

2.	Dans le champ Expression de texte, saisissez le texte.

3.	Sélectionnez le champ Nord.

4.	Positionnez le pointeur sur la fenêtre de graphiques. Le pointeur de Remplissage du champ s'affiche.

5.	Cliquez sur l'emplacement dans la fenêtre de graphiques auquel vous voulez afficher le texte. Les coordonnées sont remplies dans le champs Nord et Est.

6.	Cliquez sur OK pour ajouter le texte à la base de données.

ASTUCE: Vous pouvez utiliser le remplissage du champ pour saisir les temps d'une observation (en cliquant sur l'observation).


Configurer la mise en page graphiquement (report/impression)

 Vue d'ensemble et plus amples informations

1.	Sélectionnez Fichier / Mise en page. Le dialogue Mise en page s'affiche.

2.	Cliquez sur Meilleure adaptation.

3.	Cliquez sur le champ Origine nord ou Origine est. Le pointeur de Remplissage du champ s'affiche.

4.	Cliquez dans la fenêtre de graphiques à la position requise pour le coin en haut à gauche de la surface du report.

5.	Configurez la rotation  pour mieux s'adapter les données du projet dans le modèle de plan sélectionné.

6.	Dans la fenêtre de graphiques, cliquez et faites glisser déplacer la boîte de report sur les données du projet.

7.	Changez l'échelle à une valeur standard. Par exemple, si la meilleure adaptation du report est à 1:435, changez-la à 1:500.

8.	Cliquez sur OK pour appliques les paramètres de Mise en page.


Sélectionner un modèle de géoïde pour le projet

 Vue d'ensemble et plus amples informations

Pour sélectionner un modèle de géoïde pour un projet, utilisez l'assistant Système de coordonnées. Ce s'applique même si vous voulez changer le modèle de géoïde pour le système de coordonnées courant.

1.	Sélectionnez Fichier / Propriétés du projet. Le dialogue Propriétés du projet s'affiche.

2.	Sélectionnez la tabulation Système de coordonnées.

3.	Dans le groupe Paramètres du système de coordonnées, cliquez sur Changer. Le dialogue Sélectionner système de coordonnées s'affiche.

4.	Sélectionnez l'option Nouveau système et cliquez sur Suivant. Le dialogue Sélectionner le Type de système de coordonnées s'affiche.

5.	Sélectionnez le type de système de coordonnées approprié. Le dialogue Sélectionner le modèle du géoïde s'affiche à la fin de l'assistant de sélection de système de coordonnées. Sélectionnez le modèle de géoïde requis. La liste des modèles de géoïde disponibles est lue de la base de données de système de coordonnées. Le modèle de géoïde par défaut pour le système de coordonnées est sélectionné par défaut.

6.	Cliquez sur Finir.

Le modèles de géoïde sélectionné est appliqué au projet.

NOTE: Vous pouvez sélectionner le modèle de géoïde par défaut pour un système de coordonnées utilisant l'utilitaire Coordinate System Manager.


Couper ou copier des points avec attributs entre les projets

 Vue d'ensemble et plus amples informations

1.	Du projet de source, sélectionnez les points que vous voulez copier au projet de destination. Utilisez la commande Sélectionner / Par caract. de point (disponible dans la Vue en plan) pour sélectionner les points que vous spécifiez.

2.	Pour copier les points requis au nouveau projet, sélectionnez Editer / Copier.

3.	Dans le projet de destination, sélectionne Editer / Coller pour coller les points.

4.	Si les définitions d'attribut sont différentes dans les deux projets, un dialogue s'affiche.

5.	Cliquez sur Oui  pour coller les données communes aux deux bibliothèques 'attributs et caractéristiques. Les données d'attribut que ne sont pas collées sont enregistrées dans le rapport Paste.rpt (Coller.rpt). Ce rapport est stocké dans le dossier de rapports du projet dans lequel vous collez.

NOTE: Seulement des points peuvent être coupés et copiés entre le projets. Si des observations étaient utilisées pour positionner les points, les coordonnées dérivées sont copiées. Si les points étaient dérivés des coordonnées GPS et des observations, la coordonnée WGS-84 est copiée. Cette coordonnée WGS-84 copiée peut être utilisée dans la calibration du site GPS dans le nouveau projet.


Ajouter des attributs à un point

 Vue d'ensemble et plus amples informations

Pour ajouter des attributs, vous devez configurer le projet pour des attributs.

1.	Sélectionnez le point auquel vous voulez ajouter des attributs. Pour ce faire, utilisez l'un des suivants:

  La souris

  Le menu Sélectionner

2.	Ouvrez la fenêtre Propriétés.

3.	Dans la tabulation Attributs, cliquez sur l'icône Menu Page [image]. Le menu de raccourcis s'affiche.

4.	Cliquez sur Insérer. La liste des codes de caractéristiques avec attributs disponibles s'affiche.

5.	Sélectionnez le code de caractéristiques avec attributs que vous voulez ajouter. Les attributs du code de caractéristiques sélectionné s'affichent.

6.	Saisissez des valeurs pour chaque attribut.  Trimble Geomatics Office vérifie chaque valeur saisie. Cette vérification assure que la valeur est valide selon la bibliothèque d'attributs et caractéristiques utilisée pour configurer le projet pour des attributs.

7.	Répétez les étapes ci-dessus pour tout autre code de caractéristiques avec attributs que vous voulez ajouter au point.


Afficher les attributs

 Vue d'ensemble et plus amples informations

1.	Sélectionnez le(s) point(s). Pour ce faire, utilisez l'un des suivants:

  La souris

  Le menu Sélectionner

2.	Ouvrez la fenêtre Propriétés.

3.	Sélectionnez la tabulation Attributs.

Vous ne pouvez afficher qu'un code de caractéristiques avec attributs à la fois. Pour afficher un différent code de caractéristiques avec attributs, utilisez la liste déroulante.

NOTE: La tabulation Attributs ne s'affiche que si vous avez configurer une bibliothèque d'attributs et caractéristiques pour le projet dans le dialogue Propriétés du projet / tabulation Caractéristiques.


Editer les attributs

 Vue d'ensemble et plus amples informations

1.	Cliquez sur l'attribut. Si l'attribut est un menu, une flèche s'affiche à droite du champ.

2.	Cliquez sur la flèche pour afficher la liste des options de menu pour l'attribut.

Tous les autres types d'attribut nécessitent une entrée au clavier.

NOTE: Lorsque vous éditer des informations d'attribut,  Trimble Geomatics Office vérifie la définition de la bibliothèque d'attributs et caractéristiques pour assurer que des valeurs valides ont été saisies.


Insérer une ligne

 Vue d'ensemble et plus amples informations

1.	Sélectionnez la Vue en plan. Effectuez l'une des choses suivantes:

  Sélectionnez Affichage / Plan.

  Cliquez sur l'outil Vue en plan.

  Appuyez sur Alt+2.

2.	Effectuez l'une des choses suivantes:

  Sélectionnez Insérer / Lignes.

  Sélectionnez l'outil Insérer ligne.

Le dialogue Insérer lignes affiche la tabulation Ligne sélectionnée.

3.	Dans le champ De point, effectuez l'une des choses suivantes:

  Utilisez le remplissage du champ pour sélectionner le premier point requis pour définir la ligne.

  Saisissez le nom du point pour le premier point requis pour définir la ligne.

4.	Dans le champ Vers point, effectuez l'une des choses suivantes:

 Utilisez le remplissage du champ pour sélectionner le deuxième point requis pour définir la ligne.

  Saisissez le nom du point pour le deuxième point requis pour définir la ligne.

5.	Dans le champ Couche, effectuez l'une des choses suivantes:

  Sélectionnez la couche à laquelle vous voulez ajouter la ligne.

  Saisissez un nouveau nom de couche pour créer une nouvelle couche.

6.	Dans le champ Style de ligne, sélectionnez le style de ligne requis pour afficher la ligne.

7.	Pour ajouter la ligne, effectuez l'une des choses suivantes:

  Cliquez sur Insérer.

  Appuyez sur Entrée.

Le point que vous avez sélectionné dans le champ Vers point s'affiche dans le champ De point. Vous pouvez continuer l'ajout des lignes sans la nécessité de saisir le point "de" à nouveau.

La ligne s'affiche dans la fenêtre de graphiques (à condition que la couche sélectionnée pour la ligne ait sa propriété visible activée).

8.	Cliquez sur Fermer pour arrêter l'insertion des lignes.


Insérer une courbe

 Vue d'ensemble et plus amples informations

1.	Sélectionnez la Vue en plan. Effectuez l'une des choses suivantes:

  Sélectionnez Affichage / Plan.

  Cliquez sur l'outil Vue en plan.

  Appuyez sur Alt+2.

2.	Sélectionnez Insérer / Courbes. Le dialogue Insérer lignes s'affiche avec la tabulation Courbe sélectionnée.

3.	Dans le champ De point, effectuez l'une des choses suivantes:

  Utilisez le remplissage du champ et cliquez sur les points appropriés.

  Utilisez le clavier pour saisir le nom du point.

4.	Dans le champ Vers point, effectuez l'une des choses suivantes:

  Utilisez le remplissage du champ et cliquez sur les points appropriés.

  Utilisez le clavier pour saisir le nom du point.

5.	Dans les champs Tangente d'entrée et Tangente de sortie, laissez les valeurs comme <Automatique> pour saisir automatiquement les valeurs de tangente basées sur les points utilisés pour définir la courbe. Vous pouvez saisir vos propres valeurs de tangente.

6.	Dans le champ Coefficient de lissage, saisissez une valeur entre 0 et 1. Le Coefficient de lissage définit combien de largeur il y a dans la courbe. Le plus haut le coefficient de lissage, la plus grande la largeur de courbe.

7.	Dans le champ Couche, effectuez l'une des choses suivantes:

  Sélectionnez la couche à laquelle vous voulez ajouter la courbe.

  Saisissez un nouveau nom de couche pour créer une nouvelle couche.

8.	Dans le champ Style de ligne, sélectionnez le style de ligne requis pour afficher la courbe.

9.	Pour ajouter la courbe, effectuez l'une des choses suivantes:

  Cliquez sur Insérer.

  Appuyez sur Entrée.

La courbe s'affiche dans la fenêtre de graphiques (à condition que la couche sélectionnée pour la courbe ait sa propriété visible activée). Chaque fois que vous ajoutez une courbe à partir d'un point utilisé dans une autre courbe,  Trimble Geomatics Office recalcule la courbe existante afin d'assurer qu'une courbe lisse soit affichée.

10.	Cliquez sur Fermer pour arrêter l'insertion des courbes.


Insérer un arc

 Vue d'ensemble et plus amples informations

1.	Sélectionnez la Vue en plan. Effectuez l'une des choses suivantes:

  Sélectionnez Affichage / Plan.

  Cliquez sur l'outil Vue en plan.

  Appuyez sur Alt+2.

2.	Sélectionnez Insérer / Arcs. Le dialogue Insérer lignes s'affiche avec la tabulation Arc sélectionnée.

3.	Dans le champ De point, effectuez l'une des choses suivantes:

  Utilisez le remplissage du champ et cliquez sur le point approprié.

  Utilisez le clavier pour saisir le nom du point.

4.	Dans le champ Vers point, effectuez l'une des choses suivantes:

  Utilisez le remplissage du champ et cliquez sur le point approprié.

  Utilisez le clavier pour saisir le nom du point.

5.	Dans le champ Point, pour utiliser un point existant comme le centre de l'arc, effectuez l'une des choses suivantes:

  Saisissez un nom de point existant.

  Utilisez le remplissage du champ pour sélectionner un point existant.

Les coordonnées du point sont copiées aux champs Centre nord et Centre est.

6.	Dans le champs Centre nord et Centre est, cliquez sur un emplacement dans la fenêtre de graphiques pour utiliser des coordonnées comme le centre de l'arc. Les coordonnées sont copiées à ces champs.

Si vous saisissez un point central et des coordonnées du centre, le point central est utilisé.

7.	Dans le champ Petit arc, sélectionnez l'option Faux si vous voulez créer un grand arc. Autrement, sélectionnez Vrai. Il y a toujours deux arcs possibles entre deux points, lesquels ensemble définissent un cercle. Ce paramètre définit quel arc vous voulez tracer.

8.	Dans le champ Couche, effectuez l'une des choses suivantes:

  Sélectionnez la couche à laquelle vous voulez ajouter l'arc.

  Saisissez un nouveau nom de couche pour créer une nouvelle couche.

9.	Dans le champ Style de ligne, sélectionnez le style de ligne requis pour afficher l'arc.

10.	Pour ajouter la courbe, effectuez l'une des choses suivantes:

  Cliquez sur Insérer.

  Appuyez sur Entrée.

La courbe s'affiche dans la fenêtre de graphiques (à condition que la couche sélectionnée pour l'arc ait sa propriété visible activée).

11.	Cliquez sur Fermer pour arrêter l'insertion des arcs.


Insérer du texte

 Vue d'ensemble et plus amples informations

1.	Sélectionnez la Vue en plan. Effectuez l'une des choses suivantes:

  Sélectionnez Affichage / Plan.

  Cliquez sur l'outil Plan.

  Appuyez sur Alt+2.

2.	Effectuez l'une des choses suivantes:

  Sélectionnez Insérer / Texte.

  Cliquez l'outil Insérer texte.

Le dialogue Insérer texte s'affiche.

3.	Dans le champ Expression de texte, saisissez le texte que vous voulez afficher. Pour utiliser les codes de champ (par exemple, le titre du projet) cliquez avec le bouton droit de la souris pour afficher un menu des raccourcis et sélectionnez Champs.

4.	Dans le champ Nord, saisissez la valeur nord pour la position du texte. Sinon, utilisez le remplissage du champ en cliquant sur la position dans la fenêtre de graphiques.

5.	Dans le champ Est, saisissez la valeur est pour la position du texte. Sinon, utilisez le remplissage du champ en cliquant sur la position dans la fenêtre de graphiques.

6.	Dans le champ Orientation, saisissez l'orientation de l'entité de texte. Si vous ne spécifiez pas une orientation, le texte s'affiche horizontalement à travers de la fenêtre.

7.	Dans le champ Type d'ancrage, spécifiez où le texte est positionné par rapport aux valeurs Nord et Est et sélectionnez une valeur de la liste.

8.	Dans le champ Couche, sélectionnez la couche à laquelle vous voulez ajouter du texte de la liste. Pour saisir du texte dans une nouvelle couche, saisissez un nouveau nom.

9.	Dans le champ Style de texte, sélectionnez le style de texte pour l'entité de texte de la liste. Si vous voulez le style de texte par défaut configuré pour la couche à laquelle vous ajoutez du texte, sélectionnez l'option <Par couche>.

10.	Cliquez sur Insérer pour ajouter le texte à la base de données et tenir le dialogue ouvert pour l'ajout du texte supplémentaire.

11.	Cliquez sur Fermer pour arrêter l'insertion du texte.


Créer un nouveau modèle de projet

 Vue d'ensemble et plus amples informations

1.	Effectuez l'une des choses suivantes:

  Cliquez l'outil Nouveau.

  Sélectionnez Fichier / Nouveau projet.

Le dialogue Nouveau projet s'affiche.

2.	Dans le groupe Nouveau, sélectionnez l'option Modèle. Le dossier Projet dans le groupe Détails change au dossier de modèles par défaut.

3.	Dans le champ Nom, saisissez un nom pour le modèle du projet.

4.	Dans le champ Modèle, sélectionnez un modèle de la liste et cliquez sur OK. Le dialogue Propriétés du projet s'affiche.

5.	Sélectionnez les propriétés du projet requises comme les défauts pour tout projet utilisant ce modèle et cliquez sur OK.

Un message s'affiche. Pour continuer de travailler dans le modèle, cliquez sur Oui. Pour fermer le modèle du projet, cliquez sur Non.

6.	Si vous sélectionnez Oui, vous pouvez continuer de travailler dans le modèle du projet. Pour terminer le travail dans le modèle du projet, sélectionnez Fichier / Fermer.

7.	Lorsque vous créez un nouveau projet, vous verrez le nouveau modèle dans la liste Modèles dans le dialogue Nouveau projet.


Créer un style de point

 Vue d'ensemble et plus amples informations

1.	Dans La Vue en plan, sélectionnez Editer / Styles. Le dialogue Styles CAD s'affiche.

2.	Dans la tabulation Styles de point, cliquez sur Nouvelle. Le dialogue Style de point s'affiche.

3.	Complétez le dialogue Créer style de point.

4.	Cliquez sur OK.

Les Style de point est ajouté à la base de données. Il apparaît dans le dialogue Styles CAD − tabulation Styles de point.


Ouvrir la fenêtre Propriétés

 Vue d'ensemble et plus amples informations

Effectuez l'une des choses suivantes:

  Sélectionnez Editer / Propriétés.

  Cliquez l'outil Editer les propriétés [image].

  Choisissez Propriétés du menu de raccourcis

  Double-cliquez sur une entité graphique.

  Appuyer sur Alt + Entrée.


Exporter un fichier de dictionnaire d'attributs (.ddf) au Trimble Survey Controller

 Vue d'ensemble et plus amples informations

NOTE: Vous ne pouvez transférer les fichiers .ddf que vers un périphérique exécutant Trimble Survey Controller de version 7.00 ou ultérieure.

1.	Sélectionnez Fichier / Exporter. Le dialogue Exporter s'affiche.

2.	Dans la tabulation Levé, sélectionnez l'option Fichier de dictionnaire d'attributs au Survey Controller (*.ddf) et cliquez sur OK. Le dialogue Sélectionner s'affiche.

3.	Sélectionnez le fichier que vous voulez transférer au logiciel Trimble Survey Controller et cliquez sur Ouvrir. Le dialogue Enregistrer s'affiche.

4.	Sélectionnez un périphérique Survey Controller et cliquez sur Ouvrir. Le dialogue Connexion s'affiche.

Lorsque la connexion au périphérique est effectuée, Mémoire principale (C:) apparaît dans la boîte. Si une carte PC est insérée actuellement dans le périphérique, Carte PC (A:) apparaît aussi.

NOTE: Si votre périphérique Trimble n'a pas une carte PC, ce dialogue n'est pas disponible.

5.	Effectuez l'une des choses suivantes:

  Sélectionnez l'option Mémoire principale (C:) pour stocker le fichier dans la mémoire principale du logiciel Trimble Survey Controller.

  Sélectionnez l'option Carte PC (A:) si vous avez une carte PC insérée et vous voulez stocker le fichier sur la carte PC.

6.	Cliquez sur Ouvrir. La mémoire principale ou la carte PC est lue. Les fichiers de dictionnaire de données existants s'affichent.

7.	Le nom interne du fichier .ddf sélectionné pour l'exportation s'affiche dans le champ Nom du fichier. C'est le nom que vous verrez dans  Trimble Survey Controller.

8.	Editez ce nom si nécessaire.

9.	Cliquez sur Enregistrer. Le dialogue Téléchargement s'affiche et la barre de progression indique le progrès du transfert.

Lorsque le transfert est terminé, le dialogue Téléchargement disparaît et vous retournez à la fenêtre de graphiques. Maintenant vous pouvez utiliser le fichier .ddf avec  Trimble Survey Controller.

NOTE: Lorsque vous transférez les données vers ou depuis  Survey Controller de version 7.60, utilisez le format de fichier .dc de version 7.50.


Faire glisser-déplacer un fichier dans un projet

Vue d'ensemble et plus amples informations
Pour faire glisser-déplacer un fichier dans un projet :

1.	Ouvrez le projet Trimble Geomatics Office auquel vous voulez transférer le(s) fichier(s).

2.	Ouvrez l'Explorateur de Microsoft Windows. Pour plus amples informations, référez-vous à votre documentation du système d'exploitation Microsoft Windows.

3.	Trouvez le dossier contenant le(s) fichier(s) à importer.

4.	Cliquez avec le bouton droit de la souris sur le(s) fichier(s) à importer.

5.	Faites glisser les fichiers sur la fenêtre de graphiques Trimble Geomatics Office.

6.	Relâchez le bouton droit de la souris pour voir le menu de raccourcis.

7.	Sélectionnez l'option de fichier appropriée.

8.	Cliquez avec le bouton gauche de la souris. Les fichiers sont importés.

Pour plus amples informations concernant les actions possible qui se produisent lors de l'importation, référez-vous à Importation − Vue d'ensemble. Les fichiers sont stockées dans le dossier Fichiers de données\Fichiers Trimble pour le projet.


Sélectionner les points par nom

1.	Choisissez Sélectionner / Points. Le dialogue Sélectionner points s'affiche.

2.	Sélectionnez la tabulation Général.

3.	Dans le champ Correspondance de nom, saisissez le nom du point que vous voulez sélectionner. Vous pouvez également utiliser des limites et des caractères génériques pour sélectionner des points multiples.

4.	Cliquez OK. Le(s) point(s) correspondant à votre nom sont sélectionnés. La barre d'état indique combien de points sont sélectionnés.

NOTE: Toutes les entités sélectionnées actuellement avant que vous utilisez le dialogue Sélectionner points ne sont pas conservées dans le groupe de sélection à moins que vous ne sélectionnez la case à cocher Ajouter à la sélection courante.


Sélectionner les points par qualité de point

1.	Choisissez Sélectionner / Points. Le dialogue Sélectionner points s'affiche.

2.	Sélectionnez la tabulation Général.

3.	Dans le champ Avec qualité de point, sélectionnez la combinaison de composant et qualité que vous voulez sélectionner. Vous pouvez sélectionner des entrées multiples de la liste.

4.	Cliquez sur OK.

Le(s) point(s) correspondant à votre sélection sont sélectionnés. La barre d'état indique combien de points sont sélectionnés.

NOTE: Toutes les points sélectionnés actuellement avant que vous utilisez le dialogue Sélectionner points ne sont pas conservées dans le groupe de sélection à moins que vous ne sélectionnez la case à cocher Ajouter à la sélection courante.


Sélectionner des observations d'un point sélectionné

1.	Dans la Vue de levé, choisissez Sélectionner / Observations. Le dialogue Sélectionner observations s'affiche.

2.	Dans le champ De point, saisissez le nom du point, ou saisissez une étendue pour utiliser des points multiples dans la sélection. Vous pouvez également utiliser le remplissage du champ et les caractères génériques.

3.	Cliquez sur OK.

Les observations des points correspondant à votre sélection sont sélectionnées. La barre d'état indique combien d'observations sont sélectionnées.

NOTE: Toutes les entités sélectionnées actuellement avant que vous utilisez le dialogue Sélectionner points ne sont pas conservées dans le groupe de sélection à moins que vous ne sélectionnez la case à cocher Ajouter à la sélection courante


Désactiver les observations

1.	De la Vue de levé, sélectionnez une observation.

2.	Effectuez l'une des choses suivantes:

  Ouvrez la fenêtre Propriétés. Cliquez sur l'icône Résumé [image] et sélectionnez l'option Désactiver du champ Etat.

  	Sélectionnez Editer / Activer/Désactiver observation.

NOTE: La désactivation des observations nécessitent un Recalcul pour mettre à jour les informations de levé. Lorsque vous déplacez un point, toute observation au point est désactivé automatiquement.


Importer les études du Trimble Survey Controller au Trimble Geomatics Office

1.	Effectuez l'une des choses suivantes:

  Sélectionnez Fichier / Importer.

  Cliquez sur l'outil Importer.

2.	Dans la tabulation Levé, sélectionnez l'option Périphériques de levé. Le dialogue Transfert de données s'affiche.

3.	Sélectionnez un périphérique Trimble Survey Controller.

4.	Cliquez sur Ouvrir. Le dialogue Connexion s'affiche. Lorsqu'un connexion est effectuée, le dialogue Ouvrir se met à jour pour afficher le périphérique sélectionné.

5.	Sélectionnez les fichiers que vous voulez transférer de l'une des options suivantes:

  l'option Mémoire principal (C:) pour les fichiers stocké dans la mémoire principale du Trimble Survey Controller.

  l'option Carte PC (A:) pour les fichiers stockés sur une Carte PC insérée dans le Survey Controller.

NOTE: Si votre périphérique Trimble n'a pas une carte PC, ce dialogue n'est pas disponible.

6.	Cliquez sur Ouvrir. Le dialogue Réception s'affiche et la barre de progression indique le progrès du transfert. Lorsque le transfert est terminé, le dialogue Réception disparaît et la connexion au périphérique est interrompue.


Masquer (ou afficher) une barre d'outils

1.	Effectuez l'une des choses suivantes:

  Cliquez avec le bouton droit de la souris dans la zone de la barre d'outils. Un menu de raccourcis s'affiche.

  Sélectionnez Affichage / Barres d'outils. Un menu s'affiche.

La liste des barres d'outils disponibles s'affiche. Chaque barre d'outils visible actuellement a une coche à côté de son nom.

2.	Sélectionnez la barre d'outils requise. La barre d'outils disparaît de (ou apparaît à nouveau dans) la fenêtre de graphiques Trimble Geomatics Office.

3.	Pour réafficher la barre d'outils, répétez les étapes ci-dessus.


Transférer les détails d'antenne au Trimble Survey Controller

1.	Sélectionnez Fichier / Exporter. Le dialogue Exporter s'affiche.

2.	Dans la tabulation Levé, sélectionnez l'option Détails d'antenne au Survey Controller. Le dialogue Transfert de données Enregistrer sous s'affiche.

3.	Sélectionnez un périphérique Trimble Survey Controller et cliquez sur Ouvrir. Le dialogue Connexion s'affiche. Lorsqu'une connexion est effectuée, le dialogue Enregistrer sous s'affiche.

4.	Effectuez l'une des choses suivantes:

  Sélectionnez l'option Mémoire principale (C:) pour stocker le fichier dans la mémoire principale du logiciel Trimble Survey Controller.

  Sélectionnez l'option Carte PC (A:) si vous avez une carte PC insérée et vous voulez stocker le fichier sur la carte PC.

NOTE: Si votre périphérique Survey Controller n'a pas une carte PC, ce dialogue n'est pas disponible.

5.	Cliquez sur Ouvrir. Le répertoire sélectionné est lu. Tous les fichiers Survey Controller dans le répertoire sont affichés. Le nom du projet Trimble Geomatics Office s'affiche dans le champ Nom du fichier. C'est le nom de l'étude du logiciel Trimble Survey Controller que sera exportée au logiciel Trimble Survey Controller. Vous pouvez éditer ce nom si nécessaire.

6.	Cliquez sur Enregistrer. Le dialogue Téléchargement s'affiche et la barre de progression indique le progrès du transfert.

Le fichier Antenna.ini (qui se trouve dans le dossier \Program Files\Common Files\Trimble\Config) est transféré au logiciel Trimble Survey Controller.

Lorsque le transfert est terminé, le dialogue Téléchargement disparaît et la connexion au périphérique est interrompue.


Exporter les notes d'attribut

1.	Sélectionnez Fichier / Exporter. Le dialogue Exporter s'affiche.

2.	Dans la tabulation CAD / ASCII, sélectionnez l'option Notes d'attribut du fichier DC (Survey Controller V4) et cliquez sur OK. Le dialogue Enregistrer sous s'affiche.

3.	Saisissez le nom du fichier.

4.	Cliquez sur Enregistrer.


Afficher des informations à propos du Trimble Geomatics Office

1.	Du menu Aide, sélectionnez la commande A propos de Trimble Geomatics Office. Le dialogue A propos de Trimble Geomatics s'affiche.

2.	Cliquez sur OK.


Inverser la direction d'une observation

1.	De la Vue de levé, sélectionnez l'observation que vous voulez appliquer dans la direction opposée.

2.	Sélectionnez Editer / Inverser flux d'observation.

3.	La direction de l'observation est inversée. Lorsque l'observation est appliquée dans la direction opposée, les coordonnées calculées et les qualités du point peuvent changer.

ATTENTION: Si le point à partir duquel vous voulez faire provenir l'observation n'a pas une position connue (tel qu'un point observé RTK), l'observation n'est pas appliquée.

NOTE: Il est impossible d'inverser les observations terrestres collectées en utilisant une station totale.


Afficher les erreurs et les avertissements

1.	Effectuez l'une des choses suivantes:

 Dans la Vue de levé, double-cliquez sur un point ou une observation avec un indicateur d'avertissement.

  Double-cliquez sur l'icône d'indicateur dans la barre d'état pour sélectionner tous les points et les observations avec des indicateurs d'avertissement.

La fenêtre Propriétés s'affiche avec l'entité/les entités sélectionnées.

2.	Dans la fenêtre Propriétés, sélectionnez le point ou l'observation que vous voulez afficher (si vous avez plus d'une entité dans votre sélection).

3.	Pour afficher la page d'Avertissement, cliquez sur le bouton du menu Avertissement [image].

Le message affiche le type d'erreur que s'est produit.

4.	Pour afficher des détails supplémentaires d'un point, sélectionnez l'outil Rapport de dérivation du point [image].

Le Rapport de dérivation du point s'affiche.


Transférer les fichiers de grille nationaux UK (*.pgf) au Trimble Survey Controller

NOTE: Vous ne pouvez transférer des fichier .pgf qu'au Trimble Survey Controller de version 7.5 ou ultérieur:

1.	Démarrez l'utilitaire Trimble Data Transfer. Effectuez l'une des choses suivantes:

  Sélectionnez Utilitaires / Transfert de données lorsqu'aucun projet n'est ouvert.

  Sélectionnez Démarrer / Programs / Trimble / Utilitaires Trimble / Transfert de données à tout moment.

Le dialogue Transfert de données s'affiche.

2.	Sélectionnez la tabulation Envoyer.

3.	De la liste Périphérique, sélectionnez le périphérique Survey Controller auquel vous voulez exporter des fichiers. L'utilitaire Trimble Data Transfer essaie de connecter au périphérique sélectionné. Lorsque la connexion est réussie, le câble dans l'icône en haut à droite indique 'Connecté à <périphérique>', <périphérique> étant le type de périphérique (tel qu'un Survey Controller).

4.	Cliquez sur Ajouter. Le dialogue Ouvrir s'affiche. La liste Rechercher dans indique le dernier dossier à partir duquel vous avez sélectionné des fichiers pour ce périphérique, et la liste principale indique les dossiers et les fichiers pour ce dossier.

5.	De la liste Fichier de type, sélectionnez l'option Fichiers de grille nationaux UK (*.PGF).

6.	Utilisez la liste Rechercher dans et la boîte de liste pour changer le dossier et le lecteur sélectionnés et sélectionnez les fichiers à transférer.

Pour transférer les fichiers .pgf existants fournis avec  Trimble Geomatics Office Sélectionnez le fichier North.pgf ou East.pgf du Program Files\Common Files\Trimble\Geodata.

7.	Vous pouvez envoyer les fichiers à la mémoire principale ou à une carte PC s'il y en a installé. Sélectionnez la destination de la liste Destination.

8.	Vérifiez que le champ Format du fichier est configuré sur Fichier d'étude v.7.50, sélectionnez cette option de la liste.

9.	Cliquez sur Sélectionner. Le dialogue Ouvrir ferme et vous retournez au dialogue Transfert de données. Les fichiers sélectionnés s'affichent dans la liste Fichiers à envoyer.

10.	Cliquez sur Transférer. Le dialogue Conversion s'affiche pendant la conversion des fichiers du format de fichier du périphérique. Lorsque tous les fichiers sont convertis, le dialogue Envoi s'affiche, indiquant le progrès de l'exportation des fichiers sélectionnés.

Lorsque le transfert est terminé, le dialogue Transfert terminé s'affiche indiquant les résultats du transfert.

NOTE: Lorsque vous transférez les données vers ou depuis  Survey Controller de version 7.60, utilisez le format de fichier .dc de version 7.50.


Calibrer utilisant les coordonnées de calibration enregistrées et les coordonnées ajustées (dérivées du module Ajustement de ré
Calibrer utilisant les coordonnées de calibration enregistrées et les coordonnées ajustées (dérivées du module Ajustement de réseau)

1.	Après avoir complété un ajustement contraint imperceptiblement et l'établissement des coordonnées WGS-84 sur chacun des points dans l'ajustement de réseau, enregistrez ces coordonnées pour une utilisation comme des coordonnées GPS observées pour une calibration du site. Les coordonnées de calibration consistent en des Latitudes et Longitudes WGS-84 et les Hauteurs d'ellipsoïde.

Pour enregistrer les coordonnées de calibration:

  Sélectionnez Ajustement / Coordonnées de calibration / Enregistrer.

Le dialogue Enregistrer coordonnées de calibration s'affiche.

NOTE: Cette commande n'est disponible que si WGS-84 est sélectionné comme votre datum d'ajustement et si seulement des observations GPS étaient comprises dans l'ajustement précédent.

NOTE: Si les paramètres de transformation d'ajustement étaient créés lors d'un ajustement, ils sont écartés lorsque vous effectuez une calibration du site GPS.  Trimble Geomatics Office ne permet que l'un ou l'autre.

2.	Sélectionnez l'une des options suivantes:

  L'option Créer coordonnées sous des points existants

  L'option Créer coordonnées sous des nouveaux points avec suffixe

3.	Cliquez sur OK. Les coordonnées de calibration sont enregistrées.

NOTE: Une coordonnée WGS-84 est créée pour chaque point dans l'ajustement. Vous pouvez afficher ces coordonnées dans la fenêtre Propriétés. Cette coordonnée ne peut pas être éditée.

4.	Effectuez un ajustement contraint dans le datum du projet ( référez-vous aux rubriques d'aide d'Ajustement de réseau appropriées pour comment le faire). Les valeurs ajustées sont enregistrées dans la base de données.

NOTE: Une coordonnée de grille ajustée est créée pour chaque point non contraint dans l'ajustement. Cette coordonnée peut être affichée dans la fenêtre Propriétés.

5.	Effectuez une calibration du site GPS utilisant les paires de points dérivées de l'Ajustement de réseau. Pour ce faire, sélectionnez Levé / Calibration du site GPS. Le dialogue Calibration du site GPS s'affiche.

6.	Dans le groupe Composants de calibration, sélectionnez les paramètres de calibration que vous voulez calculer.

7.	Cliquez sur Liste points. Le dialogue Liste points s'affiche.

8.	Utilisez le dialogue Liste points pour sélectionner vos paires de points (point GPS et point de grille) pour la calibration. Effectuez l'une des choses suivantes:

  Utilisez le remplissage du champ ou saisissez le nom de la coordonnée de calibration enregistrée dans le champ Point GPS.

  Utilisez le remplissage du champ ou saisissez le nom de la coordonnée ajustée correspondante dans le champ Point de grille.

Si la coordonnée contrainte au minimum a été enregistrée sous le même point, le champ de point de grille sera rempli automatiquement. Lorsque le point GPS est saisi, les valeurs indiquées sont à partir des coordonnées de calibration enregistrées. Lorsque le point de grille est saisi, les valeurs ajustées sont sélectionnées.

9.	Répétez l'étape ci-dessus pour sélectionner toutes les paires de points requises dans la calibration.

10.	Lorsque vous avez sélectionné toutes les paires de points, cliquez sur Fermer pour retourner au dialogue Calibration du site GPS.

11.	Cliquez sur Calculer pour calculer les paramètres de calibration. Les résultats de la calibration s'affichent dans le groupe Résumé de calibration.

12.	Cliquez sur OK pour appliquer la calibration au projet. Le système de coordonnées est mis à jour avec les paramètres de calibration. Maintenant vous pouvez transférer la calibration au logiciel Trimble Survey Controller.


Rapporter et exporter des événements

Vous pouvez définir un rapport ou un format d'exportation pour enregistrer l'heure d'enregistrement d'un événement par le récepteur.

Pour rapporter sur des événements, créez un rapport de base de données personnalisé:

1.	Sélectionnez les points avec des enregistrements d'événement dans  Trimble Geomatics Office. Pour ce faire

a. Choisissez Sélectionner / Points.  Le dialogue Sélectionner points s'affiche

b. Sélectionnez la tabulation GPS

c.  De la liste Méthode sur le terrain, sélectionnez Evénements

d. Cliquez sur OK

Tous les points ayant des enregistrements d'événement sont sélectionnés

2.	Sélectionnez Rapports / Rapports supplémentaires. Le dialogue Rapports supplémentaires s'affiche.

3.	Effectuez l'une des choses suivantes:

  Cliquez sur Editer pour modifier un format de rapport existant

  Cliquez sur Nouveau pour créer un nouveau format de rapport.

Le dialogue Définir le format du rapport s'affiche.

4.	Dans le champ Corps du format, cliquez avec le bouton droit de la souris pour accéder au menu de raccourcis et puis sélectionnez Expert / Sous-enregistrements / Obs point / Heure.

Pour exporter des événements pour les définitions d'exportation personnalisées:

1.	Sélectionnez Fichier/Exporter. Le dialogue Exporter s'affiche

2.	Dans la tabulation Personnalisé, effectuez l'une des choses suivantes:

  Cliquez sur Editer format pour modifier un format existant.

  Cliquez sur Nouveau format pour créer un nouveau format de rapport.

Le dialogue Définir le format d'exportation ASCII s'affiche.

3.	Dans le champ Corps du format, cliquez avec le bouton droit de la souris pour accéder au menu de raccourcis et sélectionnez Expert / Sous-enregistrements / Obs point / Heure.


Mise en forme de l'heure

Le format d'affichage de l'heure typique pour les événements est comme suit: 09:12:55 8 Fév 2000

Pour modifier cette mise en forme, cliquez avec le bouton droit de la souris à côté du code de champ pour accéder au menu de raccourcis et puis sélectionnez Expert / Plus / Mises en forme. Sélectionnez un exemple de mise en forme de l'heure ou créez votre propre mise en forme. Pour plus amples informations, référez vous à Mises en forme de l'heure .

NOTE: Pour afficher les heures des événements à la milliseconde plus près, ajoutez la mise en forme de l'heure suivante au code de champ, <H:mm:ss.nnn>, où nnn est le nombre de secondes décimales. Par exemple, [Heure d'obs. point:<H:mm:ss.nnn>]. Pour augmenter la précision de l'heure, augmentez le nombre de secondes décimales. Par exemple, nnnn effectue la sortie les premières 4 secondes décimales.


Copier les données d'observation entre les projets

Pour copier les données d'observation entre les projets du logiciel Trimble Geomatics Office, utilisez le format Trimble Data Exchange. Vous ne pouvez pas utiliser la fonction copier et coller.

Pour copier les données d'observation entre les projets:

1.	Dans votre projet Trimble Geomatics Office courant, sélectionnez les données d'observation à copier.

2.	Exportez les données de votre projet courant en utilisant le format Trimble Data Exchange:

a.	Sélectionnez Fichier / Exporter. Le dialogue Exporter s'affiche.

b.	Dans la tabulation Levé, sélectionnez le format Trimble Data Exchange et cliquez sur OK. Le dialogue Enregistrer sous s'affiche.

c.	Enregistrez le fichier.

3.	Créez un nouveau projet dans  Trimble Geomatics Office.

4.	Importez les données d'observation au nouveau projet utilisant le format Trimble Data Exchange:

a.	Sélectionnez Fichier / Importer. Le dialogue Importer s'affiche.

b.	Dans la tabulation Levé, sélectionnez le format Trimble Data Exchange et cliquez sur OK. Le dialogue Ouvrir s'affiche.

c.	Sélectionnez le fichier.

Les données d'observation sont importées dans votre projet Trimble Geomatics Office.

NOTE: Si vous voulez créer une copie de toutes les observations et tous les points dans le projet, créez une copie du projet Trimble Geomatics Office


Importer les données terrestres manuellement

Pour importer les données terrestres manuellement au logiciel Trimble Geomatics Office, utilisez le format Softdesk Fieldbook. Ce format est développé par Autodesk.

Créez un fichier sous le format Softdesk Fieldbook, utilisant vos données terrestres:

1.	Importez le fichier au logiciel Trimble Geomatics Office:

a.	Sélectionnez Fichier / Importer. Le dialogue Importer s'affiche.

b.	Dans la tabulation Levé, sélectionnez les fichiers d'enregistreur de données tiers. Le dialogue Ouvrir s'affiche.

c.	Sélectionnez le fichier que vous voulez importer.

d.	Dans le champ Type de fichiers, sélectionnez les fichiers du format Autodesk FBK fieldbook (*.FBK) et cliquez sur OK.

Le fichier est importé dans votre projet Trimble Geomatics Office.

NOTE: Le format Autodesk Softdesk Fieldbook est décrit ci-dessous. Ce format est pris des fichiers d'aide du programme AutoCAD Land Development Desktop.

Considérez le suivant:

  Toute ligne dans le format que ne commence pas avec l'une des commandes reconnues est ignorée.

  Les éléments entre parenthèses () sont facultatifs.

   Trimble Geomatics Office requiert que vous utilisez un numéro de point.

  La description de point (descript) doit être comprise entre double guillemets (") car elle fait partie de la définition du fichier de format Fieldbook. Il signifie que des espaces peuvent être utilisés dans la description du point.


Commandes de création de point

AD (point) [angle] [distance] (descript)

AD [AV] (point) [angle] [distance] [angle vert] (descript)

AD [DV] (point) [angle] [distance] [distance vert] (descript)

BD (point) [gisement] [quadrant] [distance] (descript)

BD [AV] (point) [gisement] [quadrant] [distance] [angle vert] (descript)

BD [AV] (point) [gisement] [quadrant] [distance] [dist vert] (descript)

DD (point) [angle] [distance] [angle vert] (descript)

DD [AV] (point) [angle] [distance] [angle vert] (descript)

DD [DV] (point) [angle] [distance] [distance vert] (descript)

F1/F2 (point) [angle] [distance] (descript)

F1/F2 [AV] (point) [angle] [distance] [angle vert] (descript)

F1/F2 [DV] (point) [angle] [distance] [distance vert] (descript)

NE (point) [Nord] [Est] (descript)

NEZ (point) [Nord] [Est] (elév) (descript)

NE SS (point) [Nord] [Est] (eléev) (descript)

PRISME [hauteur]

DEPORT PT (point) [déport] (en avant)

STADIA (point) [angle] [distance] [tige] (angle vert) (descript)

ZD (point) [azimut] [distance] (descript)

ZD [AV] (point) [azimut] [distance] [angle vert] (descript)

ZD [DV] (point) [azimut] [distance] [distance vert] (descript)


Commandes de l'emplacement du point

AZ [point 1] [point 2] [azimut]

AV [point] (orientation)

STN [point] (hauteur inst.) (descript)


Commandes de correction d'équipement

ANGLES [DROIT, GAUCHE] [ZENITH, NADIR, HORIZ]

ATMOS [ACTIVE, DESACTIVE]

CR [ACTIVE, DESACTIVE]

DEPORT EDM  [déport]

ANGLE HORIZ [DROIT, GAUCHE]

PRESS [pression] [POUCE, MBAR, MM]

FE [facteur]

CONSTANTE PRISME  [constante]

DEPORT PRISME  [déport]

TEMP [température] [F, C, K]

UNITES [METRE, PIED [DMS, DEGDEC, GRAD, MILS, RADIANS]

ANGLE VERT [ZENITH, NADIR, HORIZ]


Ajouter les coordonnées dans la fenêtre Propriétés

 Trimble Geomatics Office vous permet d'ajouter les coordonnées d'un point dans la fenêtre Propriétés, utilisant l'outil Ajouter coordonnée [image] dans la page Résumé.

Pour ce faire:

1.	Double-cliquez sur le point dans le projet. La fenêtre Propriétés s'affiche.

2.	Dans le page Résumé, sélectionnez [image] et puis sélectionnez le type de coordonnée qu'il faut ajouter; c'est à dire, de grille, locale ou WGS 84.

Dans la structure d'arborescence de la fenêtre Propriétés, une nouvelle coordonnée apparaît sous le point sélectionné. Un préfixe est ajouté au nom pour indiquer si la coordonnée est une coordonné de grille(G), locale (L) ou WGS-84 (W). La page Résumé de la nouvelle coordonnée affiche tous les champs comme nuls.

NOTE: Vous ne pouvez saisir qu'une coordonnée WGS-84 et une coordonnée de grille ou locale pour chaque point. Lorsque vous sélectionnez une coordonnée WGS-84, cette option n'est plus disponible dans la liste Ajouter coordonnée. Si un point a déjà une coordonnée WGS-84 et une coordonnée de grille ou locale, le bouton Ajouter coordonnée n'est plus disponible.

3.	Pour compléter les champs Nord, Est, Elévation et Hauteur de la coordonnée, cliquez sur chaque champ et puis cliquez sur le bouton Insérer valeur courante [image]. Chaque champ est rempli avec la position de point et la qualité courante. Vous éditez ces informations comme nécessaire.

ASTUCE: N'ajoutez une coordonnée que si vous êtes sûr que vous voulez utiliser cette coordonnée au lieu des observations. Les recalculs utilisent les coordonnées entrées au clavier plutôt que les observations de la même qualité. Ainsi, il est assuré qu'un point du projet est utilisé plutôt qu'une observation d'implantation.

NOTE: Des coordonnées supplémentaires peuvent être générées par  Trimble Geomatics Office et stockées pour un point. Ces coordonnées peuvent comprendre:

  Des coordonnées dérivées d'un ajustement

  Des coordonnées WGS-84 enregistrées pour une calibration


Editer les coordonnées dans la fenêtre Propriétés

 Vue d'ensemble et plus amples informations

 Trimble Geomatics Office vous permet d'éditer les coordonnées d'un point dans la fenêtre Propriétés.

Pour ce faire:

1.	Double-cliquez sur le point dans le projet. La fenêtre Propriétés s'affiche.

2.	Dans la structure d'arborescence de la fenêtre Propriétés, développez la structure d'arborescence du point et sélectionnez les coordonnées entrées au clavier  (WGS-84 ou de grille/locale) que vous voulez éditer.

3.	Editez les composants de la coordonnée et la qualité de la coordonnée.

NOTE: Pour que le champ de coordonnées puisse être édité, le paramètre d'affichage de coordonnée et le paramètre de Type de coordonnée doivent être identiques.

4.	Après l'édition de la coordonnée, sélectionnez Levé / Recalculer pour effectuer le Recalcul.

Pour plus amples informations concernant l'ajout et l'édition des coordonnées dans la fenêtre Propriétés, référez-vous à la rubrique fenêtre Propriétés - Coordonnées .


Importer les fichiers de niveaux numériques

Le dialogue Importation de niveau numérique s'affiche lorsque vous importez des Fichiers de niveaux numériques (*.dat, *.raw) ou vous téléchargez les fichiers utilisant un niveau numérique DiNi (utilisant des périphériques de levé). Pour ce faire:

1.	Sélectionnez Fichier / Importer. Le dialogue Importer s'affiche.

2.	Dans la tabulation Levé, sélectionnez Fichiers de niveaux numériques (*.dat, *.raw) et cliquez sur OK. Le dialogue Ouvrir s'affiche.

3.	Sélectionnez le Fichier de niveau numérique. Le dialogue Importation de niveau numérique s'affiche.

4.	Dans le dialogue Importation de niveau numérique, définissez les points de station en sélectionnant la case à cocher dans la colonne Points de station.

Les élévations delta sont calculées entre les points de station. Lorsque les points de station sont importés, ils seront fusionnés avec d'autres points du même nom. Assurez-vous que les noms de point sont uniques. Vous pouvez éditer les noms de point dans le dialogue Importation de niveau numérique.

NOTE: Les points qui ne sont pas définis comme points de station ne seront pas importés.

5. 	Dans le dialogue Importation de niveau numérique, saisissez une élévation pour les points de début. Cette élévation est utilisée pour calculer les élévations des autres points. Pour éditer les élévations de point de début, cliquez sur tout point dans la boucle. Dans le groupe Elévation, éditez l'élévation et la qualité et cliquez sur Entrée. L'élévation est calculée pour tous les points de station. Utilisez ces élévations calculées pour la vérification avec des valeurs connues et validez les données.

NOTE: Les élévations avec le symbole d'élévations sont les seules élévations importées. Tous les autres points de station auront des élévation delta importées Leurs élévations seront calculées lors du recalcul.

ASTUCE: Toute modification effectuée dans le dialogue Importation de niveau numérique n'aura un effet qu'aux données importées. Les modifications n'auront pas d'effet au Fichier de niveau numérique (*.dat, *.raw).

6.	Cliquez sur OK. Le Fichier de niveau numérique est importé.

NOTE: Il se peut que les observations importées n'auront pas des coordonnées horizontales, donc elles ne s'afficheront pas dans la fenêtre de graphiques. Pour afficher les données de niveau dans la fenêtre Propriétés, sélectionnez les observations de niveaux à l'aide d'un des outils de sélection et ouvrez la fenêtre de Propriétés. Pour ce faire, effectuez l'une des choses suivantes:

  Sélectionnez Editer / Propriétés

  Sélectionnez l'outil de la fenêtre Propriétés [image] dans la barre d'outils.

ASTUCE: Lorsque vous collectez les données de niveaux numériques, Trimble vous recommande de stocker les points que vous voulez importer au logiciel Trimble Geomatics Office avec un préfixe ou suffixe unique. Par exemple, stockez tous les points que vous voulez utiliser comme des points de station avec le préfixe STN. Ainsi, il sera plus facile de sélectionner des points utilisant le bouton Sélectionner par filtre dans le dialogue Importation de niveau numérique.


Afficher une carte d'arrière plan dans  Trimble Geomatics Office

 Vue d'ensemble et plus amples informations

Pour afficher une carte d'arrière plan dans  Trimble Geomatics Office:

1.	Sélectionnez Affichage / Options. Le dialogue Options d'affichage s'affiche.

2.	Dans la tabulation Carte d'arrière plan, cliquez sur Ajouter. Le dialogue Ajouter s'affiche.

3.	Sélectionnez le fichier de Carte d'arrière plan et cliquez sur OK.

NOTE:  Trimble Geomatics Office peut afficher des fichiers d'arrière plan dans la fenêtre de graphiques. Vous pouvez importer des fichiers au format Drawing Exchange Format (.dxf), Windows Bitmap (.bmp) ou Tagged Image File Format (.tif) à afficher. Les fichiers Windows Bitmap (.bmp) et Tagged Image File Format (.tif) doivent être géoréférencés afin d'être affichés correctement. Le format de fichier ESRI World est utilisé pour géoréférencer les fichiers d'arrière plan. Ces fichiers doivent se trouver dans le même dossier que les fichiers d'arrière plan correspondants.

Le fichier Carte d'arrière plan s'affiche dans  Trimble Geomatics Office.

Pour plus amples informations concernant l'utilisation de la Carte d'arrière plan dans  Trimble Geomatics Office, référez-vous à Options d'affichage − Carte d'arrière plan.


Afficher les lignes de grille dans  Trimble Geomatics Office

1.	Sélectionnez Affichage / Options. Le dialogue Options d'affichage s'affiche.

2.	Dans la tabulation Lignes de grille, sélectionnez la case à cocher Afficher lignes de grille. Les options de ce groupe deviennent disponibles.

3.	Sélectionnez l'option Nombre fixe de lignes de grille ou Intervalle de grille, et saisissez les valeurs appropriées dans chaque champ.

4.	Sélectionnez le type de ligne de grille et la couleur.

5.	Si vous voulez étiqueter les lignes de grille, sélectionnez la case à cocher Etiqueter lignes de grille.

6.	Cliquez sur OK.


Sélectionner des lignes de base indépendantes

Un groupe de lignes de base indépendantes consiste en le nombre minimum de lignes de base requises pour connecter toutes les occupations de point dans une session GPS sur le terrain spécifique. Pour tout groupe d'occupations de point GPS simultanées, il y a n – 1 lignes de base indépendantes. Par exemple, dans une session sur le terrain avec cinq récepteurs, il y a 10 lignes de base possibles, mais seulement 4 lignes de base indépendantes. Dans le diagramme ci-dessous, les lignes de base indépendantes sont représentées par des lignes à tirets.

Le concept d'un groupe de lignes de base indépendantes applique en principe aux levés réalisés utilisant les techniques statique et statique rapide, et est particulièrement important pour des projets qui incluent un ajustement de réseau.

L'utilisation des groupes de lignes de base indépendantes assure qu'un ajustement de réseau n'est pas influencé par un sens de redondance artificiel (Degrés de liberté augmentées), que pourraient résulter en une sous-estimation de l'erreur dans les coordonnées ajustées.

Les lignes de base dérivées des sessions GPS sur le terrain utilisant plus de deux récepteurs peuvent être formées dans les groupes de lignes de base indépendantes multiples. Le choix des lignes de base comprises dans le groupe indépendant peut être basé sur la qualité des solutions de lignes de base, ou sur la réalisation d'une géométrie de réseau souhaitée.

Pour spécifier un groupe de lignes de base indépendantes, vous pouvez effectuez l'une des choses suivantes:

  Sélectionnez et traitez seulement un groupe de lignes de base indépendant à partir de toutes les lignes de base potentielles pour une session GPS sur le terrain.

  Traitez toutes les lignes de base potentielles pour une session GPS sur le terrain et sélectionnez un groupe indépendant au temps d'enregistrer les résultats dans le projet.

 Traitez toutes les lignes de base potentielles pour une session GPS sur le terrain et enregistrez tous les résultats dans le projet, puis supprimez des lignes de base dépendantes hors de considération par l'ajustement de réseau:

Pour ce faire, utilisez l'outil d'affichage de Ligne de base dépendante [image] pour lier les observations dans la fenêtre de graphiques et Timeline. Lorsque vous sélectionnez cet outil,  n'affiche que les lignes de base post traitées dans l'affichage Timeline courant, et les lignes de base post traitées observées au même temps. Toutes les autres lignes de base post traitées dans la Vue de levé sont masquées. Utilisez l'outil Affichage de ligne de base dépendante pour afficher les lignes de base observées au même temps, et puis empêcher l'utilisation des lignes de base dépendantes dans un ajustement.

Pour ce faire, effectuez l'une des choses suivantes:

  Dans la fenêtre Propriétés, changez l'état des lignes de base dépendantes à Désactivé.

 Dans la fenêtre Propriétés, changez l'état des lignes de base dépendantes à Activer comme vérification.

 Dans la fenêtre Propriétés, effacez la case à cocher Utiliser dans Ajustement de réseau pour les lignes de base dépendantes.

  Utilisez le dialogue Observations pour effacer la case à cocher Utiliser pour les lignes de base dépendantes.

La meilleure façon pour effectuer ces opérations est d'uitliser le dialogue Edition multiple. Par exemple, pour changer l'état des lignes de base dépendantes à Désactivé:

1.	Sélectionner l'outil Affichage de ligne de base dépendante.

2.	Sélectionnez les lignes de base dépendantes dans la fenêtre de graphiques.

3.	Sélectionnez Editer / Edition multiple. Le dialogue Edition multiple s'affiche.

4.	Sélectionnez la case à cocher Configurer état d'observation sur.

5.	De la liste sélectionnez Désactivé et cliquez sur OK.

Les observations sélectionnées auront un état désactivé maintenant.


Importer les fichiers Trimble 3600/5600 GDM

NOTE: Lorsque les données de levé sont comprises dans un fichier area ainsi que job, importez le fichier Area en premier.

1.	Sélectionnez Fichier / Importer. Le dialogue Importer s’affiche.

2.	Dans la tabulation Levé, sélectionnez Fichiers Trimble 3600/5600 GDM (*.job, *.are) et puis cliquez sur Options. Le dialogue Paramètres s’affiche.

3.	Spécifiez les paramètres pour l’unité de contrôle Trimble 3600/5600 GDM et les paramètres de points en double.

4.	Cliquez sur OK pour accepter toute modification et retourner au dialogue Importer.

5.	Cliquez sur OK. Le dialogue Ouvrir s’affiche.

6.	Sélectionnez le fichier à importer et cliquez sur Ouvrir.

NOTE: Trimble Geomatics Office supprose que tous les fichiers 3600/5600 GDM sont définis commes un systèmes de coordonnées à facteur d’échelle seul. Pour plus amples informations, référez-vous à Facteur d’échelle seul.

Voir aussi


Exporter à un fichier Trimble 3600/5600 GDM Area (*.are)

1.	Sélectionnez Fichier / Exporter. Le dialogue Exporter s’affiche.

2.	Dans la tabulation Levé, sélectionnez Fichiers Trimble 3600/5600 GDM (*.job, *.are) et puis cliquez sur Options. Le dialogue Paramètres s’affiche.

3.	Spécifiez les paramètres pour l’unité de contrôle Trimble 3600/5600 GDM.

4.	Cliquez sur OK pour accepter toute modification et retourner au dialogue Exporter.

5.	Cliquez sur OK. Le dialogue Enregistrer sous s’affiche.

6.	Spécifiez un nom de fichier et un emplacement et cliquez sur Enregistrer. (Par défaut le nom du projet détermine le nom du fichier).

NOTE: Les coordonnées exportées dans fichier Area utilisent le système de coordonnées défini dans le projet Trimble Geomatics Office.

Voir aussi


Exporter un fichier Pcode (*.pco) vers un instrument Trimble 3600/5600

1.	Sélectionnez Fichier / Exporter. Le dialogue Exporter s’affiche.

2.	Dans la tabulation Levé, sélectionnez Fichier Pcode au Trimble 3600/5600 GDM (*.pco) et cliquez sur OK.

Le dialogue Sélectionner s’affiche.

3.	Sélectionnez un fichier Pcode et cliquez sur Ouvrir. Le dialogue Transfert de données enregistrer sous s’affiche.

4.	Sélectionnez un périphérique Trimble 3600/5600 GDM et cliquez sur Ouvrir. Le dialogue Connexion s’affiche. Lorsqu’une connexion est effectuée le dialogue Enregistrer s’affiche.

5.	Cliquez sur Enregistrer. Un dialogue indique la progression du transfert de fichier. Lorsque le transfert est terminé la connexion au périphérique est interrompue.

Voir aussi


Importer - Vue d'ensemble

Plus amples informations

Comment ...	Dialogue	Voir aussi

Vue d'ensemble

Vous pouvez importer des données de divers formats de fichier dans un projet de Trimble Geomatics Office. Par exemple, vous pouvez importer des fichiers contenant des points de contrôle, fichiers de données brutes GPS (*.dat), fichiers de Survey Controller (*.dc), et les formats personnalisés.

Pour une description des formats d’importation pris en charge, référez-vous au dialogue Importer.

Vous pouvez importer des données à partir :

  d’un périphérique exécutant Trimble Survey Controller

  d’un récepteur GPS de Trimble

  d’un fichier sur un ordinateur

  d’une station totale

  d’un niveau numérique

Lorsqu’un fichier est importé, les points qu’il contient sont ajoutés à la base de données. Si un point importé a le même nom qu’un point existant de la base de données, la façon dont ils sont fusionnés dépend de l’option configurée dans le dialogue Options de points en double.

Ce qui se passe lorsque vous importez un fichier dépend du type de fichier. Cependant, pour tous les types de fichier, un recalcul est effectué automatiquement et un rapport d’importation (dénommé <nom du fichier importé>.html) est créé et enregistré dans le dossier Rapports du projet.

Toutes les entités provenant d’un même fichier sont enregistrées dans un groupe de sélection. Dans le cas de fichiers .dc, le groupe de sélection est nommé conformément au nom interne de l’étude du fichier. Pour les autres types de fichier, le nom donné au groupe de sélection est celui du fichier d’importation. Pour la gestion des groupes de sélection, utilisez la commande Sélectionner / Groupes de sélection.

ASTUCE : Pour afficher uniquement des données dans un groupe de sélection, utilisez l’option N’afficher que les données du groupe de sélection dans le dialogue Filtres d’affichage.

ASTUCE : Pour inverser l’opération d’importation, utilisez la commande Annuler. Si vous avez fermé le projet après l’importation, utilisez l’option De source dans le dialogue Sélectionner points / tabulation Général et supprimez la sélection pour annuler l’opération d’importation.

NOTE : Si vous importez un fichier plus d’une fois dans le même projet, le fichier est ajouté au dossier approprié, avec un tiret et un chiffre à la fin du nom de fichier. Par exemple, si vous importez deux fois un fichier de données brutes GPS de Trimble (.dat) appelé MonFichierGPS, un dossier dénommé MonFichierGPS est alors créé (lors de la première importation du fichier). Dans ce dossier, on trouvera MonFichierGPS.dat et MonFichierGPS-1.dat. Les données en double sont identifiées au moment de l’importation et ne sont pas ajoutées à la base de données.


Dialogue Importer

Utilisez ce dialogue pour importer des données dans votre projet.

Pour y accéder, effectuez l’une des opérations suivantes :

  Sélectionnez Fichier / Importer.

  Cliquez sur le bouton Importer [image] de la barre d’outils standard.

Les formats d’importation sont organisés sous trois tabulations :

  Levé

  CAD / ASCII

  Personnalisé

Nouveau format

Ce bouton devient disponible lorsque la tabulation Personnalisé est sélectionnée.

Pour créer un nouveau format personnalisé, cliquez sur Nouveau format.

Le dialogue Définir le format d’importation ASCII s’affiche.

Utilisez ce dialogue pour définir le format d’importation.

Editer le format

Ce bouton devient disponible lorsque la tabulation Personnalisé est sélectionnée.

Cliquez sur Editer le format pour modifier les formats d’importation ASCII existants.

Le dialogue Définir le format d’importation ASCII s’affiche.

Utilisez ce dialogue pour éditer le format d’importation.

Supprimer format

Ce bouton devient disponible lorsque la tabulation Personnalisé est sélectionnée.

Cliquez sur Supprimer pour supprimer un format d’importation personnalisé. Il vous est demandé de confirmer votre action.

Pour confirmer la suppression, cliquez sur Oui.

Personnaliser

Utiliser le bouton Personnaliser pour sélectionner la liste des options d’importation affichées dans le dialogue Importer. Sélectionnez les options d’importation requises de la liste.

Vous pouvez ajouter un filtre externe du dialogue Personnaliser. Les filtres externes sont des programmes séparés pouvant être écrits afin d’intégrer avec Trimble Geomatics Office. Tout filtre externe écrit pour une utilisation dans Trimble Geomatics Office doit se conformer aux exigences spécifiques s’il faut l’ajouter au système. Seulement les filtres installés actuellement pour la tabulation sélectionnée dans le dialogue Importer s’affichent.

Tout filtre de programme doit être compilé comme un fichier Dynamically Linked Library (*.dll) (Bibliothèque liée dynamiquement) avec des fonctions exportées spécifiques et doit être créé comme un objet COM (Module d’objet de composant).

Dès que vous sélectionnez le fichier approprié, Trimble Geomatics Office enregistre l’objet COM de filtre dans l’ordinateur et vérifie que l’interface .dll se conforme aux exigences du système. Si ce ne réussit pas, le système indique qu’il n’est pas possible d’enregistrer le fichier. SI le filtre est installé avec succès, la description courte de filtre est listée dans le dialogue.

Pour plus amples informations, référez-vous à Filtres d’importation/exportation externes.

Options

Pour le fichier importé, cliquez sur le bouton Options pour:

bmct BULLET.BMP}  spécifier comment gérer les points en double lors de l’importation.

  configurer la qualité des données importées.

  spécifier les unités des données importées.

  spécifier les paramètres dans le fichier importé.

  changer les options pour l’importation des données dans un périphérique 3300/3600/5600 Elta M5, ou 3600/5600 GDM.

Si vous sélectionnez un système de coordonnées différent de celui utilisé pour le projet, Trimble Geomatics Office lit le fichier, convertit les coordonnées au système de coordonnées du projet, stocke les points, et les affiche dans les système de coordonnées du projet. Le fichier de données ne change pas.

NOTE : Les valeurs des coordonnées affichées dans Trimble Geomatics Office pour les points importés du fichier de données peuvent différer de celles du fichier.


Formats d'importation - Tabulation Levé

 Vue d'ensemble et plus amples informations

Utilisez la tabulation Levé pour importer des données d’un Trimble Survey Controller, d’un récepteur Trimble GPS, d’un instrument 3600/5600 GDM Trimble, d’une station totale, d’un niveau numérique  ou d’un fichier de données de levé d’un ordinateur.

Pour accéder à la tabulation Importer / Levé :

1.	Sélectionnez Fichier / Importer.

2.	Sélectionnez la tabulation Levé.

Vous pouvez déplacer les formats personnalisés à cette tabulation. Pour de plus amples informations, référez-vous au fichier Asciiimp.dat.

NOTE : Vous pouvez également utiliser la commande glisser-déplacer pour importer des fichiers.

Utilisez la tabulation Levé pour importer les formats suivants :

Format	Utilisez ce format pour importer...
Fichiers de niveaux numériques (*.dat, *.raw)	des Fichiers de niveaux numériques.
Fichiers de données GPS (*.dat)	des fichiers Trimble GPS .dat de votre ordinateur.
Fichier MNT à grille (*.dtx)	un modèle numérique de terrain créé à l’aide de DTMLink.
Fichiers de feuille de données NGS (*.dat, *.dsx, *.htm, *.html)	des fichiers de données NGS (National Geodetic Survey) de format standard.
Format Trimble Data Exchange	un fichier ASCII avec des données de format Trimble Data Exchange. Ce format est utilisé principalement pour le transfert des données de levé aux autres logiciels.
Fichiers d’éphéméride précis (*.sp3, *.e18)	un fichier de format d’éphéméride précis à utiliser pour le traitement des lignes de base. Ce fichier est enregistré dans la zone des Fichiers communs. Pour le sélectionner, utilisez le dialogue Styles de traitement GPS.
Fichiers RINEX (*.Xxo, *.XXn, *.obs, *.nav)	des fichiers de format Receiver Independent Exchange. Ce format vous permet de transférer des fichiers de données brutes GPS de tout récepteur GPS, quel qu’en soit le fabricant. Vous avez besoin à la fois d’un fichier de navigation et d’un fichier d’observation dans le même dossier.
Fichiers d’enregistreur de données Sokkia (*.sdr)	un fichier de format d’enregistreur de données Sokkia d’un ordinateur.
Fichiers SSF/SSK (*.ssf, *.ssk)	des fichiers de solution cinématique post-traitée ou des fichiers de solution statique post-traitée créés à l’aide de Trimble GPSurvey.
Fichiers Survey Controller (*.dc)	un fichier de format Trimble Survey Controller d’un ordinateur. Lorsque des fichiers .dc sont importés, tous les vecteurs ou observations GPS avec antenne, instrument et hauteurs de prisme associés sont enregistrés dans le projet. Ces données sont utilisées pour effectuer des calculs.
Périphériques de levé	directement des données de stations totales, périphériques, cartes de données de levé, niveaux numériques et récepteurs GPS pris en charge.
Fichiers d’enregistreurs de données tiers	des fichiers d’enregistreurs de données tiers fournis à l’aide d’un filtre d’importation/exportation externe. Les formats tiers suivants sont fournis avec Trimble Geomatics Office : Leica GS1-16 SDMS (points à coordonnées d’importation) Données Nikon AP-700 ASCII Format de données Topcon GTS-700 et FC-5
Fichiers de données Trimble 3600/5600 GDM (*.job, *.are)	des fichiers de données Trimble 3600/5600 GDM.
Fichier d’étude TDS Survey Pro CE (*.job)	Importe les fichiers d’étude TDS Survey Pro CE
Fichier brut TDS Survey Pro CE (*.raw)	Importe les fichier bruts TDS Survey Pro CE
Fichiers Trimble 3300 / 3600 / 5600 Elta M5 (*.dat)	Importe un fichier 3300 / 3600 / 5600 Elta M5
Fichier de coord Trimble 3600 / 5600 TDS (*.cr5)	Importe les fichier de coord Trimble 3600 / 5600 TDS
Fichier brut Trimble 3600 / 5600 TDS (*.rw5)	Importe les fichiers bruts Trimble 3600 / 5600 TDS
Fichiers Trimble Elta GPS Controller M5 (*dat)	Importe les fichiers Trimble Elta GPS Controller M5


Option Fichier de feuille de données NGS (*,dat, *.dsx, *.htm, *.html)

Vous pouvez importer des fichiers de données NGS (National Geodetic Survey) de format standard.

Trimble Geomatics Office suppose que toutes les données NGS importées dans un projet sont dans un datum NAD-83 (c.-à-d. qu’elles sont compatibles avec un datum WGS-84).

Si vous importez des données NGS dans un projet qui n’est manifestement pas basé sur un datum NAD-83, Trimble Geomatics Office ne peut pas être sûr que les données NGS sont appropriées.

NOTE : Trimble Geomatics Office reconnaît un datum NAD-83 par son nom (c.-à-d. tout datum commençant par NAD-83).

Si votre datum n’est pas équivalent à NAD-83, le message d’avertissement suivant s’affiche :

Il paraît que votre projet n'est pas dans un datum NAD 1983. Il y aura des problèmes car les données importées sont dans un datum NAD 1983 et aucune conversion au datum du projet ne sera entreprise.

Voulez-vous continuer l’importation ?

AVERTISSEMENT : Si vous choisissez de poursuivre l’importation, les coordonnées sont interprétées à l’aide du datum, quel qu’il soit, défini pour le projet. Si le datum n’est pas équivalent à NAD-83, cette interprétation est incorrecte.


Formats d'importation CAD / ASCII

Pour accéder à la tabulation Importer / CAD/ASCII :

1.	Sélectionnez Fichier / Importer.

2.	Sélectionnez la tabulation CAD/ASCII.

NOTE : Tout format d’importation supportant les fichiers ASCII s’attend toujours à recevoir des données numériques sous une forme standardisée. Une forme standardisée ne comprend pas de séparateur de milles ni de points pour marquer les décimales.

Vous pouvez déplacer les formats personnalisés à cette tabulation. Pour de plus amples informations, référez-vous au fichier Asciiimp.dat.

NOTE : Vous pouvez également utiliser la commande glisser-déplacer pour importer des fichiers.

Utilisez la tabulation CAD/ASCII pour importer les formats suivants :

Format	Utilisez ce format pour importer...
Fichier AutoCAD (*.dwg, *.dxf)	des points avec coordonnées et des lignes à l’aide de fichiers ASCII et binaires .dxf et .dwg (jusqu’à .dxf Version 14).
Format standard C & G	un fichier ASCII standard de logiciel C & G.
C & G sans guillemets	un fichier ASCII de logiciel C & G, sans guillemets.
Fichiers CMM COR et LEV (*.cor, *.lev)	des fichiers de format ASCII standard utilisés par des bureaux d’aménagement du territoire. L’option de fichiers CMM COR et LEV est fournie à l’aide d’un filtre d’importation externe.
Fichier NMEA	des points de fichiers de format NMEA (chaînes GGA, GGK et PJK). NMEA est une norme développée par la National Marine Electronics Association qui définit les formats de données pour la communication de données de navigation entre les instruments de navigation maritime.
Format Pacsoft	un fichier ASCII standard de logiciel Pacsoft.
Coordonnées SDRmap	un fichier de texte de coordonnées SDRmap.
TPLANE (LLHE)	un fichier ASCII Trimble GPSurvey.
Fichier de graphiques TRIMMAP (*.tgf)	un fichier de graphiques TRIMMAP (ou SDRmap) *.tgf.
Format WILDsoft	un fichier ASCII standard de logiciel WILDsoft.
Fichiers de projet Terramodel (*.pro)	les points, les lignes et les couches des fichiers de projet Terramodel.


Formats d'importation personnalisés

Pour accéder à la tabulation Importer / Personnalisé :

1.	Sélectionnez Fichier / Importer.

2.	Sélectionnez la tabulation Personnalisé.

NOTE : Autrement, utilisez la commande glisser-déplacer pour importer des fichiers.

Vous pouvez déplacer les formats personnalisés à d’autres tabulations. Pour de plus amples informations, référez-vous au fichier Asciiimp.dat.

Utilisez la tabulation Personnalisé pour :

  importer des fichiers de données dans un format personnalisé

  ajouter de nouveaux formats d’importation personnalisés

  éditer les formats d’importation existants

  supprimer les formats d’importation personnalisés.

Les formats personnalisés suivants sont fournis :

  Points délimités

  Points à largeur fixe

  Nom,Est,Nord,Elévation,Code

  Nom,Latitude,Longitude,Hauteur,Code (Locaux)

  Nom,Latitude,Longitude,Hauteur,Code (WGS-84)

  Nom,Nord,Est,Elévation,Code

Lors de l’importation de valeurs d’un fichier ASCII, Trimble Geomatics Office peut lire les valeurs qui utilisent le symbole de groupement des chiffres et le symbole décimal tels qu’ils sont définis dans Panneau de configuration / Paramètres régionaux / tabulation Nombre de Microsoft Windows. Le système lit également les valeurs qui se présentent sous une forme standard, c’est à dire sans symbole de groupement des chiffres (séparateur de milles) et sans point pour le symbole décimal.

Tous les fichiers d’importation personnalisés sont écrits dans le fichier Asciiimp.dat.


Dialogue Définir le format d'importation ASCII

 Vue d'ensemble et plus amples informations

Utilisez ce dialogue pour définir les formats d’importation ASCII.

Pour y accéder :

1.	Sélectionnez Fichier / Importer. Le dialogue Importer s’affiche.

2.	Sous la tabulation Personnalisé, effectuez l’une des opérations suivantes :

  Pour créer un nouveau format, cliquez sur Nouveau format.

  Pour modifier un format existant, cliquez sur Editer le format.

Nom

Saisissez ou éditez le nom du format d’importation. Ce nom s’affiche dans la liste Types sous la tabulation Personnalisé du dialogue Importer. Les noms en double ne sont pas permis.

Chaîne nulle

Entrez la chaîne nulle pour la définition du format d’importation. Ce champ est optionnel. La chaîne nulle que vous spécifiez ici est interprétée lors de l’importation d’un fichier comme indiquant qu’il n’y a aucune valeur disponible pour un enregistrement spécifique.

Si Trimble Geomatics Office trouve la chaîne nulle spécifiée, il n’assigne aucune valeur au champ approprié de la base de données lorsque les données sont ajoutées à la base de données.

NOTE : Il est possible que certains types de fichier requièrent une chaîne particulière telle que "99999.99", "*****", ou similaire. En spécifiant le paramètre, vous pouvez demander à Trimble Geomatics Office d’utiliser votre préférence.

Description

Entrez une description du format d’importation. La description s’affiche sur la barre d'état lorsque le format est sélectionné dans le dialogue Importer. Ce champ est optionnel.

Extension par défaut

Entrez l’extension par défaut pour le format d’importation. Par exemple, .dat or .fbk. Ce champ est optionnel et s’il est ignoré, ".txt" est adopté.

ASTUCE : Pour accepter toutes les extensions de fichier, entrez ???.

NOTE : Trimble Geomatics Office utilise l’extension du champ Lister les fichiers de type dans le dialogue Ouvrir. Il met le(s) nom(s) du/des fichier(s) avec l’extension spécifiée dans la Liste de noms. Trimble Geomatics Office adopte automatiquement le point entre le nom et une extension.

Importer sous

Sélectionnez le type approprié d’enregistrement comme destination pour les données du/des fichier(s) importé(s). Il s’agit normalement de la Table de points.

Spécifier format automatiquement

Cliquez sur Spécifier format automatiquement pour ouvrir le dialogue Insérer format. Utilisez ce dialogue pour initialiser automatiquement la définition de l’en-tête et du corps du format. Pour de plus amples informations, référez-vous à Spécifier format automatiquement.

En-tête du format

A l’aide des codes de champ du menu de raccourcis, vous pouvez créer une définition pour qu’elle corresponde à ou saute la partie initiale fixe d’un fichier d’importation. La définition de l’en-tête est utilisée une fois (et une fois seulement) par Trimble Geomatics Office au début du fichier. Si les lignes d’en-tête d’un fichier d’importation ne correspondent pas à la définition, alors la lecture du fichier en question s’arrête. Cela s’avère utile pour la vérification des formats de fichier.

NOTE : Pour ajouter des lignes vides à la définition de l’en-tête, appuyez sur Ctrl+Entrée.

Par exemple, une définition d’en-tête composée des codes de champ [$Saute.Ligne:7] fait que Trimble Geomatics Office saute les sept premières lignes d’un fichier et interprète le reste du fichier comme étant défini par le corps du format.

Corps du format

A l’aide des codes de champ du menu de raccourcis, vous pouvez créer une définition qui correspond aux données du fichier à lire. Si des données se référant à une entité unique sont définies sur plusieurs lignes, cette situation est traitée dans la définition en appuyant sur Ctrl+Entrée pour commencer une nouvelle ligne de définition.

Pour de plus amples informations sur l’ensemble des champs disponibles dans le système, référez-vous à Codes de champ − Vue d’ensemble.

Lorsqu’un fichier est importé, Trimble Geomatics Office tente à maintes reprises d’interpréter les lignes lues dans le(s) code(s) de champ de la définition. Si le contenu du fichier d’importation ne correspond pas à la définition, le défaut de correspondance est enregistré dans le rapport d’importation puis le programme tente de trouver dans le fichier des données supplémentaires correspondant à la définition.

NOTE : Trimble Geomatics Office démarre automatiquement la définition du Corps du format sur une nouvelle ligne après la définition de l’en-tête.

Bas de page du format

A l’aide des codes de champ du menu de raccourcis, vous pouvez créer une définition de bas de page pour qu’elle corresponde à ou saute la partie finale fixe d’un fichier d’importation.

La définition du bas de page n’est utilisée qu’une seule fois par Trimble Geomatics Office à la fin du fichier. Si les lignes de bas de page d’un fichier d’importation ne correspondent pas à la définition, alors la lecture du fichier en question s’arrête. Cela s’avère utile pour la vérification des formats de fichier.

NOTE : Pour ajouter des lignes vides à la définition du bas de page, appuyez sur Ctrl+Entrée.

Trimble Geomatics Office démarre automatiquement la définition du Bas de page du format sur une nouvelle ligne après la définition du corps.

Pour des exemples d’importations ASCII personnalisées, référez-vous à Format d’importation personnalisé ASCII − Exemples.


Format d'importation personnalisé ASCII - Exemples

 Voir aussi

Ces exemples montrent comment définir certains formats d’importation ASCII.

Exemple 1

Pour lire un fichier comprenant les lignes suivantes :

Format de fichier version 1.00

Date : 8/11/99

Projet : Exemple :

1000,534247.541,263104.807,10.118,STN

1024,534220.249,263104.249,10.129,BE

1025,534222.850,263100.894,10.133,ARBRE

1026,534222.218,263095.934,10.136,ARBRE

1027,534224.816,263092.579,10.140,POTEAU

1028,534228.398,263089.101,10.732,BARRIERE

1041,534236.948,263084.989,11.054,BARRIERE

1042,534237.326,263087.964,10.982,BARRIERE

1087,534243.575,263099.773,10.154,BE

L’En-tête du format peut être définie comme :

Format de fichier version 1.00	Vérifie la première ligne du fichier
[$Saute.Ligne:2]	Saute les deux lignes suivantes dont le contenu n’est pas défini

Le Corps du format serait défini de la manière suivante :

[Nom],[Est],[Nord],[Elévation],[Code de caractéristiques]

Exemple 2

Pour lire un fichier comprenant les lignes suivantes :

1000  534247.541  263104.807      10.118 STN

1024  534220.249  263104.249      10.129 BE

1025  534222.850  263100.894      10.133 ARBRE

1026  534222.218  263095.934      10.136 ARBRE

1027  534224.816  263092.579      10.140 POTEAU

1028  534228.398  263089.101      10.732 BARRIERE

1041  534236.948  263084.989      10.054 BARRIERE

1042  534237.326  263087.964      10.982 BARRIERE

1087  534243.575  263099.773      10.154 BE

Aucune En-tête de format ne serait définie.

Le Corps du format serait défini de la manière suivante :

[Nom:4][Est:12.3][Nord:12.3][Elévation:12.3] [Code de caractéristiques]


Spécifier format automatiquement / dialogue Insérer format

Cliquez sur Spécifier format automatiquement pour initialiser automatiquement les définitions de corps de format(s) d’importation ou d’exportation défini(s) par l’utilisateur, ou un rapport de base de données. Le dialogue Insérer format s’affiche lorsque vous cliquez sur le bouton.

NOTE : Ce bouton apparaît dans les dialogues suivants :

  Fichier / Importer / dialogue Définir le format d’importation ASCII

  Fichier / Importer / dialogue Définir le format d’exportation ASCII

  Rapports / Rapports personnalisés / dialogue Définir le format de rapport

Trimble Geomatics Office crée un groupe de codes de champ par défaut pour toutes les données concernées. En utilisant ce défaut initial comme base de votre nouveau format, vous pouvez modifier la définition comme requis.

Disposition

Le Groupe de disposition vous permet de contrôler le format du rapport ou de la définition.

NOTE : Une fois que vous avez créé un rapport, vous pouvez l’éditer à l’aide d’un éditeur HTML.

Largeur fixe

Sélectionnez l’option Largeur fixe pour assigner une largeur fixe par défaut à chaque champ de la définition.

NOTE : Aucune Information d’en-tête par défaut n’est créée avec cette option, ni dans les définitions d’importation ni dans les définitions d’exportation.

Délimité

Sélectionnez l’option Délimité pour entrer les champs. Cette option vous permet de configurer les délimiteurs qui seront utilisés entre les champs du rapport ou de la définition.

NOTE : Avec cette option, des Informations d’en-tête par défaut sont créées dans tous les cas.

Séparateur champ

Utilisez le champ Séparateur champ pour spécifier le(s) caractère(s) qui sera/seront inséré(s) entre les champs du rapport ou de la définition.

ASTUCE : Le champ vous donne des valeurs par défaut mais vous pouvez entrer tout caractère que vous préférez dans ce champ.

Délimiteur texte

Utilisez le champ Délimiteur texte pour spécifier le(s) caractère(s) à ajouter avant et après chaque champ de texte du rapport ou de la définition.

ASTUCE : Le champ vous donne des valeurs par défaut mais vous pouvez entrer tout caractère que vous préférez dans ce champ.


Exporter - Vue d'ensemble

Comment ...	Dialogue	Voir aussi

Exportez des données de votre projet si vous avez besoin de données dans un format différent. Par exemple, si vous souhaitez utiliser les données d’un progiciel différent ou si vous souhaitez envoyer des fichiers à Trimble Survey Controller.

Vous pouvez exporter des données de divers formats de fichier à partir d’un projet de Trimble Geomatics Office. Par exemple, vous pouvez exporter des données vers :

  Trimble Survey Controller (si vous avez besoin de fichiers pour terminer un travail sur le terrain, par exemple bibliothèques de caractéristiques et attributs)

  un certain nombre de formats ASCII (par exemple si vous avez besoin de données dans un autre format)

  un certain nombre de formats SIG communs

  des formats personnalisés (et définir des formats personnalisés)

Pour une description des formats d’exportation pris en charge, référez-vous au dialogue Exporter.


Dialogue Exporter

Utilisez ce dialogue pour exporter des données à partir de votre projet.

Pour y accéder, effectuez l’une des opérations suivantes :

  Sélectionnez Fichier / Exporter.

  Sélectionnez l’outil Exporter [image].

NOTE : Tous les formats d’exportation qui génèrent des fichiers ASCII génèrent toujours des données numériques sous une forme standardisée (pas de séparateur de milles ni de point pour marquer les décimales). Les données exportées dans les fichiers ASCII sont exprimées dans les unités que Trimble Geomatics Office utilise pour l’affiche. Cependant, vous pouvez spécifier des unités différentes pour l’exportation dans Unités des données d’exportation. Sinon, changez les unités d’affichage Trimble Geomatics Office en sélectionnant Fichier / Propriétés du projet et puis sélectionnez la tabulation Unités et format dans le dialogue Propriétés du projet qui s’affiche.

Pour que la description d’un format d’exportation apparaisse sur la barre d'état, sélectionnez le format d’exportation. Les formats d’importation sont organisés sous les tabulations suivantes :

  Levé

  CAD / ASCII

  SIG

  Personnalisé

Nouveau format

Le bouton Nouveau format devient disponible lorsque la tabulation Personnalisé est sélectionnée.

Pour créer un nouveau format de Rapport personnalisé, cliquez sur Nouveau format et puis utilisez le dialogue Définir le format d’exportation ASCII pour définir le format d’exportation.

Editer le format

Le bouton Editer le format devient disponible lorsque la tabulation Personnalisé est sélectionnée.

Pour modifier un format d’exportation ASCII existant, cliquez sur Editer le format et puis utilisez le dialogue Définir le format d’exportation ASCII qui s’affiche pour modifier le format d’exportation. Pour de plus amples informations sur la définition d’un format personnalisé, référez-vous à la rubrique Définir le format d’exportation ASCII − Exemples.

Supprimer format

Le bouton Supprimer format devient disponible lorsque la tabulation Personnalisé est sélectionnée.

Pour supprimer un format personnalisé, cliquez sur Supprimer.

Personnaliser

Utiliser le bouton Personnaliser pour sélectionner la liste des options d’importation affichées dans le dialogue Exporter. Sélectionnez les options d’exportation requises de la liste.

Vous pouvez ajouter un filtre externe du dialogue Personnaliser. Les filtres externes sont des programmes séparés pouvant être écrits afin d’intégrer avec Trimble Geomatics Office. Tout filtre externe écrit pour une utilisation dans Trimble Geomatics Office doit se conformer aux exigences spécifiques s’il faut l’ajouter au système. Seulement les filtres installés actuellement pour la tabulation sélectionnée dans le dialogue Exporter s’affichent.

Tout filtre de programme doit être compilé comme un fichier Dynamically Linked Library (*.dll) (Bibliothèque liée dynamiquement) avec des fonctions exportées spécifiques et doit être créé comme un objet COM (Module d’objet de composant).

Dès que vous sélectionnez le fichier approprié, Trimble Geomatics Office enregistre l’objet COM de filtre dans l’ordinateur et vérifie que l’interface .dll se conforme aux exigences du système. Si ce ne réussit pas, le système indique qu’il n’est pas possible d’enregistrer le fichier. SI le filtre est installé avec succès, la description courte de filtre est listée dans le dialogue.

Pour plus amples informations, référez-vous à Filtres d’importation/exportation externes.

Options

Pour le fichier importé, cliquez sur le bouton Options pour:

  spécifier les unités des données exportées.

  spécifier les paramètres dans le fichier exporté.

  changer le système de coordonnées du fichier de données exporté.

  changer les options pour l’exportation des données dans un périphérique,  3300/3600/5600 Elta M5, ou 3600/5600 GDM.

Lorsqu’un format d’exportation ne peut pas être configuré, ce bouton n’est pas disponible.

Pour définir les détails de configuration relatifs au fichier spécifique à exporter, cliquez sur Options. Par exemple, dans le cas du format AutoCAD (*.dxf ou *.dwg), vous pouvez sélectionner le format de données Options d’exportation AutoCAD (.dwg, Ascii .dxf, Binaire .dxf), le numéro de version d’AutoCAD ainsi que référencer un dessin de prototype.

Système de coordonnées pour les données d’exportation

Pour sélectionner le système de coordonnées du fichier de données à exporter, cliquez sur Changer  dans le dialogue Options. Ce bouton est disponible si le format souligné actuellement n’a pas une définition de système de coordonnées compris dans le fichier. Si vous sélectionnez un système de coordonnées qui diffère du système utilisé pour le projet, Trimble Geomatics Office convertit les coordonnées des points à exporter au système de coordonnées sélectionné, écrit le fichier d’exportation, et stocke les points dans le système de coordonnées sélectionné.

NOTE: Les valeurs de coordonnées des points affichés dans Trimble Geomatics Office peut différer de celles écrites dans le fichier.

Exporter

Si une sélection courante a été effectuée à l’aide des outils Sélection, alors vous pouvez choisir d’exporter les données sélectionnées ou la base de données entière.

Deux options sont disponibles :

Sélection	Exporte les points sélectionnés
BD entière	Exporte tous les points de la base de données
Lorsqu’il n’y a pas de sélection courante, les boutons de sélection ne sont pas disponibles et l’exportation est appliquée à la base de données entière.

Trier données

Vous pouvez choisir l’ordre dans lequel les données sont exportées :

1.	Sélectionnez la case à cocher Trier données.

2.	Sélectionnez un ordre de triage dans la liste. Choisissez entre :

Ordre de base de données	Elévation
Nom	Qualité
Heure de début	Type de données
Temps d’occupation
NOTE : Les mêmes options de triage sont disponibles dans la fenêtre Propriétés.

ASTUCE : Si vous ajoutez vos propres interrogations de triage, celles-ci sont ajoutées à la liste. Pour de plus amples informations, référez-vous à Interrogations de base de données – Vue d’ensemble.


Formats d'exportation - Tabulation Levé

 Vue d'ensemble et plus amples informations

Utilisez la tabulation Levé du dialogue Exporter pour exporter les données de levé vers un fichier ou un périphérique.

Pour accéder à cette tabulation :

1.	Sélectionnez Fichier / Exporter.

Le dialogue Exporter s'affiche.

2.	Sélectionnez la tabulation Levé.

Vous pouvez déplacer les formats personnalisés à cette tabulation. Pour de plus amples informations, référez-vous au fichier Asciiexp.dat.

Cette tabulation contient les formats suivants :

Option	Utilisez cette option pour exporter...
Détails d’antenne au Survey Controller	de nouveaux détails d’antenne vers Trimble Survey Controller. Par exemple, si Trimble lance une nouvelle antenne, utilisez cette option d’exportation pour transférer les détails d’antenne vers votre  Trimble Survey Controller.
Carte d’arrière-plan au  Trimble Survey Controller (*.dxf)	Un fichier dxf au Trimble Survey Controller (version 10 ou ultérieur).
Fichier de coordonnée délimité par des virgules au Trimble Survey Controller (*.csv)	Un fichier csv au Trimble Survey Controller (version 10 ou ultérieur).
Dictionnaire d’attributs au Survey Controller (*.ddf)	un fichier de Dictionnaire d’attributs (.ddf) vers un périphérique exécutant Trimble Survey Controller. Pour cela, il faut qu’un périphérique Survey Controller soit installé. Vous pouvez créer un fichier .ddf dans l’utilitaire Data Dictionary Editor. Cet utilitaire fait partie du logiciel GPS Pathfinder de Trimble. NOTE : Vous pouvez uniquement exporter un fichier .ddf vers Trimble Survey Controller (version 7.00).
Grille de datum au Survey Controller (*.cdg)	la grille de datum vers Trimble Survey Controller (version 7.00 ou ultérieure). Vous pouvez réduire la taille du fichier de grille de datum transféré (sous-grille). Les versions antérieures de Trimble Survey Controller ne prennent pas les grilles de datum en charge. Pour cela, il faut qu’un périphérique Survey Controller soit installé. NOTE : Vous ne pouvez diviser en sous-grilles que le fichier de grille de datum utilisé dans le système de coordonnées du projet courant, mais vous pouvez exporter toute grille de datum combinée.
Fichier de description au périphérique TDS (*.txt)	un fichier de description (*.txt) ver un périphérique exécutant le logiciel TDS. Pour cela, il faut d’abord installer un périphérique TDS. Vous pouvez créer un fichier de description manuellement, ou en exporter un de l’utilitaire Feature Code Editor (Editeur de codes de caractéristiques).
Points MNT au fichier (*.dtx)	un fichier de Modèle Numérique de Terrain (*.dtx) compatible avec Trimble Survey Controller des points sélectionnés dans le projet.
Points MNT au Survey Controller	un fichier de Modèle Numérique de Terrain (*.dtx) compatible avec Trimble Survey Controller et  points sélectionnés pour l’exportation directe vers Trimble Survey Controller. Pour cela, un périphérique Survey Controller doit être installé.
Bibliothèque de caractéristiques et attributs au fichier SDR33 (*.sdr)	un fichier SDR33 contenant des enregistrements de notes pour chaque code de caractéristiques de la bibliothèque que vous sélectionnez. Vous ne pouvez pas exporter d’attributs à l’aide de cette option.
Bibliothèque de caractéristiques et attributs au Survey Controller (*.fcl)	une bibliothèque de caractéristiques et attributs vers un périphérique exécutant Trimble Survey Controller. Le fichier peut ou non contenir des attributs. (Si des attributs doivent être exportés, sélectionnez la case à cocher). Pour cela, un Survey Controller doit être installé. Aucune autre information de la bibliothèque de caractéristiques et attributs (par exemple des styles de point) n’est transférée à Trimble Survey Controller.
Bibliothèque de caractéristiques et attributs au périphérique TDS (*.fcl)	un fichier de description (*.txt) basé sur la Bibliothèque de caractéristiques et attributs sélectionnée vers un périphérique exécutant le logiciel TDS. Pour cela, d’abord il faut installer un périphérique TDS. Les attributs ne sont pas exportés au logiciel TDS.
Fichier de géoïde au Survey Controller (*.ggf)	un fichier de Grille de géoïde vers un périphérique exécutant Trimble Survey Controller (version 6 ou ultérieure). Vous pouvez sélectionner la surface dans laquelle vous travaillez. Cela entraîne la réduction de la taille du fichier de grille de géoïde transféré à Trimble Survey Controller (sous-grille). Pour cela, un Survey Controller doit être installé. NOTE : Pour pouvoir diviser en sous-grilles, vous devez utiliser un système de coordonnées avec un modèle de géoïde défini. Vous ne pouvez diviser en sous-grilles que le fichier de grille de géoïde actuellement utilisé, mais vous pouvez exporter tout fichier de géoïde.
Fichier de géoïde au TDS Survey Pro (CE)	un fichier de Grille de géoïde vers un périphérique exécutant TDS Survey Pro (CE).  Vous pouvez sélectionner la surface dans laquelle vous travaillez. Cela entraîne la réduction de la taille du fichier de grille de géoïde.
Fichiers NGS Bluebook	des fichiers B, G et SERFIL. Le format NGS Bluebook est principalement utilisé pour transférer des lignes de base GPS au National Geodetic Survey américain pour inclusion dans le réseau de contrôle national.
Fichier Pcode au 3600/5600 GDM (*pco)	un fichier pcode vers un 3600/5600 GDM.
Format Trimble Data Exchange	des données de levé de votre projet Trimble Geomatics Office. Utilisez ce format pour copier des données de levé entre les projets Trimble Geomatics Office. Ce format est utilisé principalement pour exporter des données vers des logiciels tiers d’ajustement de réseau. Le format a été défini par Trimble et est utilisé pour produire un fichier ASCII contenant des stations et des coordonnées, des vecteurs GPS traités, des observations terrestres, des observations laser ainsi que des informations sur les azimuts et le nivellement. Consultez le revendeur de votre logiciel d’ajustement de réseau pour vérifier que ce dernier prend ce format en charge.
Fichiers SDR (*.sdr)	un fichier .sdr à l’aide du projet courant de la version configurée. Cliquez sur Options pour modifier la configuration.
Points de contrôle 2D STAR*NET	des points de contrôle de format 2D STAR*NET. Seuls les points de la base de données dont le Nord/Est ont une qualité de Contrôle sont exportés. L’Elévation peut avoir n’importe quelle qualité. Lorsque vous travaillez avec le progiciel STAR*NET, la meilleure méthode consiste à combiner les fichiers de points de contrôle et d’observations en un fichier unique à l’aide d’un éditeur de texte approprié.
Points de contrôle 3D STAR*NET	des points de contrôle de format 3D STAR*NET. Seuls les points de la base de données dont le Nord/Est et l’élévation ont une qualité de Contrôle sont exportés. Lorsque vous travaillez avec le progiciel STAR*NET, la meilleure méthode consiste à combiner les fichiers de points de contrôle et d’observations en un fichier unique à l’aide d’un éditeur de texte approprié.
Observations STAR*NET	toutes les observations conventionnelles de la base de données du projet de format d’observation STAR*NET. Lorsque vous travaillez avec le progiciel STAR*NET, la meilleure méthode consiste à combiner les fichiers de points de contrôle et d’observations en un fichier unique à l’aide d’un éditeur de texte approprié.
Périphérique de levé	directement des données vers des stations totales, périphériques, cartes de données de levé et récepteurs GPS pris en charge.
Survey Controller (*.dc)	un fichier .dc à l’aide du projet courant de la version configurée. Cliquez sur Options pour modifier la configuration.
Fichiers Trimble 3300 / 3600 /5600 Elta M5	un fichier Trimble 3300 / 3600 /5600 Elta M5
Fichiers Trimble 3600/5600 GDM (*.are)	un fichier area Trimble 3600/5600 GDM
Fichiers Trimble Elta GPS Controller M5 (*.dat)	un fichier de coordonnées Trimble Elta GPS Controller M5
Fichiers d’étude TDS Survey Pro (CE) (*.job)	un fichier d’étude vers un périphérique TDS Survey Pro CE.
Fichiers de coordonnées 3600/5600 TDS (*.cr5)	un fichier de coordonnées vers un périphérique Trimble 3600/5600 TDS.
Fichiers Trimble Data Exchange Format (*.asc)	un fichier  TDEF


Formats d'exportation CAD / ASCII

 Vue d'ensemble et plus amples informations

Utilisez la tabulation CAD/ASCII pour l'exportation des données vers une variété de formats ASCII et CAD.

Pour accéder à la tabulation Exporter / CAD/ASCII :

1.	Sélectionnez Fichier / Exporter. Le dialogue Exporter s’affiche.

2.	Sélectionnez la tabulation CAD/ASCII.

Vous pouvez déplacer les formats personnalisés à cette tabulation. Pour de plus amples informations, référez-vous au fichier Asciiexp.dat.

NOTE : Tous les formats d’exportation qui génèrent des fichiers ASCII génèrent toujours des données numériques sous une forme standardisée (pas de séparateur de milles ni de point pour marquer les décimales). Les valeurs sorties vers un fichier ASCII ont les mêmes unités que celles actuellement utilisées pour l’affichage dans le système, à moins que vous ne changiez les unités à l’aide de la liste Unités des données d’exportation. Pour modifier les unités d’affichage du système, sélectionnez Fichier / Propriétés du projet. Le dialogue Propriétés du projet s’ouvre. Sélectionnez la tabulation Unités et format.

La tabulation CAD / ASCII contient les options suivantes :

Option	Utilisez cette option pour exporter...
Fichiers AutoCAD (*.dxf, *.dwg)	un fichier d’échange de dessin ou un fichier de dessin pour AutoCAD. Pour configurer le format de données, la version et le dessin de prototype, cliquez sur Options.
Format standard C&G	un fichier de format ASCII standard de logiciel C&G.
C&G sans guillemets	un fichier de format ASCII de logiciel C&G, sans guillemets.
Fichier CMM (*.cor, *.lev)	les fichiers de format ASCII standard utilisés par les bureaux d’aménagement du territoire. L’option de fichier CMM (*.cor, *.lev) est fournie à l’aide d’un filtre d’exportation externe.
Notes d'attribut du fichier DC (Survey Controller V4)	un fichier .dc Trimble Survey Controller version 4 dont les valeurs d’attribut sont séparées par deux points (:). Les attributs collectés à l’aide d’une bibliothèque de caractéristiques et attributs ne sont pas exportés dans ce format.
Format Pacsoft	un fichier de format ASCII Pacsoft.
Format SDMS	des données de format Survey Data Management System de AASHTO.
Coordonnées SDRmap	un fichier de coordonnées SDRmap.
Fichier fieldbook Autodesk (*.fbk)	un fichier de format « fieldbook » d’Autodesk.
TERRAMODEL – P,N,E,Z,20D – fichier (*.pts)	un fichier de format de base délimité par virgules.
TPLAN (LLHE)	un fichier  Trimble GPSurvey de format ASCII. Les coordonnées de latitude et de longitude WGS-84 sont exportées à l’aide de cette option.
Fichier de graphiques TRIMMAP (*.tgf)	un fichier de graphiques TRIMMAP (ou SDRmap).
Format WILDsoft	un fichier de format ASCII WILDsoft.


Formats d'exportation - Tabulation Personnalisé

Pour accéder à la tabulation Exporter / Personnalisé :

1.	Sélectionnez Fichier / Exporter. Le dialogue Exporter s’affiche.

2.	Sélectionnez la tabulation Personnalisé.

Vous pouvez déplacer les formats personnalisés aux autres tabulations. Pour de plus amples informations, référez-vous au fichier Asciiexp.dat.

Utilisez la tabulation Personnalisé pour :

  exporter des fichiers de données dans un format personnalisé

  ajouter de nouveaux formats d’exportation personnalisés

  éditer les formats d’exportation existants

  supprimer les formats d’exportation personnalisés.

Les formats personnalisés suivants sont fournis :

  Points délimités

  Points à largeur fixe

  Nom,Est,Nord,Elévation,Code

  Nom,Latitude,Longitude,Hauteur (degrés décimaux locaux)

  Nom,Latitude,Longitude,Hauteur (degrés décimaux WGS-84)

  Nom,Latitude,Longitude,Hauteur,Code (Locaux)

  Nom,Latitude,Longitude,Hauteur,Code (WGS-84)

  Nom,Nord,Est,Elév,Lat,Long,Hauteur,Code (Locaux)

  Nom,Nord,Est,Elév,Lat,Long,Hauteur,Code (WGS-84)

  Nom,Nord,Est,Elévation,Code

  Point sur route

  Points – Implantés

  Route – Point d’entrée en terre

  Route – Eléments géométriques

Les formats définis par l’utilisateur apparaissent dans l’ordre alphabétique.

NOTE : Les valeurs exportées vers un fichier ASCII sont toujours écrites sous une forme standardisée. C’est à dire sans symbole de groupement des chiffres (séparateur de milles) et sans point pour le symbole décimal. Les options Panneau de configuration / Paramètres régionaux /Nombre de Microsoft Windows ne sont pas utilisées.

Tous les fichiers d’exportation personnalisés sont écrits dans le fichier Asciiimp.dat.


Dialogue Options de bibliothèque de caractéristiques et attributs

Utilisez ce dialogue pour configurer la version de Trimble Survey Controller vers laquelle vous exportez.

Version

Si vous sélectionnez une version antérieure à la version 7.00, la case à cocher Exporter attributs du dialogue Sélectionner options de bibliothèque de caractéristiques et attributs n’est pas disponible. Cela est dû au fait que les versions antérieures ne prennent pas en charge les attributs.


Dialogue Options d'exportation AutoCAD

Utilisez ce dialogue pour configurer la version AutoCAD vers laquelle vous exportez.

Format données

Sélectionnez le format de sortie des données parmi les choix disponibles :

  DWG (un fichier binaire de dessin AutoCAD)

  ASCII DXF (un fichier standard AutoCAD Data eXchange)

  Binaire DXF (un fichier standard AutoCAD Data eXchange en format binaire)

NOTE : Le format de données est déterminé par vos exigences (ou celles de votre client).

Version

Sélectionnez la version AutoCAD appropriée pour le fichier d’exportation. Celle-ci est déterminée par le système vers lequel vous exportez.

Dessin du prototype

Utilisez ce champ pour référencer un dessin de prototype AutoCAD duquel vous souhaitez hériter des définitions de bloc pour les utiliser dans les fichiers *.dxf et *.dwg créés par le système.

Au moment où Trimble Geomatics Office crée un fichier AutoCAD, il essaie de faire correspondre les noms des symboles utilisés dans le projet aux blocs définis dans le dessin de prototype spécifié. Tous les espaces dans les noms de symbole sont assortis avec les soulignements des noms de bloc AutoCAD.

Lorsqu’un nom de symbole correspond à un nom de bloc du dessin du prototype, le bloc approprié est inséré dans le fichier d’exportation avec la mise à l’échelle et la rotation requises. Tout symbole ne pouvant être assorti à aucun bloc du dessin du prototype est inséré en tant que point. Le dessin de prototype fourni (Symbols.dxf) comprend les définitions de bloc pour tous les symboles par défaut fournis avec le système.

NOTE : Les définitions de bloc du dessin du prototype sont interprétées comme étant du type 200 unités par 200 unités, sachant que les unités sont les unités d’affichage au moment de l’exportation. Il faut faire correspondre la définition 200-par-200 à la grille utilisée par l’utilitaire Trimble Symbol and Line Type Editor. Si vous souhaitez créer votre propre dessin de prototype avec des définitions de bloc, assurez-vous que les blocs sont basés sur une définition de 200-par-200 pour obtenir une mise à l’échelle des symboles correcte.

Parcourir

Cliquez sur Parcourir pour parcourir votre système et trouver les fichiers AutoCAD dont vous avez besoin.


Dialogue Définir le format d'exportation ASCII

Utilisez ce dialogue pour définir les formats d’exportation ASCII.

Nom

Entrez le nom du format d’exportation. Ce nom s’affiche dans la liste Types sous la tabulation Personnalisé du dialogue Exporter. Les noms en double ne sont pas autorisés.

Chaîne nulle

Entrez la chaîne nulle pour la définition du format d’exportation. Ce champ est optionnel. Chaque fois qu’un champ de base de données auquel aucune valeur n’a été assignée est rencontré, Trimble Geomatics Office extrait la chaîne nulle spécifiée dans le format de fichier.

NOTE : Il est possible que certains types de fichier requièrent une chaîne particulière telle que "99999.99", "*****", ou similaire. En spécifiant le paramètre, vous pouvez demander à Trimble Geomatics Office d’utiliser votre préférence.

Description

Entrez une description du format d’exportation. Cette description apparaît sur la barre d'état. Ce champ est optionnel.

Extension par défaut

Entrez l’extension par défaut pour le format d’exportation. Par exemple, dat ou fbk. Ce champ est optionnel et s’il est ignoré, "*.txt" est adopté.

NOTE : Trimble Geomatics Office spécifie l’extension dans le champ Enregistrer sous type du dialogue Enregistrer sous. Le nom de fichier du nouveau fichier peut être saisi dans le champ Nom du fichier. Tout fichier du dossier sélectionné possédant l’extension appropriée est affiché de manière à ce qu’il puisse être écrasé si nécessaire. Trimble Geomatics Office adopte automatiquement le point dans une extension.

Exporter de

Sélectionnez l’une des tables ou interrogations primaires comme source pour le(s) fichier(s) exporté(s). Les enregistrements de point sont la source par défaut. Cela vous permet d’exporter tout enregistrement de la table de points.

Spécifier format automatiquement

Cliquez sur Spécifier format automatiquement pour initialiser l’en-tête et le corps automatiquement. Pour de plus amples informations, veuillez vous référer à Spécifier format automatiquement.

En-tête du format

Entrez le texte, les définitions de codes de champ, ou les deux (disponibles dans le menu de raccourcis), que Trimble Geomatics Office va utiliser pour créer la partie initiale fixe du fichier d’exportation. Ce champ est optionnel.

Pour les informations détaillées sur l’ensemble des codes de champ disponibles dans le système, veuillez vous référer à Codes de champ − Vue d’ensemble.

Pour ajouter des lignes vides à la définition de l’en-tête, appuyez sur Entrée.

Corps du format

Entrez la définition de code de champ appropriée (disponible dans le menu de raccourcis) pour indiquer quelles données Trimble Geomatics Office doit exporter de la base de données. Une définition de sortie doit être entrée dans ce champ. Si les données d’une entité unique doivent être exportées sur des lignes multiples, utilisez la définition. Pour cela, appuyez sur Entrée pour commencer une nouvelle ligne de définition.

NOTE : Pour de plus amples informations sur l’ensemble des codes de champ disponibles dans le système, veuillez vous référer à Codes de champ − Vue d’ensemble.

Bas de page du format

Entrez (à partir du menu de raccourcis) le texte, les définitions de codes de champ, ou les deux, que Trimble Geomatics Office va utiliser pour créer la partie finale fixe du fichier d’exportation. Ce champ est optionnel.

Pour les informations détaillées sur l’ensemble des codes de champ disponibles dans le système, veuillez vous référer à Codes de champ − Vue d’ensemble.

Pour ajouter des lignes vides à la définition du bas de page, appuyez sur Entrée.

Pour des exemples d’exportations ASCII personnalisées, référez-vous à la rubrique Définir le format d’exportation ASCII − Exemples.


Définir le format d'exportation ASCII - Exemples

 Voir aussi

Ces exemples montrent comment créer certains formats d’exportation ASCII, à condition que le projet contiennent les points suivants :

Nom : 1000	Est : 534247.541	Nord : 263104.807	Elév : 10.118	Code : STN

Nom : 1024	Est : 534220.249	Nord : 263104.249	Elév : 10.129	Code : BE

Nom : 1025	Est : 534222.850	Nord : 263100.894	Elév : 10.133	Code : ARBRE

Nom : 1026	Est : 534222.218	Nord : 263095.934	Elév : 10.136	Code : ARBRE

Nom : 1027	Est : 534224.816	Nord : 263092.579	Elév : 10.140	Code : POTEAU

Nom : 1028	Est : 534228.398	Nord : 263089.101	Elév : 10.732	Code : BARRIERE

Nom : 1041	Est : 534236.948	Nord : 263084.989	Elév : 11.054	Code : BARRIERE

Nom : 1042	Est : 534237.326	Nord : 263087.964	Elév : 10.982	Code : BARRIERE

Nom : 1087	Est : 534243.575	Nord : 263099.773	Elév : 10.154	Code : BE

Exemple 1

Le format d’exportation :

En-tête du format :

Format de fichier version 1.50

Date : [$Date.Système]

Projet : [$Nom.Projet]

Corps du format :

[Nom],[Est:0.3],[Nord:0.3],[Elévation:0.3],[Code de caractéristiques]

Bas de page du format :

[$Système.A propos de]

Ce corps de format produit le fichier d’exportation suivant (en supposant que la date, les paramètres de format de date de Microsoft Windows et le nom du projet sont corrects) :

Format de fichier version 1.50

Date : 8/11/96

Projet : Exemple :

1000,534247.541,263104.807,10.118,STN

1024,534220.249,263104.249,10.129,BE

1025,534222.850,263100.894,10.133,ARBRE

1026,534222.218,263095.934,10.136,ARBRE

1027,534224.816,263092.579,10.140,POTEAU

1028,534228.398,263089.101,10.732,BARRIERE

1041,534236.948,263084.989,11.054,BARRIERE

1042,534237.326,263087.964,10.982,BARRIERE

1087,534243.575,263099.773,10.154,BE

Trimble Geomatics Office, version 1.50

Exemple 2

Le format d’exportation :

Pas d’en-tête de format

Corps du format :

[Nom:4][Est:12.3][Nord:12.3][Elévation:12.3] [Code de caractéristiques]

Ce corps de format produit le fichier d’exportation suivant :

1000  534247.541  263104.807      10.118 STN

1024  534220.249  263104.249      10.129 BE

1025  534222.850  263100.894      10.133 ARBRE

1026  534222.218  263095.934      10.136 ARBRE

1027  534224.816  263092.579      10.140 POTEAU

1028  534228.398  263089.101      10.732 BARRIERE

1041  534236.948  263084.989      10.054 BARRIERE

1042  534237.326  263087.964      10.982 BARRIERE

1087  534243.575  263099.773      10.154 BE

NOTE : Trimble Geomatics Office suppose qu’une nouvelle ligne est insérée à la fin de l’en-tête et du corps. Pour ajouter une/des nouvelle(s) ligne(s) supplémentaire(s), appuyez sur Entrée. Le point d’insertion descend d’une ligne, retourne dans la marge gauche et crée une ligne vide pour que vous puissiez insérer du texte supplémentaire.


Option de format SDMS

Le format Survey Data Management System (SDMS), défini par l’American Association of State Highway and Transportation Officials (AASHTO), consiste en un format spécial pour les données de levé. Vous exportez des données de Trimble Geomatics Office dans ce format.

Cette option permet d’exporter les points et leurs notes d’attribut. Ces notes peuvent contenir des caractéristiques de point supplémentaires. Ces notes d’attribut sont saisies dans le périphérique exécutant Trimble Survey Controller lors d’un levé. La fonction de note d’attribut spéciale de Trimble Survey Controller apporte une assistance lors de ce procédé. Par ailleurs, tout enregistrement de notes entré avec un identificateur à deux caractères suivi par deux points (:) ou un espace entraîne l’exportation des détails d’attribut dans un fichier SDMS.

Par exemple, une note contenant I0:informations ou I0 informations est exportée vers un fichier SDMS comme I0:informations, « informations » étant le texte associé à l’identificateur SDMS I0.

Afin que les notes d’attribut soient associées aux points corrects dans le fichier SDMS créé, vous devez choisir lors de l’exportation le groupe de sélection automatiquement créé lorsque votre fichier Trimble Survey Controller *.dc a été importé dans le projet. Ce groupe de sélection contient toutes les données (y compris les notes enregistrées) dans l’ordre de levé comme indiqué dans le fichier .dc.


Filtres externes - Vue d'ensemble

Trimble Geomatics Office offre la possibilité d’ajouter des filtres externes d’importation et d’exportation. Vous pouvez écrire ces filtres, indépendamment du programme principal du système, comme des bibliothèques dynamiquement liées (.dll) avec des fonctions exportées spécifiques. Les fichiers .dll doivent être créés comme modules OLE (Object Linking and Embedding). Ces modules sont également appelés modules COM (Component Object Module).

Un filtre externe doit comprendre les fonctions exportées suivantes :

Fonction GetGUID

Déclaration de fonction en Pascal :	fonction GetGUID : TGUID; stdcall; export;

Déclaration de fonction en C/C++ :	GUID __stdcall __export GetGUID()

Cette fonction doit communiquer l’identificateur OLE de classe de composant unique (GUID - Globally Unique IDentifier : identificateur global unique) qui a été assigné au programme. Cet identificateur unique est obtenu à l’aide d’un programme approprié qui génère un identificateur unique, généralement sur la base des informations telles que l’heure et la date de l’ordinateur, toute ID de carte de réseau ou toute information similaire. Le principe consiste à avoir une ID unique au monde et l’on espère que cela pourra être réalisé en utilisant des nombres suffisamment grands (composés de 32 caractères) pour qu’il soit statistiquement impossible de donner au hasard deux nombres qui aient la même valeur. 07F750A3-594B-11D1-8E4C-0000C04FCAF9 est un exemple de GUID.

Fonction Convertir

Déclaration de fonction en Pascal :	fonction Convertir (const FileName: WideString): WordBool;

Déclaration de fonction en C/C++ :	BOOL __stdcall __export Convert(BSTR FileName)

Il s’agit de la fonction qui réalise le travail effectif du filtre. Le programme de filtrage doit être conçu de manière à convertir les données d’un fichier source en données d’un format approprié sauvegardées dans un fichier de destination. Cette fonction permet à Trimble Geomatics Office de spécifier le nom du fichier de destination pour le fichier que le filtre doit créer. Dans le cas d’un filtre d’importation, Trimble Geomatics Office attend du filtre qu’il présente le dialogue standard Fichier / Ouvrir permettant la sélection du fichier source. Le nom du fichier de destination, défini par le système et devant être créé par le programme de filtrage, est transféré au filtre par le biais de la fonction Convertir. Le fichier créé par le programme de filtrage d’importation doit avoir un format valide .dc (format de fichier Survey Controller ou SDR).

Dans le cas d’un filtre d’exportation, Trimble Geomatics Office attend du filtre qu’il présente le dialogue standard Fichier / Enregistrer permettant la sélection du fichier de destination. Trimble Geomatics Office crée un fichier source pour le filtre conformément au format de fichier défini dans la fonction Format décrite ci-dessous. Le nom de fichier du fichier source est transféré à la fonction Convertir afin que le filtre sache avec quel fichier travailler. Trimble Geomatics Office extrait les données de la base de données conformément aux critères du groupe de sélection définis pour l’opération d’exportation. La définition du code de champ, telle qu’elle est définie dans la fonction Format, contrôle le format des données écrites dans le fichier source par Trimble Geomatics Office. Le programme de filtrage d’exportation doit lire le fichier source, convertir les données dans le format d’exportation approprié puis les écrire dans le fichier de destination spécifié.

Fonction GetLongDesc

Déclaration de fonction en Pascal :	fonction GetLongDesc : WideString; safecall;

Déclaration de fonction en C/C++ :	BSTR __stdcall __export GetLongDesc()

Cette fonction est utilisée pour fournir du texte qui s’affiche sur la barre d'état de Trimble Geomatics Office lorsque l’option de filtres d’importation ou d’exportation est sélectionnée dans le dialogue Importer ou Exporter. La définition de cette fonction est optionnelle et, si elle n’est pas définie, alors rien ne s’affiche sur la barre d'état lorsque l’option de filtres est sélectionnée.

Fonction GetShortDesc

Déclaration de fonction en Pascal :	fonction GetShortDesc : WideString; safecall;

Déclaration de fonction en C/C++ :	BSTR __stdcall __export GetShortDesc()

Cette fonction est utilisée pour fournir du texte pour le nom du filtre d’importation ou d’exportation affiché dans le dialogue Importer ou Exporter de Trimble Geomatics Office afin de permettre la sélection du filtre.

Fonction Format

Déclaration de fonction en Pascal :	Fonction Format : WideString; safecall;

Déclaration de fonction en C/C++ :	BSTR __stdcall __export Format()

Cette fonction est requise uniquement dans les programmes de filtrage d’exportation et permet au programme de spécifier le format du fichier source que Trimble Geomatics Office va créer. La chaîne de texte communiquée par cette fonction doit consister en codes de champ valides Trimble Geomatics Office. Si aucune fonction Format n’est définie dans un programme de filtrage d’exportation, Trimble Geomatics Office adopte un format de ‘[Nom],[Nord],[Est],[Elévation],[Code de caractéristiques]’ créant ainsi un fichier avec des valeurs de Nom, Nord, Est, Elévation et de Code de caractéristiques séparées par virgules, qui seront lues par le filtre d’exportation.


Option Fichiers de données GPS (*.dat)

Un fichier .dat contient les données GPS brutes d’un récepteur GPS. Ces fichiers peuvent comprendre des données cinématiques post-traitées et statiques. Les fichiers GPS .dat ne contiennent pas d’informations sur le système de coordonnées.

Vous pouvez transférer un fichier .dat à un projet Trimble Geomatics Office :

  d’un Trimble Survey Controller (avec fichier .dc)

  d’un récepteur GPS Trimble Survey

  d’un ordinateur

NOTE : Si vous importez un fichier .dat plus d’une fois dans un même projet, les occupations existantes de la base de données du projet ne sont pas sélectionnées par défaut dans le dialogue Entrée d’un fichier DAT.


Options Fichiers RINEX

Les fichiers de format RINEX (Receiver INdependent EXchange) est un format de fichier de données brutes GPS standard utilisé pour échanger les fichiers de fabricants de récepteurs multiples.

Pour utiliser ce format, vous avez besoin des fichiers suivants sur votre ordinateur :

Fichier RINEX	Extension du fichier	Optionnel
Fichier de données d’observation	*.obs, *.XXo	−
Fichier de messages de navigation	*.nav, *.XXn	−
Fichier météorologique	fichier *.met ou *.XXm	Ces fichiers sont optionnels.
Pour importer des fichiers RINEX de votre ordinateur dans Trimble Geomatics, effectuez l’une des opérations suivantes :

  Utilisez le dialogue Importer et sélectionner les fichiers RINEX (*.obs, *.??o).

  Utilisez la fonction glisser-déplacer.

Sélectionnez le(s) fichier(s) de données d’observation. Si les fichiers d’observation et de navigation n’ont pas le même nom, vous pouvez faire correspondre les fichiers d’observation et de navigation dans le dialogue Correspondance de fichiers de navigation. Naviguez au  fichier .nav correspondant ou choisir un fichier dans la liste des Fichiers de navigation.

Le dialogue Entrée d’un fichier DAT s’affiche et vous pouvez y vérifier les valeurs des fichiers avant que les observations GPS ne soient importées. Vérifiez que le type d’antenne et le type de récepteur ne sont pas « inconnus ».

Une fois que les fichiers RINEX sont transférés et toutes les observations GPS importées dans la base de données du projet, un groupe de sélection portant le même nom que le fichier RINEX est créé.

Lorsque des fichiers RINEX sont importés dans un projet Trimble Geomatics Office, vous pouvez utiliser Trimble Geomatics Office pour traiter les lignes de base GPS (si le module Traitement de lignes de base est installé sur votre ordinateur) et utiliser le module Timeline pour visualiser les observations GPS brutes (si le module Traitement de lignes est installé sur votre ordinateur).


Classes de Survey Controller et comment elles sont traitées par Trimble Geomatics Office

Lorsque des points sont créés dans Trimble Survey Controller, une classe de point leur est assignée. Lorsque vous importez des points dans un projet, Trimble Geomatics Office utilise cette classe ainsi que la méthode utilisée pour déterminer la position d’un point afin d’assigner une qualité de point à ce point. Le recalcul utilise la qualité pour déterminer la position calculée du point.

La table suivante indique les classes assignées aux points par Trimble Survey Controller ainsi que la façon dont elles sont traitées par Trimble Geomatics Office au moment de l’importation.

Classe	Description	Ce qui se passe au moment de l’importation
Contrôle	Une classe de contrôle est la qualité la plus élevée que vous pouvez assigner à un point dans Trimble Survey Controller.	Importation en tant que coordonnées de contrôle pour le point.
Normal	Une classe de « normal » est assignée à tous les points mesurés à l’exception des points implantés. Les points saisis au clavier peuvent également se voir assigner cette classe.	Importation des observations et des coordonnées pour le point. Les coordonnées se voient assigner une qualité de levé.
Implanté	Une classe « d’implanté » est assignée à tous les points mesurés lors de l’implantation.	Importation des observations et détails d’implantation pour un point. Vous pouvez afficher les détails d’Implanté dans la fenêtre Propriétés, tabulation Implantation.
Visée arrière	Une classe de visée arrière est assignée aux points définis comme point de visée arrière sur le terrain.	Dans Trimble Geomatics Office, une classe de visée arrière est traitée de la même manière qu’une classe de normal.
Vérification	Une classe de vérification est assignée à une observation de vérification, ou à une position saisie au clavier comme point en double et enregistrée comme point de vérification.	Importation des observations de classe « Activées comme vérification ».
Supprimé	Une classe « de supprimé » est assignée à un point écrasé ou supprimé dans Survey Controller.	Les points de classe « supprimé » ne sont pas importés dans le projet Trimble Geomatics Office. Une liste des points supprimés des fichiers .dc se trouve dans le rapport généré lors de l’importation d’un fichier.


Qualités assignées aux coordonnées et observations d'un fichier .dc à l'aide des classes Survey Controller

Lorsque des positions GPS sont enregistrées dans un logiciel Trimble Survey Controller, elles peuvent être dérivées de différentes techniques de mesure ou de la saisie au clavier de coordonnées. La qualité attribuée à un point dépend de la méthode d’observation ou de la qualité attribuée au point lors de la saisie au clavier.

La table suivante indique les qualités assignées aux coordonnées à l’aide des classes Trimble Survey Controller.

Méthode des coordonnées	Qualité assignée
Contrôle saisi au clavier	Contrôle
Contrôle saisi normal	Levé
RTK fixe	Levé
RTK flottant	Mappage
Code L1	Mappage
Autonome	Inconnu
WAAS	Mappage
Fixe à superficie large	Levé
Flottant à superficie large	Mappage
Code RTCM	Mappage
Copie	Qualité de point copié
NOTE : Dans le cas de fichiers .dc de Trimble Survey Controller version 7.5 ou ultérieure, si vous avez utilisé la touche Ici pour déterminer la position de la base, la méthode Autonome est écrite et la position importée avec une qualité inconnue. Cela signifie que tous les points mesurés de la base auront une qualité inconnue.

Si vous avez effectué une calibration du site GPS et que vous êtes sûr des résultats, vous pouvez augmenter la qualité de la base à Levé, à la suite de quoi tous les points mesurés de la base se verront attribuer cette qualité.

Dans le cas de fichiers .dc de Trimble Survey Controller de version antérieure à la version 7.5, si vous utilisez la touche Ici pour déterminer la position de la base, celle-ci se verra attribuer une qualité de Levé.

La table suivante indique les qualités assignées aux observations à l’aide des classes Trimble Survey Controller.

Méthode d’observation	Qualité assignée
Conventionnelle	Levé
Télémètre à laser	Mappage
L1 fixe	Levé
L1 flottant	Mappage
Code L1	Mappage


Importation de fichiers Leica GSI-16

Lors de l’importation de fichiers de format Leica GSI-16, Trimble Geomatics Office lit uniquement les Blocs de mesure ainsi que les mots de REM associés à ces blocs. L’opération d’importation utilise les mots spécifiques (enregistrements) disponibles pour obtenir des informations sur la station. Les Index de mots suivants sont lus et interprétés lors de l’opération d’importation :

Index	Signification	Index	Signification
11	Id Point (identificateur de Bloc de mesure)	82	Nord du point
21	Angle horizontal	83	Elévation du point
22	Angle vertical	84	Est de la station
31	Distance inclinée	85	Nord de la station
32	Distance horizontale	86	Elévation de la station
33	Distance verticale	87	Hauteur de réflecteur
71…79	Mots de remarque (utilisés pour les codes et les notes)	88	Hauteur d'instrument
81	Est du point
Sur la base de ces Index de mots, Trimble vous recommande de configurer sur votre instrument Leica les masques suivants de données de station et d’enregistrement de données de mesure :

Masque de données de station	Masque de données de mesure
Id Point (11)	Id Point (11)
Est de la station (84)	Angle horizontal (21)
Nord de la station (85)	Angle vertical (22)
Elévation de la station (86)	Distance inclinée (31)
Hauteur d’instrument (88)	Constante ppm et prisme (51)
Code (71)	Hauteur de réflecteur (87)
Note (72)	Code (71)
	Note (72)
NOTE : Lors de l’importation, suppose que la première observation enregistrée après un enregistrement de station est une observation au point de visée arrière. Le premier mot de remarque (71) est interprété comme le code de point, et tous les autres mots de remarque (72...79) sont interprétés comme Enregistrements de notes.


Option Fichiers de niveaux numériques (*.dat, *.raw)

Les Fichiers de niveaux numériques contiennent des données enregistrées dans un niveau numérique.

Les fichiers de données des niveaux numériques suivants sont pris en charge :

  Niveau numérique DiNi Trimble

•	Leica NA 2002/3003

Lorsqu’un Fichier de niveaux numériques est importé, le dialogue Importation de niveaux numériques s’affiche.

NOTE: Il faut transférer les fichiers de données d’un niveau numérique Zeiss DiNi sous le format M5.


Exportation de fichiers NGS Bluebook

L’Exportation de fichiers NGS Bluebook vous permet de générer des fichiers National Geodetic Survey (NGS) Bluebook à partir de Trimble Geomatics Office.

Les fichiers Bluebook sont définis par le gouvernement des Etats-Unis et sont requis pour les levés soumis au National Spatial Reference System (système national de référence spatiale). Les règles relatives à ces levés sont reprises dans un libre blanc, et soumettre ces levés au système s’appelle le « bluebooking ».

L’utilitaire NGS Bluebook crée les fichiers bluebook suivants pour les lignes de base GPS :

  NGS

  B – informations sur le projet

  G – informations sur les lignes de base

  SERFIL – un fichier ASCII comprenant le numéro de série et l’ID de la station ainsi que les noms des stations d’un projet existant

Le dialogue Fichiers NGS Bluebook utilise les données associées au groupe de lignes de base indépendantes traitées sélectionnées dans votre projet. Avant d’exporter vos fichiers NGS, vous devez terminer l’ensemble des travaux sur le terrain, le traitement et le contrôle de la qualité. Une fois que vous avez sélectionné les lignes de base et initié l’exportation, vous ne pouvez pas ajouter d’autres lignes de base au dialogue Bluebooking sans avoir à recommencer le processus d’exportation depuis le début.

NOTE : Pour une description des données requises pour chaque champ du dialogue Exportation de fichiers NGS Bluebook, référez-vous à « Base de données de formats d’entrée et spécifications du National Geodetic Survey (NGS) ». La référence NGS est disponible à l’adresse suivante : http://www.ngs.noaa.gov/FGCS/BlueBook/


Dialogue Fichiers NGS Bluebook

Le filtre d’exportation de fichiers NGS Bluebook vous permet de générer des fichiers National Geodetic Survey (NGS) Bluebook à partir de Trimble Geomatics Office.

Le dialogue Fichiers NGS Bluebook utilise les données associées au groupe de lignes de base indépendantes traitées sélectionnées dans votre projet. Avant d’exporter vos fichiers NGS, vous devez terminer l’ensemble des travaux sur le terrain, le traitement et le contrôle de la qualité. Une fois que vous avez sélectionné les lignes de base et initié l’exportation, vous ne pouvez pas ajouter d’autres lignes de base au dialogue Fichiers NGS Bluebook sans avoir à recommencer le processus d’exportation depuis le début.

Pour accéder au dialogue Fichiers NGS Bluebook :

1.	Sélectionnez Fichier / Exporter. Le dialogue Exporter s'affiche.

2.	Sous la tabulation Levé, sélectionnez Fichiers NGS Bluebook.

NOTE : Si vous avez auparavant sauvegardé une session de « bluebooking », il vous est demandé si vous souhaitez charger les informations sauvegardées.

Utilisez les tabulations du dialogue Exportation de fichiers NGS Bluebook pour entrer des informations sur le projet.

Tabulation projet

Cette tabulation contient des informations générales sur le projet.

Tabulation station

Cette tabulation contient une liste de toutes les stations associées aux lignes de base sélectionnées :

  Remplissez l’ID à 4 caractères et le numéro de série de la station à l’aide des boutons de Création automatique. L’ID est créée à l’aide des quatre premiers caractères du nom de la station. Les numéros de série sont créés par séquence en commençant à 0001 ou par le premier numéro disponible.

  La colonne Code état/pays est remplie automatiquement à partir des informations qui se trouvent sous la tabulation Projet.

  Sélectionnez l’ordre dans la liste.

  La colonne Type indique par défaut un schéma principal de positionnement relatif par satellite interférométrique A puisque les données soumises au « bluebooking » proviennent de méthode de collecte GPS.

  Utilisez le bouton SERFIL pour importer les ID à 4 caractères et les numéros de série des stations existantes à l’aide d’un SERFIL d’un projet précédent. Le SERFIL doit contenir le nom de station pour chaque station. Le format est légèrement différent du format NGS SERFIL, mais NGS peut quand même utiliser le fichier. Le format suivant est celui qui doit être utilisé :

ID à 4 caractères   Numéro de série de la station   Nom de la station

COWB   0004   COWBOY

Tabulation Informations session

Cette tabulation contient des informations sur chaque session de terrain des lignes de base sélectionnées. Les couleurs en alternance représente chaque session de ligne de base individuelle.

  L’Heure de début représente l’heure de début la plus tardive de tous les récepteurs d’une session, et il est possible qu’elle ne représente pas l’heure de début de chaque ligne de base individuelle. L’heure doit uniquement être utilisée comme référence lorsqu’elle permet de faire correspondre les sessions aux feuilles de terrain.

  La colonne Récepteur est remplie automatiquement avec le type et le numéro de série du récepteur.

  La colonne Antenne est remplie automatiquement avec le type d’antenne. Cependant, vous devez entrer le numéro de série. Une fois que le numéro de série de chaque antenne a été entré, il peut être sélectionné dans une liste déroulante.

  La colonne Hauteur d’antenne (mètres/APC L1) est remplie automatiquement avec la mesure verticale vraie vers le centre de phase L1, laquelle est calculée à partir de la hauteur d’antenne saisie dans le champ ou lors de l’entrée.

Tabulations Conditions météorologiques avant la session et Conditions météorologiques après la session

Les tabulations Conditions météorologiques avant la session et Conditions météorologiques après la session contiennent des informations sur les conditions météorologiques pour chaque occupation associée aux lignes de base sélectionnées. Les champs sont requis lors de la réalisation d’un levé d’ordre B ou supérieur. L’astérisque est automatiquement supprimé de l’en-tête de la tabulation lorsqu’un levé de 1er ordre est sélectionné dans la tabulation Projet.

Ce champ...	prend implicitement...
Bulbe sec	Celsius
Bulbe humide	Celsius
Humidité	Pourcentage
Pression	Millibars
  Le champ Code conditions météo est rempli en entrant le code directement ou à l’aide du bouton Editer champ courant.

  Le bouton Editer champ courant vous permet d’afficher et de sélectionner les codes de la description pour chaque variable.

  Le champ Copier à conditions météorologiques avant (après) la session est utilisé pour copier les mêmes données que celles de la tabulation Conditions météorologiques avant la session à la tabulation Conditions météorologiques après la session et vice versa.

Tabulation Divers

La tabulation Divers contient des informations générales sur le traitement.

Equipement

La bouton Equipement vous permet d’afficher, d’ajouter et d’éditer des informations sur chaque récepteur et antenne. Seul l’équipement associé au projet est listé.

OK

En cliquant sur le bouton OK, vous fermez le dialogue Fichiers NGS Bluebook et sauvegardez les informations dans chaque tabulation. Lorsque vous entrez une grande quantité de données, vous pouvez sauvegarder votre travail et y revenir plus tard.

Exporter

Le bouton Exporter n’est disponible qu’une fois que tous les champs requis ont été remplis. Le bouton Exporter vous permet d’entrer un nom et de sélectionner un emplacement pour chaque fichier. L’emplacement par défaut est Trimble Geomatics Office\Projects\<Nom du projet>\Export.

NOTE : Au fur et à mesure que les champs requis sont remplis, le signe * dont leurs noms sont assortis est supprimé. Lorsqu’il n’y a plus de champ à remplir dans le dialogue, le bouton Exporter devient disponible.

NOTE : Pour une description des données requises pour chaque champ du dialogue Exportation de fichiers NGS Bluebook, référez-vous à « Base de données de formats d’entrée et spécifications du National Geodetic Survey (NGS) ».  La référence NGS est disponible à l’adresse suivante : http://www.ngs.noaa.gov/FGCS/BlueBook/


Format Trimble Data Exchange

 Comment

Le format Trimble Data Exchange a été conçu par Trimble Navigation Limited pour réaliser les opérations suivantes :

  Transférer des données de levé entre les projets Trimble Geomatics Office

  Exporter des données de levé à d’autres applications pour des ajustements de réseau ou autres opérations géodésiques.

NOTE : Vérifiez que votre logiciel d’ajustement de réseau prend en charge ce format.

Pour importer ou exporter des données à l’aide de ce format, sélectionnez l’option Format Trimble Data Exchange dans le dialogue Importer ou Exporter.

Les informations suivantes sont enregistrées dans un fichier ASCII :

Stations et coordonnées	Vecteurs GPS
Observations terrestres	Observations laser
Informations de nivellement	Informations sur les azimuts
Le fichier est organisé selon les sections suivantes :

[Généralités]

Cette section fournit des informations générales sur le contenu du fichier, le projet auquel il appartient et les unités utilisées pour les différents champs.

[Stations]

Cette section comprend le nom et les coordonnées de tous les points sélectionnés pour le transfert. Les coordonnées de cette section sont toujours dans le datum local. Les observations individuelles se trouvent dans des enregistrements de station qui comprennent la latitude, la longitude, le nord et l’est. La hauteur orthométrique et la hauteur d’ellipsoïde peuvent également apparaître dans chaque enregistrement.

L’enregistrement de station contient également des informations permettant de déterminer quels composants de coordonnées sont identifiés comme contrôles, ainsi que des informations sur le code de caractéristiques assigné à la station.

[GPS]

Cette section comprend les lignes de base de vecteur GPS sélectionnées pour le transfert. Les observations individuelles se trouvent dans les enregistrements de vecteurs qui comprennent :

Noms de station et vers station	Composants de vecteur
Information sur la covariance	Hauteurs d'antenne
Type de traitement	Ratio
RMS	Variance de référence
Heure et date des observations
[Terrestre]

Cette section comprend les observations terrestres sélectionnées pour le transfert. Les observations individuelles se trouvent dans les enregistrements TerrObs qui comprennent :

Point d'instrument	Point de visée arrière
Point de visée avant	Angle horizontal et erreur
Angle vertical et erreur	Distance inclinée et erreur
Constante prisme	Hauteur d'instrument
Heure et date d’observation (si disponibles)	Hauteur de destination
[Laser]

Cette section comprend les observations laser sélectionnées pour le transfert. Les observations individuelles se trouvent dans les enregistrements LaserObs qui comprennent :

Point d'instrument	Déclinaison magnétique
Azimut magnétique	Distance inclinée
Angle vertical	Hauteur de destination
Hauteur d'instrument	Point de destination
Heure et date d’observation (si disponibles)
[Niveau]

Cette section comprend les observations de nivellement sélectionnées pour le transfert. Les observations individuelles se trouvent dans les enregistrements LevelObs qui comprennent :

Points au début et à la fin de la série de niveaux	Delta élévation et erreur
Distance (si disponible)	Nombre de tours (si disponible)
Heure et date d’observation (si disponibles)

[Azimut]

Cette section contient les enregistrements d’azimuts qui comprennent les observations individuelles.

NOTE: Lorsque vous exportez les fichiers Trimble Data Exchange Format de Trimble Total Control, Trimble Geomatics Office n’importe les données que de la section de coordonnées entrées au clavier. Dans ce cas, les coordonnées ayant les qualités de coordonnées de contrôle ou ajustées sont importées avec la qualité de contrôle.


Fichiers de projet Terramodel (*.pro)

Trimble Geomatics Office prend en charge le format de fichier de projet Terramodel (*.pro).

Utilisez Trimble Geomatics Office pour importer un fichier de projet Terramodel (*.pro) dans un projet Trimble Geomatics Office:

1.	Sélectionnez Fichier / Importer.

le dialogue Importer s’affiche.

2.	Sélectionnez la tabulation CAD / ASCII.

3.	Sélectionnez l’option Fichiers de projet Terramodel (*.pro).

Le dialogue Ouvrir s’affiche.

4.	Sélectionnez le fichier à importer et cliquez sur Ouvrir.

NOTE: Seulement les points, les lignes et les couches sont importés des fichiers de projet Terramodel (*.pro).


Logiciel TDS - Vue d'ensemble

Voir aussi

Trimble Geomatics Office prend en charge les fichiers de logiciel TDS suivants:

  TDS Survey Pro CE

  TDS 3600 sur carte

  TDS 5600 sur carte

Vous pouvez importer et exporter les fichiers TDS de votre PC, ou les transférer directement vers et depuis le logiciel TDS utilisant les options d’importation et d’exportation Périphériques de levé.

NOTE: Trimble Geomatics Office ne prend pas en charge les fichiers de contrôles TDS Il faut importer les points de tout fichier de contrôle dans l’étude principale avant le transfert des fichiers. Pour plus amples informations, référez-vous à votre documentation TDS.

NOTE: Trimble Geomatics Office ne prend pas en charge les caractéristiques et les attributs TDS.

TDS Survey Pro CE

Trimble Geomatics Office prend en charge les fichiers suivants de TDS Survey Pro CE:

Nom	Extension	Importation	Exportation	Description
Etude	*.job			Comprend des points et des informations de coordonnées. Importez un fichier d’étude (*.job) vous ne voulez importer des points que d’un projet TDS Survey Pro CE.
Brut	*.raw		X	Comprend des informations d’observation. Importez un fichier brut (*.raw) lorsque vous voulez importer des points ainsi que des observations d’un projet TDS Survey Pro CE.
Description	*.txt	X		Un fichier de texte simple de codes de caractéristiques pour vous aider à entrer les descriptions de point dans TDS Survey Pro CE.
Grille de géoïde	*.ggf	X		Les séparations de l’ellipsoïde de référence WGS84 (TDS Survey Pro CE de version 3.5 ou ultérieure)

Pour exporter:

  une définition de route (*.job, *.rd5, *.tp5) de RoadLink, référez-vous à l’Aide de RoadLink.

  un MNT (*.dtm) de DTMLink, référez-vous à l’Aide DTMLink.

Pour plus amples informations concernant ces fichiers, référez-vous à la documentation TDS.

En plus Trimble Geomatics Office fournit du support  de système de coordonnées pour toutes les versions du logiciel TDS Survey Pro CE. Ce support comprend la prise en charge des fonctions géodésiques améliorées dans le TDS Survey Pro CE 3.5 ou ultérieure, lors de l’utilisation avec un récepteur Trimble GPS. Tout fichier importé du TDS Survey Pro CE comprend une définition de système de coordonnées complète, identique à la définition dans un fichier Survey Controller (*.dc).

NOTE: Les observations GPS ne sont importées du TDS Survey Pro CE 3.5 ou plus récente que lorsqu’il est utilisé avec un récepteur Trimble GPS. Les observations terrestres de tout type de station totale sont prises en charge par toutes les versions.

Trimble 3600/5600 TDS

Trimble Geomatics Office prend en charge les fichier suivants du logiciel Trimble 3600/5600 TDS:

Nom	Extension	Importation	Exportation	Description
Coordonnées	*.cr5			Comprend des points et des informations de coordonnées. Importez un fichier de coordonnées (*.cr5) lorsque vous ne voulez importer que les points d’un projet Trimble 3600/5600 TDS.
Brut	*.rw5		X	Comprend des informations d’observations. Importez un fichier brut (*.rw5) lorsque vous voulez importez des points ainsi que des observations d’un projet Trimble 3600/5600 TDS.
Description	*.txt	X		Un fichier de texte simple de codes de caractéristiques pour vous aider à entrer les descriptions de point dans TDS Survey Pro CE.
Pour exporter:

  une définition de route (*.cr5, *.rd5, *.tp5) de RoadLink, référez-vous à l’Aide de RoadLink.

  un Modèle numérique de terrain (*.dtm) de DTMLink, référez-vous à l’Aide DTMLink.

Pour plus amples informations concernant ces fichiers, référez-vous à la documentation TDS.

 Les fichiers importés du logiciel Trimble 3600/5600 TDS ne comprennent pas des définitions de système de coordonnées complètes. Les fichiers sont importés utilisant des systèmes de coordonnées à facteur d’échelle seule, identiques aux fichiers Survey Controller (*.dc).

Lors de l’importation des fichiers, Trimble Geomatics Office définit le système de coordonnées utilisant le facteur d’échelle utilisé le plus récemment dans l’étude TDS Survey Pro CE.

NOTE: Lors de l’importation des fichiers de coordonnées Trimble 3600/5600 TDS, Trimble Geomatics Office importe le fichier de coordonnées (*.cr5), et puis importe les données and du fichier brut (*.rw5) associé s’il est dans le même dossier sur votre ordinateur. Si vous ne voulez importer que des coordonnées, vérifiez qu’un fichier brut avec le nom que le fichier de coordonnées importé n’existe pas dans le même dossier sur votre ordinateur. Il est également possible d’importer les fichiers bruts individuellement.


Fichiers d'étude TDS Survey Pro CE (*.job)

Trimble Geomatics Office prend en charge les données dans le format de fichier d’étude TDS Survey Pro CE (*.job).

Utilisez Trimble Geomatics Office pour:

 Importer  un fichier d’étude TDS Survey Pro CE dans un projet Trimble Geomatics Office.

 Exporter un projet Trimble Geomatics Office and un fichier d’étude TDS Survey Pro CE.

Vous pouvez également transférer les fichier directement vers et depuis le logiciel TDS Survey Pro CE utilisant les options d’importation et d’exportation Périphériques de levé.

Pour importer un fichier d’étude TDS Survey Pro CE:

1.	Sélectionnez Fichier / Importer.

Le dialogue Importer s’affiche.

2.	Sélectionnez la tabulation Levé.

3.	Sélectionnez l’option Fichier d’étude Survey Pro CE (*.job) et cliquez sur OK.

Le dialogue Ouvrir s’affiche.

4.	Sélectionnez le fichier à importer et cliquez sur Ouvrir.

NOTE: Le format de fichier d’étude TDS Survey Pro CE comprend des points et des coordonnées associées. Utilisez cette option lorsque vous ne voulez importer que des points d’un projet TDS Survey Pro CE.

Pour exporter le projet Trimble Geomatics Office courant dans un fichier d’étude TDS Survey Pro CE:

1.	Sélectionnez Fichier / Exporter.

Le dialogue Exporter s’affiche.

2.	Sélectionnez la tabulation Levé.

3.	Sélectionnez l’option Fichier d’étude TDS Survey Pro CE (*.job) et cliquez sur OK.

Le dialogue Enregistrer sous s’affiche.

4.	Sélectionnez un dossier dans lequel il faut enregistrer le fichier.

5.	Entrez un nom dans le champ Nom du fichier et cliquez sur Enregistrer.

NOTE: Si des points en double existent dans la sélection exportée, le fichier d’étude TDS Survey Pro CE ne sera pas créé. Vérifiez que tous les points exportés aient des noms uniques.


Fichiers bruts TDS Survey Pro CE (*.raw)

Utilisez Trimble Geomatics Office pour importer un fichier brut TDS Survey Pro CE dans un projet Trimble Geomatics Office. Vous pouvez également transférer des fichiers directement du logiciel TDS Survey Pro CE utilisant l’option d’importation Périphériques de levé. Le format de fichier brut TDS Survey Pro CE comprend toutes les informations dans un projet Survey Pro CE. Utilisez cette option lorsque vous voulez importer des points ainsi que des observations d’un projet TDS Survey Pro CE.

Pour importer un fichier brut TDS Survey Pro CE:

1.	Sélectionnez Fichier / Importer.

Le dialogue Importer s’affiche.

2.	Sélectionnez la tabulation Levé.

3.	Sélectionnez l’option Fichier brut Survey Pro CE (*.raw) et cliquez sur OK.

Le dialogue Ouvrir s’affiche.

4.	Sélectionnez le fichier à importer et cliquez sur Ouvrir.


Fichiers de coord Trimble 3600/5600 TDS (*.cr5)

Utilisez Trimble Geomatics Office pour:

 Importer  un fichier de coordonnées Trimble 3600/5600 TDS dans un projet Trimble Geomatics Office.

 Exporter un projet Trimble Geomatics Office dans un fichier de coordonnées Trimble 3600/5600 TDS.

Vous pouvez également transférer les fichier directement vers et depuis le logiciel Trimble 3600/5600 TDS utilisant les options d’importation et d’exportation Périphériques de levé.

Pour importer un fichier de coordonnées Trimble 3600/5600 TDS:

1.	Sélectionnez Fichier / Importer.

Le dialogue Importer s’affiche.

2.	Sélectionnez la tabulation Levé.

3.	Sélectionnez l’option Fichier de coordonnées Trimble 3600/5600 TDS (*.cr5) et cliquez sur OK.

Le dialogue Ouvrir s’affiche.

4.	Sélectionnez le fichier à importer et cliquez sur Ouvrir.

NOTE: Lors de l’importation des fichiers de coordonnées Trimble 3600/5600 TDS, Trimble Geomatics Office importe le fichier de coordonnées, puis importe le fichier de données brutes (*.rw5) associé s’il est dans le même dossier sur votre ordinateur. Si le fichier de données brutes n’est pas dans le même dossier que le fichier de coordonnées choisi, seulement des points sont importés.

Pour exporter le projet Trimble Geomatics Office courant dans un fichier de coordonnées Trimble 3600/5600 TDS:

1.	Sélectionnez Fichier / Exporter.

Le dialogue Exporter s’affiche.

2.	Sélectionnez la tabulation Levé.

3.	Sélectionnez l’option Fichier de coord Trimble 3600/5600 TDS (*.cr5) et cliquez sur OK.

Le dialogue Enregistrer sous s’affiche.

4.	Sélectionnez un dossier dans lequel il faut enregistrer le fichier.

5.	Entrez un nom dans le champ Nom du fichier et cliquez sur Enregistrer.

NOTE: Si des points en double existent dans la sélection exportée, ou un point a un nom alphanumérique, le fichier de coordonnées Trimble 3600/5600 TDS ne sera pas créé. Vérifiez que tous les points exportés aient des noms uniques.


Fichier brut Trimble 3600/5600 TDS (*.rw5)

Utilisez Trimble Geomatics Office pour importer un fichier brut Trimble 3600/5600 TDS dans un projet Trimble Geomatics Office. Vous pouvez également transférer des fichiers directement vers et depuis le logiciel Trimble 3600/5600 TDS utilisant les options d’importation et d’exportation Périphériques de levé. Le format de fichier brut Trimble 3600/5600 TDS comprend presque toutes les informations dans un projet Survey Pro CE. Utilisez cette option lorsque vous voulez importer des points ainsi que des observations d’un projet Trimble 3600/5600 TDS.

Pour importer un fichier brut Trimble 3600/5600 TDS:

1.	Sélectionnez Fichier / Importer.

Le dialogue Importer s’affiche.

2.	Sélectionnez la tabulation Levé.

3.	Sélectionnez l’option Fichier brut Trimble 3600/5600 TDS (*.cr5) et cliquez sur OK.

Le dialogue Ouvrir s’affiche.

4.   Sélectionnez le fichier à importer et cliquez sur Ouvrir.


Fichiers de description TDS (*.txt)

Le logiciel TDS Survey Pro CE et Trimble 3600/5600 TDS utilise les fichiers de description afin d’entrer automatiquement les descriptions des points stockés dans une étude. Pour plus amples informations concernant les fichiers de description, référez-vous à la documentation TDS.

Utilisez Trimble Geomatics Office pour:

 Exporter un fichier de description file (*.txt) de l’Editeur de codes de caractéristiques.

 Exporter un fichier de description (*.txt) existant au logiciel TDS Survey Pro CE ou Trimble 3600/5600 TDS.

 Exporter un fichier de description (*.txt) d’une Bibliothèque de codes de caractéristiques au logiciel TDS Survey Pro CE ou Trimble 3600/5600 TDS.

Pour exporter un fichier de description de l’Editeur de codes de caractéristiques, référez-vous à l’Aide de l’Editeur de codes de caractéristiques.

Pour exporter un fichier de description existant au logiciel TDS Survey Pro CE ou Trimble 3600/5600 TDS:

1.	Sélectionnez Fichier / Exporter.

Le dialogue Exporter s’affiche.

2.	Sélectionnez la tabulation Levé.

3.	Sélectionnez l’option Fichier de description au périphérique TDS (*.txt) et cliquez sur OK.

Le dialogue Sélectionner s’affiche.

4.	Sélectionnez le fichier à exporter et cliquez sur Ouvrir.

Le dialogue Enregistrer sous s’affiche.

5.	Sélectionnez un périphérique TDS et cliquez sur Ouvrir.

Les fichier est exporté.

Pour exporter un nouveau fichier de description d’une Bibliothèque de codes de caractéristiques au logiciel TDS Survey Pro CE ou Trimble 3600/5600 TDS:

1.	Sélectionnez Fichier / Exporter.

Le dialogue Exporter s’affiche.

2.	Sélectionnez la tabulation Levé.

3.	Sélectionnez l’option Bibliothèque d’attributs et caractérises au périphérique TDS (*.txt) et cliquez sur OK.

Le dialogue Sélectionner bibliothèque d’attributs et caractéristiques s’affiche.

4.	Sélectionnez la bibliothèque à exporter et cliquez sur OK.

Le dialogue Enregistrer sous s’affiche.

5.	Sélectionnez un périphérique TDS et cliquez sur Ouvrir.

Les fichier est exporté.


Zeiss - Vue d'ensemble

Trimble Geomatics Office prend en charge les fichiers M5 des instruments Zeiss suivants:

  Trimble 3300

  Trimble 3600 Elta

  Trimble 5600 Elta

 Trimble Elta GPS Controller

  Niveau numérique Trimble DiNi

Vous pouvez également importer ou exporter les fichiers M5 de votre PC, ou transférer les fichiers directement vers ou depuis l’instrument utilisant les options d’importation et d’exportation Périphériques de levé.

Les fichiers Zeiss sont pris en charge par Trimble Geomatics Office comme suit:

Format de fichier	Pris en charge
Elta M5  (Seulement en anglais et en allemand)	Oui
Rec 500, R4 et R5	Nom.  Le CD Trimble Geomatics Office comprend un programme pour la conversion entre les formats de fichier M5 et Rec 500 si nécessaire.
Elta Codelist	Non

Exportation

Trimble Geomatics Office exporte les noms de point de grille, les codes, les descriptions et les coordonnées aux fichiers M5. D’autres données, telles que les observations, ne sont pas exportées.

NOTE: Les coordonnées dans le fichier exporté sont écrites avec le même nombre de décimales que celles utilisées par le projet Trimble Geomatics Office, mais avec un maximum de trois décimales pour l’instrument Trimble 3300 et quatre décimales pour les instruments Trimble 3600/5600 Elta et Trimble Elta GPS Controller.

Importation

NOTE: Le format de fichier M5 ne prend pas en charge les définitions de système de coordonnées complètes. Les fichiers sont importés utilisant des systèmes de coordonnées à facteur d’échelle seul, identique à l’importation d’un système de coordonnées à facteur d’échelle seule dans un fichier Survey Controller (*.dc). Vous pouvez entrer un facteur d’échelle lorsque vous définissez le système de coordonnées d’importation dans Trimble Geomatic Office.

Unités

Le format M5 format a des unités de "m" (pour les Mètres et les Mètres namibiens) ou "pds" (pour les Pieds internationaux et les Pieds américains).

Lors de l’exportation à ou l’importation d’un fichier M5, si le projet et l’instrument n’utilisent pas les mêmes unités, Trimble Geomatics Office suppose que "m" dans le fichier M5 soient Mètres et que “pds” soient Pieds internationaux.

Niveau numérique Trimble DiNi

NOTE: Trimble Geomatics Office n’exporte pas à l’instrument de niveau numérique Trimble DiNi.

Pour plus amples informations concernant l’importation d’un instrument de niveau numérique Trimble DiNi, référez-vous à l’option d’importation Fichiers de niveau numérique (*.dat, *.raw).


Fichiers Trimble 3300/3600/5600 Elta M5 (*.dat)

Utilisez Trimble Geomatics Office pour:

 Importer  des fichiers Trimble 3300/3600/5600 Elta M5 (*.dat) dans un projet Trimble Geomatics Office.

 Exporter un projet Trimble Geomatics Office dans un fichier de coordonnées Trimble 3300/3600/5600 Elta M5 (*.dat).

Vous pouvez également transférer les fichiers directement vers et depuis le logiciel sur carte Trimble 3300, 3600 Elta ou 5600 Elta utilisant les options d’importation et d’exportation Périphériques de levé.

Vérifiez que les Options de fichier M5 pour les fichiers Trimble 3300/3600/5600 Elta M5 (*.dat) soient configurées avant l’importation ou l’exportation des fichiers Elta M5.

Pour importer les fichiers Elta M5 (*.dat):

1.	Sélectionnez Fichier / Importer.

Le dialogue Importer s’affiche.

2.	Sélectionnez la tabulation Levé.

3.	Sélectionnez l’option Trimble 3300/3600/5600 Elta M5 (*.dat).

4.	Cliquez sur Options pour ouvrir le dialogue Fichiers M5 et configurez le fichier.

5.	Cliquez sur OK pour accepter les paramètres et retourner au dialogue Importer.

6.	Cliquez sur OK.

Le dialogue Ouvrir s’affiche.

7.	Sélectionnez le fichier à importer et cliquez sur Ouvrir.

NOTES:

 	 Les coordonnées (relatives) locales (par exemple, “neh” ou “xys”) dans le fichier M5 sont converties aux coordonnées absolues lors de l’importation.

  	Après l’importation d’un fichier M5, toutes les constantes de prisme apparaissent dans Trimble Geomatics Office comme "?". Pour le changer, dans le champ Constante prisme dans la fenêtre Propriétés, entrez la différence entre la constante de prisme entrée dans le champ et la constante de prisme courante. Par exemple, si vous avez entré une constante de prisme de "20", et la constante de prisme courante est "5", entrez "–15".

  	Trimble Geomatics Office ne prend pas en charge des deltas d’implantation orthogonaux. Ceux-ci sont convertis aux deltas d’implantation polaires lors de l’importation. Trimble vous recommande de configurer l’instrument afin de stocker seulement les deltas d’implantation de coordonnées de grille.

  	Trimble Geomatics Office ne prend pas en charge les angles horizontaux qui s’augmentent à gauche. N’utilisez pas cette configuration sur l’instrument.

Configuration de l’instrument

Lors de l’importation des observations à 2 face moyennes d’un instrument Trimble 3600/5600 Elta, vérifiez que l’instrument soit configuré pour le stockage des Valeurs uniques seulement. Si des Valeurs uniques ainsi que moyennes sont sélectionnées, Trimble Geomatics Office calculera des valeurs incorrectes. En outre, si seulement Moyenne est sélectionnée, Trimble Geomatics Office affiche les observations moyennes comme des observations de Face 1.

Trimble Geomatics Office suppose que les instruments Trimble 3600/5600 Elta ont les corrections atmosphériques activées tout le temps. Ne désactivez pas ces corrections sur l’instrument.

Vous devez utiliser l’option de stockage R-MC sur les instruments Trimble 3600/5600 Elta  (lorsque possible) et l’option de stockage MEM/3 sur l’instrument Trimble 3300 afin d’assurer que vous ayez des données de coordonnées et d’observation pour vos calculs.

Stationnement

Afin d’assurer que Trimble Geomatics Office interprète les données correctement, avant la mesure, toujours effectuez une routine de stationnement sur l’instrument.

Les enregistrements de visée arrière supprimés de la Configuration de station sont pourtant importés dans Trimble Geomatics Office. Désactivez ces observations au moyen de la fenêtre Propriétés.

Trimble Geomatics Office ne prend pas en charge les visées arrières multiples. Lorsque les points de visée arrière multiples sont mesurés dans la routine de stationnement, Trimble Geomatics Office utilisera le dernier point comme la visée arrière.

Routines Elta non prises en charge

Les routines d’instrument Elta suivantes font la sortie des données spéciales qui ne sont pas prises en charge par Trimble Geomatics Office. Néanmoins, dans la plupart des situations Trimble Geomatics Office effectuera l’importation des observations de ces routines:

  Plan vertical

  Superficie

  Distances de liaison

  Transformations

  Roadline Lite

  Reprises multiples

  3D Plane

Pour exporter le projet Trimble Geomatics Office courant dans un fichier Elta M5 (*.dat):

1.	Sélectionnez Fichier / Exporter.

Le dialogue Exporter s’affiche.

2.	Sélectionnez la tabulation Levé.

3.	Soulignez l’option Fichiers Trimble 3300/3600/5600 Elta M5 (*.dat).

4.	Cliquez sur Options pour ouvrir le dialogue Options de fichier M5  et configurer le fichier.

5.	Cliquez sur OK pour accepter les paramètres et retourner au dialogue Exporter.

6.	Cliquez sur OK.

Le dialogue Enregistrer sous s’affiche.

7.	Sélectionnez un dossier dans lequel il faut enregistrer le fichier.

8.	Entrez un nom du fichier dans le champ Nom du fichier et cliquez sur Enregistrer.


Fichier Trimble Elta GPS Controller M5 (*.dat)

Dans Trimble Geomatics Office, vous pouvez:

  Importer les fichiers Elta GPS Controller M5 dans un projet Trimble Geomatics Office.

  Exporter d’un projet Trimble Geomatics Office dans un fichier de coordonnés Elta GPS Controller M5.

Vous pouvez également transférer les fichiers directement vers et depuis le logiciel Trimble Elta GPS Controller M5 utilisant les options d’importation et d’exportation Périphériques de levé.

Vérifiez que les Options de fichier M5 pour les fichiers Trimble Elta GPS Controller M5 soient configurées avant l’importation ou l’exportation des fichiers Elta GPS Controller M5.

Pour importer les fichiers Elta GPS Controller M5:

1.	Sélectionnez Fichier / Importer.

Le dialogue Importer s’affiche.

2.	Sélectionnez la tabulation Levé.

3.	Sélectionnez l’option Fichiers Trimble Elta GPS Controller M5 (*.dat).

4.	Cliquez sur Options pour ouvrir le dialogue Options de fichier M5 et configurez le fichier.

5.	Cliquez sur OK pour accepter les paramètres et retourner au dialogue Importer.

6.	Cliquez sur OK.

Le dialogue Ouvrir s’affiche.

7.	Sélectionnez le fichier à importer et cliquez sur Ouvrir.

NOTE: Trimble Geomatics Office importe toutes les coordonnées dans le fichier M5 comme des points à grille seuls. Afin d’assurer que les coordonnées dans Trimble Geomatics Office correspondent aux coordonnées sur le contrôleur Trimble Elta GPS, vous devez effectuez une calibration (transformation) sur le terrain au début de votre levé.

Pour exporter le projet Trimble Geomatics Office courant dans un fichier Elta GPS Controller M5:

1.	Sélectionnez Fichier / Exporter.

Le dialogue Exporter s’affiche.

2.	Sélectionnez la tabulation Levé.

3.	Soulignez l’option Fichiers Trimble Elta GPS Controller M5 (*.dat).

4.	Cliquez sur Options pour ouvrir le dialogue Options de fichier M5  et configurer le fichier.

5.	Cliquez sur OK pour accepter les paramètres et retourner au dialogue Exporter.

6.	Cliquez sur OK.

Le dialogue Enregistrer sous s’affiche.

7.	Sélectionnez un dossier dans lequel il faut enregistrer le fichier.

8.   Entrez un nom du fichier dans le champ Nom du fichier et cliquez sur Enregistrer.


Dialogue Options de fichier M5

Utilisez le dialogue Options de fichier M5 pour configurer les fichiers suivants pour l’importation et l’exportation:

  Fichiers Trimble 3300/3600/5600 Elta M5

  Fichiers Trimble Elta GPS Controller M5

Spécifiez le type d’instrument, les marques de fichier, les paramètres de coordonnées et les paramètres de points en double  (pour l’importation).

Pour accéder au dialogue d’options:

1.	Sélectionnez Fichier / Exporter (ou Importer).

2.	Sélectionnez la tabulation Levé.

3.	Sélectionnez l’une des options suivantes:

  Fichiers Trimble 3300/3600/5600 Elta M5 (*.dat)

  Fichiers Trimble Elta GPS Controller M5 (*.dat)

4.	Cliquez sur Options.

Marques de fichier

Sélectionnez l’une des options suivantes:

  Défaut. (Trimble recommande cette option)

Cliquez sur le bouton Expert pour afficher les Paramètres de format de ligne et l’Aperçu de format de l’option Défaut.

NOTE: Les valeurs par défaut sont différentes selon le type d’instrument sélectionné.

  Défini de l’utilisateur

L’option Défini de l’utilisateur vous permet de changer les paramètres de format de ligne et d’afficher les modifications dans la section Aperçu de format. Sélectionnez Défini de l’utilisateur, et cliquez sur Expert. Dans le dialogue Options de marquage avancées:

a.	Spécifiez la position du caractère de début du nom de point, du code et de la description.

b.	Spécifiez les largeurs de champ du nom de point, du code et de la description.

c.	Entrez un Numéro de marque.

NOTE: Si les paramètres de format de ligne définis par l’utilisateur sont non valides, Trimble Geomatics Office demande des paramètres de configuration valides.

Paramètres de coordonnées

Dans les listes déroulantes ordre de coordonnées et Elévation, sélectionnez l’ordre des coordonnées et les étiquettes de coordonnés dans le fichier à importer ou exporter.

NOTES:

Les champs disponibles dépendent du type d’instrument sélectionné.

Les noms de point, les codes et les descriptions qui sont plus longs que l’espace disponible seront tronqués lors de l’exportation.

Les points avec des noms alphanumériques ne sont pas exportés à l’instrument Trimble 3300.


Points - Vue d'ensemble

Les points sont les éléments principaux d’un projet Trimble Geomatics Office. Des points sont ajoutés au projet à chaque fois que vous importez des données, ou vous pouvez ajouter un point à l’aide du dialogue Insérer point ou du dialogue Insérer points par azimut et distance.

Les points sont visibles dans la vue de levé ainsi que dans la Vue en plan. Pour afficher les détails d’un point, sélectionnez-le et ouvrez la fenêtre Propriétés.

Un point comporte une position calculée déterminée par un recalcul. Cette position est la meilleure position pour le point et elle est déterminée en utilisant toute observation vers le point ou coordonnée saisie au clavier.

Dans la Vue de levé, les points apparaissent sous forme de points, toute observation entre eux étant également représentée. Dans la Vue en plan, les points sont affichés selon leur style de point. La base de données autorise l’utilisation de points avec des noms en double.


Dialogue Insérer points

Utilisez ce dialogue pour insérer des points dans la base de données. Ces champs changent en fonction de l’option que vous sélectionnez dans le champ Type de coordonnées.

Pour y accéder, effectuez l’une des opérations suivantes :

  Dans la Vue de levé, sélectionnez Insérer / Points.

  A partir de la Vue en plan, effectuez l’une des opérations suivantes :

  Sélectionnez Insérer / Points.

  Cliquez sur l'outil Points [image].

Insérer

Cliquez sur Insérer pour insérer le point dans la base de données. Le dialogue demeure disponible afin que vous puissiez ajouter un (nouveau) point supplémentaire.

NOTE : Lorsque vous ajoutez la première position de grille ou GPS à un projet dont la projection par défaut n’est pas définie (Transverse Mercator), Trimble Geomatics Office présente un dialogue approprié pour vous aider à initialiser correctement la projection.

Fermer

Cliquez sur Fermer pour terminer l’insertion de points et quitter le dialogue.

Nom

Entrez un nom à l’aide du clavier. Vous pouvez entrer un nom comprenant jusqu’à 16 caractères.

Vous ne pouvez pas utiliser les caractères suivants dans les noms :

  accolades {}

  crochets []

  parenthèses ()

  signe de pourcentage %

  signe égal =

  guillemets ″ ″, «  » ou ‘ ’

NOTE : Lors de l’insertion de plusieurs points à la suite, les noms sont automatiquement incrémentés lorsque le nom du point précédent se termine par un nombre.

NOTE : Pour configurer le nom du point par défaut dans le dialogue Propriétés du projet, sélectionnez la tabulation Unités et format et utilisez Démarrer le nom du point auto du champ.

Groupe Détails de coordonnées

Sélectionnez la qualité appropriée pour chaque composant de coordonnées (horizontal, d’élévation, de hauteur). Ces qualités sont utilisées par un recalcul pour déterminer la meilleure position pour le point.

Le groupe Détails des coordonnées change suivant que vous sélectionnez ou non l’option WGS-84, Grille ou Local dans ce dialogue. Chaque groupe Détails de coordonnées différent correspond à un affichage des coordonnées.

Si vous avez sélectionné l’option Grille, les champs suivants s’affichent :

Nord	Utilisez le remplissage du champ ou entrez une valeur au clavier. Si vous utilisez le remplissage du champ, les champs Nord et Est sont remplis avec la position sur laquelle vous avez cliqué.
Est	Utilisez le remplissage du champ ou entrez une valeur au clavier. Si vous utilisez le remplissage du champ, les champs Nord et Est sont remplis avec la position sur laquelle vous avez cliqué.

Si vous sélectionnez l’option WGS-84 ou Local, et avez configuré le paramètre Affiche WGS-84 sous (dans la tabulation Unités et format du dialogue Propriétés du projet) sur L,L,H, les champs suivants s’affichent :

Latitude	Utilisez le remplissage du champ ou entrez une valeur au clavier. Si vous utilisez le remplissage du champ, les champs Latitude et Longitude sont remplis avec la position sur laquelle vous avez cliqué.
Longitude	Utilisez le remplissage du champ ou entrez une valeur au clavier. Si vous utilisez le remplissage du champ, les champs Latitude et Longitude sont remplis avec la position sur laquelle vous avez cliqué.

Les champs suivants s’affichent quelle que soit l’option sélectionnée :

Elévation	Entrez une valeur au clavier ou cliquez sur un point existant. Si vous cliquez sur un point existant, la valeur d’Elévation du point en question est transférée dans le champ Elévation.
Hauteur	Utilisez le remplissage du champ ou entrez une valeur au clavier. Si vous cliquez sur un point existant, la valeur de Hauteur du point en question est transférée dans le champ Hauteur.

ASTUCE : Pour placer un point le long d’une ligne d’observation de la Vue de levé ou de toute ligne de la Vue en plan, maintenez la touche Ctrl enfoncée lors de l’utilisation du remplissage du champ.

NOTE : Si le paramètre Afficher WGS-84 sous de la tabulation Unités et format du dialogue Propriétés du projet est configuré sur X,Y,Z, alors les champs X,Y,Z s’affichent. Utilisez le remplissage du champ ou entrez une valeur au clavier. Si vous utilisez le remplissage du champ, les champs sont remplis avec la position sur laquelle vous avez cliqué.

Type de coordonnées

Dans ce champ, sélectionnez l’option (WGS-84, Local ou Grille) que vous souhaitez utiliser pour enregistrer les coordonnées dans la base de données. La sélection du type de coordonnées entraîne la modification de l’affichage des coordonnées qui reprend les paramètres du type de coordonnées.

NOTE : Vous ne pouvez éditer les champs du groupe Détails de coordonnées que si les paramètres du type de coordonnées et de l’affichage des coordonnées sont identiques.

Groupe Détails CAD

Ce groupe contient tous les champs d’un point qui sont visibles dans la Vue en plan.

Code de caractéristiques

Entrez une valeur au clavier ou cliquez sur un point existant. Si vous cliquez sur un point existant, la valeur de Code de caractéristiques du point en question est transférée dans le champ Code de caractéristiques.

Description

Entrez une valeur au clavier ou cliquez sur un point existant. Si vous cliquez sur un point existant, la valeur de Description du point en question est transférée dans le champ Description.

Couche

Effectuez l'une des opérations suivantes :

  Sélectionnez une couche existante dans la liste.

  Entrez une nouvelle couche à l’aide du clavier. Le dialogue Nouvelle couche s'affiche.

  Cliquez sur un point existant. Si vous cliquez sur un point existant, la valeur de Couche du point en question est transférée dans le champ Couche.

Style de point

Effectuez l'une des opérations suivantes :

  Sélectionnez un style de point existant dans la liste.

  Entrez un nouveau style de point à l’aide du clavier. Le dialogue Créer style de point s'affiche.

  Cliquez sur un point existant. Si vous cliquez sur un point existant, la valeur de Style de point du point en question est transférée dans le champ Style de point.


Commande Points par azimut et distance

Utilisez cette commande pour insérer des points par un azimut et une distance d’un point connu.

Pour y accéder, sélectionnez Insérer / Points par azimut et distance.

Le dialogue Insérer points par azimut et distance s’affiche.


Dialogue Insérer points par azimut et distance

Utilisez ce dialogue pour insérer des points par un azimut et une distance d’un point connu.

Pour y accéder, effectuez l’une des opérations suivantes :

  Sélectionnez Insérer / Points par azimut et distance.

  Dans la Vue en plan, cliquez sur l’outil Ajouter points par azimut et distance [image].

NOTE : Les qualités de ce nouveau point sont les mêmes que celles du point du champ De point. Si aucune delta élévation n’est spécifiée, les champs Elévation et Hauteur du nouveau point sont inconnus.

Nom de point

Entrez un nom de point à l’aide du clavier. Vous pouvez entrer un nom comprenant jusqu’à 16 caractères.

Vous ne pouvez pas utiliser les caractères suivants dans les noms :

  accolades {}

  crochets []

  parenthèses ()

  signe de pourcentage %

  signe égal =

  guillemets ″ ″, «  » ou ‘ ‘

De point

Entrez le nom du point existant à l’aide du clavier ou cliquez sur un point existant.

Azimut

Le champ dépend de la configuration du type d’azimut. Entrez une valeur d’azimut dans le format fixé pour le Format d’angle (en degrés) sous la tabulation Unités et format.

Distance

Ce champ dépend de la configuration du champ Afficher distance sous de la tabulation Unités et format. Si vous ne spécifiez pas d’unités, les unités courantes du projet sont utilisées ; cependant, vous pouvez spécifier m, pdi (pieds internationaux) ou pdl (pieds américains).

Delta élévation

Cette valeur est ajoutée à l’élévation du point De point pour déduire une élévation pour le nouveau point. Une modification se soldant par un nombre positif signifie que l’élévation du nouveau point est supérieure à celle du point De point. Un résultat négatif signifie que l’élévation du nouveau point est inférieure à celle du point De point. Si vous ne spécifiez pas d’unités, les unités courantes du projet sont utilisées ; cependant, vous pouvez spécifier m, pdi (pieds internationaux) ou pdl (pieds américains). Si vous ne spécifiez aucune delta élévation, l’élévation du nouveau point est nulle.

Type d'azimut

Utilisez ce groupe pour sélectionner l’azimut de grille ou l’azimut avant de section normale.

 Grille – Sélectionnez cette option pour afficher l’Azimut de grille

  Avant section normale – Sélectionnez cette option pour afficher l’Azimut avant de section normale

Méthode

Utilisez ce groupe pour sélectionner la méthode utilisée par le logiciel pour préparer le dialogue après qu’un point a été inséré.

  Séquentielle – Lorsqu’un nouveau point est ajouté, cette option insère le nom du nouveau point dans le champ De point. Utilisez cette option pour entrer un cheminement.

  Radiale – Conserve le même nom pour le champ De point. Utilisez cette option pour entrer des rayonnements vers les points à partir du même point.

Joindre

Utilisez cette case à cocher pour tracer une ligne entre le point « de » et le nouveau point. La ligne est ajoutée à la couche courante et tracée dans le style de ligne par défaut de la couche courante. Elle n’est visible que dans la Vue en plan.


Dialogue Insérer azimut

Un azimut est utilisé pour orienter les observations terrestres. Utilisez ce dialogue pour insérer un azimut entre deux points.

Pour y accéder :

  Dans la vue de levé, sélectionnez Insérer / Azimut.

De point et Vers point

Entrez un nom de point existant dans ces champs à l’aide du clavier ou du remplissage du champ.

Azimut

Si vous entrez des points « de » et « vers » existants, une valeur calculée d’azimut est indiquée. Le cas échéant, vous pouvez éditer cette valeur et la qualité de l’azimut. L’azimut affiché dans ce champ dépend de l’affichage d’azimut que vous spécifiez ci-dessous.

Si un azimut existe déjà pour votre paire de points, il vous est demandé si vous souhaitez l’écraser.

Type d'azimut

Utilisez ce champ pour déterminer comment l’azimut est enregistré dans la base de données, c’est à dire soit comme azimut de grille soit comme azimut avant de section normale.

Grille/Avant section normale

Sélectionnez l’une de ces options pour déterminer comment vos coordonnées sont affichées dans le dialogue Insérer azimut, c’est à dire pour déterminer l’affichage d’azimut.

NOTE : Pour que vous puissiez éditer les champs de ce dialogue, le type d’azimut et l’affichage d’azimut doivent être identiques.

NOTE : Si le point « vers » ne comporte pas de coordonnée, la conversion entre l’azimut de grille et l’azimut avant de section normale ne peut pas être calculée et l’azimut est donc affiché comme étant Nul.

Insérer

Utilisez ce bouton pour insérer l’azimut dans le projet. L’azimut est affiché entre les deux points de définition et avec une flèche indiquant la direction.

Fermer

Utilisez ce bouton pour fermer le dialogue.


    Commande Lignes

Utilisez cette commande pour insérer des lignes dans la base de données.

Pour y accéder, effectuez l’une des opérations suivantes :

  Sélectionnez Insérer / Lignes.

Le dialogue Insérer lignes s’affiche avec la tabulation Ligne sélectionnée.


    Dialogue Insérer lignes

Utilisez ce dialogue pour insérer de nouvelles lignes (ligne, courbe ou arc) dans la base de données.

Pour y accéder, effectuez l’une des opérations suivantes :

Lignes	Sélectionnez l'outil Insérer lignes [image].   Sélectionnez Insérer / Lignes.
Courbes	Sélectionnez l’outil Insérer courbes [image].    Sélectionnez Insérer / Courbes.
Arcs	Sélectionnez l'outil Insérer arcs [image].    Sélectionnez Insérer / Arcs.

Le dialogue Insérer lignes s’affiche avec la tabulation appropriée sélectionnée.

Pour compléter ce dialogue, utilisez le remplissage du champ ou entrez les détails au clavier. Pour ajouter rapidement des lignes, double-cliquez sur le remplissage du champ pendant que vous vous trouvez dans le champ Vers point.

Vous pouvez modifier le type de ligne à tout moment. Dans ce cas, il est possible de joindre des séquences de lignes, courbes, arcs ou de toute combinaison de ces éléments.

Vous pouvez ajouter trois types de ligne à la base de données :

  lignes

  arcs

  courbes

Insérer

Cliquez sur Insérer pour ajouter des lignes dans le projet et gardez le dialogue ouvert pour ajouter de nouvelles lignes.

La valeur du point « de » devient la valeur du point « vers » prêt pour la prochaine ligne.


    Lignes - Vue d'ensemble

Plus amples informations

Comment ...	Dialogue	Voir aussi
Vue d'ensemble

Vous pouvez insérer des lignes entre les points dans Trimble Geomatics Office. Insérez des lignes dans votre projet pour définir les limites de lignes (par exemple lignes de barrière, bâtiments et pipelines).

Des lignes s’affichent dans la fenêtre graphique lorsque vous visualisez votre projet dans la Vue en plan. Pour afficher les propriétés de la ligne, sélectionnez-la et ouvrez la fenêtre Propriétés.

Les lignes sont affichées selon leur style de ligne. Vous pouvez utiliser les styles de ligne Trimble ou créer les vôtres.

Les lignes peuvent être exportées dans certains formats (par exemple les formats AutoCAD et TRIMMAP).


    Arcs - Vue d'ensemble

Plus amples informations

Comment ...	Dialogue	Voir aussi
Vue d'ensemble

Vous pouvez créer des arcs afin d’afficher des arcs circulaires réguliers entre les points.

Des arcs s’affichent dans la fenêtre graphique lorsque vous visualisez votre projet dans la Vue en plan. Pour afficher les propriétés de l’arc, sélectionnez-le et ouvrez la fenêtre Propriétés.

Les arcs sont affichés selon leur style de ligne. Vous pouvez utiliser les styles de ligne Trimble ou créer les vôtres.

Les arcs sont exportés dans certains formats (par exemple les formats AutoCAD et TRIMMAP).


    Courbes - Vue d'ensemble

Plus amples informations

Comment ...	Dialogue	Voir aussi
Vue d'ensemble

Insérez des courbes pour créer des courbes aussi bien adaptées que possible entre les points. Utilisez les courbes pour définir les limites qui ne sont pas des lignes droites (par exemple cours d’eau ou chemins).

Des courbes s’affichent dans la fenêtre graphique lorsque vous visualisez votre projet dans la Vue en plan.

Les courbes sont affichées selon leur style de ligne. Vous pouvez utiliser les styles de ligne Trimble ou créer les vôtres.

Chaque fois que vous insérez une courbe à partir d’un point utilisé dans une autre courbe, Trimble Geomatics Office recalcule la courbe existante pour garantir la régularité de la courbe affichée.

Les courbes sont exportées dans certains formats (par exemple les formats AutoCAD et TRIMMAP).


    Dialogue Insérer lignes - Tabulation Ligne

Utilisez ce dialogue pour insérer des lignes dans la base de données.

Pour y accéder à partir de la Vue en plan, effectuez l'une des opérations suivantes :

  Sélectionnez Insérer / Lignes.

  Sélectionnez l'outil Insérer lignes [image].

Le dialogue Insérer lignes s’affiche avec la tabulation Ligne sélectionnée.

De point

Effectuez l'une des opérations suivantes :

  Utilisez le remplissage du champ pour sélectionner le premier point que vous souhaitez utiliser pour définir la ligne.

  Entrez le nom de point du premier point que vous souhaitez utiliser pour définir la ligne.

Vers point

Effectuez l'une des opérations suivantes :

  Utilisez le remplissage du champ pour sélectionner le second point que vous souhaitez utiliser pour définir la ligne.

  Entrez le nom de point du second point que vous souhaitez utiliser pour définir la ligne.

Couche

Effectuez l'une des opérations suivantes :

  Sélectionnez la couche dans laquelle vous souhaitez ajouter la ligne.

  Entrez un nouveau nom de couche pour créer une nouvelle couche. Le dialogue Nouvelle couche s'affiche.

Style de ligne

Sélectionnez le style de ligne que vous souhaitez utiliser pour afficher la ligne. Les styles disponibles sont les styles de ligne du dialogue Styles CAD.

Insérer

Cliquez sur Insérer pour insérer la ligne. Le point sélectionné dans le champ Vers point est sélectionné par défaut dans le champ De point. Vous pouvez modifier cette sélection si nécessaire.

Fermer

Cliquez sur Fermer pour arrêter l’insertion de lignes.


    Commande Arcs

Utilisez cette commande pour insérer des arcs dans la base de données.

Pour y accéder à partir de la Vue en plan, effectuez l'une des opérations suivantes :

  Sélectionnez Insérer / Arcs.

  Sélectionnez l'outil Insérer arcs [image].

Le dialogue Insérer lignes s’affiche avec la tabulation Arc sélectionnée.


    Dialogue Insérer lignes - Tabulation Arc

Utilisez ce dialogue pour insérer des arcs dans la base de données.

Pour y accéder à partir de la Vue en plan, effectuez l'une des opérations suivantes :

  Sélectionnez Insérer / Arcs.

  Sélectionnez l'outil Insérer arcs [image].

Le dialogue Insérer lignes s’affiche avec la tabulation Arc sélectionnée.

De point

Effectuez l'une des opérations suivantes :

  Utilisez le remplissage du champ et cliquez sur le point approprié.

  Utilisez le clavier pour entrer le nom du point.

Vers point

Effectuez l'une des opérations suivantes :

  Utilisez le remplissage du champ et cliquez sur le point approprié.

  Utilisez le clavier pour entrer le nom du point.

Point central

Pour utiliser un point existant comme centre de l’arc, effectuez l’une des opérations suivantes :

  Entrez le nom d’un point existant.

  Utilisez le remplissage du champ pour sélectionner un point existant.

Les coordonnées du point sont copiées dans les champs Centre nord et Centre est.

Centre nord, Centre est

Pour utiliser des coordonnées comme centre de l’arc, utilisez le remplissage du champ pour cliquer sur une position de la fenêtre graphique. Les coordonnées sont copiées dans ces champs.

NOTE : Si vous entrez à la fois un point central et des coordonnées centrales, le point central est utilisé.

Petit arc

Sélectionnez l’option Vrai si vous souhaitez créer un petit arc. Sinon sélectionnez Faux. Il existe deux arcs possibles entre deux points, ces deux arcs formant un cercle. Ce paramètre vous permet de définir l’arc que vous souhaitez tracer. Cela est expliqué dans la figure suivante :

NOTE : Le point central d’une ligne droite entre 2 points ou les coordonnées des champs Centre nord et Centre est constituent le centre de l’arc.

Couche

Effectuez l'une des opérations suivantes :

  Sélectionnez la couche dans laquelle vous souhaitez ajouter l’arc.

  Entrez un nouveau nom de couche pour créer une nouvelle couche. Le dialogue Nouvelle couche s'affiche.

Style de ligne

Sélectionnez le style de ligne que vous souhaitez utiliser pour afficher l’arc. Les styles disponibles sont les styles d’arc du dialogue Styles CAD.

Insérer

Cliquez sur Insérer pour insérer l’arc. Le point sélectionné dans le champ Vers point est sélectionné par défaut dans le champ De point. Vous pouvez modifier cette sélection si nécessaire.

Fermer

Cliquez sur Fermer pour arrêter l’insertion d’arcs.


    Commande Courbes

Utilisez cette commande pour insérer des courbes dans la base de données.

Pour y accéder à partir de la Vue en plan, effectuez l'une des opérations suivantes :

  Sélectionnez Insérer / Courbes.

  Sélectionnez l’outil Insérer courbes [image].

Le dialogue Insérer lignes s’affiche avec la tabulation Courbe sélectionnée.


    Dialogue Insérer lignes - Tabulation Courbe

Utilisez ce dialogue pour insérer des courbes dans la base de données.

Pour y accéder à partir de la Vue en plan, effectuez l'une des opérations suivantes :

  Sélectionnez Insérer / Courbes.

  Sélectionnez l’outil Insérer courbes [image].

Le dialogue Insérer lignes s’affiche avec la tabulation Courbe sélectionnée.

De point

Effectuez l'une des opérations suivantes :

  Utilisez le remplissage du champ et cliquez sur les points appropriés.

  Utilisez le clavier pour entrer le(s) nom(s) des points.

Vers point

Effectuez l'une des opérations suivantes :

  Utilisez le remplissage du champ et cliquez sur les points appropriés.

  Utilisez le clavier pour entrer le(s) nom(s) des points.

Tangente d’entrée, Tangente de sortie

Pour entrer automatiquement les valeurs de tangente sur la base des points utilisés pour définir la courbe, laissez les valeurs réglées sur <Automatique>. Vous avez la possibilité d’entrer vos propres valeurs de tangente si nécessaire.

Coefficient de lissage

Entrez une valeur entre 0 et 1. Le Coefficient de lissage définit le degré de courbure de la courbe.

Plus le coefficient de lissage est important, plus la courbure est importante.

Couche

Effectuez l'une des opérations suivantes :

  Sélectionnez la couche dans laquelle vous souhaitez insérer la courbe.

  Entrez un nouveau nom de couche pour créer une nouvelle couche. Le dialogue Nouvelle couche s'affiche.

Style de ligne

Sélectionnez le style de ligne que vous souhaitez utiliser pour afficher la courbe. Les styles disponibles sont les styles de ligne du dialogue Styles CAD.

Insérer

Cliquez sur Insérer pour insérer la courbe. Le point sélectionné dans le champ Vers point est sélectionné par défaut dans le champ De point. Vous pouvez modifier cette sélection si nécessaire.

Fermer

Cliquez sur Fermer pour arrêter l’insertion de courbes.


    Commande Texte

Utilisez cette commande pour ouvrir le dialogue Insérer texte.

Pour y accéder à partir de la Vue en plan, effectuez l'une des opérations suivantes :

  Sélectionnez Insérer / Texte.

  Cliquez sur l’outil Insérer texte [image].


    Dialogue Insérer texte

Utilisez ce dialogue pour insérer du nouveau texte dans la base de données.

Pour y accéder à partir de la Vue en plan, effectuez l'une des opérations suivantes :

  Sélectionnez Insérer / Texte.

  Cliquez sur l’outil Insérer texte [image].

Expression de texte

Entrez le texte que vous voulez afficher. Pour utiliser les codes de champ (par exemple, le titre du projet), cliquez avec le bouton droit de la souris et sélectionnez Champs dans le menu de raccourcis.

Nord

Pour entrer une valeur de Nord pour l’emplacement du texte, effectuez l’une des opérations suivantes :

  Utilisez le remplissage du champ en cliquant sur une position de la fenêtre graphique.

  Entrez la valeur de Nord à l’aide du clavier.

Est

Pour entrer une valeur d’Est pour l’emplacement du texte, effectuez l’une des opérations suivantes :

  Utilisez le remplissage du champ en cliquant sur une position de la fenêtre graphique.

  Entrez la valeur d’Est à l’aide du clavier.

Orientation

Affiche une orientation pour l’entité de texte. Si vous ne spécifiez aucune orientation, le texte est affiché horizontalement dans la fenêtre.

Type d'ancrage

Spécifiez où le texte se trouve pour les valeurs de Nord et d’Est. Sélectionnez une valeur dans la liste.

Couche

Sélectionnez dans la liste la couche à laquelle vous souhaitez ajouter du texte. Pour entrer du texte dans une nouvelle couche, saisissez un nouveau nom. Le dialogue Nouvelle couche s'affiche.

Style de texte

Sélectionnez dans la liste le style de texte pour l’entité de texte. Si vous souhaitez utiliser le style de texte par défaut pour une couche à laquelle vous ajoutez du texte, sélectionnez l’option <Par couche>.

Pour éditer et créer des styles de texte, sélectionnez Editer / Styles.

Insérer

Cliquez sur Insérer pour ajouter le texte au projet et gardez le dialogue ouvert pour ajouter du texte supplémentaire.


Texte - Vue d'ensemble

Plus amples informations

Comment ...	Dialogue	Voir aussi
Vue d'ensemble

Vous pouvez ajouter du texte dans un projet. Par exemple, ajoutez du texte pour décrire des figures ou pour afficher des titres de projet.

Le texte s’affiche dans la fenêtre graphique. Pour afficher les propriétés du texte, sélectionnez-le et ouvrez la fenêtre Propriétés. Vous pouvez également faire glisser du texte jusqu’à un nouvel emplacement de la fenêtre graphique ou encore faire tourner le texte.

Le texte est exporté dans certains formats (par exemple les formats AutoCAD et TRIMMAP).


    Annotations - Vue d'ensemble

Plus amples informations

Comment ...	Dialogue	Voir aussi
Vue d'ensemble

Les annotations correspondent à du texte associé aux entités (points, lignes et arcs). Le texte qui s’affiche est défini par un modèle d’annotation.

Les modèles d'annotation définissent :

  ce qui est annoté

  la position de l’annotation par rapport à l’entité

Les modèles d’annotation peuvent comporter des combinaisons de texte et de codes de champ définis par l’utilisateur. Les codes de champ sont développés à chaque fois que le champ qui est annoté change. Par exemple, si un point est annoté avec son élévation et que l’élévation du point change, l’annotation est mise à jour.

Les annotations sont enregistrées dans le projet.

Modèles d'annotation

Trimble vous fournit certains modèles d’annotation prédéfinis. Si vous traitez des codes de caractéristiques et que la bibliothèque de caractéristiques et attributs contient des modèles d’annotation, ces derniers sont ajoutés à la liste de modèles d’annotation. Les modèles d’annotation de la bibliothèque de caractéristiques et attributs ne possédant pas de paramètre de style de texte, le style de texte standard est utilisé par défaut lorsque les modèles d’annotation sont ajoutés au projet Trimble Geomatics Office.

Ajouter des annotations

Pour ajouter des annotations à partir de la Vue en plan, effectuez l'une des opérations suivantes :

  Sélectionnez Insérer / Annotations.

  Sélectionnez Editer / Modèles d’annotation. Cliquez sur Nouveau.

  Utilisez le traitement des codes de caractéristiques pour annoter des entités. Pour de plus amples informations, référez-vous à l'Aide de l’utilitaire Trimble Feature and Attribute Editor.

NOTE : Pour éditer des annotations existantes, utilisez la fenêtre Propriétés.


    Commande Annotations

Utilisez cette commande pour annoter les points, lignes ou arcs sélectionnés.

Pour y accéder, cliquez sur Insérer / Annotations.

Le dialogue Annoter s'affiche.


    Dialogue Annoter

 Vue d'ensemble et plus amples informations

Utilisez ce dialogue pour annoter les points, lignes ou arcs sélectionnés.

Pour y accéder, cliquez sur Insérer / Annotations.

La position d’insertion des annotations pour des entités au format papier est déterminée dans le dialogue Fichier / Mise en page en ce qui concerne les dimensions de l’entité et la configuration de l’échelle de report. Un espace vide correspondant à un tiers de la hauteur des caractères est ajouté entre l’annotation et l’entité annotée.

Modèle d'annotation

Sélectionnez un modèle dans la liste.

Conserver annotations existantes

Sélectionnez cette case à cocher pour conserver des annotations existantes ayant été assignées aux entités sur le point d’être annotées. Les nouvelles annotations sont ajoutées à celles qui existent, même si elles se trouvent au même emplacement.

ASTUCE : Vous pouvez utiliser cette fonction pour fournir des annotations multiples pour les entités réparties entre différentes couches dans le dialogue Modèles d’annotation.

Si vous désélectionnez cette case à cocher, les annotations assignées aux entités sur le point d’être annotées sont supprimées de la base de données avant que la nouvelle annotation ne soit ajoutée.

Annoter

Il existe deux méthodes permettant d’annoter les entités :

Sélection	Sélectionnez cette option pour annoter les entités sélectionnées.
BD entière	Sélectionnez cette option pour annoter toutes les entités.
Ces boutons sont disponibles uniquement lorsque aucune sélection n’a été faite. Dans ce cas, toutes les entités sont annotées.

Supprimer

Pour supprimer toutes les annotations des entités sélectionnées, effectuez l’une des opérations suivantes :

  Cliquez sur Supprimer.

  Sélectionnez les annotations puis Editer / Supprimer.

Modèles

Cliquez sur Modèles pour afficher le dialogue Modèles d’annotation. Utilisez ce dialogue pour sélectionner les Modèles d’annotation que vous souhaitez créer, modifier ou supprimer. Pour de plus amples informations, référez-vous au dialogue Modèles d’annotation.


    Dialogue Modèles d'annotation

 Vue d'ensemble et plus amples informations

Utilisez ce dialogue pour gérer les modèles d’annotation.

Pour y accéder :

1.	Sélectionnez Insérer / Annotations.

2.	Cliquez sur Modèles.

Un modèle d’annotation définit le style, la position et le contenu d’une ou plusieurs annotations. Une fois assignée, chaque annotation correspond à du texte d’information isolé qui ne dépend pas du modèle d’annotation. Les annotations sont enregistrées dans le projet.

Trimble Geomatics Office utilise les codes de champ pour spécifier les valeurs qui figurent dans l’annotation. Une annotation n’est pas figée — tous les codes de champ sont re-développés à chaque fois qu’une entité parente change.

Modèles d'annotation

Cette liste reprend tous les modèles d’annotation actuellement dans le projet.

Pour modifier ou supprimer un modèle d’annotation existant, sélectionnez le nom du modèle approprié.

Nouveau

Cliquez sur Nouveau pour créer un nouveau modèle d’annotation. Le dialogue Modèles d’annotation s'affiche.

Editer

Cliquez sur Editer pour modifier un modèle d’annotation existant. Le dialogue Modèles d’annotation s'affiche.

Supprimer

Cliquez sur Supprimer pour supprimer le Modèle d’annotation sélectionné. Cette opération le supprime du projet.


    Dialogue Modèles d'annotation

 Vue d'ensemble et plus amples informations

Utilisez les tabulations de ce dialogue pour configurer le modèle d’annotation créé ou modifié.

Pour y accéder :

1.	Effectuez l'une des opérations suivantes :

  Sélectionnez Insérer / Annotations. Cliquez sur Modèles.

  Sélectionnez Editer / Modèles d’annotation.

Le dialogue Modèles d’annotation s'affiche.

2.	Cliquez sur Nouveau ou Editer.

Utilisez les quatre tabulations pour définir le modèle d’annotation :

  Paramètres

  Points

  Lignes

  Arcs


    Dialogue Modèles d'annotation - Tabulation Paramètres

 Vue d'ensemble et plus amples informations

Utilisez la tabulation Paramètres pour éditer les paramètres généraux du modèle d’annotation.

Pour y accéder :

1.	Sélectionnez Insérer / Annotations.

2.	Cliquez sur Modèles.

3.	Cliquez sur Nouveau ou Editer.

4.	Sélectionnez la tabulation Paramètres.

Nom

Entrez un nom pour afficher ou renommez le nom courant du modèle.

Style de texte

Définissez le style de texte de l’annotation.

Pour définir des styles de texte d’annotation, sélectionnez Editer / Styles dans la Vue en plan. Si vous sélectionnez l’option <Par couche>, le style de texte est hérité du paramètre de style de texte de la couche à laquelle l’annotation est ajoutée.

Pour configurer les couches par défaut, sélectionnez Editer / Couches dans la Vue en plan.

Orientation d'annotation point

Entrez l’Orientation d’annotation point dont vous avez besoin. Vous pouvez utiliser toute orientation entre 00°00'00" et 360°00'00". L’orientation par défaut est 90°00'00".

Couche

Placer dans	Utilisez cette option pour placer les annotations dans une couche spécifiée. Vous pouvez sélectionner une couche dans la liste des couches existantes ou entrer un nom de nouvelle couche. Si vous entrez un nom de nouvelle couche, cette couche est automatiquement créée lorsque le modèle est appliqué.

Placer dans la couche courante

Utilisez cette option pour placer les annotations dans la couche courante. La couche courante est affichée dans l’outil Sélection de couche de la Vue en plan et dans le dialogue Couches.

Placer dans la couche d'entité annotée

Utilisez cette option pour placer les annotations dans la couche d’entité annotée.


    Dialogue Modèles d'annotation - Tabulation Points

 Vue d'ensemble et plus amples informations

Utilisez la tabulation Points pour définir les annotations des points lorsque les points sont annotés à l’aide de ce modèle.

Pour y accéder :

1.	Sélectionnez Insérer / Annotations.

2.	Cliquez sur Modèles.

3.	Cliquez sur Nouveau ou Editer.

4.	Sélectionnez la tabulation Points.

Le texte d’annotation de chacune des quatre positions (Gauche, Sur, Droite, Flottant) peut comporter au maximum 100 caractères alphanumériques. Les valeurs de chaque position peuvent consister en texte défini par l’utilisateur et en codes de champ. Les codes de champ sont utilisés pour extraire les détails d’annotation de la base de données. Par exemple, le code de champ ‘[nom]’ est utilisé pour annoter un point avec son nom — dans le cas présent, à droite du point.

Vous pouvez annoter un point avec toute combinaison des quatre positions par défaut. Pour déplacer une annotation d’une position par défaut, utilisez la souris pour déplacer l’annotation et la mettre à un emplacement plus approprié. L’annotation reste liée au point.

Position	L’annotation s’affiche :

Gauche	à gauche du point.

Sur	sur le point.

Droite	à droite du point.

Flottant	à un emplacement éloigné du point mais lié au point.


    Dialogue Modèles d'annotation - Tabulation Lignes

Utilisez la tabulation Lignes pour définir les annotations des lignes lorsque les lignes sont annotées à l’aide de ce modèle.

Pour y accéder :

1.	Sélectionnez Insérer / Annotations.

2.	Cliquez sur Modèles.

3.	Cliquez sur Nouveau ou Editer.

4.	Sélectionnez la tabulation Lignes.

Le texte d’annotation de chacune des sept positions (voir ci-dessous) peut comporter au maximum 100 caractères alphanumériques. Les valeurs de chaque emplacement peuvent consister en texte défini par l’utilisateur et en codes de champ. Les codes de champ sont utilisés pour extraire les détails d’annotation de la base de données. Par exemple, le code de champ ‘[azimut]’ est utilisé pour annoter une ligne avec sa valeur d’azimut (dans le cas présent au-dessus du centre de la ligne) et le code de champ ‘[distance]’ pour annoter une ligne avec sa valeur de distance (dans le cas présent sous le centre de la ligne).

Vous pouvez annoter une ligne avec toute combinaison des sept positions par défaut. Pour déplacer une annotation d’une position par défaut, utilisez la souris pour déplacer l’annotation et la mettre à un emplacement plus approprié. L’annotation reste liée à la ligne.

Position	L’annotation s’affiche :

En haut à gauche	au-dessus et à l’extrémité gauche de la ligne.

En haut au centre	au-dessus et au centre de la ligne.

En haut à droite	au-dessus et à l’extrémité droite de la ligne.

En bas à gauche	au-dessous et à l’extrémité gauche de la ligne.

En bas au centre	au-dessous et au centre de la ligne.

En bas à droite	au-dessous et à l’extrémité droite de la ligne.

Flottant	à un emplacement éloigné de la ligne mais lié à la ligne.


    Dialogue Modèles d'annotation - Tabulation Arcs

Utilisez la tabulation Arcs pour définir les annotations des arcs lorsque les arcs sont annotés à l’aide de ce modèle.

Pour y accéder :

1.	Sélectionnez Insérer / Annotations.

2.	Cliquez sur Modèles.

3.	Cliquez sur Nouveau ou Editer.

4.	Sélectionnez la tabulation Arcs.

Le texte d’annotation de chacune des sept positions (voir ci-dessous) peut comporter au maximum 100 caractères alphanumériques. Les valeurs de chaque emplacement peuvent consister en texte défini par l’utilisateur et en codes de champ. Les codes de champ sont utilisés pour extraire les détails d’annotation de la base de données. Par exemple, le code de champ ‘[longueur]’ est utilisé pour annoter un arc avec sa valeur de longueur (dans le cas présent au-dessus du centre de l’arc) et le code de champ ‘[rayon]’ pour annoter un arc avec sa valeur de rayon (dans le cas présent sous le centre de l’arc).

Vous pouvez annoter un arc avec toute combinaison des sept positions par défaut. Pour déplacer une annotation d’une position par défaut, utilisez la souris pour déplacer l’annotation et la mettre à un emplacement plus approprié. L’annotation reste liée à l’arc.

Position	L’annotation s’affiche :

En haut à gauche	au-dessus et à l’extrémité gauche de l’arc.

En haut au centre	au-dessus et au centre de l’arc.

En haut à droite	au-dessus et à l’extrémité droite de l’arc.

En bas à gauche	au-dessous et à l’extrémité gauche de l’arc.

En bas au centre	au-dessous et au centre de l’arc.

En bas à droite	au-dessous et à l’extrémité droite de l’arc.

Flottant	à un emplacement éloigné de l’arc mais lié à l’arc.


Commande Points

Utilisez cette commande pour insérer de nouveaux points dans la base de données.

Pour accéder à cette commande, sélectionnez Insérer / Points.

Le dialogue Insérer points s'affiche.


Ajouter des entités au projet

Vous pouvez ajouter des entités dans le projet à partir de Trimble Geomatics Office. Une entité est définie comme un point, une ligne, un arc, une courbe, une annotation ou du texte. Vous pouvez ajouter tous ces types d’entité à l’aide du menu Insérer de la Vue en plan. Vous pouvez également ajouter des points dans la vue de levé à l’aide du menu Insérer.

Pour obtenir des informations sur chaque type d’entité, utilisez les liens suivants :

  points

  lignes

  arcs

  annotations

  texte


 Commande Lignes pour les observations courantes

Utilisez cette commande pour créer des lignes CAD pour les types d’observation sélectionnés. Pour y accéder à partir de la Vue en plan, sélectionnez Insérer / Lignes d’observation.

Le dialogue Insérer lignes pour les observations s’affiche.


 Dialogue Insérer lignes pour les observations courantes

Utilisez ce dialogue pour créer des lignes CAD pour toutes les observations du projet.

Pour y accéder à partir de la Vue en plan, sélectionnez Insérer / Lignes pour les observations courantes.

Type d'observation

Affiche le type d’observation. Les types d’observation possibles sont les suivants :

Vecteurs RTK
Vecteurs post-traités	Statique, FastStatic et Cinématique post-traitée
Observations terrestres	Conventionnelle, Par télémètre à laser et Réduite
Couche CAD

Sélectionnez la couche pour ajouter des lignes au type d’observation. Les lignes sont tracées à l’aide du style de ligne par défaut de la couche que vous avez sélectionnée. Si vous entrez une nouvelle couche à l’aide du clavier, le dialogue Nouvelle couche s'affiche.


Propriétés du projet - Vue d'ensemble

Plus amples informations

Comment...	Dialogue

Vue d'ensemble

Les propriété d'un projet comprennent tous les paramètres généraux du projet (par exemple, le système de coordonnées et les unités d'affichage). Vous pouvez les modifier à tout moment. Lorsque vous modifiez une propriété du projet, le projet est mis à jour pour refléter cette modification. Certaines propriétés peuvent modifier les données du projet (par exemple, la modification du système de coordonnées).  Donc, il faut faire attention lors de la modification d'une propriété du projet que modifie les données existantes.

Il est impossible d'annuler une modification aux propriétés d'un projet.


Dialogue Propriétés du projet

Utilisez ce dialogue pour configurer les paramètres généraux du projet.

Pour y accéder:

  Sélectionnez Fichier / Propriétés du projet.

Le dialogue Propriétés du projet a les tabulations suivantes:

  tabulation Système de coordonnées

  tabulation Rapport en cours

  tabulation Détails du projet

  tabulation Recalculer

  tabulation Unités et format

  tabulation Caractéristiques


Dialogue Propriétés du projet - tabulation Détails du projet

Utilisez cette tabulation pour saisir les données concernant le projet/modèle que vous voulez inclure dans des rapports ou des reports.

Pour y accéder:

1.	Sélectionnez Fichier / Propriétés du projet. Le dialogue Propriétés du projet s'affiche.

2.	Sélectionnez la tabulation Détails du projet.

NOTE: Dans la liste Détails du projet, cliquez avec le bouton droit de la souris pour accéder à un menu de raccourcis que vous permet de renommer ou supprimer des éléments existants, et d'insérer des nouveaux éléments.

Tous les champs dans cette tabulation sont facultatifs.

Titre

Saisissez un titre pour le projet/modèle courant dans ce champ.

Description

Saisissez une description pour le projet/modèle si requis.

Référence

Saisissez un numéro de référence du projet afin de vous permettre de lier le projet à votre procédé d'évaluation des coûts au bureau.

Topographe sur le terrain

Saisissez le nom du topographe qui a effectué le travail sur le terrain pour le projet, si requis.

Opérateur sur ordinateur

Saisissez le nom de l'opérateur sur ordinateur pour le projet, si requis.

Date

Les paramètres par défaut permettent à ce champ d'enregistrer la date de création du projet. C'est effectué automatiquement utilisant le code de champ de la date dans ce champ dans un projet de modèle.

Datum vertical

Saisissez le nom du datum vertical.

Equipement

Saisissez le(s) numéro(s) de série de votre équipement.

NOTE: Lorsqu'un modèle est ouvert, les définitions de code de champ ne sont pas développées.  Vous pouvez utiliser cette option pour saisir des définitions de code de champ appropriées dans vos modèles. Lors de l'ouverture du projet pour la première fois, les définitions de code de champ sont développées.


Dialogue Propriétés du projet - tabulation Système de coordonnées

 Vue d'ensemble et plus amples informations

Utilisez cette tabulation pour sélectionner et afficher le système de coordonnées du projet.

Pour y accéder:

1.	Sélectionnez Fichier / Propriétés du projet. Le dialogue Propriétés du projet s'affiche.

2.	Sélectionnez la tabulation Système de coordonnées.

Paramètres du système de coordonnées

Ce groupe affiche un résumé du système de coordonnées courant. Il affiche le site, le système, la zone, et le modèle de géoïde du projet.

Le bouton Changer ouvre le dialogue Sélectionner système de coordonnées. Ce dialogue est la première page d'un assistant que vous utilisez pour sélectionner un système de coordonnées.

NOTE: Pour créer et éditer les systèmes de coordonnées, utilisez l'utilitaire Trimble Coordinate System Manager .

Paramètres du site local

Ce groupe affiche un résumé de l'emplacement du site. Il affiche la latitude, la longitude et la hauteur du projet ainsi que la méthode d'affichage du système de coordonnées.

Le bouton Changer ouvre le dialogue Paramètres du site local. Dans ce dialogue vous pouvez changer l'emplacement du projet et spécifier s'il faut utiliser des coordonnées au sol.

Détails

Le bouton Détails ouvre le dialogue Détails du système de coordonnées. Utilisez ce dialogue pour afficher les paramètres complets de la projection, la transformation de datum, l'ajustement, le modèle de géoïde et le site local du système de coordonnées courant.

Si vous effectuez une calibration du site GPS dans le projet, les détails du système de coordonnées sont une combinaison des paramètres du système de coordonnées sous-jacente et la calibration du site GPS (l'ajustement horizontal et l'ajustement vertical).

Si vous effectuez un ajustement de réseau les paramètres sont utilisés comme l'ajustement.

Un système de coordonnées peut avoir une calibration du site ou des paramètres de transformation d'ajustement de réseau mais pas les deux.

Enregistrer sous site

Le bouton Enregistrer sous site ouvre le dialogue Enregistrer sous site. Utilisez ce dialogue pour enregistrer les détails du système de coordonnées (y compris toute calibration du site GPS courante) dans la base de données du système de coordonnées comme une définition du site.

Une fois que vous nommer la définition du site, vous pouvez l'utiliser comme la définition du système de coordonnées pour les projets futurs. Pour éditer le site, utilisez l'utilitaire Trimble Coordinate System Manager.

NOTE: Les paramètres d'ajustement de réseau ne sont pas enregistrés dans un site.


Dialogue Détails du système de coordonnées - tabulation Projection

Pour accéder à cette tabulation:

1.	Sélectionnez Fichier / Propriétés du projet. Le dialogue Propriétés du projet s'affiche.

2.	Dans la tabulation Système de coordonnées, cliquez sur Détails. Le dialogue Détails du système de coordonnées s'affiche.

3.	Sélectionnez la tabulation Projection.

Pour afficher les paramètres de projection, utilisez cette tabulation. Elle affiche le type de projection (par exemple, Mercator transversal ou Lambert). Les champs qui s'affichent dépendent de la projection.

Le champ Azimut sud fait référence à la référence d'azimut. La plupart des pays utilisent le Nord comme la référence d'azimut (Nord = 0°). Cependant, certains pays, tels que l'Afrique du sud et Hawaii, utilisent le Sud comme la référence d'azimut (Sud = 0°). Cette boîte est sélectionnée si vous travaillez dans une région qui utilise cette méthode lorsque vous configurez les paramètres de projection dans l'assistant Système de coordonnées ou définissez le système de coordonnées d'azimut sud dans l'utilitaire Trimble Coordinate System Manager™.

Les options de Nord ou Sud et Est ou Ouest positifs font référence à la direction dont les coordonnées s'augmentent.


Dialogue Détails du système de coordonnées - tabulation Transformation de datum

 Voir aussi

Pour accéder à cette tabulation:

1.	Sélectionnez Fichier / Propriétés du projet. Le dialogue Propriétés du projet s'affiche.

2.	Dans la tabulation Système de coordonnées, cliquez sur Détails. Le dialogue Détails du système de coordonnées s'affiche.

3.	Sélectionnez la tabulation Transformation de datum.

Pour afficher les paramètres de transformation de datum, utilisez cette tabulation. Elle affiche la méthode de transformation et les paramètres utilisés pour la transformation de l'ellipsoïde WGS-84 à l'ellipsoïde local. Les paramètres de l'ellipsoïde local sont affichés dans le groupe Ellipsoïde local.

Les options de transformation de datum disponibles sont:

  Molodensky (Trois paramètres)

  Sept paramètres

  Régression multiple

  Grille de datum

La table suivante indique les champs disponibles pour chaque option:


Dialogue Détails du système de coordonnées - tabulation Ajustement

Pour accéder à cette tabulation:

1.	Sélectionnez Fichier / Propriétés du projet. Le dialogue Propriétés du projet s'affiche.

2.	Dans la tabulation Système de coordonnées, cliquez sur Détails. Le dialogue Détails du système de coordonnées s'affiche.

3.	Sélectionnez la tabulation Ajustement.

Pour afficher les paramètres d'ajustement, utilisez cette tabulation. L'ajustement est sous la forme d'une calibration du site GPS ou des paramètres de transformation de datum appliqués aux données de levé à partir de:

  une calibration du site GPS calculée

  une calibration importée d'un fichier Survey Controller (*.dc)

  une définition du site

  un ajustement de réseau (si vous avez le module d'Ajustement de réseau installé)

NOTE: Les champs Pente nord et Pente est dans le groupe Ajustement vertical calibration sont affichés en parties par million (ppm).


Dialogue Détails du système de coordonnées - tabulation Modèle de géoïde

Pour accéder à cette tabulation:

1.	Sélectionnez Fichier / Propriétés du projet. Le dialogue Propriétés du projet s'affiche.

2.	Dans la tabulation Système de coordonnées, cliquez sur Détails. Le dialogue Détails du système de coordonnées s'affiche.

3.	Sélectionnez la tabulation Modèle de géoïde.

Pour afficher les détails du modèle de géoïde, utilisez cette tabulation. Elle affiche le nom et le fichier de définition pour tout modèle de géoïde utilisé pour déterminer les élévations des positions GPS.

NOTE: Pour afficher la qualité du modèle de géoïde, utilisez la tabulation Propriété du projet / Recalculer.


Dialogue Détails du système de coordonnées - tabulation Site local

Pour accéder à cette tabulation:

1.	Sélectionnez Fichier / Propriétés du projet. Le dialogue Propriétés du projet s'affiche.

2.	Dans la tabulation Système de coordonnées, cliquez sur Détails. Le dialogue Détails du système de coordonnées s'affiche.

3.	Sélectionnez la tabulation Site local.

Cette tabulation indique:

  la latitude, la longitude, et la hauteur de l'emplacement du projet

  le facteur d'échelle au sol, le déport de constante nord, le déport de constante est, et si des coordonnées au sol sont utilisées.


Dialogue Paramètres du site local

Utilisez le dialogue Paramètres du site local pour éditer l'emplacement du projet, et spécifiez si des coordonnées au sol sont utilisées dans le projet. Pour l'accéder:

1.	Sélectionnez Fichier / Propriétés du projet. Le dialogue Propriétés du projet s'affiche.

2.	Sélectionnez la tabulation Système de coordonnées.

3.	Du groupe Paramètres du site local, cliquez sur Changer.

Emplacement du projet

Utilisez ce groupe pour définir l'emplacement du projet, utilisant un point au centre du projet. L'emplacement du projet peut être défini sous forme de coordonnées de grille, locales, ou WGS-84.

NOTE: Vous pouvez saisir des coordonnées de grille en sélectionnant l'emplacement dans la fenêtre de graphiques.

Il faut définir l'emplacement du projet au début du projet, parce que le logiciel utilise cette information dans ses calculs. Il utilise:

  la latitude du projet pour calculer le rayon de la terre

  la latitude et la longitude du projet pour définir le facteur d'échelle de la grille

  le hauteur du projet comme l'élévation par défaut et pour calculer le facteur d'élévation.

Si le système de coordonnées n'est pas défini, le dialogue de Définition de projection par défaut s'affiche. Saisissez une latitude et une longitude d'origine avant de sélectionner l'option Utiliser coordonnées au sol.

Utiliser coordonnées au sol

Sélectionnez cette case à cocher si vous voulez utiliser des coordonnées au sol dans votre projet. Cependant, d'abord assurez-vous de définir le système de coordonnées et l'emplacement du projet.

Lorsque cette case à cocher est sélectionnée, tous les éléments dans le groupe Utiliser coordonnées au sol deviennent disponibles. La case à cocher Calculer le facteur d'échelle de la position du projet est vide la première fois que vous utilisez le dialogue Paramètres du site local.

NOTE: Le logiciel ne peut calculer le facteur d'échelle au sol que si les champs de nord, d'est et d'élévation sont saisis dans le groupe Emplacement du projet.

NOTE: Si vous effectuez une calibration du site GPS et puis sélectionnez la case à cocher Utiliser coordonnées au sol, la calibration définie dans le projet devient non valide. Un message vous avertit que de continuer avec l'opération supprimera la calibration.

Affichage de coordonnées au sol

L'option Affichage de coordonnées au sol vous permet de modifier la façon dans laquelle les coordonnées au sol sont affichées dans un projet Trimble Geomatics Office. Avec cette modification, il sera plus facile de différencier entre les coordonnées de projection et les coordonnées au sol. Sélectionnez l'une des choses suivantes:

  Déport nord/Déport est − ajoute un déport aux coordonnées qui sont affichées

  Coordonnées à 4 chiffres − affiche 4 chiffres dans l'affichage des coordonnées au sol. Par exemple, si la coordonnée du projet est 264150.000, la coordonnée au sol est affichée comme 4510.000.

  Coordonnées à 5 chiffres − affiche 5 chiffres dans l'affichage des coordonnées au sol. Par exemple, si la coordonnée du projet est 264150.000, la coordonnée au sol est affichée comme  64510.000.


Dialogue Enregistrer sous site

Utilisez ce dialogue pour enregistrer le système de coordonnées courant comme un site.

Pour l'accéder:

1.	Sélectionnez Fichier / Propriétés du projet. Le dialogue Propriétés du projet s'affiche.

2.	Cliquez sur Enregistrer sous site.

Nom du site

Saisissez le nom de la définition du site enregistrée.

OK

Cliquez sur OK pour enregistrer le site. Le site est enregistré dans la base de données de système de coordonnées. Tout site enregistré apparaît dans la tabulation Sites de l'utilitaire Trimble Coordinate System Manager™. Vous pouvez utiliser la définition du site comme la définition du système de coordonnées pour les projets futurs.

ASTUCE: Lors de l'utilisation de l'utilitaire Coordinate System Manager, cliquez sur la tabulation Sites. Le site s'affiche en bleu.

NOTE: Vous ne pouvez pas enregistrer des paramètres de transformation d'ajustement comme un site.


Dialogue Propriétés du projet - tabulation Unités et format

Utilisez cette tabulation pour configurer les valeurs d'unité utilisées partout dans Trimble Geomatics Office (par exemple, l'affichage à l'écran, l'importation, l'exportation et les rapports).

Pour y accéder:

1.	Sélectionnez Fichier / Propriétés du projet. Le dialogue Propriétés du projet s'affiche.

2.	Sélectionnez la tabulation Unités et format.

Coordonnée

Configurez les unités d'affichage des valeurs de coordonnées (Nord et Est). Les options sont Pieds internationaux, Links, Mètres, Pieds américains ou Mètres namibiens.

Elévation/Hauteur

Configurez les unités d'affichage des élévations. Les options sont Pieds internationaux, Links, Mètres, Pieds américains ou Mètres namibiens.

Décimales de coordonnées

Configurez le nombre de décimales à afficher pour les coordonnées. Vous pouvez spécifier jusqu'à cinq décimales.

Ordre des coordonnées

Configurez l'ordre des coordonnées. Les options sont Nord en premier ou Est en premier.

Afficher la position du curseur sous

Configurez l'affichage de la position du curseur dans la barre d'état. Les options sont Grille, Lat long locales et Lat long WGS.

Distance

Configurez l'affichage des distances. Les options sont Chains, Pieds internationaux, Milles internationales, Kilomètres, Links, Mètres, Pieds américains ou Mètres namibiens.

Décimales de distance

Configurez le nombre de décimales à afficher pour les distances. Vous pouvez spécifier jusqu'à cinq décimales.

Afficher distance sous

Configurez le type de distances à afficher. Les options sont:

  Distances ellipsoïdales (distances au niveau de la mer)

C'est la distance entre deux points de base de données calculés sur la surface d'ellipsoïde de référence (à une hauteur zéro).

  Grille

C'est la distance pythagoricienne entre les coordonnées de projection des deux points de fin. C'est à dire, aucun ajustement de projection n'est appliqué signifiant que le facteur d'échelle de la ligne n'est pas appliquée à la ligne.

  Au sol

C'est la distance parallèle à l'ellipsoïde calculée à la hauteur moyenne au-dessus de l'ellipsoïde. C'est à dire, la distance ellipsoïdale calculée mathématiquement ajustée pour la hauteur moyenne des points de fin. Si des coordonnées au sol sont utilisées, la distance de grille devrait être égale à la distance au sol. C'est décrit comme:

Où

H	représente la hauteur de la base de données

R	représente le rayon de la terre

DE	représente la distance ellipsoïdale

Surface

Configurez les unités d'affichage des surfaces. Les options sont Acres, Hectares, Pieds carrés et Mètres carrés.

Décimales des surfaces

Configurez le nombre de décimales à afficher pour les surfaces. Vous pouvez spécifier jusqu'à cinq décimales.

Unités d'angle

Configurez les unités d'angle utilisées pour toutes les directions horizontales et les azimuts. Les options sont Degrés, Grads, et Mils.

Format d'angle (degrés)

Configurez le format des angles ou des azimuts exprimés comme degrés. Les options sont:

  DDD.dddd	Degrés décimaux

  DDDMM'SS	Degrés, minutes, et secondes

  DDDMM'SS.s	Degrés, minutes, et secondes décimales

  DDDMM.mm	Degrés et minutes décimales

Format d'angle (grads / mils)

Configurez le format des angles ou des azimuts exprimés comme grads ou mils. Les options sont trois, quatre ou cinq décimales.

Format d'azimut

Configurez le format des valeurs d'azimut. Les options sont:

  Gisement quadrant (présenté sous forme de N6207'40"W)

  Cercle entier (présenté sous forme de 29752'20")

Afficher WGS-84 sous

Configurez l'option Afficher WGS-84 sous sur L,L,H ou X,Y,Z. La configuration ici détermine comment les coordonnées WGS-84 sont affichées pour les points dans la fenêtre Propriétés, lorsque vous insérez un point ou éditez les coordonnées d'un point utilisant l'utilitaire Trimble Coordinate System Manager™. Cette configuration est également utilisée si vous créez un format d'importation, un rapport ou un format d'exportation avec les valeurs WGS-84.

Format lat, long

Configurez le format des valeurs de Latitude et Longitude. Les options sont:

  DDD.ddddddddd	Degrés décimaux

  DDDMM'SS.sssss	Secondes décimales

  DDDMM.mmmmmmm	Minutes décimales

Format d'échelle

Configurez l'affichage et l'interprétation des valeurs d'échelle. Les options sont:

  1 pc à X pd	Unités de mesure légale

  1:X	Unités de mesure métrique

Afficher séparateur de milles

Configurez l'affichage de séparateur de milles.

Les options sont Oui ou Non. Si vous sélectionnez Oui, le séparateur de milles comme configuré dans votre système Microsoft Windows est appliqué aux valeurs numériques. Pour plus amples informations, référez-vous aux Paramètres régionaux dans le Panneau de configuration de Microsoft Windows.

NOTE: Lorsque ce paramètre d'affichage est configuré sur Oui, la présence du séparateur de milles est attendu dans les fichiers ASCII importés et il est utilisé dans les fichiers ASCII et les rapports exportés. Ce peut causer l'échec d'importation des fichiers et produire des fichiers exportés que ne se conforment pas au format attendu. (Considérez, par exemple, un fichier délimité par des virgules avec des valeurs numériques utilisant un séparateur de milles de virgule). Le code de champ modificateur "standard" peut être utilisé dans les définitions de format pour forcer le traitement de la valeur numérique à supprimer des séparateurs de mille et de toujours attendre le décimal d'être un point.

Unités d'affichage

Détermine si les unités de coordonnées sont affichées dans la fenêtre de Propriétés et la barre d'état. Le paramètre par défaut est Oui.

Afficher nuls sous

Configurez l'affichage des valeurs nulles.

Commencer nom du point auto de

Configurez le nom de numéro de point auto. Cette valeur est utilisée lorsque les coordonnées sont ajoutées ou créées par le logiciel. Le nom de point incrémente automatiquement.

Format de pente

Configurez le format de pente. Il est utilisé dans Trimble RoadLink™. Les options sont Pourcentage ou Ratio.

Format de pente de talus

C'est utilisé et affiché dans Trimble RoadLink. Les options sont Pourcentage ou Ratio.

Unités de ratio

Configurez l'ordre du format de ratio utilisé et affiché dans Trimble RoadLink. Les options sont Valeur V.:Valeur H. ou Valeur H.:Valeur V. Par exemple, Valeur V.:Valeur H. s'affiche comme 1:100. Valeur H.:Valeur V. s'affiche comme 100:1.

Format de station

Configurez le format d'affichage des valeurs de station partout dans Trimble RoadLink. Les options de format sont:

  1000.0

  10+00.0

  1+000.0

  1/000.0

Unités de volume

Configurez les unités de volume utilisées et affichées dans Trimble RoadLink. Les options sont mètres cubes, pieds cubes ou yards cubes.

Type d’affichage de l’heure GPS

Détermine le format sous lequel il faut afficher l’heure. Les options sont  Heure locale, Heure GPS et Heure UTC.


Dialogue Propriétés du projet - tabulation Rapport en cours

Utilisez cette tabulation pour configurer les paramètres de report du projet.

Pour y accéder:

1.	Sélectionnez Fichier / Propriétés du projet. Le dialogue Propriétés du projet s'affiche.

2.	Sélectionnez la tabulation Rapport en cours.

Afficher rapport généré

L'option Afficher rapport généré contrôle l'affichage de tout rapport généré par des opérations tels que l'importation des données, le recalcul et le traitement de codes de caractéristiques. Les rapports sont toujours créés pour ces opérations et stockés dans le dossier approprié du dossier Rapport du projet.

Ces options contrôlent l'affichage des rapports après une opération de traitement:

Toujours	Un rapport de tout traitement effectué est toujours affiché—lorsqu'il n'y a aucune erreur ou aucun avertissement produit lors du traitement, le rapport consiste en seule une en-tête avec toute information relative à l'opération de traitement effectuée.
Si d'erreur	Si une erreur s'est produite lors du traitement, le rapport sur ce traitement, y compris les détails de (des) erreur(s) rencontrée(s), est présenté automatiquement.
Demande si erreur	Si une erreur s'est produite lors du traitement, un avertissement s'affiche pour indiquer qu'une erreur s'est produite, et vous pouvez afficher le rapport détaillant la (les) erreur(s).
Jamais	Vous n'êtes pas averti qu'une erreur s'est produite lors du traitement et le rapport fournissant les détails ne s'affiche pas.
Demander toujours	Vous êtes demandé pour l'affichage d'un rapport sur tout traitement effectué.


Dialogue Propriétés du projet - tabulation Recalculer

Utiliser cette tabulation pour recalculer les paramètres du projet.

Pour y accéder:

1.	Sélectionnez Fichier / Propriétés du projet. Le dialogue Propriétés du projet ouvre.

2.	Sélectionnez la tabulation Recalculer.

Observations de prise de vue isolée multiples

Utilisez ce groupe pour spécifier comment un recalcul détermine la position des points de prise de vue isolée observés multiples fois. Un point de prise de vue isolée n'a pas des:

  observations y provenant

  coordonnées de contrôle

  observations statique ou statique rapide y entrant

Utiliser meilleure observation

Sélectionnez cette option si vous ne voulez pas que le recalcul fait la moyenne des observations aux point de prise de vue isolée. La meilleure observation d'un point sera utilisée pour déterminer sa position. Cette option est sélectionnée par défaut.

Calculer la moyenne à partir du même type d'observation (GPS, Conventionnelle, Réduite, Laser)

Sélectionnez cette option si vous voulez que le recalcul fait la moyenne des coordonnées dérivées pour chaque observation du même type. Un recalcul ne fait jamais la moyenne des coordonnées dérivées de types d'observation différents. Par exemple, les coordonnées déterminées à partir d'une observation conventionnelle ne seront jamais moyennées avec celles déterminées par une observation GPS. Seuls les points de prise de vue isolée sont moyennés.


Vérification de tolérance

Sélectionnez ces cases à cocher pour utiliser la vérification de tolérance lors du procédé de recalcul. Un recalcul utilise les valeurs de tolérance pour déterminer les coordonnées moyennes lorsque l'option Calculer la moyenne à partir du même type d'observation est sélectionnée.

Vous pouvez configurer les tolérances des positions de qualité Levé, Mappage et Inconnue.

Les valeurs de tolérance sont utilisées pour déterminer lorsqu'une erreur de fermeture est affichée pour un point ayant des coordonnées ou des observations multiples. Si les coordonnées dérivées de toutes les observations et les coordonnées dont hors de la tolérance, une erreur de fermeture est rapporté et un indicateur d'avertissement apparaît sur le point. Une erreur hors de tolérance est affichée dans la fenêtre Propriétés / Page d'avertissement.

Horizontal

Saisissez la valeur de tolérance pour le composant horizontal d'une position.

Vertical

Saisissez la valeur de tolérance pour le composant vertical d'une position.

Réduire les observation terrestres à l'ellipsoïde (correction au niveau de la mer) Sélectionnez cette case à cocher si vous voulez que les distances observées conventionnellement soient corrigées à l'ellipsoïde lors du recalcul.

Il faut appliquer les corrections au niveau de la mer à moins que vous savez que les mesures de distance sur le terrain stockées dans la base de données ont déjà eu cette correction appliquée, ou les coordonnées entrées sont au sol et pas les coordonnées à grille. Si cette correction n'est pas appliquée et les données conventionnelles ont été collectées à une hauteur considérable au-dessus du niveau de la mer, les coordonnées calculées ne seront pas correctes.

Cette correction n'est pas applicable aux données GPS car ces données sont toujours collectées en fonction de l'ellipsoïde WGS-84.

NOTE: Si vous combinez des mesures GPS et terrestres, vous devez réduire les observations à l’ellipsoïde.

NOTE: Vous pouvez utilisez l'option Afficher distance sous dans la tabulation Unités et format pour contrôler comment les distances entre les points de base de données sont rapportées. Lorsque vous configurez cette option sur les distances au sol ce vous permet d'afficher ce qui est effectivement le composant horizontal de toute distance au sol mesurée.

Pour plus amples informations, référez-vous à la rubrique Correction au niveau de la mer.

Qualité du modèle de géoïde

Sélectionnez la qualité du modèle de géoïde utilisé pour le projet. Si c'est dans l'ordre d'un précision de 5 m, sélectionnez Mappage, Si vous savez que les modèle de géoïde est d'une précision de levé, sélectionnez l'option Levé. Recalculer utilise cette qualité pour déterminer la qualité des élévations (pour un point GPS) ou de la hauteur (pour les points terrestres) dérivés d'un modèle de géoïde. Ce champ n'est pas disponible s'il n'y a aucun géoïde utilisé dans le système de coordonnées du projet.


Dialogue Propriétés du projet - tabulation Caractéristiques

Utilisez cette tabulation pour spécifier les paramètres de caractéristiques et attributs du projet.

Pour y accéder:

1.	Sélectionnez Fichier / Propriétés du projet. Le dialogue Propriétés du projet s'affiche.

2.	Sélectionnez la tabulation Caractéristiques.

Codification lors de l'importation

Sélectionnez la case à cocher Codification lors de l'importation  pour traiter les codes de caractéristiques automatiquement lors de l'importation d'un fichier de périphérique. Sélectionnez la bibliothèque d'attributs et caractéristiques que vous voulez utiliser pour traiter les codes de caractéristiques automatiquement.

NOTE: Si vous sélectionnez cette option vous pouvez toujours traiter les codes de caractéristiques plus tard utilisant le menu Outils dans la Vue en plan, et sélectionnez n'importe quelle bibliothèque d'attributs et caractéristiques à utiliser. (Il n'est pas nécessaire d'utiliser la bibliothèque que vous avez sélectionné ici.)

Bibliothèque

Saisissez le nom de la bibliothèque d'attributs et caractéristiques que vous voulez utiliser ou cliquez sur Parcourir pour la trouver. Ce champ n'est pas disponible à moins que vous ne sélectionnez la case à cocher Codification.

Utilisez attributs

Pour configurer le projet/modèle pour des attributs, sélectionnez la case à cocher Utiliser attributs.

Ce champ ne fait référence qu'à la bibliothèque utilisée pour les définitions d'attribut.

Bibliothèque

Pour sélectionner la bibliothèque, saisissez le chemin et le nom dans le champ Bibliothèque, ou cliquez sur Parcourir.

Parcourir

Cliquez sur Parcourir et rechercher le fichier de Dictionnaire d'attributs (*.ddf) ou le fichier de Bibliothèque d'attributs et caractéristiques (*.fcl). Le dossier par défaut est le dossier système Trimble Geomatics Office.

Si vous n'avez aucun point dans la base de données contenant des caractéristiques, vous pouvez modifiez la bibliothèque d'attributs et caractéristiques associée au projet.

Cependant, une fois que vous importez un fichier Survey Controller (*.dc) contenant des attributs, ou vous ajoutez des attributs utilisant la fenêtre Propriétés, vous ne pouvez pas changer la bibliothèque d'attributs et caractéristiques utilisée pour configurer le projet pour des attributs. Ainsi il est assuré que vous avez des données d'attributs consistantes dans le projet.


Fichiers Trimble Geomatics Office - Vue d'ensemble

Trimble Geomatics Office installé sur votre ordinateur comprend certains fichiers qui stockent des divers paramètres concernant le système et chaque projet.

AVERTISSEMENT: Ne pas éditer ces fichiers à moins que ce ne soit nécessaire. Il se peut que des erreurs s'introduisent et le logiciel ne fonctionnera pas comme attendu. Si vous voulez éditer ces fichiers, d'abord faites une copie de sauvegarde.

Les fichiers sont divisés dans les sections suivantes:

Fichier Trimble

Ces fichiers se trouvent dans le dossier Program Files\Common Files\Trimble:

  Current.csd

  Antenna.ini

  Receiver.ini

Fichiers système Trimble Geomatics Office

Ces fichiers se trouvent dans le dossier Program Files\Trimble\Trimble Geomatics Office\System:

  Asciiexp.dat

  Fichiers Asciiimp.dat

  Asciirpt.dat

  Fichiers html

  Symbols.sym

  Linetype.ltp

  Default.fcl

  Fichiers frm

Fichiers du projet

Ces fichiers se trouvent dans le répertoire racine de chaque projet (en général: C:\Trimble Geomatics Office\Projets\<nom du projet>:

  Options.ini

  TGO_v150.mdb

  TGO_v150.dbs


Fichier Options.ini

 Voir aussi

Un fichier Options.ini est associé à chaque projet. Il se trouve dans le dossier de nom du projet. Ce fichier fournit un contrôles des données affichées dans la fenêtre Propriétés, le résumé de points et les en-têtes de rapport.

Le fichier Options.ini est divisé en trois sections:

  [Résumé]

  [Rapport]

  [Etiquettes]

Configuration de la définition du résumé

La définition du Résumé dans le fichier Options.ini vous permet de spécifier les détails de points affichés sur la barre d'état pendant la sélection des points, ainsi que l'Info-bulle que s'affiche lorsque le pointeur de la souris est positionné sur un point. Par défaut c'est configuré sur [Nom] : [Code de caractéristiques] afin que les points sont résumés avec leur nom et leur code de caractéristiques.

Si vous préférez que les points sont résumés avec leur nom et leur style de point séparés d'un '/' définissez la section [Résumé] du fichier Options.ini comme:

Point=[Nom]/[Style de point]

Vous pouvez également assigner des résumé spécifiques pour les entités Ligne, Arc, Courbe et Texte. Par exemple, si vous voulez que l'Azimut et la Distance sont rapportés comme le résumé d'une ligne, ajoutez la définition suivante dans la section [Résumé]:

Ligne=Az: [Azimut] Dist: [Distance]

Vous pouvez inclure de texte avec les codes de champ dans la spécification du résumé.

Par défaut, le résumé des lignes, des arcs et des courbes consiste en les résumés des points des points de début et de fin, séparés par une virgule.

Configuration de l'en-tête du rapport

La section [Rapport] vous permet de spécifier une en-tête de rapport standard que peut être comprise dans tous les rapports produits utilisant un seul code de champ dénommé [$Rapport.En-tête]. L'en-tête du rapport peut consister en une combinaison de codes de champ et de texte, Utilisez ‘\n’ pour indiquer la position du début de la nouvelle ligne. Ainsi vous pouvez diviser l'en-tête sur plusieurs lignes. Par exemple, vous pouvez inclure votre nom de société dans l'en-tête du rapport standard.

Configuration des options d'étiquette

La section [Etiquettes] vous permet d'ajouter des nouvelles options d'étiquette qui s'affichent dans le dialogue Etiquettes.

Vous pouvez utiliser une combinaison de codes de champ et de texte pour définir une définition d'étiquette. Utilisez le caractère ‘|’ pour séparer chaque option d'étiquette et à nouveau pour séparer le nom de l'option d'étiquette et quel texte est en fait affiché comme l'étiquette. Par exemple, directement après la dernière option d'étiquette, saisissez ce qui suit à l'aide du clavier:

|Description|[description]

Ainsi une nouvelle option d'étiquette sera ajoutée dans le dialogue Etiquettes dénommée ‘Description’. Lors de sa sélection, chaque point élection, chaque point étiqueté aura sa valeur de champ de description affichée.

NOTE: Vous ne pouvez pas définir les options d'étiquette des lignes, des arcs ou des courbes.


TGO v160.mdb

 Voir aussi

AVERTISSEMENT: Trimble Navigation Limited ne supporte l'utilisation ou la modification de cette base de données qu'en utilisant Trimble Geomatics Office propriétaire de Trimble. Trimble ne supporte ni autorise l'utilisation ou la modification, ni garantit le fonctionnement, de cette base de données en utilisant d'autres applications informatiques. Trimble peut modifier le format de cette base de données sans préavis le rendant incompatible avec d'autres applications informatiques.

Chaque projet a un fichier Microsoft Access Database (*.mdb) dénommé TGO_v160.mdb. Il se trouve dans le dossier du projet. Ce fichier contient toutes les données du projet, y compris les tables et les interrogations. Si vous avez Microsoft Access (Access 2000 ou plus récente) vous pouvez:

  afficher les données du projet

  créer vos propres interrogations


Fichier TGO_v160.dbs

La première fois que vous fermez un nouveau projet Trimble Geomatics Office, un fichier schéma de base de données (*.dbs) est créé. Il est basé sur le fichier Microsoft Access Database (*.mdb).

Le fichier .dbs et le fichier .mdb se trouvent dans le dossier racine du projet.

Si vous ajoutez une interrogation personnalisée à la base de données vous devez supprimer ce fichier .dbs file après la création de l'interrogation.

Un nouveau fichier .dbs (contenant la nouvelle interrogation) est créé lorsque vous ouvrez le projet à nouveau et l'interrogation devient disponible à utiliser.


Fichier Antenna.ini

Ce fichier liste les antennes GPS disponibles dans Trimble Geomatics Office. Il est utilisé par d'autres logiciels Trimble ainsi que ce logiciel.


Fichier Receiver.ini

Ce fichier comprend une liste des récepteurs disponibles. Vous ne devriez pas éditer ce fichier.


Fichier Asciiexp.dat

Ce fichier se trouve dans le dossier Program Files\Trimble\Trimble Geomatics Office\System.

Il contient tous les formats ASCII et d'exportation personnalisée de Trimble Geomatics Office. Le fichier est lu chaque fois que vous accédez au dialogue Exporter.

Pour accéder au dialogue, effectuez l'une des choses suivantes:

  Sélectionnez l'outil Exporter [image].

  Sélectionnez Fichier / Exporter.

Les symboles suivants indique dans quelle tabulation le format apparaît. Seules des définitions ASCII sont comprises dans ce fichier:

Symbole	Tabulation
=	Levé
+	CAD/ASCII
*	SIG
Aucun	Personnalisé

Vous ne pouvez éditer que les formats disponibles dans la tabulation Personnalisé. Donc, si vous voulez éditer le format de l'un des formats disponibles dans la tabulation CAD/ASCII tab, supprimez le signe plus (+) du début de la définition du format.

Par exemple, la ligne suivante est comprise dans le fichier Asciiexp.dat:

+coordonnées SDRmap

Pour le rendre un format pouvant être édité, supprimez le signe plus (+).

La prochaine fois que vous accéder au dialogue Exporter, le format s'affiche dans la tabulation Personnalisé et le bouton Editer devient disponible.

De la même façon, vous pouvez changer un format personnalisé à un format que ne peut pas être édité par l'insertion de l'un des symboles ci-dessus au début du nom de format.

Par exemple la ligne suivante est inclue dans le fichier Asciiexp.dat:

Points délimités

Changez cette ligne à:

+Points délimités

La prochaine fois que vous accéder au dialogue Exporter, le format s'affiche dans la tabulation CAD/ASCII tab et le bouton Editer n'est pas disponible.


Fichier Asciiimp.dat

Ce fichier se trouve dans le dossier Program Files\Trimble\Trimble Geomatics Office\System. Il contient tous les formats ASCII et d'importation personnalisée de Trimble Geomatics Office. Le fichier est lu chaque fois que vous accédez au dialogue Importer.

Pour accéder au dialogue, effectuez l'une des choses suivantes:

  Sélectionnez l'outil Importer [image].

  Sélectionnez Fichier / Importer.

Les symboles suivants indique dans quelle tabulation le format apparaît. Seules des définitions ASCII sont comprises dans ce fichier:

Symbole	Tabulation
=	Levé
+	CAD / ASCII
Aucun	Personnalisé

Vous ne pouvez éditer que les formats d'importation disponibles dans la tabulation personnalisé. Donc, si vous voulez éditer le format de l'un des formats disponibles dans la tabulation CAD/ASCII tab, supprimez le signe plus (+) du début de la définition du format.

Par exemple, la ligne suivantes est comprise dans le fichier Asciiimp.dat file:

+format WILDsoft

Pour le rendre un format pouvant être édité, supprimez le signe plus (+).

La prochaine fois que vous accéder au dialogue Importer, le format s'affiche dans la tabulation Personnalisé et le bouton Editer devient disponible. De la même façon, vous pouvez changer un format personnalisé à un format que ne peut pas être édité par l'insertion de l'un des symboles ci-dessus au début du nom de format.

Par exemple la ligne suivante est inclue dans le fichier Asciiexp.dat:

Points délimités

Changez cette ligne à:

+Points délimités

La prochaine fois que vous accéder au dialogue Importer, le format s'affiche dans la tabulation CAD/ASCII tab et le bouton Editer n'est pas disponible.


Fichier Asciirpt.dat

Ce fichier se trouve dans le dossier Program Files\Trimble\Trimble Geomatics Office\System. Il contient tous les formats de rapports de base de données de Trimble Geomatics Office. Ce fichier est lu chaque fois que vous accéder au dialogue Rapports supplémentaires.

Pour accéder au dialogue Rapports supplémentaires, sélectionnez Rapports / Rapports supplémentaires.

Vous pouvez changer tout format de base de données du système dans le dialogue Rapports supplémentaires à un format pouvant être édité en supprimant le signe égal (=) du nom de format dans le fichier.

Par exemple, la ligne suivantes est comprise dans le fichier Asciirpt.dat:

=Points

Pour changer ce format à un format pouvant être édité, supprimes le signe égal (=).

Ce format s'affichera dans le dialogue Rapports supplémentaires maintenant et il peut être édité. De la même façon, vous pouvez changer un format personnalisé du dialogue Rapports supplémentaires à un format que ne peut pas être édité par l'insertion d'un signe égal (=) au début du nom de format.

Par exemple, la ligne suivantes est comprise dans le fichier Asciirpt.dat:

Liste de point à largeur fixe

Changez cette ligne à:

=Liste de point à largeur fixe

Le format s'affiche dans le dialogue Rapports supplémentaires maintenant et ne peut pas être édité.


Fichiers .html

Un nombre de fichiers de modèle au langue Hypertext Markup (.html) sont installés dans le dossier Program Files\Trimble\Trimble Geomatics Office\System. Ces modèles sont utilisés pour créer des rapports de système/ Vous ne devriez pas supprimer ces fichiers..

Les rapports (par exemple, Recompute.html) générés par Trimble Geomatics Office sont stockés dans le dossier de Rapports du projet approprié. Pour ouvrir un rapport. html, double cliquez sur le rapport dans l'Explorateur de Microsoft Windows. Il ouvre dans l'afficheur HTML par défaut.

NOTE: Vous pouvez modifier ces fichiers .html pour personnaliser les informations continues dans les rapports du système. Cependant, Trimble ne supporte ni garantit que les fichiers .html modifiés fonctionneront comme attendu. Trimble vous recommande de faire une copies de sauvegarde de tout fichier .html avant de le modifier.


Fichier symbols.sym

Le fichier Symbols.sym se trouve dans le dossier Program Files\Trimble\Trimble Geomatics Office\System.

Utilisez l'utilitaire Trimble Symbol Editor™ pour afficher et éditer ce fichier. Le fichier contient des symboles. Dans Trimble Geomatics Office, ou dans l'utilitaire Trimble Feature and Attribute Editor™, vous pouvez créer des nouveaux styles de point utilisant les symboles disponibles dans ce fichier.

Vous pouvez également créer vos propres fichiers.sym. Pour plus amples informations concernant l'utilisation des fichiers .sym, référez-vous au chapitre Utilitaires dans le Manuel de l'utilisateur du logiciel Trimble Geomatics Office.


Fichier Linetype.ltp

Le fichier Linetype.ltp se trouve dans le dossier Program Files\Trimble\Trimble Geomatics Office\System. Utilisez l'utilitaire Trimble Line Type Editor™ pour afficher et éditer ce fichiers.

Le fichier contient des types de lignes. Dans Trimble Geomatics Office, ou dans l'utilitaire Trimble Feature and Attribute Editor™, vous pouvez créer des nouveaux styles de lignes utilisant les types de ligne disponibles dans ce fichier.

Vous pouvez également créer vos propres fichiers .ltp. Pour plus amples informations concernant l'utilisation des fichiers .ltp, référez-vous au chapitre Utilitaires dans le Manuel de l'utilisateur du logiciel Trimble Geomatics Office.


Fichier Default.fcl

Le fichier Default.fcl se trouve dans le dossier Program Files\Trimble\Trimble Geomatics Office\System. Utilisez l'utilitaire Trimble Line Type Editor™ pour afficher et éditer ce fichiers. Tous les fichiers .fcl sont les bibliothèques d'attributs et caractéristiques.

Ce fichier contient des codes de caractéristiques prédéfinis (avec et sans attributs), des styles de point, des styles de ligne, des styles d'annotation et des codes de contrôle.

Pour utiliser ce fichier:

1.	Transférez-le au Trimble Survey Controller pour collecter des informations de codes de caractéristiques.

2.	Utilisez le fichier dans Trimble Geomatics Office pour traiter les codes de caractéristiques et configurer le projet pour des attributs si vous collectez des informations d'attribut.

Vous pouvez créer vos propres bibliothèques d'attributs et caractéristiques à l'aide de l'utilitaire Trimble Feature and Attribute Editor™. Pour plus amples informations concernant la création des bibliothèques d'attributs et caractéristiques, référez-vous à l'aide du Trimble Feature and Attribute Editor.


Fichier current.csd

Le fichier current.csd se trouve dans le dossier Program Files\Common Files\Trimble\Geodata.

Current.csd est la base de données de système de coordonnées de Trimble Geomatics Office. Ce fichier est utilisé lorsque vous sélectionnez un système de coordonnées pour un projet.

Pour afficher et éditer ce fichier, utilisez l'utilitaire Trimble Coordinate System Manager™. Pour plus amples information concernant la création des systèmes de coordonnées, référez-vous à l'aide du Trimble Coordinate System Manager.


Fichiers .frm

Les modèles de modèle de plan sont stockées sous forme des fichiers .frm. Ils se trouvent dans le dossier Program Files\Trimble\Trimble Geomatics Office\System.

Lorsque vous configurez la mise en page pour l'impression ou le traçage, vous devez sélectionner un modèle de plan à utiliser.

NOTE: Les fichiers .frm sont les fichiers texte. Pour les afficher et les éditer, utilisez un éditeur de texte.


Microsoft Access 97
Microsoft Access 2000

AVERTISSEMENT: Trimble Navigation Limited ne supporte l'utilisation ou la modification de cette base de données qu'en utilisant le logiciel Trimble Geomatics Office propriétaire de Trimble. Trimble ne supporte ni autorise l'utilisation ou la modification, ni garantit le fonctionnement, de cette base de données en utilisant d'autres applications informatiques. Trimble peut modifier le format de cette base de données sans préavis le rendant incompatible avec d'autres applications informatiques.

Trimble Geomatics Office utilise une base de données Microsoft Access 2000. Si vous avez une version précédente du logiciel Microsoft Access, vous ne pouvez pas ouvrir le fichier <projet>.mdb. Si vous avez Microsoft Access 2000 (ou plus récent) vous pouvez utiliser ce logiciel pour afficher vos données de projet et créer des nouvelles interrogations.


Interrogations de base de données - Vue d'ensemble

Plus amples informations

 Comment ...

AVERTISSEMENT: Trimble Navigation Limited ne supporte l'utilisation ou la modification de cette base de données qu'en utilisant le logiciel Trimble Geomatics Office propriétaire de Trimble. Trimble ne supporte ni autorise l'utilisation ou la modification, ni garantit le fonctionnement, de cette base de données en utilisant d'autres applications informatiques. Trimble peut modifier le format de cette base de données sans préavis le rendant incompatible avec d'autres applications informatiques.

Vue d'ensemble

Trimble Geomatics Office utilise une base de données Microsoft Access 2000. Vous pouvez utiliser Microsoft Access pour créer vos propres interrogations.

NOTE: Trimble ne prend en charge des interrogations définies de l'utilisateur.

Les détails fournis ici fournissent des informations limitées concernant la conception des interrogations à être utilisées dans Trimble Geomatics Office. S'il faut revoir ces techniques, nous suggérons que vous consultez la documentation fournie avec Microsoft Access.

Définition des interrogations Microsoft Access

En général, les interrogations que vous voulez définir d'être utilisées dans les projets peuvent être prédéfinis dans les modèles de projet.

Vous pouvez exporter les définitions d'interrogation aux autres bases de données Microsoft Access. Donc, si des définitions d'interrogation utiles ont été créées dans un projet particulier, vous pouvez exporter ces définitions au(x) modèle(s) du projet principal plus tard afin qu'elles soient disponibles dans d'autres projets Trimble Geomatics Office.

La clé pour l'utilisation des interrogations Microsoft Access dans Trimble Geomatics Office est que le(s) champ(s) d'ID spécifié(s) dans l'interrogation définissent les types d'entité qui doivent être sélectionnées par l'interrogation.

Deux conditions sont relatives à l'utilisation des champs d'ID dans une interrogation de sélection:

  L'interrogation doit être définie avec la première colonne de sortie (sélectionnez la case à cocher Afficher pour la colonne) étant un champ d'ID ou une table entière (l'option * utilisée).

  S'il faut concevoir l'interrogation pour sélectionner des types d'entité multiples (par exemple, la sélection des tous les points associés aux entités Arc), le champs d'ID relatifs à chacune des entités à sélectionner doivent être les seuls champs de sortie dans l'interrogation.

Définition des interrogations pour Trimble Geomatics Office

Si vous voulez définir une interrogation à être utilisée dans Trimble Geomatics Office, vous devez préfixer le nom de l'interrogation correctement afin que l'interrogation apparaît dans la liste requise.

NOTE: Dans les interrogations de rapport et d'exportation, la table de données de groupe de sélection doit être référencée dans le nom de l'interrogation. Par exemple, pour exporter une liste de points, l'interrogation devrait être dénommée Exporter(Point)_Liste points.

Si vous créez une interrogation pour les besoins de sélection et vous voulez qu'elle s'affiche dans la liste Sélectionner / Par interrogation, l'interrogation doit avoir le préfixe Sélect_. Toute interrogation avec le préfixe Sélect_ sera disponible aussi lors de la création des rapports personnalisés.

Si vous voulez concevoir une interrogation seulement pour des rapports, utilisez le préfixe Rapport(table de données de groupe de sélection)_. Les interrogations avec le préfixe Rapport(table de données de groupe de sélection)_ ne s'afficheront pas dans la liste Sélectionner / Par interrogation.

Si vous voulez ajouter une interrogation pour filtrer des éléments dans la liste de sélection de la fenêtre Propriétés, utilisez le préfixe SS_. L'interrogation s'affiche dans la liste lorsque vous cliquez sur l'outil Filtrer dans la fenêtre de Propriétés.


Transfert des paramètres aux interrogations dans Microsoft Access

AVERTISSEMENT: Trimble Navigation Limited ne supporte l'utilisation ou la modification de cette base de données qu'en utilisant le logiciel Trimble Geomatics Office propriétaire de Trimble. Trimble ne supporte ni autorise l'utilisation ou la modification, ni garantit le fonctionnement, de cette base de données en utilisant d'autres applications informatiques. Trimble peut modifier le format de cette base de données sans préavis le rendant incompatible avec d'autres applications informatiques.

Afin de pouvoir utiliser les mêmes interrogations personnalisées pour la sélection des entités différentes aux temps différents, il est possible de spécifier les paramètres comme partie de la définition d'interrogation avec le logiciel Microsoft Access.

Lorsqu'un paramètre est spécifié, Trimble Geomatics Office vous permet de saisir la valeur appropriée au temps du commencement de l'interrogation. Un exemple de l'utilisation d'un paramètre dans une interrogation est la provision d'une interrogation pour la sélection des entités de points selon leur code de caractéristiques.

Par exemple, vous pouvez sélectionner tous les points avec le code de caractéristique "Arbre" pendant une étape et puis tous les points avec le code de caractéristiques "Poteau" pendant une étape plus tard.

Un paramètre peut être spécifié pour une colonne dans une définition d'interrogation par l'entrée (c'est à dire, []) du message que vous voulez afficher, ainsi que la condition de comparaison à remplir (c'est à dire, =, >, <, Tel que), dans le champ de critères.

Par exemple, dans le cas d'une interrogation pour la sélection des points par leur code de caractéristiques, le champ Code de caractéristiques doit être ajouté à la définition d'interrogation. Le champ Critères doit définir la condition à remplir, par exemple: "Tel que [Code]".

Dans cet exemple l'opérateur de comparaison de chaînes Tel que est utilisé pour la vérification des codes correspondants.

Un autre perfectionnement est possible dans la spécification des paramètres d'interrogation. Ce vous permet de remplir le champ d'entrée de paramètres de Trimble Geomatics Office en cliquant sur les entités dans le modèle graphique. Ce perfectionnement est défini en suivant le message du paramètre avec deux points, puis le nom du champ de base de données rempli. Dans l'exemple de code ci-dessus ce serait:
Tel que [Code:Code de caractéristiques]

En plus de la définition de la condition du paramètres dans la champ Critère, vous devez également saisir la définition du paramètre dans le dialogue de l'interrogation. Ce dialogue apparaît du menu de raccourcis lorsque le pointeur est dans la partie supérieure du dialogue d'interrogation.

NOTE: Le champ Paramètre doit contenir exactement le même texte que le texte saisi entre les crochets du champ Critères.

Le champ Type de données de chaque paramètre est configuré selon le type de données saisi. Dans l'exemple ci-dessus, étant donné que le code de caractéristiques est un champ du type texte, c'est configuré sur Texte. Si un paramètre du type numéro réel telle qu'une hauteur de point est spécifié, le champ Type de données devrait être configuré sur Double.

Les détails fournis ici fournissent des informations limitées concernant la conception des interrogations d'être utilisées dans Trimble Geomatics Office. S'il faut revoir ces techniques, nous suggérons que vous consultez la documentation fournie avec Microsoft Access.


Utilisation des interrogations personnalisées pour filtrer un groupe de sélection

AVERTISSEMENT: Trimble Navigation Limited ne supporte l'utilisation ou la modification de cette base de données qu'en utilisant le logiciel Trimble Geomatics Office propriétaire de Trimble. Trimble ne supporte ni autorise l'utilisation ou la modification, ni garantit le fonctionnement, de cette base de données en utilisant d'autres applications informatiques. Trimble peut modifier le format de cette base de données sans préavis le rendant incompatible avec d'autres applications informatiques.

A l'aide de Microsoft Access, vous pouvez générer des interrogations plus compliquées qui fonctionnent avec la table Données de groupe de sélection, et permettent en effet l'interrogation de sélection de filtrer un groupe de sélection de base de données.

La table Données de groupe de sélection contient les ID d'enregistrement de toutes les entité dans un groupe de sélection.

Afin d'obtenir cette fonctionnalité de filtrage, la table Données de groupe de sélection doit être comprise dans la définition d'interrogation, ainsi que la table contenant toutes les entités à sélectionner (par exemple, la table Point). Dans ce  cas le champ d'ID d'enregistrement dans la table Données de groupe de sélection doit être lié au champ d'ID approprié dans l'autre table comprise dans l'interrogation. Cette liaison des champs dans les deux tables peut être effectuées en faisant glisser déplacer le nom du champ approprié dans une table sur le nom du champ dans l'autre table.

De la même façon que les autres interrogations de sélection discutées ci-dessus, le premier champ affiché dans l'interrogation doit être un champ d'ID ou une table entière. Pour que l'interrogation ne sélectionne que des entités d'un table spécifique, l'interrogation doit définir le nom de la table contenant les entités à sélectionner dans la table Données de groupe de sélection. Ce signifie que si des points sont sélectionnés, le champ Nom de table dans la table Données de groupe de sélection doit être ajouté à l'interrogation, avec les Critères configurés sur 'Point'.

La table Données de groupe de sélection contient les ID d'enregistrement de toutes les entités dans une groupe de sélection. Chaque groupe de sélection est assigné un numéro d'identification unique listé dans la colonne ID nom du groupe de sélection de la table Données de groupe de sélection. Le numéro 0 indique que l'entité est un membre du groupe de sélection courant, ainsi l'interrogation peut utiliser cette valeur pour assurer que seul le groupe de sélection courant est filtré. On le fait par l'ajout d'une autre colonne à l'interrogation spécifiant l'ID nom du groupe de sélection et configurant le champ Critères sur 0. La configuration de Critères sur une autre valeur vous permet d'effectuer la sélection courante basée sur un groupe de sélection existant si nécessaire.

Les détails fournis ici fournissent des informations limitées concernant la conception des interrogations d'être utilisées dans Trimble Geomatics Office. S'il faut revoir ces techniques, nous suggérons que vous consultez la documentation fournie avec Microsoft Access.


Modèles du projet - Vue d'ensemble

Plus amples informations

Comment...	Dialogue

Vue d'ensemble

Lors de la création d'un nouveau projet, vous devez sélectionner un modèle. Un modèle contient tous les paramètres, les données et les structures de fichier. Les informations dans le modèle sont copiées au projet lorsque vous le créez. Par exemple, si vous avez un fichier Survey Controller (*.dc) dans le Dossier d'entrée du modèle, le projet aura le même fichier .dc dans don Dossier d'entrée.  Toute édition effectuée au projet n'est pas effectuée au modèle.

Les modèles sont utilisés pour:

  la sélection des styles CAD à utiliser

  l'ajout des points de contrôle à la base de données

  l'ajout de tout fichier requis

  la sélection du système de coordonnées approprié

  la sélection des unités requises

  la sélection des paramètres de points en double

Trimble fournit de nombreux modèles avec Trimble Geomatics Office. Par exemple, le modèle métrique vous permet d'utiliser des mesures métriques partout dans le projets créés en utilisant ce modèle.

Les modèles sont stocké sans le dossier Program Files\Trimble\Trimble Geomatics Office\Template.


Commande Rapport de recalcul

Utilisez cette commande pour afficher le Rapport de recalcul. Le rapport est créé à partir du recalcul de données de levé le plus récent.

Pour y accéder, sélectionnez Rapports / Rapport de recalcul.

Le rapport est affiché dans votre afficheur HTLM par défaut.


Rapports du système - Vue d'ensemble

 Voir aussi

Les rapports du système sont générés à la suite d’une opération de traitement ou du calcul de données de levé. Ces rapports sont créés dans le dossier Rapport. Lorsqu’un nouveau rapport est généré, il écrase tout rapport existant ayant le même nom dans le dossier, sauf pour le Rapport d’importation.

Les rapports suivants sont automatiquement créés par le système après une opération de traitement :

  Rapport d’importation

  Rapport de recalcul

  Rapport du traitement de codes de caractéristiques

  Rapport du collage du projet

Pour être averti de la création de ces rapports, sélectionnez l’option Afficher rapport généré dans le dialogue Propriétés du projet − Tabulation Rapports.

Vous pouvez générer les rapports du système suivants après avoir effectué une opération de levé. Ces rapports fournissent les détails des calculs de levé ou la modification des données de levé.

  Recalcul

  Rapport de dérivation du point

  Rapport du traitement de lignes de base GPS (si le module Traitement de lignes de base WAVE est installé sur votre ordinateur)

  Rapport des fermetures de boucle GPS

  Rapport d’ajustement de réseau (si le module Ajustement de réseau est installé sur votre ordinateur)

  Rapport de calibration du site GPS

  Rapport de transformation de coordonnées

  Rapport d’ajustement d’élévation

  Rapport d’inverse

  Rapport Renommer Points

  Rapport de niveau

NOTE : La disponibilité de certains rapports dépend des modules (Traitement de lignes de base et Ajustement de réseau) qui sont installés sur votre ordinateur.


Rapport de base de données - Vue d'ensemble

 Voir aussi

Les rapports de base de données sont des rapports que vous générez à partir des enregistrements et interrogations de table primaire définis dans la base de données. C’est à dire, qui font un rapport sur les informations de la base de données du projet.

Ces rapports s’avèrent utiles si vous voulez obtenir un résumé du projet ou fournir des rapports aux clients.

Les rapports de base de données se décomposent en deux catégories :

  Rapports de base de données du système

  Rapports personnalisés

Vous pouvez accéder à ces rapports à partir du dialogue Rapports supplémentaires.

Vous pouvez utiliser ou modifier un rapport de base de données existant fourni par Trimble Geomatics Office, ou encore personnaliser votre propre format de rapport.


Rapports - Vue d'ensemble

Trimble Geomatics Office génère tous les rapports dans le format HyperText Markup Language (HTML) et les affiche dans un afficheur HTML installé sur votre ordinateur. Votre afficheur HTML par défaut est généralement un logiciel de navigation.

Tous les rapports sont créés dans le dossier Rapports du répertoire Trimble Geomatics Office\Projects. Les rapports sont organisés en dossiers avec le nom du type de rapport. Lorsqu’un nouveau rapport est généré, il écrase tout rapport existant ayant le même nom dans le dossier, sauf pour le Rapport d’importation et le Rapport d’ajustement de réseau.Deux types de rapports peuvent être générés :

  Rapports du système

  Rapports de base de données

Vous pouvez accéder aux rapports habituellement utilisés à partir du menu Rapports. La disponibilité de certains rapports du système dépend des modules (Traitement de lignes de base et Ajustement de réseau) qui sont installés sur votre ordinateur et du fait qu’un rapport a déjà été généré ou pas.

NOTE : Les rapports Trimble Geomatics Office ne peuvent pas être affichés dans Microsoft Internet Explorer version 2 ou 3. Si une de ces deux versions est installée sur votre ordinateur, Trimble vous recommande d’installer Microsoft Internet Explorer version 4 ou 5 à partir du Web.


Logiciel de navigation Microsoft Internet Explorer - Afficheur de rapports

Si aucun afficheur HTML n’est installé sur votre ordinateur, vous pouvez vous procurer le logiciel de navigation Microsoft Internet Explorer sur le Web. Le logiciel de navigation Microsoft Internet Explorer vous permet d’afficher et d’imprimer le rapport que vous avez créé. Vous pouvez utiliser d’autres fonctions pratiques de Microsoft Internet Explorer pour satisfaire vos exigences de rapports.

Personnaliser l’apparence du rapport

Les rapports de Trimble Geomatics Office sont affichés dans un format standardisé. Les polices de caractères, leurs tailles, les couleurs et les fonds sont les mêmes pour tous les rapports. Vous pouvez substituer vos propres préférences à ces paramètres. Procédez de cette manière si vous souhaitez afficher le rapport dans un format plus facile à lire pour vous.

ASTUCE : Pour modifier le fond du rapport, remplacez le fichier backgrnd.gif par un autre fichier graphique CompuServe Graphics Interchange (GIF) du même nom que vous trouverez dans le dossier Rapports.

Vous pouvez configurer les options d’impression de manière à ce que le fond du rapport ne soit pas imprimé. Par défaut, cette option est désactivée.

L’ouvrir dans une autre application informatique

De nombreuses applications informatiques prennent en charge les documents de format HTML. Par ailleurs, vous pouvez enregistrer le rapport sous un fichier Texte (*.txt) qui pourra être ouvert dans une autre application telle qu’un éditeur de texte, une feuille de calcul ou un programme de traitement de texte. Vous pouvez également copier une sélection du rapport dans le presse-papiers et le coller dans une autre application informatique.

Ces fonctions s’avèrent utiles lorsque vous souhaitez insérer un rapport dans un document, éditer le contenu du rapport ou utiliser les données pour effectuer des calculs supplémentaires dans une autre application.

Pour de plus amples informations sur l’utilisation des fonctions de Microsoft Internet Explorer pour vos rapports, référez-vous à Internet Explorer Help.


Dialogue Définir le format du rapport

Utilisez ce dialogue pour définir le format d’un rapport.

Pour y accéder :

1.	Sélectionnez Rapports / Rapports supplémentaires.

Le dialogue Rapports supplémentaires s’affiche.

2.	Effectuez l'une des opérations suivantes :

  Cliquez sur Nouveau pour définir un nouveau format de rapport.

  Choisissez un format de rapport existant dans la liste Rapports et cliquez sur Editer pour modifier le format sélectionné.

NOTE : Vous pouvez éditer les Points, l’Intervalle d’élévation, les Détails du système de coordonnées ou les Rapports de comparaison de points.

Nom

Entrez le nom du format de rapport. Ce nom s’affiche dans la liste Rapports du dialogue Rapports personnalisés. Les noms en double ne sont pas autorisés.

Chaîne nulle

La Chaîne nulle indique la chaîne nulle qui a été configurée dans Fichier / Propriétés du Project / tabulation Unités et format. La chaîne spécifiée dans ce champ est utilisée dans les champs du rapport personnalisé généré qui ont une valeur nulle. Ce champ est optionnel.

Description

Vous pouvez entrer une description du format de rapport dans le champ Description. Cette description s’affiche sur la barre d'état. Ce champ est optionnel.

Extension par défaut

L’extension par défaut n’est pas utilisée dans la définition d’un rapport de format.

Rapport sur

Sélectionnez l’une des tables ou interrogations primaires comme source pour le rapport. L’enregistrement Informations sur les points est la source par défaut. Cela vous permet de faire un rapport sur tout enregistrement de la table de points.

Spécifier format automatiquement

Cliquez sur Spécifier format automatiquement pour initialiser l’en-tête et le corps automatiquement. Pour de plus amples informations, référez-vous à Spécifier format automatiquement.

En-tête du format

Entrez le texte fixe et toutes les définitions de codes de champ (disponibles dans le menu de raccourcis) qui formeront l’en-tête du rapport. Ce champ est optionnel et il n’est pas nécessaire d’y spécifier une définition.

Pour ajouter des lignes vides à la définition de l’en-tête, appuyez sur Entrée.

Corps du format

Entrez les définitions de codes de champ appropriées (disponibles dans le menu de raccourcis) pour indiquer sur quelles données du projet Trimble Geomatics Office doit faire un rapport.

Par exemple, pour faire un rapport de liste de points à largeur fixe, utilisez le format suivant :

[Nom:10][Première ordonnée:14.3][Deuxième ordonnée:14.3][Elévation:14.3][Code de caractéristiques:20]

L’en-tête du format ou le corps du format (mais pas les deux) peut être laissé en blanc.

Bas de page du fichier de format

Entrez le texte fixe et toutes les définitions de code de caractéristiques (disponibles dans le menu de raccourcis) qui formeront le bas de page du rapport. Ce champ est optionnel et il n’est pas nécessaire d’y spécifier une définition. Par exemple, utilisez ce bas de page pour afficher une fin de fichier (commande de fin de fichier).

Pour ajouter des lignes vides à la définition du bas de page, appuyez sur la touche Entrée.

Pour de plus amples informations sur l’ensemble des codes de champ disponibles dans le système, référez-vous à Codes de champ − Vue d’ensemble.


Rapport de recalcul - Vue d'ensemble

Le Rapport de recalcul est un rapport en langage HyperText Markup Language (HTML) qui résume les résultats du processus de recalcul. Utilisez le Rapport de recalcul pour supprimer les indicateurs d’avertissement en :

  vérifiant les détails d’occupation

  vérifiant que les points ont été nommés correctement

  désactivant la coordonnée si vous pensez qu’elle a été saisie de manière incorrecte

  procédant à de nouvelles observations si nécessaire

Le Rapport de recalcul donne des informations sur les éléments suivants :

  Erreurs de tolérance lorsque des observations ou coordonnées multiples résultent en des positions hors tolérance. (Les tolérances sont définies dans la tabulation Recalculer du dialogue Propriétés du projet.)

  Fermetures lorsque plus d’une observation ou plus d’un groupe de coordonnées saisies au clavier pourrait être utilisé pour déterminer la position calculée du point, et lorsque les positions sont dans la tolérance.

  Observations inutilisées et erreurs.

  Coordonnées de début et l’ordre dans lequel les coordonnées ont été dérivées.

  Observations et coordonnées référencées dans le rapport.

Trimble Geomatics Office crée un Rapport de recalcul chaque fois qu’il effectue un recalcul. La façon dont vous êtes averti de la création d’un Rapport de recalcul dépend du champ Afficher rapport généré de la tabulation Rapports du dialogue Propriétés du projet. Le Rapport de recalcul est toujours enregistré dans le dossier Rapports du projet. Comme pour tous les rapports du système, si un rapport existe déjà, alors l’ancien rapport est écrasé par le nouveau.

Le Rapport de recalcul comprend les parties suivantes :

  Détails du projet

  Erreurs

  Avertissements

  Fermetures

  Dérivations du point

  Points de début

  Rapport de cheminement

  Données de levé (observations et coordonnées)


Rapport de recalcul - Section Détails du projet

La section Détails du projet du Rapport de recalcul reprend le nom du projet et les détails du système de coordonnées.


Rapport de recalcul - Sections d'erreur et avertissement

Les sections d’erreur et avertissement du Rapport de recalcul donne la liste de tous les points, pour lesquels plus d’un groupe de coordonnées a été dérivé et les coordonnées sont hors tolérance. Elles indiquent également quelles observations ne sont pas utilisées et les points pour lesquels il est impossible de déterminer des coordonnées.

Utilisez ces sections pour passer en revue les erreurs et avertissements, et pour décider si vous devez éditer votre projet pour y remédier. Pour consulter la dérivation pour un point, cliquez sur le point en question. Cette opération vous amène à la section Dérivation du point.

Les tolérances horizontales et verticales sont également indiquées. Elles sont configurées dans le dialogue Propriétés du projet − tabulation Recalculer.


Rapport de recalcul - Section Dérivations du point

La section Dérivations du point du Rapport de recalcul montre la position calculée d’un point. Seules les dérivations des points comprenant des erreurs ou fermetures du Rapport de recalcul font l’objet d’un rapport. Les qualités du composant horizontal (Nord, Est), du composant d’élévation et du composant de hauteur de la position calculée sont indiquées.

Sous la position calculée et les qualités se trouve une liste des données de levé qui pourraient être utilisées pour déterminer des coordonnées pour le point.

Chaque observation indique :

  le type d’observation

  l’ID d’observation

  les points « de » et « vers »

Pour afficher les détails d’observation, cliquez sur l’ID. Ce bouton est lié à la section Données de levé du Rapport de recalcul.

Pour sélectionner l’observation dans la fenêtre graphique (et dans la fenêtre Propriétés), sélectionnez l’icône à gauche de l’ID d’observation. Ce lien vous permet de visualiser facilement les fermetures hors tolérance dans l’Affichage du levé.

Chaque coordonnée indique :

  le type de coordonnées (grille, géodésique ou solution de position)

  la source de la coordonnée

  l’ID de coordonnée

Pour afficher les détails de coordonnées, cliquez sur l’ID. Ce bouton est lié à la section Données de levé du Rapport de recalcul.

De nouveau, pour sélectionner le point dans la fenêtre graphique et dans la fenêtre Propriétés, sélectionnez l’icône à gauche de l’ID d’observation.

La colonne Utilisé pour calc. indique quels composants de position (NE,é,h) ont été calculés à partir de l’observation ou de la coordonnée. Par exemple, si une observation détermine les composants de Nord, d’Est et de hauteur du point et que la coordonnée détermine l’élévation, la colonne Utilisé pour calc. affiche NEh à côté de l’observation GPS et é à côté de la coordonnée.

Les colonnes delta nord, delta est, delta élévation et delta hauteur d’ellipsoïde indiquent la différence entre la position calculée finale et la position calculée à l’aide de l’observation ou de la coordonnée uniquement.

Si les valeurs de delta sont supérieures au paramètre de tolérance configuré pour le projet, des indicateurs s’affichent et les nombres deviennent rouges. Les observations en rouge n’indiquent pas nécessairement une observation ou des coordonnées saisies au clavier incorrectes — il se peut que le recalcul ait utilisé une observation ou un groupe de coordonnées saisies au clavier de mauvaise qualité. Vous devez déterminer la cause de l’erreur et désactiver toute observation de mauvaise qualité.

Utilisez la section de levé du rapport pour passer chaque observation en revue.

ASTUCE : Vous pouvez utiliser la fenêtre Propriétés et la commande Rapports / Rapport de dérivation du point pour afficher le rapport de dérivation de tout point du projet, y compris les points sur lesquels le Rapport de recalcul ne contient aucune information.


Rapport de recalcul - Section Rapport de cheminement

La section Rapport de cheminement du Rapport de recalcul indique comment un recalcul chemine à partir du point de début et montre les observations qu’il utilise.

Un cheminement fait l’objet d’un rapport uniquement s’il y a plus d’une étape à partir du point de début. Cela permet d’éviter que les déports ne soient rapportés.

Par exemple : si des observations sont :

  Du point de début 1 − au point 2 − au point 3 puis retour au point 1, cela fait l’objet d’un rapport.

  Du point de début 1 − au point 2, cela ne fait pas l’objet d’un rapport.


Rapport de recalcul - Section Fermetures

La section Fermetures du Rapport de recalcul contient la liste des points de station avec des observations multiples vers un point, ces points ne dépassant pas la tolérance spécifiée pour le projet. Les fermetures de données de levé d’un point sont indiquées dans les colonnes delta nord, delta est, delta élévation et delta hauteur d’ellipsoïde.

Cliquez sur un point pour afficher les détails de dérivation. Cette opération vous amène à la section Dérivations du point.


Rapport de recalcul - Section Données de levé

La section Données de levé contient toutes les observations et coordonnées des points figurant dans le Rapport de recalcul. Les données de levé sont organisées par type d’observation. Utilisez cette section lorsque vous essayez de déterminer quelles observations sont incorrectes.

Seules les observations et coordonnées référencées dans le Rapport de recalcul sont listées dans cette section.

Vous pouvez afficher toutes les données de levé dans la fenêtre Propriétés.


Rapport de recalcul - Section Points de début

La section Points de début du Rapport de recalcul indique les points à partir desquels le recalcul a commencé à générer les observations.


Rapport de recalcul - Détails supplémentaires du système de coordonnées

La section Détails supplémentaires du système de coordonnées donne la liste des informations relatives aux paramètres du site local.

Les paramètres du site local sont configurés dans le dialogue Propriétés du projet − tabulation Système de coordonnées.


Commande Rapport de dérivation du point

Utilisez cette commande pour afficher le Rapport de dérivation du point pour le(s) point(s) sélectionné(s).

Pour y accéder :

1.	Sélectionnez le(s) point(s) que vous souhaitez afficher.

2.	Sélectionnez Rapports / Rapport de dérivation du point.

Le Rapport de dérivation du point s’affiche dans l’afficheur HTML par défaut.


Rapport de dérivation du point

Le Rapport de dérivation du point est une section du Rapport de recalcul qui comprend la position calculée d’un point. Les qualités du composant horizontal (Nord, Est), du composant d’élévation et du composant de hauteur de la position calculée sont indiquées.

Le format du rapport est le même que celui de la section Dérivations du point du Rapport de recalcul.

Pour accéder au rapport :

1.	Sélectionnez le(s) point(s) que vous souhaitez afficher.

2.	Effectuez l'une des opérations suivantes :

  Sélectionnez Rapports / Rapport de dérivation du point.

  Ouvrez la fenêtre Propriétés / tabulation Levé / page Résumé et sélectionnez l’outil Dérivation du point [image].

Le rapport s’affiche dans l’afficheur HTML.

NOTE : Pour sélectionner un point dans l’Affichage du levé, cliquez sur l’icône de point ou d’observation à gauche de l’ID. Cela s’avère utile pour retourner au point dans l’Affichage du levé.


Commande Rapport d'inverse

Utilisez cette commande pour afficher le plus récent Rapport d’inverse généré à partir du dialogue Inverse après qu’une inverse a été calculée.

Pour y accéder :

  Sélectionnez Rapports / Rapport d’inverse.

Le rapport s’affiche dans l’afficheur HTML par défaut.


Rapport d'inverse

Le Rapport d’inverse peut être généré après le calcul d’une inverse et affiche des informations résumées ou détaillées sur l’inverse ou une combinaison des deux.

Le Rapport d’inverse résumé comprend l’azimut de grille, la distance de grille et la delta élévation entre les points « de » et « vers » utilisés pour définir l’inverse.

Le rapport détaillé affiche les positions des points « de » et « vers » dans les systèmes de coordonnées de grille, local (géodésique) et cartésien (WGS-84), ainsi que les paramètres de l’inverse calculée.

Utilisez le dialogue Inverse pour générer le rapport.

Après la création d’un rapport, vous pouvez afficher le rapport le plus récent à tout moment en sélectionnant Rapports / Rapport d’inverse.


Rapport Renommer points

Le rapport Renommer points est un rapport que vous pouvez générer pour afficher les noms d’origine et les nouveaux noms des points renommés à l’aide de la commande Renommer points.

Utilisez le dialogue Renommer pour générer le rapport.


Rapports d'importation

Chaque fois qu’un fichier est transféré à un projet, Trimble Geomatics Office crée un Rapport d’importation pour chaque fichier importé. Ce fichier s’appelle <nom du fichier importé>.html. Il contient les sections suivantes :

Détails du projet

Cette section contient une synthèse du nom du projet et des détails du système de coordonnées.

Messages

Cette section liste tous les messages qui ont été produits au moment de l’importation. Si aucun message n’est produit au moment de l’importation, cette section n’est pas affichée.

Rapport de recalcul

Ce rapport comprend un lien pour le Rapport de recalcul, ce rapport étant automatiquement généré après le processus d’importation. Le Rapport de recalcul résume les résultats du processus de recalcul.

Rapport du traitement des codes de caractéristiques

Ce rapport comprend un lien pour le rapport de codification lorsque des codes de caractéristiques ont été traités au moment de l’importation.

NOTE : Si le Rapport de recalcul ou le Rapport du traitement des codes de caractéristiques sont en rouge, cela signifie que des erreurs ont été rencontrées durant le processus et qu’elles sont indiquées dans les rapports.

Un Rapport de fichiers importés appelé Import.html est également créé durant le processus d’importation. Ce rapport donne la liste de tous les fichiers qui sont importés dans le projet. Tout fichier importé dans le même projet lors d’une autre session est automatiquement ajouté à ce rapport.

La façon dont vous êtes averti de la création de ces rapports est configurée dans l’option Afficher rapport généré du dialogue Propriétés du projet − tabulation Rapports.


Commande Rapports supplémentaires

Utilisez cette commande pour générer des rapports de base de données du système ou personnalisés fournis par Trimble Geomatics Office.

Pour y accéder, sélectionnez Rapports / Rapports supplémentaires. Le dialogue Rapports supplémentaires s’affiche.


Dialogue Rapports supplémentaires

Utilisez le dialogue Rapports supplémentaires pour générer des rapports de base de données du système ou personnalisés fournis par Trimble Geomatics Office. Pour accéder à ce dialogue, sélectionnez Rapports / Rapports supplémentaires.

Rapports de base de données du système

Les rapports de base de données du système sont des formats de rapport prédéfinis qui font un résumé du projet courant. Bien que vous ne puissiez généralement pas éditer les rapports, vous pouvez éditer les formats de rapport. Pour cela, éditez le fichier Asciirpt.dat qui se trouve dans le dossier Program Files\Trimble\Trimble Geomatics Office\System.

Rapports personnalisés

Les rapports personnalisés utilisent, conformément à vos spécifications, les types d’enregistrements ou d’interrogations primaires définis dans la base de données.

Vous pouvez créer, définir ou éditer un format de rapport personnalisé. Cependant, vous pouvez également empêcher que ces formats de rapport ne soient édités en éditant le fichier Asciirpt.dat qui se trouve dans le dossier Program Files\Trimble\Trimble Geomatics Office\System.

Si le nom du rapport que vous souhaitez effectuer est suivi par une ellipse, alors vous devez d’abord entrer certaines informations. Par exemple, lorsque vous choisissez d’effectuer un rapport d’élévation, Trimble Geomatics Office vous demande d’entrer les élévations minimum et maximum.

Rapport sur

Vous avez deux options pour effectuer un rapport sur des entités ou types d’enregistrement sélectionnés :

Sélection	Sélectionnez cette option pour effectuer un rapport sur les entités ou types d’enregistrement sélectionnés.
BD entière	Sélectionnez cette option pour effectuer un rapport sur toutes les entités ou types d’enregistrement.
Si rien n’est sélectionné, ces boutons ne sont pas disponibles et toutes les entités ou types d’enregistrement font l’objet d’un rapport.

NOTE : Lorsque le rapport Détails du système de coordonnées est sélectionné, cette option n’est pas disponible.

Trier données

Vous pouvez choisir l’ordre dans lequel les données font l’objet d’un rapport :

Pour cela :

1.	Sélectionnez la case à cocher Trier données.

2.	Sélectionnez un ordre de triage dans la liste.  Vous pouvez choisir de trier par :

Ordre de base de données	Elévation
Nom	Qualité
Heure de début	Type de données
Durée d’occupation

NOTE : Les mêmes options de triage sont disponibles dans la fenêtre Propriétés.

ASTUCE : Si vous créez vos propres interrogations de triage, elles s’affichent également dans la liste.

Rapports

Le groupe Rapports liste les rapports de base de données du système ou personnalisés. Un certain nombre de rapports ont déjà été définis par Trimble Geomatics Office.

OK

Génère le rapport et l’affiche dans l’afficheur HTML par défaut.

Nouveau

Crée et définit un nouveau format de rapport.

Editer

Modifie un format de rapport existant.

Supprimer

Supprime un rapport. Il vous est demandé de confirmer votre action.

NOTE : Les valeurs de séparation que vous entrez sont interprétées en fonction des unités de coordonnées de la base de données plutôt que des unités d’affichage courantes. Les unités de la base de données sont configurées sur la base des unités d’affichage au moment où les premières données ont été enregistrées dans la base de données.

ASTUCE : Vous pouvez modifier ces rapports en éditant le fichier Asciirpt.dat, qui se trouve dans le dossier Program Files\Trimble\Trimble Geomatics Office\System, à l’aide de l’éditeur de texte.


    Commande Rapport de calibration du site GPS

Utilisez cette commande pour afficher le Rapport de calibration du site GPS le plus récent généré à partir du dialogue Calibration du site GPS après un calcul de calibration.

Pour y accéder, sélectionnez Rapports / Rapport de calibration du site GPS

Le rapport s’affiche dans l’afficheur HTML par défaut.

NOTE : Cette commande n’est pas disponible dans l’Affichage du levé.

Pour accéder au Rapport de calibration du site dans l’Affichage du levé, cliquez sur Rapport dans le dialogue Calibration du site GPS.


Rapport de calibration du site GPS

Le Rapport de calibration du site GPS est généré après un calcul de calibration. Il comprend tous les paramètres calculés ainsi qu’une liste des positions de point calculées comparées à leurs positions et valeurs résiduelles individuelles connues.

Utilisez le dialogue Calibration du site GPS pour générer un rapport.


Commande Rapport de transformation de coordonnées

Utilisez cette commande pour afficher le Rapport de transformation de coordonnées le plus récent généré à partir du dialogue Transformation de coordonnées après un calcul de transformation.

Pour y accéder, sélectionnez Rapports / Rapport de transformation de coordonnées.

Le rapport s’affiche dans l’afficheur HTML par défaut.

NOTE : Cette commande de menu n’est pas disponible dans l’Affichage du levé. Pour générer le rapport à partir de l’Affichage du levé, sélectionnez la case à cocher Générer le rapport de transformation de coordonnées dans le dialogue Transformation de coordonnées.


Rapport de transformation de coordonnées

Le Rapport de transformation de coordonnées est généré après un calcul de transformation de coordonnées et montre les coordonnées d’origine et les coordonnées transformées.

Utilisez le dialogue Transformation de coordonnées pour générer le rapport.

Après la création d’un rapport, vous pouvez afficher le rapport le plus récent à tout moment en sélectionnant Rapports / Rapport de transformation de coordonnées dans la Vue en plan.


Commande Rapport d'ajustement d'élévation

Utilisez cette commande pour afficher le Rapport d’ajustement d’élévation le plus récent généré à partir du dialogue Ajustement d’élévation après qu’un ajustement a été effectué.

Pour y accéder, sélectionnez Rapports / Rapport d’ajustement d’élévation.

Le rapport s’affiche dans l’afficheur HTLM par défaut.

NOTE : Cette commande n’est pas disponible par défaut dans l’Affichage du levé. Pour générer le rapport à partir de l’Affichage du levé, sélectionnez la case à cocher Générer le rapport d’ajustement d’élévation dans le dialogue Ajustement d’élévation.


Rapport d'ajustement d'élévation

Le Rapport d’ajustement d’élévation est généré après qu’un ajustement d’élévation a été effectué et montre les coordonnées d’origine et les coordonnées ajustées.

Utilisez le dialogue Ajustement d’élévation pour générer le rapport.

Après la création d’un rapport, vous pouvez afficher le rapport le plus récent à tout moment en sélectionnant Rapports / Rapport d’ajustement d’élévation dans la Vue en plan.


Rapport du traitement de codes de caractéristiques

Le Rapport du traitement de codes de caractéristiques est généré à la suite d’une opération de traitement de codes de caractéristiques.

En configurant l’option Afficher rapport généré dans le dialogue Propriétés du projet − tabulation Général vous êtes averti lorsque le rapport est généré.

Le rapport résume l’opération de traitement et comprend des détails sur le fichier de Bibliothèque de caractéristiques et attributs (.fcl) utilisé et sur tout problème rencontré lors du traitement des codes de caractéristiques.


Rapport du collage du projet

Le Rapport du collage du projet est généré après que des points ont été copiés et collés dans un projet.

En configurant l’option Afficher rapport généré dans le dialogue Propriétés du projet − tabulation Général vous êtes averti lorsque ce rapport est généré.

Le rapport résume tous les attributs et caractéristiques assignés à un point copié, qui ne sont pas collés si les définitions d’attributs diffèrent entre les projets.


Commande Rapport de niveau

Utilisez cette commande pour afficher le Rapport de niveau du fichier sélectionné.

Pour y accéder :

1.	Sélectionnez Rapports / Rapport de niveau.

2.	Sélectionnez le fichier de niveaux.

Le Rapport de niveau du fichier de niveaux que vous avez sélectionné s’affiche dans l’afficheur HTML par défaut.


Rapport de niveau

Le Rapport de niveau comprend les observations et données de niveau de votre Fichier de niveaux.

Pour y accéder :

1.	Sélectionnez Rapports / Rapports de niveau.

2.	Sélectionnez le Fichier de niveaux.

Le rapport s’affiche dans l’afficheur HTML par défaut. Il affiche les informations suivantes :

Données de niveau

Cette section montre la série de niveaux dans un format de niveau fieldbook. Elle comprend des observations de visée arrière, de visée intermédiaire et de visée avant ainsi que les informations sur l’élévation calculée affichées dans le dialogue Importation de niveaux numériques.

Observation de niveau

La table des observations de niveau contient les informations suivantes :

  ID

  De point

  Vers point

  Qualité

  Δ Elévation −  ces valeurs ne sont pas calculées entre les points de station.

ASTUCE : Pour sélectionner l’observation de niveau dans la fenêtre graphique, sélectionnez l’ID d’observation. Ce lien vous permet d’afficher facilement l’observation de niveau dans la fenêtre Propriétés.

NOTE : Le Rapport de niveau est enregistré dans le dossier <Nom du projet>\Reports\Level.


Commande Tout

Utilisez cette commande pour sélectionner toutes les entités dans le projet courant.

Pour y accéder, choisissez Sélectionner / Tout.

Affichage du levé

Lorsque vous travaillez dans Affichage du levé, toutes les entités visibles sont sélectionnées.

Par exemple, si vous utilisez la commande Affichage / Filtres pour masquez certaines observations, puis vous utilisez la commande Sélectionner  / Tout, les observations masquées ne sont pas sélectionnées.

Vue en plan

Lorsque vous travaillez dans la Vue en plan, toutes les entités sont sélectionnées qui se trouvent dans une couche qui a:

  sa propriété Visible configurée sur activé

  sa propriété Verrouillée configurée sur désactivé.

Par exemple, lorsque vous utilisez l'outil de sélection de couche
pour activer la propriété Verrouillé d'une couche, utilisez la commande Sélectionner / Tout, toutes les entités dans la couche verrouillée ne seront pas sélectionnées.

ASTUCE: La barre d'état affiche un résumé du nombre d'entités sélectionnées.


Commande Aucun

Utilisez cette commande pour effacer la sélection.

Pour y accéder, choisissez Sélectionner / Aucun.

ASTUCE: La barre d'état affiche un message indiquant que la sélection est vide.


Commande Points

Utilisez cette commande pour ouvrir le dialogue Sélectionner points.

Pour y accéder, choisissez Sélectionner  / Points.


Dialogue Sélectionner points

Vue d'ensemble	Comment ...

Utilisez ce dialogue pour sélectionner les points selon les critères spécifiés. Vous pouvez utiliser des caractères génériques et des intervalles. Vous pouvez également saisir des critères dans plus d'une tabulation. Trimble Geomatics Office ne sélectionne que des points qui répondent à tous les critères.

Pour y accéder, choisissez Sélectionner / Points.

Lorsque vous accédez au dialogue en utilisant l'Affichage du levé, Trimble Geomatics Office ne prend en considération que les points qui sont visibles actuellement. Ce signifie que si vous avez utilisé le dialogue Filtres d'affichage pour filtrer les observations, ces points ne seront pas sélectionnés. Lors de l'utilisation de l'Affichage du levé, les informations de couche (disponibles dans la Vue en plan) sont ignorées dans la sélection de critères.

Lorsque vous accédez à ce dialogue en utilisant la Vue en plan, Trimble Geomatics Office ne prend en considération que les points qui sont visibles actuellement et qui ne sont pas verrouillés. Lors de l'utilisation de la Vue en plan, tout filtre d'affichage appliqué dans l'Affichage du levé est ignoré.

Les tabulations sont:

  Général

  Qualité GPS

  Occupation

Ajouter à la sélection courante

Si vous avez une ou plusieurs entités sélectionnées quand vous accédez à ce dialogue, cette case à cocher devient disponible. Si vous voulez effacer la sélection courante avant d'appliquer la sélection obtenue de ce dialogue, effacez cette case à cocher.


Dialogue Sélectionner points - tabulation Général

Utilisez cette tabulation pour saisir des critères concernant les champs de point généraux.

Correspondance de noms et Correspondance de codes de caractéristiques

Utilisez ces champs pour sélectionner par nom (et par code de caractéristiques). Les noms séparés par une virgule peuvent être saisis. Par exemple,  "100,250,1020" effectue la sélection de ces noms un à la fois lorsqu'ils existent dans le projet. L'ellipse (...) ou un tiret (-) effectue la sélection de chaque point avec le nom spécifie dans l'intervalle.

Avec un intervalle comprenant des noms alphanumériques, certaines entités inattendues peuvent être sélectionnées car un tri alphabétique complet est utilisé.

NOTE: Si les noms de début et de fin de l'intervalle sont numériques, un intervalle numérique est sélectionné.

Vous pouvez utiliser les caractères génériques dans ces champs.

NOTE: Lorsque le nom contient un tarit d'union, il faut comprend ce trait d'union entre guillemets. Par exemple, le nom de point "a-2" est saisi comme "a-2". Autrement Trimble Geomatics Office effectue le tri de la gamme de a à 2.

Utilisez les doubles guillemets pour éviter la correspondance de modèles dans les noms contenant des caractères génériques. Par exemple, "URS#1"..."URS#8" évite que # soit utilisé comme un caractère générique dans la gamme de sélection, et effectue la sélection des points avec des noms explicites entre URS#1 et URS#8.

Les noms ne sont pas sensibles à la casse.

Avec qualité(s) de point

De la liste, sélectionnez la (les) qualité(s) de composant de position du point dans vos critères. Vous pouvez sélectionner des qualités de point multiples.

De source(s)

De la liste, sélectionnez les source(s) des points. Vous pouvez sélectionner des source(s) multiples. Les points ayant les données de levé d'au moins une source sélectionnée seront sélectionnés.

Avec code de caractéristiques

Saisissez le(s) code(s) de caractéristiques à utiliser pour la sélection des points.

De couche(s)

Cette option n'est pas disponible lorsque vous accédez à ce dialogue de l'Affichage du levé.

Utilisez la liste pour sélectionner des couches dans vos critères. Vous pouvez sélectionner des couches multiples. Tous les points dans les couches sélectionnées seront sélectionnés.

Avec style(s) de point

De la liste, sélectionnez les styles CAD de point dans vos critères. Vous pouvez sélectionner des styles de point multiples. Tous les points avec le style de point (spécifié dans la fenêtre Propriétés – tabulation CAD sont sélectionnés.

Ajouter à la sélection courante

Si vous avez une ou plusieurs entités sélectionnées quand vous accédez à ce dialogue, cette case à cocher devient disponible. Si vous voulez effacer la sélection courante avant d'appliquer la sélection obtenue de ce dialogue, effacez cette case à cocher.


Dialogue Sélectionner points - tabulation GPS

Utilisez cette tabulation pour saisir les critères de sélection utilisant des données GPS. Les critères s'appliquent aux lignes de base GPS RTK et post traitées.

Lorsqu'un point a au moins une ligne de base le joignant que répond aux critères, le point est sélectionné.

Précision horizontale, Précision verticale

Cliquez sur le bouton moins de et égal à ou le bouton plus de et égal à pour sélectionner le signe pour vos critères.

Saisissez les valeurs à l'aide du clavier pour les valeurs de précision.

Lorsque vous laissez un des champs comme un point d'interrogation (?), le champ n'est pas utilisé dans les critères de sélection.

Type(s) de solution

Pour utiliser le(s) type(s) de solution dans vos critères, sélectionnez de la liste. Vous pouvez sélectionner des types de solution multiples. Tout point ayant des observations GPS (avec les types de solution sélectionnés) est sélectionné.

Méthode(s) sur le terrain

De la liste, sélectionnez la méthode sur le terrain que vous voulez utiliser dans vos critères. Vous pouvez sélectionner des méthodes sur le terrain multiples.

NOTE: Seuls les points qui répondent au type de solution et à la méthode sur le terrain sont sélectionnés.

Ajouter à la sélection courante

Si vous avez une ou plusieurs entités sélectionnées quand vous accédez à ce dialogue, cette case à cocher devient disponible. Si vous voulez effacer la sélection courante avant d'appliquer la sélection obtenue de ce dialogue, effacez cette case à cocher.


Dialogue Sélectionner points - tabulation Occupation

Utilisez cette tabulation pour saisir les critères de sélection utilisant les détails d'occupation. Tous les points avec au moins une occupation qui répond aux critères sont sélectionnés.

Toutes les stations de base et les installations d'instrument

Sélectionnez cette case à cocher pour choisir tous les points de stations de base GPS et les points d'installations terrestres.

Avec hauteurs de destination ou d'antenne entre

Utilisez ces champs pour saisir une hauteur d'antenne ou une portée de hauteur de destination dans vos critères. La valeur saisie dans le premier champ doit être inférieure à la valeur saisie dans le deuxième champ. Les valeurs saisies sont inclusives.

Occupé entre les heures

Sélectionnez cette case à cocher pour saisir une intervalle de date et d'heure  pour utiliser dans vos critères. Les occupations de point avec l'heure de début et l'heure d'arrêt dans cet intervalle sont sélectionnées. Le logiciel enregistre les valeurs saisies pour la prochaine fois que vous utilisez cette option.

Observé du point

Saisissez un nom de point ayant des observations. Le point peut être soit une station de base GPS, soit un point d'installation terrestre.

Utilisez le remplissage du champ pour sélectionner un point dans la fenêtre graphique.

Ajouter à la sélection courante

Si vous avez une ou plusieurs entités sélectionnées quand vous accédez à ce dialogue, cette case à cocher devient disponible. Si vous voulez effacer la sélection courante avant d'appliquer la sélection obtenue de ce dialogue, effacez cette case à cocher.


Sélection des entités - à l'aide des caractères génériques

 Vue d'ensemble et plus amples informations

Lorsque vous sélectionnez des points utilisant le dialogue Sélectionner / Points, ou que vous sélectionnez des observations utilisant le dialogue Sélectionner / Observations, vous pouvez utiliser des caractères génériques dans certains champs pour sélectionner une gamme de valeurs.

Vous pouvez utiliser les caractères génériques suivants:

Vous pouvez utiliser des caractères génériques multiples. Par exemple, 1?0*0 correspond à  1500, 1A0250, 15020.

La table suivante illustre comment vous pouvez utiliser les types de correspondance afin de tester pour les modèles.

Le nom de point "10" est considéré comme se trouvant entre "1" et "a". Il ne se trouvera pas dans l'intervalle "1" à "9" parce que tous les deux noms sont numériques.


    Commande Observations

Utilisez cette commande pour ouvrir le dialogue Sélectionner observations.

Pour y accéder, choisissez Sélectionner / Observations.


    Dialogue Sélectionner observations

Plus amples informations

Vue d'ensemble	Comment...

Pour y accéder, choisissez Sélectionner / Observations.

Utilisez ce dialogue pour sélectionner des observations utilisant les critères spécifiés. Vous pouvez utiliser des caractères génériques et des intervalles. Vous pouvez également saisir des critères dans plus d'une tabulation. Trimble Geomatics Office ne sélectionne que des observations qui répondent à tous les critères.

Les tabulations sont:

  Général

  GPS

Ajouter à la sélection courante

Si vous avez une ou plusieurs entités sélectionnées quand vous accédez à ce dialogue, cette case à cocher devient disponible. Si vous voulez effacer la sélection courante avant d'appliquer la sélection obtenue de ce dialogue, effacez cette case à cocher.


    Dialogue Sélectionner observations - tabulation Général

Utilisez cette tabulation pour saisir des critères généraux pour la sélection des observations.

De point correspondant, Vers point correspondant

Les noms séparés par une virgule peuvent être saisis. L'ellipse (...) ou un tiret (-) effectue la sélection de chaque ligne de base avec le nom spécifié dans l'intervalle indiqué par l'ellipse. Avec un intervalle comprenant des noms alphanumériques, certaines entités inattendues peuvent être sélectionnées car un tri alphabétique complet est utilisé.

NOTE: Si les noms de début et de fin de l'intervalle sont numériques, un intervalle numérique est sélectionné.

Vous pouvez utiliser des caractères génériques.

Avec distance inclinée entre

Utilisez ces champs pour saisir un intervalle de distance inclinée dans vos critères. La valeur saisie dans le premier champ doit être inférieure à la valeur saisie dans le deuxième champ. Les valeurs saisies sont inclusives.

Lorsque vous avez une observation réduite avec une distance horizontale dans l'intervalle spécifié, les observations sont sélectionnées aussi..

Types

De cette liste, sélectionnez le type d'observation . Par exemple pour pouvez sélectionner les observations RTK, Laser ou Azimut.

De source(s)

De la liste, sélectionnez la (les) source(s) d'observation. Vous pouvez sélectionner des sources multiples. Toutes les observations ayant au moins une source d'observation sont sélectionnées.

Avec état

De la liste, sélectionnez l'état d'observation dans vos critères.

Avec ID d'observation

Saisissez l'ID d'observation, ou une série d'ID d'observation, pour sélectionner la (les) observation(s). Par exemple, pour sélectionner la ligne de base GPS 8, saisissez LB8. Saisissez LB8−LB12 pour sélecxtionner toutes les observations dans cet intervalle.

Prises de vue isolées seules

Losrque cette case à cocher est sélectionnée, seules les prises de vue isolées sont sélectionnées.

Ajouter à la sélection courante

Si vous avez une ou plusieurs entités sélectionnées quand vous accédez à ce dialogue, cette case à cocher devient disponible. Si vous voulez effacer la sélection courante avant d'appliquer la sélection obtenue de ce dialogue, effacez cette case à cocher.


    Dialogue Sélectionner observations - tabulation GPS

Utilisez cette tabulation pour saisir des critères pour la sélection des lignes de base GPS.

Ratio

Sélectionnez le signe moins de ou plus de. Saisissez la valeur de ratio que vous voulez utiliser dans vos critères.

Variance de réf

Sélectionnez le signe moins de ou plus de Saisissez la variance de référence que vous voulez utiliser dans vos critères.

RMS

Sélectionnez le signe moins de ou plus de. Saisissez la valeur RMS que vous voulez utiliser dans vos critères.

Préc horz

Cliquez sur le bouton moins de ou égal à ou le bouton plus de ou égal à pour sélectionner le signe pour vos critères. Saisissez la valeur de précision horizontale que vous voulez utiliser dans vos critères.

Préc vert

Cliquez sur le bouton moins de ou égal à ou le bouton plus de ou égal à pour sélectionner le signe pour vos critères. Saisissez la valeur de précision verticale que vous voulez utiliser dans vos critères.

NOTE: Les interrogations Préc horz et Préc vert sont basées sur des valeurs exactes. Les valeurs affichées telle que les champs dans la fenêtre Propriétés utilisent les chiffres ronds.

Type(s) de solution

De la liste, sélectionnez le(s) type(s) de solution que vous voulez utiliser dans vos critères. Vous pouvez sélectionner des types de solution multiples.

Méthode(s) sur le terrain

De la liste, sélectionnez la méthode(s) sur le terrain GPS que vous voulez utiliser dans vos critères. Vous pouvez sélectionner des méthodes sur le terrain multiples.

NOTE: Seuls les observations qui répondent au type de solution et à la méthode sur le terrain sont sélectionnés.

Observé entre

Utilisez ces champs pour spécifier une intervalle de date et de temps pour vos critères. Les observations avec une heure de début et de fin dans les critères spécifiés sont sélectionnées. Le logiciel enregistre les valeurs saisies pour la prochaine fois que vous utilisez cette option.

Observé pendant plus de (sec)

Utilisez ce champ pour sélectionner les lignes de base GPS observées pendant plus du temps spécifié. Ce champ a le format hh:mm:ss.

NOTE: Vous pouvez utilissez le remplissage du champ pour spécifier le temps.

Ajouter à la sélection courante

Si vous avez une ou plusieurs entités sélectionnées quand vous accédez à ce dialogue, cette case à cocher devient disponible. Si vous voulez effacer la sélection courante avant d'appliquer la sélection obtenue de ce dialogue, effacez cette case à cocher.


    Commande Par couche

Utilisez cette commande pour sélectionner toutes les entités des couches spécifiées.

Pour y accéder:

  Choisissez Sélectionner / Par couche.

Le dialogue Sélectionner par couches s'ouvre.


    Dialogue Sélectionner par couches

Utilisez ce dialogue pour sélectionner les entités dans les couches choisies.

Pour y accéder:

  Choisissez Sélectionner / Par couche.

Couches

La liste Couches affiche les noms des couches dans le projet.

Pour sélectionner les entités qui s'affichent dans une couche, sélectionnez la case à cocher à côté du nom de couche.

Lorsqu'une couche a sa propriété Verrouillée configurée sur activé, ou sa propriété Visible configuré sur désactivé, la case à cocher des couches s'affiche grisée. Afin de permettre la sélection de la couche, utilisez l'outil Sélection de couche ou le dialogue Couche pour changer les propriétés Verrouillée et Visible. La case à cocher de la couche devient disponible.

Tout

Cliquez sur Tout pour sélectionner chaque couche affichée dans la liste Couches.

Aucun

Cliquez sur Aucun pour effacer toutes les cases à cocher de toutes les couches dans la liste Couches.


    Commande Par styles

Utilisez cette commande pour sélectionner toutes les entités utilisant les styles spécifiés.

Pour y accéder, choisissez Sélectionner / Par styles.

Le dialogue Sélectionner par styles s'ouvre.


    Dialogue Sélectionner par styles

Utilisez ce dialogue pour sélectionner toutes les entités CAD (points, lignes, arcs, courbes, annotations, texte) dans la Vue en plan ayant les styles spécifiés.

Pour y accéder, choisissez Sélectionner / Par styles.

Styles

Sélectionnez la tabulation du type de style spécifique. Les styles sont les mêmes que les styles dans le dialogue Styles CAD. Sélectionnez un style de la liste que s'affiche. Vous ne pouvez utiliser qu'une tabulation pour la sélection des entités. Lorsque vous choisissez un ou plusieurs styles d'une tabulation et puis vous choisissez une autre tabulation, toute sélection que vous avez fait dans la tabulation précédente est perdue.

Pour sélectionner les types d'entité multiples (par exemple, les points et les lignes) utilisez ce dialogue pour effectuer la sélection, enregistrez groupe de sélection, et puis sélectionnez les autres types d'entité. Enregistrez le nouveau groupe de sélection. Vous pouvez récupérer les deux groupes de sélection au même temps en utilisant Sélectionner / Groupes de sélection / Récupérer.

Utilisez l'option  <Par couche> pour sélectionner les entités avec le champ approprié configuré sur <Par couche>.

Tout

Cliquez sur Tout pour sélectionner chaque style affiché dans la liste Style.

Aucun

Cliquez sur Aucun pour effacer toutes les cases à cocher du style affiché dans la liste Styles.


    Commande Par types

Utilisez cette commande pour sélectionner toutes les entités d'un type spécifié.

Pour y accéder, choisissez Sélectionner / Par types.

Le dialogue Sélectionner par types s'ouvre.


    Dialogue Sélectionner par types

Utilisez ce dialogue pour sélectionner les entités avec un type(s) spécifié.

Pour y accéder, choisissez Sélectionner / Par types.

Types

La liste Types affiche les noms de toutes les différents types d'entité (points, lignes, courbes, arcs, annotations, texte).

Sélectionnez la case à cocher d'un type spécifique pour sélectionner les entités de ce type.

Tour

Cliquez sur Tout pour sélectionner chaque type affiché dans la liste Types.

Aucun

Cliquez sur Aucun pour effacer les cases à cocher de chaque type affiché dans la liste Types.


Commande Par interrogation

Utilisez cette commande pour sélectionner les entités correspondantes aux conditions spécifiées dans les interrogations.

Pour y accéder, choisissez Sélectionner / Par interrogation.

Le dialogue Sélectionner par interrogation s'affiche.


Dialogue Sélectionner par interrogation

 Vue d'ensemble et plus amples informations

Utilisez ce dialogue pour sélectionner une interrogation pour la sélection d'entités.

Pour  y accéder, choisissez Sélectionner / Par interrogation.

Interrogations

La liste des interrogations disponibles comprend toute interrogation personnalisée que vous avez défini dans le projet ainsi que les suivantes:

Nom de l'interrogation	Sélectionne…
Tous les événements	des observations avec des enregistrements d'événement
Points autonomes GPS	des points avec des positions autonomes. Par exemple des points dans un fichier .dat que n'ont pas été traités.
Lignes de base utilisant récepteur …	des lignes de base observées utilisant le même récepteur. Saisissez la valeur dans le dialogue Entrée de paramètres.
Points observées conventionnellement	des points observés utilisant des instruments conventionnels.
Intervalle d'élévation	des points ayant des élévations dans l'intervalle spécifié. Saisissez les valeurs dans le dialogue Entrée de paramètres
Points WGS-84 entrés	des points qui ont été des points WGS-84 entrés.
Points de télémètre laser	des points observés utilisant un télémètre laser.
Notes	tous les enregistrements de note dans le projet. Les notes peuvent être ajoutés sur le terrain lors de l'utilisation du Trimble Survey Controller. Utilisez cette option de sélection des notes pour la création des rapports personnalisés.
Coordonnées nulles	des points avec des coordonnées nulles. Ce peut se produire si vous avez observé à un point de visée arrière sans coordonnées.
Elévations nulles	des points avec des élévations nulles.
Points par description	des points ayant un champ de description le même que vous spécifiez. Saisissez les valeurs dans le dialogue Entrée de paramètres. Vous pouvez utiliser des caractères génériques.
Points observés multiples fois	des points observés plus qu'une fois. Cette interrogation utilise tous les types d'observation. Pour les observations terrestres, l'interrogation utilise des configurations de destination. Pour les occupations RTK, elle utilise des occupations GPS du point. Pour les occupations statique et statique rapide, elle utilise des occupations dans lesquelles le point est le point 'vers' pour les observations. Puis l'interrogation fait le total de tous les choses ci-dessus et sélectionner les points dans lesquels le total est plus d'un.
Points avec des noms différents dans la distance	des points ayant des noms différents que se trouvent dans la distance spécifiée. Saisissez les valeurs dans le dialogue Entrée de paramètres.
Points sans nom	des points qui n'ont pas de nom.
Vecteurs RTK avec covariances (QC2)	des vecteurs RTK ayant des enregistrements QC2. Ces points peuvent être utilisés dans un Ajustement de réseau (si vous avez le module Ajustement de réseau installé)
Lignes de base non traitées	des lignes de base pouvant être mais qui n'ont pas été traitées.

Les interrogations effectuent la sélection des entités selon la définition de l'interrogation et une description de l'interrogation s'affiche sur la barre d'état.

Par exemple, vous pouvez sélectionner des points avec un intervalle d'élévations spécifique.

Lorsque les interrogations requièrent des paramètres, le nom est suivi d'une ellipse. Si vous choisissez une telle interrogation, vous serez demandé de fournir les valeurs des paramètres.

Sélection

Dans le groupe Sélection,  vous pouvez effectuer trois opérations de sélection:

  Nouveau − crée un nouveau groupe de sélection.

  Ajouter au courant − ajoute le groupe de sélection que vous allez créer au groupe de sélection courant.

  Affiner courant − trouve les points qui sont communs au groupe de sélection que vous allez créer, utilisant le groupe de sélection courant.

Par exemple, si vous avez un projet dans lequel vous voulez rechercher toutes les lignes de base non traitées observées utilisant le récepteur GPS avec le numéro de série 1234, effectuez le suivant:

1.	Utilisez l'interrogation pour sélectionner toutes les lignes de base non traitées.

2.	Utilisez l'interrogation à nouveau pour sélectionner l'option Lignes de base utilisant le récepteur.

3.	Dans le Groupe de sélection, sélectionnez l'option Affiner courant.

4.	Saisissez le numéro de série du récepteur (1234) dans le dialogue Sélectionner par interrogation − Paramètres.

Toutes les lignes de base non traitées observées avec le récepteur 1234 seront sélectionnées.


Commande Par caract de pt

Utilisez cette commande pour sélectionner les points ayant des codes de caractéristiques avec attributs correspondant aux noms que vous spécifiez.

Pour  y accéder, choisissez Sélectionner / Par caract de pt.

Le dialogue Sélectionner par caract. de point s'affiche.


Dialogue Sélectionner par caract. de point

Utilisez ce dialogue pour sélectionner les points pour lesquels d'informations de caractéristiques et attributs ont été collectées, Pour afficher les caractéristiques et attributs collectés pour le point, utilisez la fenêtre Propriétés − tabulation Attributs.

Caractéristiques

La liste Caractéristiques affiche toutes les caractéristiques ayant des définitions d'attribut existant dans fichier Bibliothèque de caractéristiques et attributs (*.fcl) ou le fichier Dictionnaire d'attributs (*.ddf) que vous avez utilisé pour configurer le projet pour les attributs.

Sélectionnez les cases à cocher des caractéristiques. Les points avec au moins une des caractéristiques spécifiées sont sélectionnés.

Tout

Pour sélectionner toutes les cases à cocher, cliquez sur Tout.

Aucun

Pour effacer toutes les cases à cocher, cliquez sur Aucun.


Groupes de sélection - Vue d'ensemble

Plus amples informations

 Comment...

Vue d'ensemble

Un groupe de sélection est tout ce que vous avez sélectionné actuellement dans Trimble Geomatics Office. Vous pouvez enregistrer votre groupe de sélection courant à tout moment et le récupérer plus tard. Par exemple, vous pouvez enregistrer un groupe de sélection de tous les points de contrôle dans un projet et rapporter ces points à intervalles réguliers en récupérant le groupe de sélection. Ces groupes de sélection enregistrés conservent l'ordre de leurs entités. Lorsque vous supprimez des entités faisant partie d'un groupe de sélection, le groupe de sélection fonctionne toujours avec les entités restantes.

NOTE: Vous pouvez afficher les groupes de sélection utilisant l'option N'afficher que les données du groupe de sélection dans le dialogue Filtres d'affichage.

Lorsque vous importez un fichier, Trimble Geomatics Office produit automatiquement un groupe de sélection nommé. Un groupe de sélection enregistre l'ordre dans lequel les données sont collectées. C'est important pour les tâches telle que la codification. Pour les fichiers Survey Controller (*.dc), Trimble Geomatics Office fournit le nouveau groupe de sélection le nom de l'identificateur de l'étude que vous voulez importer. Pour d'autres fichiers, tel que les fichiers ASCII, Trimble Geomatics Office fournit le nouveau groupe de sélection le nom du fichier.

NOTE: Si vous importez des données RTK dans lesquelles l'initialisation était perdue pendant les premières 15 minutes, un groupe de sélection de RMS élevé est créé lors de l'importation.


Groupes de sélection - commande Enregistrer

Utilisez cette commande pour enregistrer le groupe de sélection courant

Pour  y accéder:

  Choisissez Sélectionner / Groupes de sélection / Enregistrer.

Le  dialogue Enregistrer s'affiche.

Cette commande n'est pas disponible lorsqu'il n'y a rien sélectionné actuellement.


Groupes de sélection - dialogue Enregistrer

 Vue d'ensemble t plus amples informations

Utilisez ce dialogue pour enregistrer le groupe de sélection courant.

Pour y accéder, choisissez Sélectionner / Groupes de sélection / Enregistrer.

Nom

Saisissez un nom unique pour le groupe de sélection.

OK

Cliquez sur OK pour créer le groupe de sélection.

Le groupe de sélection s'affiche dans le dialogue Récupérer lorsque vous l'ouvrez la prochaine fois. Le nouveau groupe de sélection est ajouté à la liste de sélection utilisé récemment. Pour ouvrir le dialogue Récupérer et pour afficher les groupes de sélection utilisés récemment, utilisez Sélectionner / Groupes de sélection.


Groupes de sélection - Commande Récupérer

Utilisez cette commande pour récupérer un groupe de sélection. Cette commande n'est pas disponible lorsqu'il n'y en a pas aucun groupe de sélection disponible à récupérer.

Pour y accéder:

  Choisissez Sélectionner / Groupes de sélection / Récupérer.

Le dialogue Groupes de sélection − Récupérer s'affiche.


Groupes de sélection - dialogue Récupérer

 Vue d'ensemble et plus amples informations

Utilisez ce dialogue pour récupérer un ou plusieurs groupes de sélection à la fois. Par exemple, si vous avez deux groupes de sélection enregistrés utilisez ce dialogue pour sélectionner toutes les entité dans les deux groupes de sélection.

Pour y accéder, choisissez Sélectionner / Groupes de sélection / Récupérer.

Groupes de sélection

Cette liste contient tous les groupes de sélection enregistrés pour le projet.

Tout

Cliquez sur Tout pour sélectionner chaque groupe de sélection.

Aucun

Cliquez Aucun pour effacer les cases à cocher pour tous les groupes de sélection.


Groupes de sélection - commande Editer

Utilisez cette commande pour éditer les détails des groupes de sélection enregistrés. Cette commande n'est pas disponible lorsqu'il n'y en a pas aucun groupe de sélection à éditer.

Pour y accéder, choisissez Sélectionner / Groupes de sélection / Editer.

Le dialogue Editer s'affiche.


Groupes de sélection - dialogue Editer

 Vue d'ensemble et plus amples informations

Utilisez ce dialogue pour renommer ou supprimer les groupes de sélection enregistrés.

Groupes de sélection

Soulignez le groupe de sélection que vous voulez éditer ou supprimer.

Renommer

Cliquez sur Renommer pour renommer le groupe de sélection souligné.

Supprimer

Cliquez sur Supprimer pour supprimer le groupe de sélection souligné.


Groupes de sélection - dialogue Renommer

 Vue d'ensemble et plus amples informations

Utilisez ce dialogue pour renommer le groupe de sélection souligné.

Pour y accéder:

1.	Choisissez Sélectionner / Groupes de sélection / Editer. Le dialogue Editer s'affiche.

2.	2.	Cliquez sur Renommer. Le dialogue Renommer s'affiche.

Nom

Saisissez un nom unique pour le groupe de sélection.

OK

Cliquez sur OK pour appliquer les changements.

Le groupe de sélection s'affiche dans le dialogue Editer. Le nouveau groupe de sélection s'affiche dans la liste de groupe de sélection utilisée récemment.


Sélection des entités - Vue d'ensemble

Quelques fois vous ne voulez effectuer une action que sur un sous-groupe du projet (par exemple, le traitement des codes de caractéristiques). Vous pouvez sélectionner des entités (points, lignes, arcs, courbes, texte, annotations, observations) dans le projet de deux façons:

  En utilisant la souris.

  En utilisant le menu Sélectionner.

Utilisation de la souris pour sélectionner des entités

Pour sélectionner une entité cliquez-le utilisant la souris. (Pour sélectionner des entités multiples enfoncez la touche Maj en même temps).

Ainsi vous pouvez effectuer les tâches requises sur les entités sélectionnées.

Pour effectuer une sélection inverse, appuyez sur Ctrl. Toute entité sélectionnée actuellement est désélectionnée et vice versa.

Note: Vous pouvez utiliser la souris pour afficher des entités dans la fenêtre Propriétés sans besoin de les résélectionner continûment. Appuyez sur et maintenez le bouton de la souris sur un point pour quelques secondes, puis faites glisser déplacer le pointeur sur les entités que vous voulez afficher.

Utilisation du menu Sélectionner pour sélectionner les entités

Le menu Sélectionner contient beaucoup de méthodes de sélectionner les points. Les options dans le menu Sélectionner dépendent de si vous êtes dans la Vue de levé ou la Vue en plan.


Points en double -  commande Par coordonnées

Utilisez cette commande pour sélectionner les points ayant des coordonnées identiques.

Pour y accéder, choisissez Sélectionner / Points en double / Par coordonnées.


Points en double - commande Par nom

Utilisez cette commande pour sélectionner des points ayant les noms en double.

Pour y accéder:

  Choisissez Sélectionner / Points en double / Par noms.

ASTUCE: Utilisez cette commande avant de fusionner les points en double.


Points en double - commande Par nom et distance

Utilisez cette commande pour sélectionner des points ayant les noms en double et que sont dans la distance spécifiée. Saisissez les valeurs dans le dialogue Entrée paramètres.

Pour y accéder:

Choisissez Sélectionner / Points en double / Par nom et distance.


Dialogue Sélectionner points implantés

Utilisez ce dialogue pour sélectionner les points implantés utilisant une série d'interrogations. Ces outils de sélection peuvent être utilisés avec les rapports supplémentaires et les options d'exportation personnalisées.

Pour y accéder, choisissez Sélectionner / Points implantés.

Interrogations

La liste d'interrogations disponibles comprend toute interrogation personnalisée que vous avez défini dans le projet ainsi que les suivantes:

Nom d'interrogation	Sélectionne …	Saisissez ces informations dans le dialogue Entrée de paramètre …
Tous les points implantés	tous les points implantés sur le terrain.
Points d'arc	les points qui font référence à un arc. Par exemple, les points implantés sur, de, ou vers un arc.	Nom de l'arc
Points MNT	les points qui font référence à un MNT. Par exemple, les points implantés sur, de, ou vers un MNT.	Nom du MNT
Points de ligne	les points qui font référence à une ligne. Par exemple, les points implantés sur, de, ou vers une ligne.	Nom de la ligne
Points du projet	les points qui font référence à un point du projet. Par exemple, les points implantés sur, de, ou vers un point du projet.	Nom de point du projet
Points de route	les points qui font référence à une route. Par exemple, les points implantés sur, de, ou vers une route	Nom de la route
Points d'entrée en terre pour une route	les points d'entrée en terre pour une route	Nom de la route
Points sur une route avec caractéristique	les points sur une route avec une caractéristique	Le nom de la route et la caractéristique
Tous les points avec des erreurs supérieures à	les points avec des erreurs supérieures à l'erreur horizontale et l'erreur verticale spécifiées dans le dialogue Entrée de paramètres.
Lorsqu'une interrogation requiert des paramètres, le nom est suivi d'une ellipse. Si vous choisissez l'une de ces interrogations, vous êtes demandé de fournir des valeurs de paramètres.

Sélection

Dans le groupe Sélection,  vous pouvez effectuer trois opérations de sélection:

  Nouveau − crée un nouveau groupe de sélection.

  Ajouter au courant − ajoute le groupe de sélection que vous allez créer au groupe de sélection courant.

  Affiner courant − trouve les points qui sont communs au groupe de sélection que vous allez créer, utilisant le groupe de sélection courant.

Par exemple, si vous avez un projet dans lequel vous voulez rechercher toutes les lignes de base non traitées observées utilisant le récepteur GPS avec le numéro de série 1234, effectuez le suivant:

1.	Utilisez l'interrogation pour sélectionner toutes les lignes de base non traitées.

2.	Utilisez l'interrogation à nouveau pour sélectionner l'option Lignes de base utilisant le récepteur.

3.	Dans le Groupe de sélection, sélectionnez l'option Affiner courant.

4.	Saisissez le numéro de série du récepteur (1234) dans le dialogue Sélectionner par interrogation − Paramètres.

Toutes les lignes de base non traitées observées avec le récepteur 1234 seront sélectionnées.


Sélectionner points de calibration

Utilisez cette option pour sélectionner les points de calibration qui apparaissent dans la Calibration du site GPS − Liste points dans votre projet Trimble Geomatics Office.

Pour y accéder, choisissez Sélectionner / Points de calibration.

Tous les points de calibration qui apparaissent dans le Calibration du site GPS – Liste points seront sélectionnés dans votre projet.

NOTE: S'il n'y a aucun point dans la Liste points de la calibration, l'option n'est pas disponible.


Support technique

Lorsque vous avez un problème et que vous ne pouvez pas trouver les informations requises dans la documentation du produit, mettez-vous en contact avec votre fournisseur local. Sinon, demandez du support technique en utilisant le site World Wide Web de Trimble (http://www.trimble.com/support/support.htm).


Barres d'outils - Vue d'ensemble

Une barre d'outils consiste en une ou plusieurs outils. Les barres d'outils sont séparées par des lignes verticales dans la fenêtre graphique de Trimble Geomatics Office. Un outil est un raccourci aux commandes de menu utilisées fréquemment.

Pour afficher une description de l'outil, mettez le pointeur de la souris sur l'outil et affichez:

  l'Info-bulle

  la barre d'état

Pour changer l'ordre dans lequel les barres d'outils sont positionnées dans la fenêtre de Trimble Geomatics Office, faites glisser déplacer une barre d'outils utilisant la ligne verticale séparant la barre d'outils des autres éléments dans la fenêtre.

Les barres d'outils suivantes sont disponibles dans la Vue de levé ainsi que la Vue en plan:

  Standard

  Affichages

  Afficher contrôles

Les barres d'outils suivantes ne sont disponibles que dans la Vue de levé:

  Filtres

  Timeline (lorsque vous avez le module Traitement de lignes de base installé)

  Reports Timeline (lorsque vous avez le module Traitement de lignes de base installé)

  Contrôles d'ellipse (lorsque vous avez le module Ajustement de réseau installé)

Les barres d'outils suivantes ne sont disponibles que dans la Vue en plan:

  Couches

  CAD


Barre d'outils CAD

La barre d'outils CAD fournit d'accès rapide aux caractéristiques communes du menu Insérer.

Cet outil …	ouvre ce dialogue …	Utilisez-le pour ajouter ces éléments à la base de données …
	Insérer points	enregistrements de point
	Insérer points par azimut et distance	enregistrements de point
	Insérer lignes	enregistrements de ligne
	Insérer lignes	enregistrements de courbe
	Insérer lignes	enregistrements d'arc
	Insérer texte	enregistrements de texte


Barre d'outils standard

La barre d'outils standard fournit d'accès à certaines des options principales disponibles dans le menu Fichier et le menu Editer.

Outil	Sélectionnez-le pour …
	créer un nouveau projet
	ouvrir un projet existant
	importer les données d'un fichier, un périphérique ou un récepteur GPS
	exporter les données vers un fichier, un périphérique ou un récepteur GPS
	couper la sélection et la mettre dans le presse-papiers
	copier la sélection et la mettre dans le presse-papiers
	coller le contenu du presse-papiers
	annuler la dernière action
	rétablir la dernière action annulée
	ouvrir la fenêtre Propriétés et afficher ou éditer les propriétés de votre sélection courante.


Outil Editer couches

Sélectionnez [image]  pour afficher le dialogue Couches. Utilisez ce dialogue pour gérer les couches.


    Outil Sélection de couche

La couche affichée est la couche courante. Pour changer la couche courante, cliquez sur la flèche et sélectionnez la couche de la liste déroulante.

L'icône d'ampoule indique si la couche est visible ou non. Pour changer la visibilité de toute couche, cliquez sur l'ampoule de la couche appropriée.

L'icône d'ampoule est baissée. Pour rendre la couche visible, cliquez sur l'icône d'ampoule à nouveau. Lorsqu'une ou plusieurs couches ont leur visibilité désactivée, une icône d'ampoule s'affiche dans la barre d'état.

L'icône de cadenas indique si la couche est verrouillée ou non. Si la propriété Verrouillée est activée, vous ne pouvez pas sélectionner des entités dans la couche, bien que vous pouvez les voir dans la fenêtre graphique.

Pour changer la propriété Verrouillée de toute couche, cliquez sur l'icône de cadenas de la couche appropriée. Pour rendre la couche disponible à la sélection, cliquez sur l'icône de cadenas à nouveau. Lorsqu'une ou plusieurs couches ont leur propriété Verrouillée sélectionnée, l'icône de cadenas s'affiche dans la barre d'état.

NOTE: Vous pouvez également changer ces propriétés en utilisant le dialogue Couches.


Barre d'outils Afficher contrôles

La barre d'outils Afficher contrôles fournit des outils de navigation pour aider l'affichage dans la fenêtre graphique principale.

Les boutons suivants fonctionnent dans un mode rémanent. Le mode rémanent signifie que le curseur reste comme l'outil sélectionné jusqu'à ce que vous sélectionnez un outil différent en cliquant sur un autre bouton:

Outil	Sélectionnez-le pour …
	changer au mode de sélection
	déplacer le centre de la surface de zoom
	effectuer un zoom avant d'un facteur de 2
	effectuer un zoom arrière d'un facteur de 2

Les outils suivants ne sont pas rémanentes. Ils réalisent une opération chaque fois que vous les sélectionnez:

Outil	Sélectionnez-le pour …
	effectuer un zoom aux étendues du modèle
	effectuer un zoom à la configuration précédente
	Activer ou désactiver la Barre du projet
	afficher les lignes de grille dans la fenêtre graphique. (Vous pouvez également choisir cette configuration dans le dialogue Options d'affichage.)


Barre d'outils Affichages

Utilisez les outils dans la barre d'outils Affichages pour effectuer le suivant:

Cet outil …	change à cette vue …
	Levé
	Plan


    Barre d'outils Filtres

La barre d'outils Filtres est disponible dans la Vue de levé. Elle contient les Filtres d'affichage et les outils Timeline

Outil	Sélectionnez-le pour …
	spécifier quelles observations sont à afficher, ou comment elles doivent s'afficher dans la Vue de levé.
	ouvrir la fenêtre Timeline
	lier les observations dans la fenêtre graphique et Timeline. Lorsque vous sélectionnez cet outil, le logiciel n'affiche que les lignes de base post traitées dans la vue Timeline courante, et les lignes de base post traitées observées au même temps. Toutes les autres lignes de base post traitées dans la vue de levé sont masquées. Cet outil sert à la sélection des lignes de base indépendantes.


    Barre d'outils Couches

La barre d'outils Couches est disponible dans la Vue en plan. Elle contient l'outil Editer couches et l'outil Sélection de couches. Utilisez cette barre d'outils pour gérer vos couches.


    Commande Traiter codes de caract.

Utilisez cette commande pour traiter les codes de caractéristiques.

Pour y accéder, sélectionnez Outils / Traiter codes de caract.

Le dialogue Traiter codes de caract. s'affiche.


    Dialogue Traiter codes de caract.

 Vue d'ensemble

Utilisez ce dialogue pour traiter les codes de caractéristiques.

Pour y accéder, sélectionnez Outils / Traiter codes de caract.

Bibliothèque d'attributs et caractéristiques

Sélectionnez la bibliothèque dattributs et caractéristiques avec laquelle vous voulez traiter les données.

Parcourir

Cliquez sur Parcourir pour rechercher la bibliothèque d'attributs et caractéristiques.

NOTE: Dans Trimble Geomatics Office toutes les bibliothèques d’attributs et caractéristiques sont du type de fichier .fcl. Lorsque vous sélectionnez une bibliothèque d'attributs et caractéristiques d'un système TRIMMAP ou d'un système SDRmap, Trimble Geomatics Office convertit automatiquement cette bibliothèque à un type de fichier .fcl.

Par exemple, si vous sélectionnez un fichier ‘Fclib??.dat’ (étant le modèle de nom du fichier de la bibliothèque de codes de caractéristiques d'un système TRIMMAP ou d'un système SDRmap), lorsque vous traitez cette bibliothèque Trimble Geomatics Office vous demande si vous voulez convertir la bibliothèque. Le système également demande un nom sous lequel il faut enregistrer la bibliothèque. Le nom par défaut suggéré suit le modèle Fclib??.fcl et il est stocké par défaut dans le dossier système.

Lorsque Trimble Geomatics Office rencontre aucune définition de code qu'il ne peut pas convertir, il vous avertit qu'il faut éditer la bibliothèque résultante. Lorsque toute erreur est rencontrée, le système produit un rapport.

Traiter

Les codes de caractéristiques sont traités dans l'ordre dont ils sont sélectionnés.

Sélectionnez l'option Sélection courante pour traiter les entités sélectionnées actuellement.

Choisissez l'option Groupe de sélection et choisissez un groupe de sélection à traiter de la liste.

L'option Sélection courante n'est pas disponible lorsqu'il n'y a aucun groupe de sélection disponible.

OK

Cliquez sur OK pour traiter les codes de caractéristiques.

La bibliothèque est traitée et les résultats de la codification s'affichent dans la fenêtre de graphiques. Lorsque toute erreur se produit pendant le traitement, vous pouvez afficher un rapport d'erreurs. Pour découvrir où les erreurs se sont produites, vous pouvez afficher le rapport. Le rapport est conservé dans le dossier Rapports du projet pour tout affichage ultérieur.


Trimble RoadLink - Vue d'ensemble

Trimble RoadLink™ est un utilitaire Trimble Geomatics Office. C'est une interface puissante entre les définitions de route tiers et les périphériques de levé Trimble.

Utilisez-le pour:

  saisir les données

  afficher les données de façon graphique

  éditer les données

Ainsi vous pouvez transférer la définition de route aux péripohériques de levé Trimble afin de l'utiliser pendant l'implantation.

RoadLink fournit deux méthodes pour transférer les définitions de route:

  La conversion automatique des fichiers d'axe en plan et de profil en long projet et de profil en travers fournis par un logiciel de conception tiers.

  L'entrée manuelle de la définition de route complète, les axes en plan et les profils en long projet, les sections normales et le dévers.

Un composant intégrant de Trimble Geomatics Office, RoadLink également effectue le calcul des volumes de déblai et remblai des terrassements entre la définition de route et un modèle de la surface défini dans l’utilitaire DTMLink de Trimble.


Trimble DTMLink - Vue d'ensemble

Trimble DTMLink™ est un utilitaire de Trimble Geomatics Office. Cet utilitaire produit des surfaces de terrain numériques. A l'aide de ce logiciel vous pouvez importer des surfaces définies d’un logiciel tiers, créer des nouvelles surfaces et éditer les surfaces créées précédemment.  Vous pouvez également transférer ces surfaces à un périphérique de levé Trimble pour l'implantation sur le terrain.

Le logiciel a la capacité d'inclure les limites et les lignes de changement de pente dans le modèle de la surface des courbes de niveau, et permet l'utilisation des caractéristiques d'édition de triangles avancées afin de fournir plus contrôle de la formation du modèle.

Un Modèle numérique du terrain triangulé ou à grille est créé du modèle de la surface des courbes de niveau pour l'exportation vers un périphérique ou un fichier de données.

Le modèle créé peut également être utilisé par Trimble RoadLink pour les calculs de terrassements.

Trimble DTMLink vous permet de créé les rapports de volume.

Les applications typiques des calculs de volume sont:

  les calculs de réserves

  les volumes des couches de matériaux, tels que les couches de charbon

  Les volumes des vides, tels qu'un lac derrière un barrage de retenu proposé

DTMLink utilise la méthode du calcul de volumes des prismes triangulaires individuels. Ils sont définis par les plans triangulaires formant le modèle de la surface de courbes de niveau en utilisant les surfaces de base et les hauteurs moyennes des prismes.

DTMLink vous permet de créer les surfaces de comparaison. Une surface de comparaison est un modèle de courbes de niveau de la différence entre deux surfaces. Vous pouvez utiliser cette surface conjointement avec le type de calcul de volume 'Au-dessus d'une élévation' pour calculer les volumes de déblai et de remblai entre les deux surfaces.


    Commande Démarrer

Cette commande démarre l’utilitaire DTMLink de Trimble.

Pour y accéder:

   Sélectionnez Outils / DTMLink / Démarrer.

Utilisez cette commande pour démarrer DTMLink afin d'importer un fichier AutoCAD DXF à faces 3D sans créer une nouvelle surface ou ouvrir une surface existante.

Pour l'effectuer:

1.	Sélectionnez Outils / DTMLink / Démarrer.

2.	Sélectionnez Fichier / Importer.

Pour plus amples informations, référez-vous à l'Aide de DTMLink.


    Commande Nouvelle Surface

Utilisez cette commande pour créer un nouveau modèle de la surface.

Pour y accéder, sélectionnez Outils / DTMLink / Nouvelle surface.

Le dialogue Nouvelle surface s'affiche.


    Dialogue Nouvelle surface

Utilisez ce dialogue pour créer un nouveau modèle de la surface.

Pour y accéder, sélectionnez Outils / DTMLink / Nouvelle surface.

Nom

Saisissez un nom approprié pour la nouvelle surface.

Points

Les options Points vous permettent de sélectionner quels points de la base de données du projet il faut utiliser pour la formation de la surface.

  Lorsqu'il n'y a aucune sélection courante de base de données, l'option Sélection courante n'est pas disponible.

  Lorsqu'il y a une sélection courante, l'option Sélection courante utilise les points dans la sélection courante pour la formation de la surface.

L'option BD entière utilise tous les points de base de données pour la formation de la surface.

L'option Groupe de sélection vous permet de choisir un groupe de sélection existant, contenant les points à utiliser pour la formation de la surface.

Utiliser lignes de c/p

Les options Utiliser lignes de c/p vous permettent de sélectionner les lignes de la base de données à utiliser comme des lignes de changement de pente lors de la formation de la surface.

Les lignes de changement de pente, en ce qui concerne la formation du modèle de la surface, sont des lignes à considérer d'avoir une pente constante et donc elles assistent à contrôler la formation du modèle.

Ces options ne sont disponibles que si l'option Utiliser lignes de c/p est sélectionnée. Il faut sélectionner Utiliser lignes de c/p afin de passer toute information de lignes de changement de pente à l'opération de formation de la surface.

L'option Groupe de sélection vous permet d'utiliser les lignes d'un groupe de sélection existant comme des lignes de changement de pente pendant la formation de la surface. L'option Couche vous permet d'utiliser les lignes d'une couche existante comme des lignes de changement de pente pendant la formation de la surface. Pour tous les deux cas, la liste modifiable associée fournit des listes de groupes de sélection ou de couches disponibles.

Utiliser Exclure limites

Les options Utiliser exclure limites fonctionnement de la même façon que les options Utiliser lignes de c/p discutées ci-dessus. Ces options vous permettent de spécifier que les lignes existantes dans un groupe de sélection ou une couche sont à utiliser pour la définition d'un ou plusieurs Exclure limites pendant la formation du modèle de la surface.

Un Exclure limite est un groupe de lignes définissant une figure fermée dans laquelle aucun triangle de surface ne sera formé. Si les lignes spécifiées pour la définition d'exclure limite ne forment pas des figures complètement fermées, l'opération de formation du modèle de la surface ajoute les lignes de fermeture par défaut.

Utiliser Inclure limites

Les options Utiliser Inclure limites fonctionnent dans la même façon que les options Utiliser Exclure limites décrites ci-dessus. Ces options vous permettent de spécifier que les lignes existantes dans un groupe de sélection ou une couche sont à utiliser pour la définition d'un ou plusieurs Inclure limites pendant la formation du modèle de la surface.

Un Inclure limite est un groupe de lignes définissant une figure fermée dans laquelle aucun triangle de surface ne sera formé. Si les lignes spécifiées pour la définition d'inclure limite ne forment pas des figures complètement fermées, l'opération de formation du modèle de la surface ajoute les lignes de fermeture par défaut.

NOTE: Les lignes utilisées pour des Lignes de c/p, Exclure limites et Inclure limites peuvent comprendre des arcs et des courbes. En ce cas, DTMLink interprète les arcs ou les courbes comme une série de segments de lignes droites courtes pour l'utilisation dans la formation du triangle. Le logiciel assure qu'aucun segment de ligne ne s'écarte de la ligne d'arc ou de courbe de plus de deux centimètres.

Pour plus amples informations à propos des limites et des lignes de changement de pente, référez-vous à l'Aide en ligne de Trimble DTMLink.


    Commande Surface existante

Utilisez cette commande pour éditer un modèle de surface existant.

Pour y accéder, sélectionnez Outils / DTMLink / Surface existante.

Le dialogue Surface existante s'affiche.


    Dialogue Surface existante

Utilisez ce dialogue pour éditer un modèle de surface existant. Ce dialogue vous permet de spécifier des points et des lignes dans le projet que vous voulez ajouter à une surface existante.

Pour y accéder, sélectionnez Outils / DTMLink / Surface existante.

NOTE: Lorsque vous spécifiez des points et des lignes dans ce dialogue qui sont déjà dans la surface existante, le système en reconnaît et ne les ajoute pas à la surface une deuxième fois.

Nom

Ce champ présente une liste des surfaces existantes dans le projet. Sélectionnez la surface que vous voulez charger et éditer.

Ajouter points

Sélectionnez les points de la base de données du projet qu'il faut ajouter à la surface existante.

Lorsqu'il n'y a aucune sélection de base de données courante, l'option Sélection courante n'est pas disponible.

Lorsqu'il y a une sélection courante, l'option Sélection courante ajoute les points dans la sélection courante à la surface existante.

L'option BD entière passe tous les points actuellement dans la base de données au modèle de surface. Cette option vous permet d'ajouter automatiquement au modèle tout nouveau point dans la base de données.

L'option Groupe de sélection vous permet de choisir un groupe de sélection existant contenant les points que vous voulez ajouter à la surface existante.

Ajouter lignes de c/p

Les options Ajouter lignes de c/p vous permettent des lignes de la base de données qu'il faut ajouter aux lignes de changement de pente déjà dans la surface existante . Ces options ne sont disponible que si l'option Ajouter lignes de c/p est sélectionnée.

L'option Ajouter lignes de c/p doit être sélectionné pour toute information de lignes de changement de pente à jouter à la surface existante. Les lignes de changement de pente, en ce qui concerne la formation du modèle de la surface, sont des lignes à considérer d'avoir une pente constante et donc elles assistent à contrôler la formation du modèle.

L'option Groupe de sélection vous permet d'ajouter des lignes d'un groupe de sélection existant à la surface existante sous la forme de lignes de changement de pente.

L'option Couche vous permet d'ajouter les lignes d'une couche existante à la surface existante sous la forme de lignes de changement de pente.

Dans tous les deux cas, les listes affichent les groupes de sélection ou les couches disponibles.

Ajouter exclure limites

Les options Ajouter exclure limites fonctionnent de la même façon que les options Ajouter lignes de c/p discutées ci-dessus.

Ces options vous permettent de spécifier que les lignes existantes dans un groupe de sélection ou une couche sont à ajouter comme des lignes d'exclure limite à la définition d'exclure limite dans la surface existante.

Un exclure limite est un groupe de lignes définissant une figure fermé dans laquelle aucun triangle de surface ne sera formé.

Si les lignes spécifiées pour la définition d'exclure limite ne forment pas des figures complètement fermées, l'opération de formation du modèle de la surface ajoute des lignes de fermeture par défaut.

Ajouter inclure limites

Les options Ajouter inclure limites fonctionnent de la même façon que les options Ajouter exclure limites discutées ci-dessus.

Ces options vous permettent de spécifier que les lignes existantes dans un groupe de sélection ou une couche sont à ajouter comme des lignes d'inclure limite à la définition d'inclure limite dans la surface existante.

Un inclure limite est un groupe de lignes définissant une figure fermé dans laquelle aucun triangle de surface ne sera formé. Aucun triangle n'est formé au dehors de tout inclure limite défini.

Si les lignes spécifiées pour la définition d'inclure limite ne forment pas des figures complètement fermées, l'opération de formation du modèle de la surface ajoute des lignes de fermeture par défaut.

NOTE: Les lignes utilisées pour des Lignes de c/p, Exclure limites et Inclure limites peuvent comprendre des arcs et des courbes. En ce cas, DTMLink interprète les arcs ou les courbes comme une série de segments de lignes droites courtes pour l'utilisation dans la formation du triangle. Le logiciel assure qu'aucun segment de ligne ne s'écarte de la ligne d'arc ou de courbe de plus de deux centimètres.

Pour plus amples informations à propos des limites et des lignes de changement de pente, référez-vous à l'Aide en ligne de Trimble DTMLink.


    Commande Démarrer

Cette commande démarre l’utilitaire RoadLink de Trimble. Utilisez ce logiciel pour importer ou saisir manuellement les définitions de route pour le transfert aux périphériques de levé Trimble pour l'implantation.

Pour y accéder:

  Sélectionnez Outils / RoadLink / Démarrer.


    Commande Exporter

Utilisez cette commande pour exporter les points d'un groupe de sélection, de la base de données entière ou de la sélection courante à RoadLink.

Pour y accéder:

  Sélectionnez Outils / RoadLink / Exporter.

Le dialogue Exporter s'affiche.


    Dialogue Exporter

Utilisez ce dialogue pour exporter les points d'un groupe de sélection, de la base de données entière ou de la sélection courante à RoadLink. L'affichage des points de la base de donnés existants dans RoadLink peut assister lors de la saisie ou la vérification d'une définition de route.

Pour y accéder:

  Sélectionnez Outils / RoadLink / Exporter.

Points

Sélectionnez le groupe de points.

Trois options sont disponibles:

Sélection courante	Exporte les points sélectionnés
BD entière	Exporte tous les points dans la base de données
Groupe de sélection	Choisissez le groupe de sélection de la liste.

Pour plus amples informations, référez-vous à l'Aide en ligne de RoadLink.


Marques de commerce

Le logo Globe et Triangle, Trimble, DTMLink, GPSurvey, RoadLink, Trimble Geomatics Office, Trimble Survey Controller, Trimble Survey Office, Trimble Total Control, TRIMMAP, TRIMNET, TSC1, et TSCe sont des marques de commerce de Trimble Navigation Limited

Le logo de sextant avec Trimble, Geodimeter et GPS Pathfinder sont des marques déposées de Trimble Navigation Limited enregistrées au Bureau des marques et des Brevets des Etats-Unis (the U.S. Patent and Trademark Office).

Toutes les autres marques sont des marques de commerce de leurs détenteurs respectifs

Standard Template Library, © 1994 Hewlett-Packard Company, Standard Template Library, © 1996,97 Silicon Graphics Computer Systems, Inc., Standard Template Library, © 1997 Moscow Center for SPARC Technology.


Commande Compresser et réparer le projet

Utilisez cette commande pour compresser et réparer la base de données de votre projet.

Pour y accéder:

  Sélectionnez Utilitaires / Compresser et réparer le projet.

Le dialogue Compresser et réparer le projet s'ouvre.


Dialogue Compresser et réparer le projet

Utilisez ce dialogue pour compresser et réparer la base de données de votre projet.

Compresser

La compression gagne d'espace disque et aide une meilleure exécution du logiciel. Il est recommandé de compresser la base de données de votre projet régulièrement.

Réparer

Il est possible d'endommager votre base de données si, par exemple, il y a une coupure de courant pendant l'exécution de Trimble Geomatics Office.

Lorsque vous éditez la base de données et Trimble Geomatics Office s'arrête de façon imprévue, il se peut que votre dernière modification aux données soit perdue après la réparation de la base de données. Vous devrez ouvrir le projet à nouveau, retourner à votre dernière édition, et vérifier que toutes vos modifications ont été enregistrées.

Pour utiliser le dialogue Compresser et réparer le projet:

1.	Sélectionnez Utilitaires / Compresser et réparer le projet.

2.	Sélectionnez le projet que vous voulez compresser et réparer.

Un message d'erreur s'affiche s'il n'est pas possible de compresser ou réparer la base de données de votre projet.

ASTUCE: Si votre disque dur est fragmenté sérieusement, défragmentez-le avant la compression de la base de données.


Utilitaires de Trimble Geomatics Office

Les utilitaires Trimble Geomatics Office sont disponibles avec l'option Installation standard Trimble Geomatics Office:

L'utilitaire Trimble Coordinate System Manager™

L'utilitaire Trimble Convert to RINEX™

l'utilitaire Trimble Data Transfer

L'utilitaire Trimble DC File Editor™

L'utilitaire Trimble Feature and Attribute Editor™

L'utilitaire Trimble Line Type Editor™

L’utilitaire Planning

L'utilitaire Trimble Symbol Editor™

L'utilitaire Trimble A1 Viewer

NOTE: L'utilitaire Trimble Grid Factory™ est disponible dans l'option Installation personnalisée.

Pour plus amples informations, référez-vous au Manuel de l'utilisateur de Trimble Geomatics Office.

Ajout des utilitaires

Pour ajouter des utilitaires supplémentaires au menu Utilitaires:

1.	Utilisez l'Explorateur de Windows pour trouver ...\Program Files\Trimble\Trimble Geomatics Office\Utility.

2.	Cliquez avec le bouton droit de la souris dans la partie vide de la fenêtre gauche de l'Explorateur.

3.	Sélectionnez l'option Nouveau / Raccourci.

Ainsi l'assistant Créer un raccourci s'ouvre.

4.	Pour configurer le raccourci, saisissez le chemin de l'utilitaire que vous voulez ajouter au menu Utilitaires.

5.	Cliquez sur Suivant et complétez le champ ‘Sélectionner un nom pour le raccourci’ pour spécifier le nom de l'utilitaire que vous voulez afficher dans le menu Utilitaires.

La prochaine fois que vous exécutez Trimble Geomatics Office, l'utilitaire pour le raccourci sélectionné s'affichera dans le menu Utilitaires.


Editeur de symboles

L'utilitaire Trimble Symbol Editor™ vous permet d'éditer les bibliothèques de symboles de point (*.sym) utilisées dans Trimble Geomatics Office.

Pour y accéder, effectuez l'une des choses suivantes:

  Sélectionnez Utilitaires / Editeur de symboles lorsqu'il n'y a aucun projet ouvert.

  Sélectionnez-le du menu Démarrer de Microsoft Windows à tout moment.

Pour créer ou modifier des symboles de point existants, utilisez cet utilitaire. Une fois que vous enregistrez le fichier édité comme le fichier système, les symboles que vous créez sont les symboles disponibles lorsque vous ajoutez ou modifiez les styles de point dans Trimble Geomatics Office ou dans l'utilitaire Trimble Feature and Attribute Editor.

Un style de point

  est une définition expliquant comment un point est dessiné dans Trimble Geomatics Office

  comprend un symbole créé dans l'utilitaire Symbol Editor.

La création d'un symbole dans l'utilitaire Symbol Editor ne rendra pas ce symbole disponible automatiquement comme un style de point dans Trimble Geomatics Office. Un point est assigné un style de point, pas un symbole. Un style de point doit être défini utilisant un symbole créé récemment.

Le Symbol Editor s'ouvre automatiquement avec la bibliothèque de symboles courante utilisée dans Trimble Geomatics Office. Les noms des symboles dans la bibliothèque ouverte sont présentés dans une liste sur l'écran. Utilisez cette liste pour sélectionner le symbole que vous voulez éditer.

Vous pouvez exécutez l'éditeur indépendamment de Trimble Geomatics Office. Sélectionnez l'icône de raccourci de l'utilitaire Symbol Editor du dossier Trimble Geomatics Office.

NOTE: Lorsque vous exécutez l'éditeur indépendamment de Trimble Geomatics Office, toute modification effectuée au type de symbole courant n'est pas reflétée dans des projets ouverts actuellement. Pour que ces modifications soient appliquées, fermez et ouvrez le projet à nouveau. Les modifications enregistrées deviennent disponibles dans la bibliothèque de symboles.

Pour plus amples informations, référez-vous à l'aide de l'utilitaire Trimble Symbol Editor.


Editeur de types de ligne

L'utilitaire Trimble Symbol Editor™ vous permet d'éditer les bibliothèques de types de ligne (*.ltp) utilisées dans Trimble Geomatics Office.

Pour y accéder, effectuez l'une des choses suivantes:

  Sélectionnez Utilitaires / Editeur de types de ligne lorsqu'il n'y a aucun projet ouvert.

 Sélectionnez-le du menu Démarrer de Microsoft Windows à tout moment.

Pour créer ou modifier des types de ligne existants, utilisez cet utilitaire. Une fois que vous enregistrez le fichier édité comme le fichier système, les types de ligne que vous créez sont les types de ligne disponibles lorsque vous ajoutez ou modifiez les styles de ligne dans Trimble Geomatics Office ou dans l'utilitaire Trimble Feature and Attribute Editor

Un style de ligne

  est une définition expliquant comment une ligne est dessinée dans Trimble Geomatics Office

  comprend un type de ligne créé dans l'utilitaire Line Type Editor.

La création d'un type de ligne dans l'utilitaire Line Type Editor ne le rendra pas disponible automatiquement comme un type de ligne dans Trimble Geomatics Office. Une ligne est assignée un style de ligne. Un style de ligne doit être défini utilisant le type de ligne créé récemment.

Le Symbol Editor s'ouvre automatiquement avec la bibliothèque de types de ligne courante utilisée dans Trimble Geomatics Office. Les noms des types de ligne dans la bibliothèque ouverte sont présentés dans une liste sur l'écran. Utilisez cette liste pour éditer les types de ligne.

Vous pouvez exécutez l'éditeur indépendamment de Trimble Geomatics Office. Sélectionnez l'icône de raccourci de l'utilitaire Line Type Editor du dossier Trimble Geomatics Office.

NOTE: Lorsque vous exécutez l'éditeur indépendamment de Trimble Geomatics Office, toute modification effectuée à bibliothèque de types de ligne du système n'est pas reflétée dans votre projet courant. Pour que ces modifications soient appliquées, fermez et ouvrez le projet à nouveau après avoir enregistré les modification à la bibliothèque de types de ligne. Les modifications enregistrées deviennent disponibles dans la bibliothèque de types de ligne.

Pour plus amples informations, référez-vous à l'aide de l'utilitaire Trimble Line Type Editor.


Grid Factory

Pour installer l'utilitaire Trimble Grid Factory™, utilisez l'option Installation personnalisée.

Pour y accéder une fois que c'est installé, effectuez l'une des choses suivantes:

  Sélectionnez Utilitaires / Grid Factory lorsqu'aucun projet n'est ouvert.

 Sélectionnez-le du menu Démarrer de Microsoft Windows à tout moment.

L'utilitaire Trimble Grid Factory vous permet de travailler avec des fichiers Grille de datum (*.dgf) et les fichiers Grille de géoïde (*.ggf).

Utilisez cet utilitaire pour créer des sous-grilles à partir des grilles existantes ou pour combiner des grilles multiples existantes dans une seule grille. Vous pouvez faire une sous-grille des fichiers de grille de datum et des fichiers de grille de géoïde utilisant  Trimble Geomatics Office.

Lorsque vous utilisez l'utilitaire Grid Factory pour créer des fichiers de grille de géoïde, assurez-vous que le nom saisi pour le nouveau fichier soit le même que le nom du modèle de géoïde créé dans l'utilitaire Trimble Coordinate System Manager™.

Les grilles sont grandes si la surface géographique couverte est grande.

Par exemple, le grille de géoïde OSU91 consiste en environ 2 Mb. En général les carnets de terrain et les récepteurs n'ont pas assez de mémoire disponible pour contenir cette quantité de données. La grille OSU91 couvre le globe entier mais la plupart de topographes n'ont pas besoin d'une telle couverture. En général, les topographes n'ont besoin que de quelques dizaines ou centaines de kilomètres carrés. Une grille couvrant une surface plus petite est d'une taille de quelques centaines d'octets seulement.

Avec l'utilitaire Grid Factory vous pouvez créer des fichiers de sous-grille du modèle de géoïde couvrant une surface locale. Ces fichiers sont assez petits pour être téléchargés vers un périphérique exécutant  Trimble Survey Controller (version 6 ou ultérieure) pour une utilisation dans la région locale.

Beaucoup de produits Trimble sont fournis avec des fichiers de grille pour la conversion de datum et le modelage de géoïde que vous pouvez utiliser dans l'utilitaire Grid Factory. L'utilitaire Grid Factory également importe les fichiers de grilles étranges et les convertit au format de grille Trimble.

Pour plus amples informations, référez-vous à l'aide de l'utilitaire Trimble Grid Factory.


Editeur de fichiers DC

L'utilitaire Trimble DC File Editor™ vous permet d'éditer les fichiers Survey Controller (*.dc).

Pour y accéder effectuez l'une des choses suivantes:

  Sélectionnez Utilitaires / DC File Editor lorsqu'aucun projet n'est ouvert.

 Sélectionnez-le du menu Démarrer de Microsoft Windows à tout moment.

NOTE: Trimble vous recommande de compléter toute édition dans  Trimble Geomatics Office. Ainsi vous êtes assuré que vous avez toujours le fichier de carnet de terrain d'origine auquel vous pouvez retourner lors de tout problème pendant l'édition dans  Trimble Geomatics Office.

Utilisez l'utilitaire Trimble DC File Editor pour l'édition des fichiers des périphériques suivants:

 Trimble

 Sokkia SDR (jusqu'à la version 4-03)

  Tout fichier de périphérique pouvant être converti au format SDR.

Pour éditer le contenu d'un fichier .dc, utilisez l'utilitaire Trimble DC File Editor. Par exemple, vous pouvez modifier les données erronées collectées sur le terrain ou ajouter des informations que ne sont pas disponibles sur le terrain.

NOTE: Vous ne pouvez pas éditer les attributs ou modifier les informations dans la bibliothèque d'attributs et caractéristiques.

Pour plus amples informations, référez-vous à l'aide de l'utilitaire Trimble DC File Editor.


Editeur de caractéristiques et attributs

Utilisez l'utilitaire Trimble Feature and Attribute Editor™ pour gérer les fichiers de bibliothèque d'attributs et caractéristiques (*.fcl) pour le traitement des codes de caractéristiques et la collecte de données d'attribut SIG.

Pour y accéder, effectuez l'une des choses suivantes:

  Sélectionnez Utilitaires / Feature and Attribute Editor lorsqu'aucun projet n'est ouvert.

 Sélectionnez-le du menu Démarrer de Microsoft Windows à tout moment.

Utilisez l'utilitaire Trimble Feature and Attribute Editor pour afficher ou éditer des fichiers de bibliothèques d'attributs et caractéristiques. Une bibliothèque d'attributs et caractéristiques sert à:

  le traitement de codes de caractéristiques

  la collecte des attributs

Un .fcl est un fichier texte contenant des définitions de caractéristiques, les styles CAD, et les codes de contrôle. Une définition de caractéristique peut comprendre:

  un code pour la caractéristique

  les procédures CAD à effectuer sur les points qui sont associés à cette caractéristique

  les attributs requis pour la caractéristique

Vous pouvez créer des bibliothèques d'attributs et caractéristiques. Vous pouvez également utiliser la bibliothèque d'attributs et caractéristiques par défaut de Trimble Geomatics Office sous sa forme courante, ou modifiez la bibliothèque pour s'adapter mieux à vos exigences individuelles.

Pour plus amples informations, référez-vous à l'aide de l'utilitaire Trimble Feature and Attribute Editor.


Gestionnaire du système de coordonnées

L'utilitaire Trimble Coordinate System Manager™ vous fournit d'accès à la base de données de votre système de coordonnées (Current.csd).

Pour y accéder, effectuez l'une des choses suivantes:

  Sélectionnez Utilitaires / Gestionnaire du système de coordonnées lorsqu'aucun projet n'est ouvert.

 Sélectionnez-le du menu Démarrer de Microsoft Windows à tout moment

A l'aide de l'utilitaire Trimble Coordinate System Manager vous pouvez spécifier des nouveaux:

  paramètres d'ellipsoïde

  transformations de datum

  systèmes de coordonnées

  définitions du site

  modèles de géoïde

NOTE: La plupart des utilisateurs n'a pas besoin d'utiliser l'utilitaire Trimble Coordinate System Manager parce que Trimble a fourni un groupe répandu de définitions de système de cordonnées prédéfinies. N'utilisez l'utilitaire Trimble Coordinate System Manager que si vous voulez ajouter une définition du système de coordonnées supplémentaire.

 Trimble Geomatics Office utilise la base de données du système de coordonnées pour spécifier quels systèmes, quels modèles de géoïde et quels sites sont disponibles pour la sélection dans un projet.

NOTE: Les Sites peuvent être créés dans  Trimble Geomatics Office ou dans l'utilitaire Coordinate System Manager, et ils sont stockés dans la base de données géodésique.

Pour plus amples informations, référez-vous à l'aide de l'utilitaire Trimble Coordinate System Manager ou le Manuel de l'utilisateur de Trimble Geomatics Office.


Planning

L'utilitaire Planning vous aide à planifier et programmer les sessions GPS sur le terrain. Utilisez l'utilitaire avant la planification des sessions GPS pour déterminer les conditions appropriées à la topographie GPS.

Pour y accéder à l'utilitaire Planning, effectuez l'une des choses suivantes:

  Sélectionnez Utilitaires / QuickPlan lorsqu'aucun projet n'est ouvert.

 Sélectionnez-le du menu Démarrer de Microsoft Windows à tout moment.

En général la planification de mission est la première étape dans la gestion d'un projet de levé. L'objet est de spécifier tous les aspects significatifs du projet afin qu'il peut être effectué de manière efficace sous toute condition prévisible. Si vous effectuez des levés cinématiques en temps réel (RTK):

  utilisez l'utilitaire Planning pour vérifier qu'au moins quatre satellites seront disponibles pour obtenir une initialisation de point connu et que cinq satellites seront disponibles pour une initialisation en vol (on-the fly).

  étudiez les valeurs de Dilution de précision de position (PDOP) pour confirmer que les conditions géométriques des satellites sont appropriées à un positionnement en temps réel précis.

NOTE: L'utilitaire Planning n'est pas lié directement à un projet Trimble Geomatics Office et ne stocke des descriptions de session. Vous pouvez spécifier les sessions et les utiliser pendant la même exécution du programme.

L'utilitaire Planning peut vous aider à:

  visualiser et prédire la disponibilité des satellites au moyen des tables et des représentations graphiques

  expérimenter avec la sélection des satellites, les almanachs, les fuseaux horaires, les obstructions à la visibilité du site, et les masques d'élévation

  déterminer les meilleures périodes d'observation pour une session, en tenant compte des contraintes nécessaire sur le PDOP et les heures pendant lesquelles l'équipe de terrain peut travailler.

Pour plus amples informations référez-vous à l'aide de l'utilitaire Trimble Planning.


Conversion à Rinex

Le format Receiver INdependent EXchange (RINEX) est une représentation ASCII des données GPS collectées par des récepteurs. L'utilitaire Convert to RINEX effectue la conversion d'un fichier de format .dat binaire collecté en utilisant des récepteurs de levé Trimble aux fichiers de format ASCII RINEX. Vous pouvez exécuter l'utilitaire indépendamment de Trimble Geomatics Office.

Pour accéder à l'utilitaire Convert to RINEX, effectuez l'une des choses suivantes:

  Sélectionnez Utilitaires / Convert to RINEX lorsqu'aucun projet n'est ouvert.

 Sélectionnez-le du menu Démarrer de Microsoft Windows à tout moment.

Utilisez l'utilitaire pour:

  sélectionner les fichiers pour la conversion

  configurer les paramètres de conversion

  configurer les informations du fichier de sortie pour les antennes et les en-têtes

  convertir les fichiers sélectionnés au format RINEX

L'utilitaire Convert to RINEX peut créer trois types de fichier RINEX:

  Observation (des observation GPS brutes avec des informations d'antenne de station)

  Navigation (des orbites GPS)

  Météorologique (la pression, la température, et l'humidité relative)

L'utilitaire crée les fichiers RINEX d'observation et de navigation par défaut. Vous pouvez également choisir de créer un fichier RINEX météorologique.

Pour plus amples information référez-vous à l'aide en ligne de l'utilitaire Trimble Convert to RINEX .


Afficheur A1

L'utilitaire A1 Viewer vous permet d'afficher le contenu d'un fichier GPSurvey Project Archive (*.A1), et puis extraire (enregistrer) les fichiers sélectionnés vers un fichier de destination. L'utilitaire A1 Viewer fonctionne indépendamment du logiciel GPSurvey, donc il n'est pas nécessaire de charger ce logiciel sur votre ordinateur. Après l'enregistrement des fichiers du projet GPSurvey, vous pouvez utiliser les fonctions d'importation de Trimble Geomatics Office pour importer les données dans le projet courant.

Pour accéder à l'utilitaire A1 Viewer, effectuez l'une des choses suivantes:

 Sélectionnez Utilitaires / A1 Viewer lorsqu'aucun projet n'est ouvert.

 Sélectionnez-le du menu Démarrer de Microsoft Windows à tout moment.

Importateur de coordonnées GPSurvey

L'option Importateur de coordonnées GPSurvey Coordinates Importer(Coordonnées GPSurvey [curr_net.dat] ) vous permet d'importer des coordonnées ajustées d'un fichier Curr_net.dat file. Pour accéder à ce logiciel:

1.	Sélectionnez Fichier / Importer. Le dialogue Importer s'affiche.

2.	Dans la tabulation Levé, sélectionnez Coordonnées GPSurvey (Curr_net.dat) et cliquez sur OK. Le dialogue Ouvrir s'affiche.

NOTE: Le fichier doit avoit le nom curr_net.dat afin d'être importé avec succès

3.	Naviguez au fichier et cliquez sur OK.

Maintenant vous vous importez des coordonnées ajustées d'un fichier Curr_net.dat.

Fichiers pris en charge par l'utilitaire A1 Viewer

Seuls des fichiers archivés par  GPSurvey sont disponibles à l'affichage:

  fichiers .ssf/.ssk – ces fichiers contiennent les résultats de solutions de lignes de base du processeur GPSurvey WAVE. Ils sont importées dans un projet Trimble Geomatics Office sous le format d'importation de fichiers SSF/SSK (*.ssf,*.ssf), qui est disponible dans la tabulation Levé du dialogue Importation.

  fichiers de données GPS – ces fichiers contiennent les données GPS brutes collectées sur le terrain. Les deux types de fichiers—*.dat (fichiers de données brutes GPS d'un récepteur Trimble) et *.rnx (fichiers de données RINEX brutes pour une utilisation dans le processeur WAVE)—sont disponibles lors de la sélection de ce filtre. A nouveau, les deux types de fichier (.dat et .rnx) sont importés dans un projet sous le format d'importation de fichiers Données GPS (*.dat).

NOTE: Pour afficher les fichiers *.rnx disponibles dans le dialogue Ouvrir un fichier du format d'importation de fichiers Données GPS, dans le champ Fichiers de type. sélectionnez Tous les fichiers *.*.

  fichiers de Coordonnées ajustées – ces fichiers, Curr_net.dat et Coords.log, contiennent les coordonnées ajustées d'un projet GPSurvey archivé. Le fichier Curr_net.dat file est utilisé pour importer vos coordonnées dans  Trimble Geomatics Office. Le fichier Coords.log est sous le format ASCII et n'est disponible à l'affichage qu'avec un éditeur texte. Les points dans le projet GPSurvey peuvent être sous le format de coordonnées de grille ou géographiques, et dans tout système de coordonnées. Pour que l'importation de coordonnées puisse fonctionner correctement, le système de coordonnées des propriétés du système de coordonnées du Trimble Geomatics Office doit correspondre aux propriétés du système de coordonnées du projet GPSurvey exactement.

Le fichier (Curr_net.dat) est importé dans un projet sous le format d'importation de Coordonnées GPSurvey (Curr_net.dat). Lorsque le fichier (Curr_net.dat) est lu, le software identifie quelles coordonnées étaient fixes pendant l'ajustement GPSurvey. On doit avoir au moins un composant de coordonnées (NE, e, h) fixe dans le fichier avant que tous les composants de coordonnées du même type sont importés. Par exemple, un élévation fixe est requise avant que les élévations sont importées. Tous les composants importés sont assignés une qualité de contrôle.

  fichiers .ggf – ces fichiers contiennent les données de géoïde des fichiers de Grille de géoïde locale (*.ggf) créés dans, ou importés dans un projet GPSurvey. Ils sont crées utilisant la fonction de Residual Geoid Modeling (Modelage de géoïde résiduel) dans le logiciel TRIMNet Plus. Ce filtre n'affichera pas les fichiers Global .ggf (c'est à dire, WW15MGH, GEOID99) disponibles dans GPSurvey. Pour utiliser les fichier .ggf dans  Trimble Geomatics Office, enregistrez le fichier .ggf au dossier Program Files\Common Files\Trimble\GeoData, puis utilisez l'utilitaire Coordinate System Manager pour ajouter le géoïde à la base de données du système de coordonnées. Pour plus amples informations concernant l'utilisation de cet utilitaire, référez-vous à l'aide de Coordinate System Manager.

Utilisation du mode Une hauteur dans TRIMNet Plus

Lorsque vous importez des coordonnées GPSurvey d'un projet utilisant le mode Une hauteur (One Height Mode) dans TRIMNet Plus, le champ Hauteur dans le projet Trimble Geomatics Office contient la valeur d'élévation. C'est à cause du fait que pendant un en mode Une hauteur, TRIMNet Plus utilise le champ Hauteur d'ellipsoïde pour l'élévation (hauteur orthométrique). SI vous ne connaissez pas quel mode a été utilisée pour un projet GPSurvey, ouvrez le projet dans GPSurvey et vérifier les paramètres de contrôle spéciaux dans TRIMNet Plus.

Utilisation des informations d'Ephéméride précise

La fonction d'archivé du projet GPSurvey n'archivera pas les informations d'éphéméride précise utilisées lors du traitement des lignes de base dans le logiciel WAVE. Si les informations d'éphéméride précise ont été utilisées pendant le traitement des lignes de base dans le projet GPSurvey, vous devez les importer dans  Trimble Geomatics Office de la source d'origine.


Modèles d'antenne NGS

Vous pouvez utiliser des groupes de modèles d'antenne—Trimble ou NGS— avec  Trimble Geomatics Office. Vos exigences de levé déterminent groupe de modèles d'antenne à utiliser. En général, vous utiliserez le groupe de modèles d'antenne NGS lorsque vous utilisez des types d'antenne différents aux types définis dans antenna.ini.

Le groupe d'antennes Trimble est installé par défaut. Cependant, pour installer les modèles d'antenne NGS, dans la tabulation Global du dialogue Styles du traitement de lignes de base, sélectionnez l’option NGS dans la liste Modèle d’antenne.


 Commande Vue de levé

Utilisez cette commande pour afficher le projet dans la Vue de levé.

Pour y accéder, effectuez l'une des choses suivantes:

  Sélectionnez Affichage / Levé.

  Sélectionnez l'outil Vue de levé.


 Vue de levé - Vue d'ensemble

 Voir aussi

Vous pouvez afficher un projet dans la Vue de levé ou la Vue en plan.

Utilisez la Vue de levé pour effectuer des tâches ayant rapport au levé tels que:

  vérifier les observations terrestres et GPS

  corriger les données erronées.

Certaines options de menu et certaines barres d'outils ne sont disponibles que dans la Vue de levé. Elles vous permettent de réaliser les tâches ayant rapport au levé, tels que le traitement de lignes de base, et l'Ajustement de réseau.

ASTUCE: Pour afficher les différentes parties du projet au même temps, créez des fenêtres multiples. Pour plus amples informations, référez-vous à la commande Nouvelle fenêtre.

Affichage des données du levé

Les observations terrestres et GPS sont affichées sous forme de lignes dans la Vue de levé.

Cliquez deux fois sur une ligne d'observation afin de voir l'observation utilisant la fenêtre Propriétés, et éditez l'observation si nécessaire. Des indicateurs d'erreur s'affiche sur les points ou les observations lorsqu'il y a une erreur de fermeture hors de tolérance ou lorsqu'une observation GPS échoue à un critère de rejet.

Des couleurs de ligne différentes représentent un état différent dérivé de la configuration Style de légende dans le dialogue Propriétés du projet - tabulation Rapports en cours. Les points de contrôle et les points ajustés de réseau ont également des affichages spéciaux dans la Vue de levé.

Tâches disponibles dans la Vue de levé

Les tâches suivantes ne sont disponibles que dans la Vue de levé:

  Filtres d'affichage

  Sélectionner Observations

  Les options de rapport des fermetures de boucle GPS

D'autres tâches ayant rapport au levé sont disponibles si vous avez le module Traitement de lignes de base et le module Ajustement de réseau installés. Les tables suivantes illustrent les tâches supplémentaires disponibles de chaque module.

Module Traitement de lignes de base

Module Ajustement de réseau

Changement d'affichage

Vous pouvez changer l'affichage à tout moment.

Pour changer à la Vue en plan, effectuez l'une des choses suivantes:

  Sélectionnez sur l'outil Vue en plan.

  Sélectionnez Affichage / Plan.


 Commande Vue en plan

Utilisez cette commande pour afficher le projet dans la Vue en plan.

Pour y accéder effectuez l'une des choses suivantes:

  Sélectionnez Affichage / Plan.

 Sélectionnez l'outil Vue en plan.


    Vue en plan - Vue d'ensemble

 Voir aussi

Vous pouvez afficher un projet dans la Vue en plan ou la Vue de levé.

Utilisez la Vue en plan pour afficher les caractéristiques topographiques pendant le levé sur le terrain.

Vous pouvez ajouter des points, des lignes, des styles et des annotations pour améliorer la présentation du projet. Toute modification effectuée dans la Vue en plan, n'aura aucun effet sur les observations de levé sous-jacentes.

Certaines options de menu et certaines barres d'outils ne sont disponibles que dans la Vue en plan. Ces options vous permettent d'effectuer les tâches ayant rapport à CAD (telles que la codification, le contrôle des couches et la gestion du style).

ASTUCE: Pour afficher les différentes parties du projet au même temps, créez des fenêtres multiples. Pour plus amples informations, référez-vous à la commande Nouvelle fenêtre.

Les entités (points, lignes, courbes, arcs, texte, et annotations) sont affichées selon le style leur attribué. Vous pouvez changer le style d'une entité à l'aide de codification ou en utilisant la fenêtre Propriétés. Les observations conventionnelles et GPS ne sont pas affichées. Lorsque vous ajoutez une ligne dans la Vue en plan, vous ajoutez une nouvelle entité à la base de données.

Lorsque vous changez à la Vue de levé, les lignes CAD ne s'affichent pas.

Une fois que vous avez terminé les tâches dans la Vue en plan vous pouvez imprimer ou exporter le plan à votre logiciel de dessin.

Tâches disponibles dans le Vue en plan

Les tâches suivantes ne sont disponibles que dans la Vue en plan:

  Couches

  Styles (point, ligne, texte)

  Insérer points

  Insérer lignes

  Insérer texte

  Insérer annotations

  Insérer lignes pour les observations courantes

  Sélectionner par couches

  Sélectionner par styles

  Sélectionner par types

  Sélectionner par caract. de point

  Codification

  DTMLink

  RoadLink

Changement d'affichage

Vous pouvez changer l'affichage à tout moment.

Pour changer à la Vue de levé, effectuez l'une des choses suivantes

  Sélectionnez l'outil Vue de levé.

  Sélectionnez Affichage / Levé.

  Appuyez sur Alt+1.


Commande Zoom avant

Utilisez cette commande pour agrandir une section de la fenêtre de graphiques.

Pour y accéder effectuez l'une des choses suivantes:

  Sélectionnez Affichage/ Zoom avant

  Sélectionnez l'outil Zoom avant [image].

Lorsque vous avez Navigateur de zoom affiché, la zone affichée dans la fenêtre de graphiques s'affiche sous forme d'une case noire dans la fenêtre Navigateur de zoom.

Pour utiliser l'outil Zoom avant:

1.	Déplacez le pointeur sur la fenêtre de graphiques.

Le pointeur à loupe s'affiche entourant un signe de plus. Les croisées du réticule au-dessus du pointeur à loupe indiquent le point chaud.

2.	Effectuez l'une des choses suivantes:

  Cliquez sur une position dans la fenêtre de graphiques pour effectuer un zoom avant par un facteur de deux, centré sur le point chaud.

  Cliquez sur le pointeur et le faites glisser déplacer à travers de la fenêtre de graphiques afin de produire un rectangle contenant la zone sur laquelle vous voulez effectuer un Zoom avant. La plus petite la surface, le plus grand le changement d'échelle. Le contenu de la zone contenue par le rectangle occupera la fenêtre de graphiques entière lorsque vous relâchez le bouton de la souris. Pour annuler l'opération courante de l'outil Zoom avant, appuyez sur Echap pendant que le bouton de la souris est enfoncé.

Lorsque vous avez une souris à roulette, utilisez la roulette pour effectuer le zoom avant et arrière. Pour faire une panoramique avec la roulette, maintenez la roulette enfoncée et déplacez la souris. Le contenu de la fenêtre de graphiques se déplace pendant que vous déplacez la souris.


Commande Zoom arrière

Utilisez cet outil pour augmenter la zone affichée dans une fenêtre de graphiques.

Pour y accéder, effectuez l'une de choses suivantes:

  Sélectionner Affichage / Zoom arrière.

  Sélectionnez l'outil Zoom arrière [image].

Lorsque le Navigateur de zoom est affiché, la zone affichée dans la fenêtre de graphiques s'affiche sous forme d'une case noire dans la fenêtre Navigateur de zoom. Si vous faites un zoom au-delà des étendues du projet, aucune case noire n'est affichée dans le Navigateur de zoom.

Pour utiliser l'outil Zoom arrière:

1.	Déplacez le pointeur sur la fenêtre de graphiques.

Le pointeur à loupe s'affiche entourant un signe de moins. Les croisées du réticule au-dessus du pointeur à loupe indiquent le point chaud.

2.	Effectuez l'une des choses suivantes:

 Cliquez sur une position dans la fenêtre de graphiques. L'échelle de la fenêtre de graphiques change et sera centrée sur la position du point chaud.

 Cliquez sur le pointeur et le faites glisser déplacer à travers de la fenêtre de graphiques afin de produire un rectangle contenant la zone que vous voulez, puis relâchez le bouton de la souris. L'échelle de la fenêtre de graphiques change et les informations présentées précédemment dans la fenêtre de graphiques s'affiche dans la zone rectangulaire sélectionnée. Le plus petit le rectangle, le plus grand le changement d'échelle. Pour annuler l'opération courante de l'outil Zoom arrière, appuyez sur Echap pendant que le bouton de la souris est enfoncé.

Lorsque vous avez une souris à roulette, utilisez la roulette pour effectuer le zoom avant et arrière. Pour faire une panoramique avec la roulette, maintenez la roulette enfoncée et déplacez la souris. Le contenu de la fenêtre de graphiques se déplace pendant que vous déplacez la souris.


Commande Etendue de zoom

Utilisez cette commande pour afficher tous les points visibles dans le projet courant. Lorsque vous sélectionnez la commande Etendue de zoom, la fenêtre de graphiques s'ajuste automatiquement afin que toutes les données sont visibles.

Pour y accéder, effectuez l'une des choses suivantes:

  Sélectionnez Affichage / Zoom / Etendue.

  Appuyez sur Ctrl+E.

  Sélectionnez l'outil Etendue de zoom [image].

  Positionnez le pointeur dans la fenêtre de graphiques. Cliquez avec le bouton droit de la souris, et sélectionnez Etendue de zoom du menu de raccourcis.


Commande Zoom précédent

Utilisez cette commande pour retourner à chacun de vos affichages de zoom précédents depuis le temps que vous avez accédé au projet.

Pour y accéder, effectuez l'une des choses suivantes:

  Sélectionnez Affichage / Zoom / Précédent.

  Sélectionnez l'outil Zoom précédent [image].

Lorsque le Navigateur de zoom est affiché, la zone affichée dans la fenêtre de graphiques s'affiche sous forme d'une case noire dans la fenêtre Navigateur de zoom. Si vous faites un zoom au-delà des étendues du projet, aucune case noire n'est affichée dans le Navigateur de zoom.

Lorsque vous avez plus d'une fenêtre de graphiques ouverte, une pile de zoom séparée est stockée pour chaque fenêtre.


Commande Déplacer

Utilisez cette commande pour déplacer le centre de la zone de zoom afin d'afficher une différente zone de l'affichage de l'écran.

Pour y accéder, effectuez l'une des choses suivantes:

  Sélectionnez Affichage / Déplacer.

  Sélectionnez l'outil Déplacer [image].

Le pointeur à main s'affiche.

Lorsque le Navigateur de zoom est affiché, la zone affichée dans la fenêtre de graphiques s'affiche sous forme d'une case noire dans la fenêtre Navigateur de zoom. Si vous vous déplacez au-delà des étendues du projet, aucune case noire n'est affichée dans le Navigateur de zoom.

L'action de Déplacer n'est pas rappelée comme un zoom. Donc, vous ne pouvez pas effectuer un Zoom précédent pour retourner à l'affichage initial après avoir effectué une action Déplacer.

Pour utiliser l'outil Déplacer:

1.	Déplacez le pointeur sur la fenêtre de graphiques.

Les croisées du réticule au-dessus du pointeur indique le point chaud.

2.	Cliquez et faites glisser déplacer le pointeur à la position à laquelle vous voulez vous déplacer. Le contenu de la fenêtre de graphiques se déplace en temps réel pendant que vous déplacez la souris.

Lorsque vous avez une souris à roulette, utilisez la roulette pour effectuer le zoom avant et arrière. Pour se déplacer avec la roulette, maintenez la roulette enfoncée et faites glisser déplacer la souris. Le contenu de la fenêtre de graphiques se déplace pendant que vous déplacez la souris.


Commande Déplacer vers

Utilisez cette commande pour déplacer et mettre à l'échelle une position spécifique sur la fenêtre de graphiques.

Le dialogue Déplacer vers s'affiche.


Dialogue Déplacer vers

Utilisez ce dialogue pour déplacer et mettre à l'échelle une position spécifique sur la fenêtre de graphiques.

Pour y accéder, sélectionnez Affichage / Déplacer vers.

Centre de l'affichage

Ces champs vous permettent de compléter les actions de remplissage du champ, telles que la spécification d'un nom ou d'une position pour le centre de l'affichage.

Nom du point	Utilisez ce champ pour spécifier le nom du point dans la base de données. Lorsque vous utilisez le remplissage du champ pour ce champ, les champs Nom du point, Nord et Est sont remplis.
Nord/Est	Utilisez ces champs pour déterminer le centre de l'affichage. Si vous utilisez le remplissage du champ, les champs Nord et Est sont remplis.
Rayon à afficher

Le champ Rayon à afficher fixe l'étendue de l'affichage. Lorsque vous ne spécifiez pas une unité, le groupe d'unités de distance configuré dans la tabulation Unités et format du dialogue Propriétés du projet est utilisé.


    Commande Filtres

Utilisez cette commande pour ouvrir le dialogue Filtres d'affichage.

Pour y accéder effectuer l'une des choses suivantes:

  Sélectionnez Affichage/ Filtres.

  Sélectionnez l'outil Filtres d'affichage [image].

  Cliquez sur l'icône Filtres d'affichage [image] sur la barre d'état.


    Dialogue Filtres d'affichage

Utilisez ce dialogue pour éliminer toute observation que vous ne voulez pas afficher pendant votre travail dans la Vue de levé.

NOTE: Les données masquées par le filtre d'affichage ne peuvent pas être sélectionnées utilisant une commande du menu Sélectionner.

Pour y accéder, effectuez l'une des choses suivantes:

  Sélectionnez Affichage / Filtres

  Sélectionnez l'outil Filtres d'affichage

  Appuyez sur Ctrl+F

Observations

Utilisez ce groupe pour sélectionner les groupes d'observation que vous voulez afficher. Ce sert lorsque vous avez un projet avec plusieurs types d'observation, par exemple RTK et conventionnels, et que vous ne voulez afficher que les observations RTK.

Afficher toutes les observations

Sélectionnez cette option pour afficher toutes les observations dans le projet. Cette option s'affiche si vous avez le module Ajustement de réseau installé.

N'afficher que les observations marquées pour un ajustement

Sélectionnez cette option pour n'afficher que les observations avec la case à cocher Utiliser dans Ajustement de réseau sélectionnée. (Cette case à cocher est disponible dans la page Résumé de la fenêtre Propriétés).

Cette option s'affiche si vous avez le module Ajustement de réseau installé. Lorsque vous sélectionné cette option, certains champs des Groupes d'observation ne sont pas disponibles.

N'afficher que les observations pouvant être ajustées

Sélectionnez cette option pour n'afficher que les observations pouvant être utilisées dans un Ajustement de réseau, mais qui n'ont pas nécessairement la case à cocher Utiliser dans Ajustement de réseau sélectionnée. (Cette case à cocher est disponible dans la page Résumé de la fenêtre Propriétés).

Par exemple, les observations désactivées et les observations activées comme une vérification peuvent être sélectionnées si elles peuvent être utilisées dans un ajustement de réseau.

Les lignes de base et les observations conventionnelles possibles (ne pouvant pas être utilisées dans un Ajustement de réseau) sont éliminées par filtrage.

NOTE: Cette option s'affiche si vous avez le module Ajustement de réseau installé. Lorsque vous sélectionné cette option, certains champs des Groupes d'observation ne sont pas disponibles.

Types d'observation à afficher

Le groupe Types d'observation à afficher contient les cases à cocher suivantes:

  RTK

  Cinématique PT

  Laser

  Elévations delta

  Statique

  Terrestre

  Réduit

  Azimut

  Lignes de base non traitées

Utilisez ces cases à cocher pour sélectionner quels types d'observation vous voulez afficher. Par exemple, pour n'afficher que les vecteurs RTK, sélectionnez la case à cocher RTK et effacez les autres dans le groupe.

  Observations de prise de vue isolée

  Continu

  Observations désactivées

  Observations de vérification

Utilisez ces cases à cocher pour sélectionner quelles observations des types d'observation sélectionnés vous voulez afficher. Vous devez spécifier le type d'observation dans le groupe supérieur que les observations dans le groupe inférieur doivent filtrer.

Par exemple, pour n'afficher que les observations de vérification RTK, sélectionnez les cases à cocher RTK et Observations de vérification et effacez les autres dans chaque groupe.

NOTE: Lorsqu'une observation a plus d'une propriété d'observation, vous devez sélectionner toutes les cases à cocher ayant rapport au type d'observation afin de l'afficher.

Afficher points non connectés

Sélectionnez cette case à cocher pour afficher des points sans aucune observation passant de allant vers le point, par exemple, les points saisis au clavier.

Afficher événements

Sélectionnez cette case à cocher pour afficher les événements.

N'afficher que les données du groupe de sélection

Sélectionnez cette case à cocher pour afficher les données que vous avez enregistré précédemment comme un groupe de sélection. Sélectionnez le groupe de sélection de la liste.

Utiliser Timeline pour afficher les lignes de base GPS dépendantes

Lorsque cette case à cocher est sélectionné, seules les observations affichées dans la vue Timeline s'afficheront dans la fenêtre de graphiques.

Vous pouvez utiliser cet outil pour la sélection des lignes de base indépendantes pour un ajustement de réseau.

Utilisez le bouton Affichage de ligne de base dépendante dans la barre d'outils pour activer et désactiver cette option.

NOTE: Cette option ne s'applique qu'aux données GPS post traitées. Toutes les autres données s'afficheront dans la fenêtre de graphiques avec les filtres d'affichage appropriés appliqués.

NOTE: Cette option n'est disponible que si vous avez le module Traitement de lignes de base installé.

Heure

Sélectionnez cette case à cocher pour n'afficher que les observations GPS se trouvant dans un intervalle d'heure et de date spécifique. Les champs De et Vers sont activés.

De / Vers

Utilisez le remplissage du champ (sélectionnez l'heure de la règle de temps dans Timeline ou utilisez le clavier pour saisir l'intervalle d'heure et de date des observations GPS à afficher). Vous pouvez également saisir le jour julien.


Etiquettes de point - Vue d'ensemble

Plus amples informations

Comment ...	Voir aussi
Vue d'ensemble

Pour afficher des informations dans la fenêtre de graphiques à propos des entités sans stocker ces informations dans la base de données, utilisez des étiquettes. Les étiquettes sont disponibles dans la Vue de levé ainsi que la Vue en plan.

Les paramètres d'étiquettes sont restaurés lorsque vous ouvrez un projet à nouveau entre les sessions de travail. Pour stocker des informations à propos des entités dans la base de données du projet, utilisez annotations. Vous ne pouvez ajouter des annotations que dans la Vue en plan.

Vous pouvez étiqueter les points avec des combinaisons de champs et de texte définissable de l'utilisateur. Pour ajouter les options d'étiquette personnalisées, utilisez un éditeur de texte pour éditer le fichier Options.ini . Les étiquettes toujours s'affichent à droit du point à étiqueter.

Les étiquettes sont affichées utilisant la taille et la police assignées aux Info-bulles dans vos paramètres d'Apparence Microsoft Windows.


Commande Barre du projet

Utilisez cette commande pour afficher et masquer la Barre du projet.

Pour y accéder, effectuez l'une des choses suivantes:

  Sélectionnez Affichage / Barre du projet.

  Sélectionnez l'outil Barre du projet [image].


Commande Rafraîchir

Utilisez cette commande pour rafraîchir l'écran.

Pour y accéder, effectuez l'une des choses suivantes:

  Sélectionnez Affichage / Rafraîchir

  Appuyez sur F5.

Sélectionnez cette commande pour redessiner la vue en plan lorsque la fenêtre de graphiques devient écrasée et difficile à lire.


Dialogue Options d'affichage

Utilisez ce dialogue pour personnaliser votre affichage des données. Pour y accéder, sélectionnez Affichage / Options.

Le dialogue Options d'affichage a les tabulation suivantes:

  Lignes de grille

  Légende de levé

  Données de levé

  Contrôles d'ellipse d'erreur

  Carte d'arrière plan


Dialogue Etiquettes de point

Utilisez ce dialogue pour afficher les étiquettes de point du projet.

Pour y accéder, dans le menu principal, sélectionnez Affichage / Etiquettes de point.

Etiqueter points avec

Sélectionnez les champs avec lesquels vous voulez étiqueter des points. Les options suivantes sont fournies avec  Trimble Geomatics Office. Toute option de labels personnalisée que vous ajoutez en utilisant le fichier Options.ini s'affiche dans la liste.

  Nom

  Code de caractéristiques

  Elévation

  Facteur d'échelle de grille

  Facteur d'échelle d'élévation

  Facteur d'échelle combiné

  Angle de convergence

  Qualité horizontale

  Qualité d'élévation

  Qualité de hauteur

Appliquer à

Deux options pour l'étiquetage des points sont disponibles:

  Sélection − sélectionnez cette option pour étiqueter les points sélectionnés.

  BD entière − Sélectionnez cette option pour étiqueter tous les points. Lorsqu'il n'y a rien sélectionné dans la Vue de levé, cette option est sélectionnée par défaut.

Lorsque vous étiquetez une sélection,  Trimble Geomatics Office automatiquement produit un groupe de sélection nommé des points étiquetés.

NOTE: Vous ne pouvez étiqueter qu'une sélection à la fois. Lorsque vous avez une sélection de points étiquetée et puis vous étiquetez une nouvelle sélection, le labels dans la première sélection sont effacés.

Effacer

Cliquez sur Effacer pour supprimer tous les labels courants.

NOTE: Les paramètres de label sont restaurés lorsque vous ouvrez un projet ayant des labels de point à nouveau.

ASTUCE: Pour changer la taille et la police du texte, utilisez vos paramètres d'affichage pour les info-bulles Windows standard.


Dialogue Options d'affichage - tabulation Lignes de grille

Utilisez cette tabulation pour afficher les lignes de grille dans la fenêtre de graphiques.

Pour y accéder:

1.	Sélectionnez Affichage / Options. Le dialogue Options d'affichage s'affiche.

2.	Sélectionnez la tabulation Lignes de grille.

Afficher lignes de grille

Sélectionnez cette case à cocher pour afficher les lignes de grille dans la fenêtre de graphiques. Lorsque les lignes de grille sont affichées, l'intervalle de grille s'affiche dans la barre d'état.

NOTE: Vous pouvez également afficher les lignes de grille en sélectionnant le bouton de lignes de grille [image] dans la barre d'outils.

Les éléments suivants deviennent disponibles:

Nombre fixe de lignes de grille:

Utilisez cette option pour configurer le nombre de lignes de grille affichées sur l'écran. Lorsque vous effectuez un zoom avant ou arrière, le même nombre de lignes de grille seront affichées..

NOTE: Le nombre fixe de lignes de grille s'affichera le long du côté plus court de l'écran.

Intervalle de grille

Utilisez cette option pour spécifier l'intervalle de grille. Les unités de distance seront celles spécifiées pour le projet.

NOTE: Lorsqu'il y a trop de lignes de grille pour un affichage clair sur l'écran, les lignes de grille ne sont pas affichées.

Type de ligne de grille

Utilisez cette option pour sélectionner le style de lignes de grille, par exemple, solide ou pointillé.

Couleur de ligne de grille

Utilisez cette option pour sélectionner la couleur des lignes de grille. Il se peut que vous devez changer la couleur des lignes afin qu'elles s'affichent clairement contre la couleur d'arrière plan dans la fenêtre de graphiques.

Etiqueter lignes de grille

Sélectionnez cette case à cocher pour étiqueter les lignes de grille avec les valeurs nord et est.

NOTE: Les lignes de grille seront imprimées lorsque vous sélectionnez la commande Report/Imprimer.


Dialogue Options d'affichage - tabulation Légende de levé

Utilisez cette tabulation pour changer les couleurs utilisées pour l'affichage de votre projet dans la fenêtre de graphiques Trimble Geomatics Office. Elle détermine:

  comment les types d'observation différents et les points sont affichés dans la Vue de levé

  la couleur d'arrière plan de la fenêtre de graphiques

  la couleur par défaut du texte

Pour y accéder:

1.	Sélectionnez Affichage / Options. Le dialogue Options d'affichage s'affiche.

2.	Sélectionnez la tabulation Légende de levé.

Style de légende

Dans ce champ, sélectionnez Noir ou Défaut. La table suivantes décrit les paramètres de couleur déterminés par chaque option:


Dialogue Options d'affichage - tabulation Données de levé

Utilisez cette tabulation pour changer la façon dont les données de levé sont affichées dans la fenêtre de graphiques.

Pour y accéder:

1.	Sélectionnez Affichage / Options. Le dialogue Options d'affichage s'affiche.

2.	Sélectionnez la tabulation Données de levé.

NOTE: Cette tabulation n'est visible que si vous êtes dans la Vue de levé.

Utiliser morceaux de lignes pour les prises de vue isolées

Sélectionnez cette case à cocher pour afficher des vecteurs d'observation de prise de vue isolée sous forme des morceaux de lignes.

Afficher le nombre d'observations (où > 1)

Sélectionnez cette case à cocher pour afficher le nombre d'observations sur des lignes observées multiples.

Lorsqu'une observation est filtrée, le nombre d'observations est indiqué sous forme d'un ratio de ce qui est affiché par rapport au nombre d'observations total, par exemple, 1/3.


Dialogue Options d'affichage - tabulation Contrôles d'ellipse d'erreur

Utilisez cette tabulation pour déterminer si les ellipses d'erreur, les flèches d'erreur de hauteur et les flèches d'erreur d'élévation s'afficheront dans la fenêtre de graphiques après un ajustement. Vous pouvez également déterminer l'échelle des ellipses d'erreur et la taille des coches.

Pour y accéder:

1.	Sélectionnez Affichage / Options. Le dialogue Options d'affichage s'affiche.

2.	Sélectionnez la tabulation Contrôles d'ellipse d'erreur.

NOTE: Cette tabulation n'est visible que si vous êtes dans la Vue de levé.

Afficher ellipses d'erreur de point & flèches

Sélectionnez cette case à cocher pour afficher les ellipses d'erreur de point et les flèches.

Afficher erreurs d'élévation

Sélectionnez cette case à cocher pour afficher les erreurs d'élévation.

Afficher erreurs de hauteur

Sélectionnez cette case à cocher pour afficher les erreurs de hauteur.

Echelle d'erreur d'ellipse

Le curseur d'Echelle d'erreur d'ellipse, indiqué ci-dessus, contrôle la taille des ellipses d'erreur que s'affichent dans la fenêtre de graphiques après un ajustement.

Pour rendre les ellipses plus petites, déplacez le curseur à gauche. Pour rendre les ellipses plus grandes, déplacez le curseur à droite.

Taille de la coche

Utilisez ce champ, indiqué ci-dessous, pour déterminer la distance entre les coches qui s'affichent sur les barres d'échelle d'ellipse d'erreur dans la fenêtre de graphiques.

Les échelles de barre devient plus longues lorsqu'elles sont augmentées de taille, pendant que la distance entre les coches change pour correspondre à la valeur que s'affiche dans le champ Taille de coche.


Dialogue Affichage d'options - tabulation Carte d'arrière plan

 Trimble Geomatics Office peut afficher les fichiers d'arrière plan dans la fenêtre de graphiques. Vous pouvez importer les fichiers d'arrière plan tels que les Drawing Exchange Format (.dxf), Windows Bitmap (.bmp), ou Tagged Image File Format (.tif) pour les afficher. Les fichiers Windows Bitmap (.bmp), ou Tagged Image File Format (.tif) doivent être géoréférencés afin qu'ils s'affichent correctement.

Le format World ESRI. Un fichier World est un fichier de texte ASCII avec une extension .twf ou .wld.  Trimble Geomatics Office a les exigences suivantes pour les fichiers World. Ils doivent:

  être stockés dans le même emplacement que le fichier .dxf, .bmp ou.tif relatif

  avoir le même système de coordonnées que votre projet Trimble Geomatics Office

  avoir les mêmes unités que votre projet Trimble Geomatics Office

NOTE:  Trimble Geomatics Office prend en charge les fichiers TIFF non compressés, et les schèmes ce compression suivants: CCITT, PackBits, Thunder, Next, LogLuv. Cependant, il ne prend  pas  en charge les fichier compressés suivants: OJPEG, JPEG, ZIP, LZW, PixarLog.

NOTE:  Trimble Geomatics Office également ne prend pas en charge les fichiers  compressés RLE .bmp.

Utilisez la tabulation Carte d'arrière plan pour sélectionner une carte d'arrière plan à afficher.

Pour y accéder:

1.	Sélectionnez Affichage / Options. Le dialogue Options d'affichage s'affiche.

2.	Sélectionnez Carte d'arrière plan.

Noms de fichier

Ce groupe affiche une liste des fichiers sélectionnés à utiliser comme des cartes d'arrière plan.

Ajouter

Utilisez le bouton Ajouter pour sélectionner les fichiers d'arrière plan. Vous pouvez sélectionner un fichier .dxf, .bmp, ou .tif.

NOTE: Assurez-vous que le fichier World relatif soit stocké dans le même emplacement. Il n'est pas nécessaire de sélectionner ce fichier de la liste.

Supprimer

Utilisez le bouton Supprimer pour supprimer un fichier d'arrière plan de la liste.

Fichier de format World

Un exemple d'un fichier de format World est illustré ci-dessous:

20.17541308822119

0.00000000000000

0.00000000000000

−20.17541308822119

424178.11472601280548

4313415.90726399607956

Ces 6 valeurs sont utilisées pour calculer une transformation image au monde réel utilisant une transformation affine à six paramètres sous la forme de:

x1 = Ax + By + C

y1 = Dx + Ey + F

Où

x1	coordonnée x calculé du pixel sur la carte

y1 	coordonnée y calculé du pixel sur la carte

x 	numéro de la colonne d'un pixel dans l'image

y 	numéro de la ligne d'un pixel dans l'image

A 	échelle x; dimension d'un pixel dans les unités de carte dans la direction x

B, D	termes de rotation

C, F	termes de translation; x,y coordonnées de carte du centre du pixel en haut à gauche

E 	négatif de l'échelle y; dimension d'un pixel dans les unités de carte dans la direction y

Les paramètres de transformation sont stockés dans le fichier Word dans l'ordre suivant:

20.17541308822119 − A

0.00000000000000 − D

0.00000000000000 − B

-20.17541308822119 − E

424178.11472601280548 − C

4313415.90726399607956 − F

NOTE:  Trimble Geomatics Office ne prend en charge la rotation des images. Si votre fichier World a des termes de rotation non zéro (paramètres D et B), ils ne seront pas utilisés.

NOTE: Le signe de l’échelle x (A) ou l’échelle y (E) change selon l’orientation du système de coordonnées. C’est du fait que les origines d'une image et un système de coordonnées géographique sont différentes. L'origine d'une image se trouve dans le coin en haut à gauche, pendant que l'origine du système de coordonnées géographique change selon la direction des coordonnées.

Par exemple, pour un système de coordonnées NE, l’origine du système de coordonnées de carte se trouve dans le coin en bas à gauche. Les valeurs de ligne dans l’image s’augmentent du coin en haut à gauche vers le bas, pendant que les valeurs de coordonnées y dans la carte s’augmentent du coin en bas à gauche vers le haut. Donc, pour un système NE, l'échelle y (E) est négative. La table suivante fournit un résumé du signe des échelles x et y qu’il faut utiliser pour les systèmes de coordonnées NE, SE et NO:

	NE	SO	NO
x(A)	+ve	-ve	-ve
y(E)	-ve	+ve	-ve


Garantie

Garantie limitée du logiciel et du firmware

Trimble garantit que ce produit de logiciel Trimble (le “Logiciel”) sera substantiellement conforme aux spécifications publiées pour le Logiciel pour une période de quatre-vingt dix jours, à compter de la date de livraison.

Recours de garantie

La seule responsabilité de Trimble et votre recours exclusif sous la garantie détaillée ci-dessus sera, à l'option de Trimble, de réparer ou remplacer tout produit ou logiciel que ne se conforme pas à une telle garantie ("Produit non conforme") ou de rembourser le prix d'achat que vous avez payé pour tout produit non conforme, sur le retour du produit non conforme à Trimble.

Exclusions de la garantie

Cette garantie ne sera appliquée que dans le cas et dans la mesure que: (i) les Produits et le Logiciel sont installés, configurés, interfacés, stockés et utilisés  de façon correcte et selon les spécifications et le manuel de l'utilisateur applicables de Trimble, et (ii) les Produits et le Logiciel ne sont pas modifiés ou employés improprement. La garantie précédente ne s'appliquera pas à, et Trimble ne sera pas responsable pour, toute demande de contravention de la garantie basée sur (i) des défauts ou des problèmes de fonctionnement qui se produisent à cause de la combinaison ou l'utilisation du Produit ou du Logiciel avec des produits, des informations des systèmes ou des périphériques qui ne sont pas fabriqués, fournis ou spécifiés par Trimble; (ii) le fonctionnement du Produit ou du Logiciel sous toute spécification autre que, ou en plus des spécifications standard de Trimble pour ses produits; (iii) la modification ou l'utilisation non autorisée du Produit ou du Logiciel; (iv) les dommages à cause de la foudre ou d'autres décharges électriques, ou l'immersion dans l'eau normale ou salée ou la pulvérisation; ou (v) l'usure normal des pièces de consommation (par exemple, les batteries).

LA GARANTIE CI-DESSUS DECLARE LA RESPONSABILITE ENTIERE DE TRIMBLE ET VOS RECOURS EXCLUSIFS POUR LES PRODUITS ET LE LOGICIEL. SAUF STIPULE DANS CETTE GARANTIE, TRIMBLE FOURNIT LES PRODUITS ET LE LOGICIEL "TELS QUELS", SANS GARANTIE EXPRESSE OU LIMITEE, ET L'EXCLUSION EXPRESSE DES GARANTIES SUGGEREES DE VALEUR COMMERICALE ET DE CAPACITE POUR UN BUT PARTICULIER. LES GARANTIES EXPRESSES INDIQUEES SONT AU LIEU DE TOUTE OBLIGATION OU RESPONSABILITE DE LA PART DE TRIMBLE SURVENANT OU RELATIVEMENT A, TOUT PRODUI OU LOGICIEL. CERTAINS ETATS N'AUTORISENT PAS LES LIMITATIONS SUR LA DUREE D'UNE GARANTIE IMPLIQUEE, DONC LA GARANTIE CI-DESSUS PEUT NE PAS S'APPLIQUER DANS VOTRE CAS.

Limitation de responsabilité

A L'ETENDUE MAXIMUM PERMISE PAR LA LOI APPLICABLE, TRIMBLE NE SERA PAS RESPONSABLE A VOUS-MEMES POUR TOUTE PLAINTE DE DOMMAGE SPECIAL, INDIRECT OU CONSEQUENTIEL DE TOUT TYPE ET SOUS TOUTE CIRCONSTANCE OU THEORIE LEGALE AYANT RAPPORT AUX PRODUITS OU AU LOGICIEL, ET INDIFFERENT AUX RAPPORTS COMMERCIAUX QUE POURRAIENT AVOIR DEVELOPPES ENTRE VOUS ET TRIMBLE. A CAUSE DU FAIT QUE CERTAINS ETATS N'AUTORISENT PAS L'EXCLUSION OU LA LIMITATION DES DOMMAGES ACCIDENTELS OU CONSEQUENTIELS, LA LIMITATION CI-DESSUS PEUT NE PAS APPLIQUER DANS VOTRE CAS.

DANS AUCUN CAS, LA SEULE RESPONSABILITE DE TRIMBLE ET VOTRE SEUL RECOURS SOUS CET ACCORD OU EN CAS D'UNE INFRACTION DE L'ACCORD, SERA LIMITE  AU REMBOURSEMENT DU PRIX D'ACHAT OU DE DROIT DE LICENCE PAYE POUR LES PRODUITS OU LE LOGICIEL.


Bienvenue dans l'Aide Trimble Geomatics Office™.

Cette Aide est conçue afin de fournir les informations requises pour vous permettre d'utiliser le potentiel et la puissance totale du logiciel Trimble Geomatics Office. Elle décrit comment installer et utiliser Trimble Geomatics Office.

  Utilisation de l'Aide

  Trimble Geomatics Office − Vue d'ensemble

  A propos de Trimble

  Feedback


Commande Nouvelle fenêtre

Utilisez cette commande pour ouvrir une autre fenêtre pour le projet courant.

Pour l'accéder, sélectionnez Fenêtre / Nouvelle fenêtre.

Chaque nouvelle fenêtre est ajoutée à la liste de fenêtres dans le menu Fenêtre.

La nouvelle fenêtre a le même contenu que la fenêtre active. Vous pouvez ouvrir des fenêtres multiples afin d'afficher des parties ou des vues différentes du projet au même temps. Par exemple, vous pouvez ouvrir des fenêtres différentes et afficher une différente couche du projet dans chaque fenêtre.

Lorsque vous modifiez le contenu d'une fenêtre, toutes les autres fenêtres contenant le même projet refléteront ces modifications à condition qu'elles affichent cette ces informations.

Lorsque vous ouvrez une nouvelle fenêtre, elle devient la fenêtre active et s'affiche en dessus de toutes les autres fenêtres ouverts.


Commande Cascade

Utilisez cette commande pour faire se chevaucher multiples fenêtres du projet ouvertes.

Pour l'accéder, sélectionnez Fenêtre / Cascade.

Les fenêtres se chevauchent afin que vous pouvez voir la barre de titre de chaque fenêtre.


Commande Mosaïque

Utilisez cette commande pour réorganiser toutes les fenêtres du projet ouvertes actuellement.

Pour l'accéder, sélectionnez Fenêtre / Mosaïque.

Les fenêtres sont organisées sous forme d'une mosaïque.


Commande Réorganiser les icônes

Utilisez cette commande pour réorganiser toute icône des fenêtres réduites de façon ordonnée sur l'écran.

Pour l'accéder, sélectionnez Fenêtre / Réorganiser les icônes.

Les icônes sont réorganisées en bas de la fenêtre.


Commande Fractionné

Utilisez cette commande pour afficher deux parties d'un projet simultanément.

Pour l'accéder, sélectionnez Fenêtre / Fractionné.

La division dans la fenêtre peut être redimensionnée en cliquant sur et déplaçant la barre de division. Les modifications effectuées à une partie se produisent simultanément dans l'autre.


Commande Liste d'affichage

Utilisez cette commande pour ouvrir une fenêtre dans la liste d'affichage. La liste affiche toutes les fenêtres ouvertes.

Pour l'accéder, sélectionnez du menu de liste d'affichage Fenêtre.


Site World Wide Web

Pour une vue interactive de Trimble, visitez notre site sur le World Wide Web
www.trimble.com/


  Fenêtre Propriétés - Vue d'ensemble

La fenêtre Propriétés vous permet d’afficher des informations sur toutes les entités (points, observations, arcs, lignes, courbes, texte, annotation, notes).

Utilisez la fenêtre Propriétés pour afficher et éditer les propriétés d’une entité sélectionnée. Vous pouvez afficher les propriétés d’une seule unité à la fois. Toute modification effectuée sur les entités dans la fenêtre Propriétés est appliquée immédiatement.

La taille de la fenêtre Propriétés peut être modifiée en cliquant sur le cadre de la fenêtre et en le faisant glisser. Cliquez sur le bouton du coin supérieur droit pour fermer la fenêtre.

Vous pouvez laisser la fenêtre Propriétés ouverte pendant que vous utilisez d’autres parties de Trimble Geomatics Office. La fenêtre Propriétés est automatiquement mise à jour pour afficher les informations correctes sur le groupe de sélection.

Si vous avez défini un groupe de sélection courant et que vous sélectionnez une entité à afficher dans la liste de sélection de la fenêtre Propriétés, la sélection courante de la fenêtre graphique se met à jour pour inclure uniquement l’entité que vous affichez. La liste de sélection de la fenêtre Propriétés conserve le groupe de sélection d’origine. Toute entité sélectionnée dans la liste de sélection de la fenêtre Propriétés devient le groupe de sélection courant de la fenêtre graphique.

La vue arborescente figurant dans la fenêtre Propriétés indique les noms et valeurs des articles sélectionnés à l’écran. Cette vue arborescente indique également la relation entre les entités. Par exemple, vous pouvez afficher les points « de » et « vers » de toute entité d’observation ou de type de ligne en cliquant sur le signe plus (+) et en sélectionnant le point que vous souhaitez afficher et éditer.

L’affichage peut être réduit ou agrandi au niveau des nœuds :

  Cliquez sur le signe plus (+) pour agrandir l’affichage de l’arborescence.

  Cliquez sur le signe moins (−) pour réduire l’affichage de l’arborescence.

Les informations et tabulations disponibles sur le côté droit de la fenêtre Propriétés dépendent du type d’entité actuellement sélectionné.

Outils de sélection de la fenêtre Propriétés

Les outils de sélection de la fenêtre Propriétés vous permettent de configurer votre liste de sélection.

Sélection	Icône	Cliquez sur ce bouton pour...
Groupe les données de levé		grouper les données de levé par source, type de données ou direction d’observation.
Trier		trier les entités de la liste de sélection. Sélectionnez l’option souhaitée dans la liste. Vous pouvez choisir de trier les points par ordre de base de données, élévation, nom, qualité, heure de début, durée d’occupation ou type de données.
Filtrer		conserver uniquement des points, observations, observations balisées ou le groupe de sélection courant. Vous pouvez également supprimer la sélection courante. NOTE : Vous pouvez ajouter vos propres interrogations à la liste de filtres.

Indicateurs de qualité

Les indicateurs de qualité sont des petits symboles qui s’affichent à droite de chaque champ de la page Résumé. Ces symboles indiquent la qualité des données de levé. La qualité peut être de type contrôle fixe, contrôle, levé, mappage, inconnu ou ajusté. Les indicateurs de qualité sont également utilisés dans le Rapport de dérivation du point et le Rapport de recalcul.

Indicateurs de géoïde

Les indicateurs de géoïde montrent si le composant d’élévation ou de hauteur d’un point a été dérivé du géoïde défini pour votre projet. L’indicateur de géoïde s’affiche avec l’indicateur de qualité.

Par exemple, si vous mesurez un point RTK, il comporte une hauteur WGS-84 et une élévation dérivées du géoïde. Si vous collectez un point terrestre, il comporte une élévation et une hauteur WGS-84 dérivées du géoïde.

Les indicateurs de géoïde sont également utilisés dans le Rapport de dérivation du point et le Rapport de recalcul.

La qualité du géoïde du modèle de géoïde est déterminée sous la tabulation Recalculer du dialogue Propriétés du projet.

Outil Remontage auto

Utilisez cet outil [image]  pour réduire la fenêtre Propriétés lorsque vous sélectionnez les entités dans la fenêtre graphique.

Verrouiller la fenêtre ouverte

Utilisez cet outil [image] afin d’assurer que lorsque la fenêtre Propriétés est ouverte elle reste à la même taille.


Fenêtre Propriétés - Tabulation Levé - Vue d'ensemble

Utilisez la tabulation Levé pour afficher et éditer les propriétés relatives au levé d’une entité de la sélection. Vous pouvez afficher les propriétés des entités figurant dans la Vue de levé, parmi lesquelles :

  Points

  Observations GPS

  Observations conventionnelles

  Observations de niveau

  Observations par télémètre à laser

  Observations réduites (déports)

Pages

Les informations sur le levé sont organisées en pages. Vous pouvez afficher une seule page à la fois. Vous pouvez accéder aux pages à l’aide des boutons de menu disponibles. Les boutons de menu permettent d’accéder à toutes les informations sur le levé et dépendent du type d’entité actuellement sélectionné.

Suppression de points et d’observations

Pour supprimer un point ou une observation, utiliser le bouton Supprimer [image] de la fenêtre Propriétés. Si une observation ou un point ne peut être supprimé, le bouton Supprimer n’est pas disponible.

NOTE : Une coordonnée fixe ne peut pas être supprimée. « Annuler l’état fixe » de la coordonnée avant de la supprimer.


Fenêtre Propriétés - Tabulation Levé - Points

 Vue d'ensemble

Utilisez la tabulation Levé de la fenêtre Propriétés pour afficher et éditer les informations relatives au levé d’un point sélectionné. Les informations sur le levé sont organisées en pages.

Vous pouvez accéder aux pages suivantes en cliquant sur le bouton de menu approprié :

Résumé
Avertissement
Occupations et configurations
Estimation d'erreur


Fenêtre Propriétés - Tabulation levé - Page Résumé (Points)

Plus amples informations

 Comment…

Utilisez la page Résumé pour afficher la position dérivée et les qualités du point sélectionné dans la fenêtre Propriétés.

Pour y accéder à partir de la fenêtre Propriétés :

  Dans la tabulation Levé, cliquez sur le bouton de menu Résumé [image].

Nom de point

Affiche le nom du point. Entrez une valeur dans ce champ pour modifier le nom.

Champs de coordonnées

Affiche la position dérivée du point ainsi que les qualités assignées à chaque composant. Les champs de coordonnées affichés dépendent de l’option d’affichage configurée.

Grille / Local / WGS-84

Sélectionnez l’option d’affichage des coordonnées pour le point.

NOTE : Si l’option d’affichage est configurée sur WGS-84, les champs de coordonnées affichés dépendent de ce qui est configuré pour le paramètre Afficher WGS-84 sous de la tabulation Unités et format du dialogue Propriétés du projet. Par exemple, si le champ Afficher WGS-84 sous est configuré sur L,L,H, les champs de coordonnées de la page Résumé sont affichés sous Latitude, Longitude, Hauteur. Si le champ Afficher WGS-84 sous est configuré sur X,Y,Z, les champs WGS-84 sont affichés sous X, Y, Z.

Rapport de dérivation

Sélectionnez l’outil Rapport de dérivation [image] pour afficher le Rapport de dérivation du point.

Ajouter coordonnée

Sélectionnez l’outil Ajouter coordonnée [image] lorsque vous souhaitez saisir au clavier une coordonnée de grille, locale ou WGS-84 saisie au bureau. (Vous ne devriez effectuer cette opération que lorsque vous voulez que le logiciel utilise votre coordonnée saisie au bureau plutôt que les coordonnées dérivées des observations.) La coordonnée saisie au clavier comportera une source saisie au bureau.

Seules une coordonnée de grille/locale et une coordonnée WGS-84 saisie au bureau peuvent être associées à un point. Si une coordonnée WGS-84 et une coordonnée de grille ou locale existent déjà, l’outil Ajouter coordonnée n’est pas disponible.

NOTE : Lorsque vous ajoutez une coordonnée, les champs des composants de la coordonnée ainsi que la qualité sont nuls.  Pour entrer la position et la qualité du point courant, sélectionnez l’outil Insérer valeur courante [image].


Fenêtre Propriétés - Tabulation Levé - Page Avertissement

Utilisez la page Avertissement pour afficher et supprimer les messages associés aux indicateurs d’avertissement qui se trouvent sur des points ou observations de la Vue de levé.

Pour cela :

1.	Effectuez l'une des opérations suivantes :

  Double-cliquez sur un point ou une observation avec indicateur d’avertissement que vous voulez afficher.

  Double-cliquez sur l’icône de l’indicateur d’avertissement de la barre d'état pour sélectionner tous les points avec des indicateurs d’avertissement.

La fenêtre Propriétés s’affiche.

2.	Si votre sélection comporte plus d’une entité, sélectionnez l’entité que vous souhaitez afficher.

3.	Pour accéder à la page Avertissement, cliquez sur le bouton de menu Avertissement [image].

Vous devez remédier aux conditions ayant provoqué l’apparition des indicateurs d’avertissement ou supprimer ces derniers avant d’exporter des coordonnées. Lorsque vous remédiez à une Fermeture hors tolérance en désactivant une mauvaise observation ou en renommant le point parce que l’observation était vers un point physique différent, l’indicateur d’erreur disparaît.

La table suivante décrit les messages qui peuvent s’afficher avec des indicateurs d’avertissement :

Supprimer ce message

Sélectionnez cette case à cocher pour supprimer l’indicateur d’avertissement associé au point ou à l’observation sélectionné si vous décidez que l’erreur est acceptable pour le levé. L’indicateur devient blanc dans la fenêtre Propriétés et disparaît de la fenêtre graphique.

NOTE : Si vous supprimez tous les indicateurs d’avertissement, l’icône de l’indicateur sur la barre d'état devient blanche.

Si vous avez sélectionné un point, vous pouvez sélectionner l’outil Dérivation du point [image] pour afficher le Rapport de dérivation du point.

Utilisez ce rapport pour déterminer comment la position du point a été dérivée et pourquoi l’erreur s’est produite.


Fenêtre Propriétés - Tabulation Levé - Pages Occupations et configurations

Utilisez les pages Occupations et configurations pour afficher et corriger les détails de l’occupation GPS ou de la configuration terrestre du point ou de l’observation sélectionné dans la fenêtre Propriétés.

Les pages suivantes sont disponibles en fonction du type d’entité sélectionné et de la méthode de terrain utilisée :

Pour accéder à ces pages à partir de la fenêtre Propriétés:

1.	Sélectionnez l’observation.

2.	Cliquez sur le bouton de menu Occupations et configurations [image].

NOTE : S’il y a plus d’une occupation ou configuration pour le point, un menu de raccourcis s’affiche. Sélectionnez la page que vous souhaitez afficher. Au fur et à mesure que vous maintenez la souris sur différentes occupations et configurations du menu de raccourcis, les détails sont mis à jour dans la page.

Un point peut comporter plus d’une occupation. Cependant, une observation ne comporte qu’un seul point « de » et qu’un seul point « vers ».


Fenêtre Propriétés - Tabulation Levé - Page Occupations GPS

Utilisez la page Occupations GPS pour afficher et corriger les détails sur l’occupation GPS du point sélectionné. Le contenu de la page est identique si le point a été utilisé comme station de base ou position mobile.

Pour y accéder à partir de la fenêtre Propriétés:

1.	Sélectionnez la tabulation Levé.

2.	Effectuez l'une des opérations suivantes :

  Si un point est sélectionné, cliquez sur le bouton de menu Occupations et configurations [image].

NOTE : S’il y a plus d’une occupation ou configuration pour le point, un menu de raccourcis s’affiche. Sélectionnez la page que vous souhaitez afficher.

  Si une observation GPS est sélectionnée, cliquez sur le bouton de la page Occupations de base [image] pour afficher les détails sur l’occupation de base, ou cliquez sur le bouton de menu Occupations mobile [image] pour afficher les détails sur l’occupation mobile.

Point

Affiche le nom du point objet de l’occupation.

Heure

Affiche la durée de l’occupation ainsi que l’heure à laquelle l’occupation a commencé.

NOTE : Lors de l’importation de données, les heures de Début et de Fin sont converties de temps GPS en TUC (temps universel coordonné - c.-à-d. en temps moyen de Greenwich). Les heures sont ensuite affichées comme heures locales à l’aide de la configuration du fuseau horaire courant de Microsoft Windows.

ASTUCE : Positionnez le pointeur sur le champ pour afficher une Info-bulle sur l’heure d’arrêt de l’occupation.

Détails d'antenne

Le groupe Détails d’antenne contient des détails sur l’antenne GPS utilisée pour l’occupation relative au point observé.

Hauteur

Affiche la hauteur de l’antenne. Si la hauteur saisie lors de l’occupation est incorrecte, utilisez le clavier pour entrer une valeur afin de modifier ce champ.

Type

Affiche le type d’antenne connectée au récepteur. Si l’antenne saisie lors de l’occupation est incorrecte, sélectionnez un type d’antenne dans la liste afin de modifier ce champ.

Mesuré à

Affiche jusqu’où la hauteur de l’antenne a été mesurée. Si la méthode saisie lors de l’occupation est incorrecte, sélectionnez une option dans la liste afin de modifier ce champ.


Fenêtre Propriétés - Tabulation Levé - Page Configuration d'instrument

Utilisez la page Configuration d’instrument pour afficher et corriger les détails de configuration de l’instrument conventionnel utilisé pour l’observation.

Pour y accéder à partir de la fenêtre Propriétés:

1.	Sélectionnez la tabulation Levé.

2.	Avec un point sélectionné, cliquez sur le bouton de menu Occupations et configurations [image].

NOTE : S’il y a plus d’une occupation ou configuration pour le point, un menu de raccourcis s’affiche. Sélectionnez la page Configuration d’instrument que vous souhaitez afficher. Toutes les occupations du point, GPS et terrestres, peuvent être affichées dans cette page.

Point d'instrument

Affiche le nom du point à partir duquel l’observation a été effectuée.

Hauteur d'instrument

Affiche la hauteur de l’instrument au moment où l’observation a été effectuée. Si la hauteur saisie lors de la configuration est incorrecte, utilisez le clavier pour entrer une valeur afin de modifier ce champ.

Point VA

Affiche le nom du point de visée arrière observé.

Azimut VA

Affiche l’azimut calculé vers la visée arrière observée.

NOTE : Si le point de visée arrière comporte des coordonnées, celles-ci sont utilisées pour déterminer l’azimut vers le point de visée arrière. S’il existe également un azimut saisi au clavier, la qualité de cet azimut saisi au clavier est comparée à celle de l’azimut calculé à l’aide des coordonnées du point de visée arrière, et l’azimut de la plus haute qualité est utilisé. Si la qualité de l’azimut calculé et celle de l’azimut saisi au clavier sont les mêmes, alors l’azimut calculé est utilisé.


Fenêtre Propriétés - Tabulation Levé - Page Configuration de destination

Utilisez la page Configuration de destination pour afficher et corriger les détails de configuration de destination utilisés pour l’observation conventionnelle.

Pour accéder è cette page de la fenêtre Propriétés:

1.	Dans la tabulation Levé, sélectionnez l’observation conventionnelle.

2.	Cliquez sur le bouton de menu Configuration de destination [image]  pour afficher les détails de configuration de destination.

Point VAv

Affiche le nom du point de visée avant.

Hauteur de destination

Affiche la hauteur de la destination au moment où l’observation de visée avant a été effectuée. Si la hauteur saisie lors de la configuration est incorrecte, utilisez le clavier pour entrer une valeur afin de modifier ce champ.

Constante prisme

Affiche la constante prisme de la destination qui est appliquée à l’observation. Si la constante n’a pas été saisie ou si la constante saisie lors de la configuration est incorrecte, utilisez le clavier pour entrer une valeur afin de modifier ce champ.

Le champ peut afficher ‘?’ parce que l’observation a déjà eu la constante de prisme appliquée dans le fichier, et le fichier ne spécifie pas la valeur de la constante. S’il y avait une erreur pendant la configuration, et la constante de prisme a été entrée incorrectement, il faut appliquer la différence entre la constante de prisme entrée dans le champ et la constante que devrait été entrée. Par exemple, si vous avez entré “20” dans le champ, et vous devriez avoir entré “5”, alors entrez une correction de “–15” dans le champ de constante prisme.


Fenêtre Propriétés - Tabulation Levé - Page Coordonnées GPS (APC)

Utilisez la page Coordonnées GPS (APC) pour afficher une solution de position GPS. Une solution de position peut être obtenue à partir de :

  GPS différentiel

  Wide Area Augmentation System, WAAS (système de renforcement à couverture étendue)

  A l’aide de la touche [image] du Trimble Survey Controller

  Positions autonomes

VRS – superficie plus large RTK (fixe et flottant)

Les solutions de position sont affichées pour un point si elles sont de qualité Levé ou Mappage. Vous pouvez avoir plus d’une solution de position mais ces données ne peuvent pas être éditées.

Pour accéder à cette page à partir de la fenêtre Propriétés:

1.	Dans la fenêtre graphique, double-cliquez sur le point comprenant une Coordonnée GPS (APC).

2.	Agrandissez le point dans l’arborescence de la fenêtre Propriétés.

3.	Sélectionnez le SN?soln de l’arborescence [image], où ? indique le numéro de coordonnée.

La page Coordonnées GPS (APC) s’affiche.

Point

Affiche le nom du point.

Latitude, Longitude, Hauteur

Affiche les coordonnées WGS-84 et les qualités de point déterminées pour le point.

NOTE : La hauteur affichée est la hauteur du Centre de phase d’antenne de l’antenne GPS utilisée pour l’occupation. La hauteur d’antenne n’est pas appliquée.

Type de processeur

Affiche la méthode de traitement. Par exemple, WAAS, superficie large RTK.

Etat

Affiche l’état de la solution de position. L’état est pris en compte par le recalcul, ou lorsqu’un calcul de levé est effectué. Pour modifier ce champ, cliquez sur la flèche et sélectionnez l’option de la liste déroulante que vous souhaitez appliquer.


Fenêtre Propriétés - Tabulation Levé - Page Coordonnées

Plus amples informations

 Voir aussi

Vous pouvez utiliser la page Coordonnées pour afficher les coordonnées d’un point sélectionné. Vous pouvez également l’utiliser pour modifier les qualités et l’état des coordonnées.

Un point peut comporter une coordonnée WGS-84, une coordonnée de grille/locale et des coordonnées ajustées. Une coordonnée ajustée peut se produire après que vous avez effectué un ajustement de réseau (si le module Ajustement de réseau est installé sur votre ordinateur).

Utilisez la page Résumé pour afficher les coordonnées du point.

Pour accéder à la page Coordonnées :

1.	Dans la fenêtre graphique, double-cliquez sur le point. La fenêtre Propriétés s’affiche.

2.	Agrandissez l’arborescence pour le point et sélectionnez la coordonnée.

Les coordonnées du point sont affichées.

Champs de coordonnées

Affiche les valeurs des composants de coordonnées du point ainsi que les qualités qui leur sont assignées. Les champs de coordonnées affichés dépendent de l’option d’affichage configurée et peuvent être édités uniquement si l’affichage courant est le même que celui du type de coordonnées enregistré.

Grille / Local / WGS-84

Sélectionnez l’option d’affichage des coordonnées pour le point.

NOTE : Si l’option d’affichage est configurée sur WGS-84, les champs de coordonnées affichés dépendent de ce qui est configuré pour le paramètre Afficher WGS-84 sous de la tabulation Unités et format du dialogue Propriétés du projet. Par exemple, si le champ Afficher WGS-84 sous est configuré sur X,Y,Z, le champ des coordonnées de la page Résumé est affiché sous X,Y,Z. Si le champ Afficher WGS-84 sous est configuré sur L,L,H, les champs de coordonnées de la page Résumé sont affichés sous Latitude, Longitude, Hauteur.

Commande Sélecteur de qualité

Utilisez la commande Sélecteur de qualité pour assigner une qualité au composant de coordonnées. Les qualités sont listées dans l’ordre descendant :

  Contrôle

  Levé

  Mappage

 Inconnu

Etat

Affiche l’état du point sélectionné. L’état d’un point est pris en compte par le recalcul, ou lorsqu’un calcul de levé est effectué. Pour modifier ce champ, cliquez sur la flèche et sélectionnez l’option de la liste déroulante que vous souhaitez appliquer.

Désactivez une coordonnée si vous ne souhaitez pas qu’elle soit utilisée pour déduire la position du point.

Activez une coordonnée comme vérification si vous souhaitez que la coordonnée soit utilisée uniquement s’il n’y a aucune autre observation ou coordonnée qui puisse être utilisée pour déduire la position du point.

Le Rapport de recalcul indique les coordonnées activées comme vérifications mais pas les coordonnées désactivées.

Type de coordonnées

Les coordonnées sont enregistrées dans la base de données sous le type de coordonnées que vous sélectionnez. Si vous enregistrez la coordonnée sous un type différent, toute coordonnée existante du même type est écrasée.


Fenêtre Propriétés - Tabulation Levé - Observations GPS

Vous pouvez afficher les informations sur tout type d’observation GPS, tel que RTK, Statique, Statique rapide et vecteurs cinématiques post-traités.

Utilisez la tabulation Levé de la fenêtre Propriétés pour afficher et éditer les informations relatives au levé de l’observation GPS sélectionnée. Les informations sur le levé sont organisées en pages.

Vous pouvez accéder aux pages suivantes en cliquant sur le bouton de menu approprié :

Résumé
Avertissement
Occupation de base
Occupation mobile
Observations
Statistiques d'observation


Fenêtre Propriétés - Tabulation Levé - Page Résumé (Observations GPS)

La page Résumé fait une synthèse des détails d’observation pour l’observation GPS sélectionnée dans la fenêtre Propriétés.

1.	Sélectionnez une observation GPS dans la Vue de levé.

2.	Pour y accéder à partir de la fenêtre Propriétés, cliquez sur le bouton de menu Résumé [image].

De

Affiche le nom du point de référence à partir duquel l’observation est effectuée.

Vers

Affiche le nom du point mobile vers lequel l’observation est effectuée.

Heure

Affiche la durée de l’observation et le moment auquel elle a commencé.

NOTE : Lors de l’importation de données, les heures de Début et de Fin sont converties de temps GPS en TUC (temps universel coordonné − c.-à-d. en temps moyen de Greenwich). Les heures sont ensuite affichées comme heures locales à l’aide de la configuration du fuseau horaire courant de Microsoft Windows.

ASTUCE : Positionnez le pointeur sur le champ pour afficher une Info-bulle sur l’heure d’arrêt de l’occupation.

Type de processeur

Affiche le type de processeur de l’observation sélectionnée. La classification se rapporte au processeur utilisé pour déterminer la solution d’observation. Le champ du type de processeur peut être :

  RTK

  Statique ou Statique rapide

  Cinématique post-traitée

  Non traité − signifie que la solution d’observation peut être déterminée par le processeur de lignes de base mais ne l’a pas été.

Méthode sur le terrain

Affiche la méthode de levé GPS utilisée pour l’observation.

  Topo

  Continu

  Contrôle observé

  Rapide

  Statique ou Statique rapide

  Evénement

  Arrêt et Marche

Etat

Affiche l’état de l’observation sélectionnée. L’état d’une observation est pris en compte par le recalcul, ou lorsqu’un calcul de levé est effectué. Utilisez ce champ pour désactiver de mauvaises observations (telles que celles entraînant une erreur de fermeture).

Pour modifier ce champ, cliquez sur la flèche et sélectionnez l’option de la liste déroulante que vous souhaitez appliquer à l’observation.

Désactivez une observation si vous ne souhaitez pas qu’elle soit utilisée pour déduire la position du point.

Activez une observation comme vérification si vous souhaitez que l’observation soit utilisée uniquement s’il n’y a aucune autre observation ou coordonnée qui puisse être utilisée pour déduire la position du point.

Le Rapport de recalcul indique les observations activées comme vérifications mais pas les observations désactivées.

Utiliser dans Ajustement de réseau

Désélectionnez cette case à cocher si vous ne souhaitez pas que l’observation sélectionnée soit utilisée dans l’ajustement de réseau (si le module Ajustement de réseau est installé sur votre ordinateur), mais qu’elle soit utilisée lorsque la base de données est calculée par un recalcul.

Si une observation ne comporte pas d’informations sur la covariance ni d’informations sur les erreurs standard, cette option est désactivée.

Si l’observation est désactivée, cette option n’est pas disponible.


Fenêtre Propriétés - Tabulation Levé - Page Observations (Observations GPS)

Utilisez la page Observations pour sélectionner une solution active et afficher les composants de ligne de base de l’observation sélectionnée :

1.	Sélectionnez la ligne de base dans la Vue de levé.

2.	Pour y accéder à partir de la fenêtre Propriétés, cliquez sur le bouton de menu Observations [image].

Solution active

Affiche la solution de ligne de base active. Si vous avez traité la ligne de base plus d’une fois (lorsque le module Traitement de lignes de base est installé sur votre ordinateur) en utilisant des styles de traitement différents et que vous avez choisi de ne pas écraser les solutions de ligne de base en double dans le dialogue Traitement GPS, la ligne de base peut comporter plus d’une solution. Cependant, la ligne de base ne peut comporter qu’une seule solution active. Il s’agit de la solution utilisée dans un recalcul et un ajustement de réseau. Cliquez sur la flèche et sélectionnez la meilleure solution de la liste déroulante que vous souhaitez activer pour l’observation.

ASTUCE : Maintenez le pointeur sur les solutions pour afficher une Info-bulle sur les résultats de ligne de base.

Type de solution

Affiche le type de solution d’observation de l’observation sélectionnée.

dX, dY, dZ

Affiche les valeurs delta de la ligne de base GPS en termes de système de coordonnées cartésien WGS-84 lorsque l’option d’affichage est configurée sur XYZ.

Azimut avant SN, Distance d’ellipsoïde, Delta hauteur

Affiche les composants de ligne de base GPS en termes d’ellipsoïde local lorsque l’option d’affichage est configurée sur Géodésique.

NOTE : L’azimut avant SN fait référence à l’azimut (géodésique) avant de section normale dans la direction allant du point de référence au point mobile.

dN, dE, dH

Affiche le delta Nord, delta Est et les Valeurs de hauteur sur l’horizon géodésique local lorsque l’option d’affichage est configurée sur NEH.

Afficher détails de sol au sol

Sélectionnez cette case à cocher pour afficher les valeurs des composants de ligne de base réduites aux hauteurs au sol des marques occupées par les récepteurs aux points mobiles et de référence.

Si la case à cocher est désélectionnée, les valeurs des composants de ligne de base sont indiquées du centre de phase d’antenne de référence au centre de phase d’antenne mobile. La figure suivante illustre cette différence.


Fenêtre Propriétés - Tabulation Levé - Page Observations (Solution de position)

Utilisez la page Observation pour sélectionner la solution active et pour afficher les composants de ligne de base de l’observation sélectionnée:

1.	Sélectionnez la ligne de base dans l’Affichage du levé.

2.	Pour y accéder, dans la fenêtre Propriétés, cliquez sur le bouton du menu Observation [image].

Type de processeur

Affiche la méthode de traitement.  Par exemple, WAAS, ou de superficie large RTK.

Méthode sur le terrain

Affiche la méthode de champ utilisée, par exemple, continu.

Type de solution

Affiche le type de solution d’observation de l’observation sélectionnée, par exemple, L1 flottant.


Fenêtre Propriétés - Tabulation Levé - Page Statistiques d'observation (Observations GPS)

La page Statistiques d’observation affiche les statistiques de la solution appropriée ainsi que les indicateurs de qualité en fonction du type d’observation de la ligne de base. Les statistiques d’une observation RTK sont différentes de celles d’une observation statique/post-traitée.

Utilisez cette page pour vérifier la qualité de la solution.

1.	Sélectionnez la ligne de base dans la Vue de levé.

2.	Pour y accéder à partir de la fenêtre Propriétés, cliquez sur le bouton de menu Statistiques d’observation [image].

Si l’observation affichée est une observation RTK, les champs suivants s’affichent dans la page Statistiques d’observation :

Nombre de mesures

Affiche le nombre de mesures (époques) effectuées pour réaliser l’observation.

Nombre minimum de satellites

Affiche le nombre minimum de satellites poursuivis par les récepteurs de référence et les récepteurs mobiles lors de l’observation.

RMS

Affiche la valeur moyenne quadratique (Root Mean Square : RMS). La valeur RMS indique la précision des mesures utilisées pour déterminer la solution de ligne de base.

Distance inclinée

Affiche la distance inclinée de l’observation.

Précision horizontale

Affiche la précision horizontale des mesures utilisées pour déterminer la solution d’observation.

Précision verticale

Affiche la précision verticale des mesures utilisées pour déterminer la solution d’observation.

PDOP/RDOP

Affiche la valeur de dilution de précision de position (PDOP) pour les lignes de base cinématiques, ou la valeur de dilution de précision relative (RDOP) pour les lignes de base statiques. La RDOP baisse avec le temps.

Si l’observation affichée est une observation post-traitée, les champs suivants s’affichent dans la page Statistiques d’observation :

Ratio

Affiche le ratio des deux variances les mieux adaptées aux mesures et générées dans la recherche de nombres entiers effectuée par le processeur.

Variance de référence

Affiche la variance de référence. La variance de référence est un indicateur du niveau d’adaptation à la solution calculée des données observées pour la ligne de base.

RMS

Affiche la valeur moyenne quadratique. La valeur RMS exprime la précision des mesures utilisées pour déterminer la solution de ligne de base.

Distance inclinée

Affiche la distance inclinée de l’observation.

Précision horizontale

Affiche la précision horizontale des mesures utilisées pour déterminer la solution d’observation.

Précision verticale

Affiche la précision verticale des mesures utilisées pour déterminer la solution d’observation.

RDOP

Affiche la valeur de dilution de précision relative.


Fenêtre Propriétés - Tabulation Levé - Page Statistiques d'observation (Solution de position)

La page Statistiques d’observation affiche les statistiques de solution et le type d’indicateur de qualité appropriés.

Utilisez cette page pour vérifier la qualité de la solution:

1.	Sélectionnez la ligne de base dans l’Affichage du levé.

2.	Pour y accéder, dans la fenêtre Propriétés, cliquez sur le bouton de menu Statistiques d’observations [image].

Précision horizontale

Affiche la précision horizontale des mesures utilisées pour déterminer la solution de position.

Précision verticale

Affiche la précision verticale des mesures utilisées pour déterminer la solution de position.

RMS

Affiche la valeur moyenne quadratique (RMS). La valeur RMS indique la précision des mesures utilisées pour déterminer la solution de ligne de base.

PDOP

Affiche la valeur de Position Dilution of Precision (PDOP) des lignes de base.

Nombre minimum de satellites:

Affiche le nombre minimum de satellites poursuits pendant l’observation.

Nombre de positions GPS

Affiche le nombre de positions GPS utilisées dans la solution.


Fenêtre Propriétés - Tabulation Levé - Observations conventionnelles

Vous pouvez afficher les informations sur toute observation terrestre mesurée sur le terrain à l’aide d’un instrument conventionnel, tel qu’une station totale ou un théodolite.

Utilisez la tabulation Levé de la fenêtre Propriétés pour afficher les informations relatives au levé de l’observation conventionnelle sélectionnée ainsi que pour éditer les détails de configuration. Les informations sur le levé sont organisées en pages.

Vous pouvez accéder aux pages suivantes en cliquant sur le bouton de menu approprié :

Résumé
Configuration de destination
Observations
Statistiques d'observation
NOTE : Dans l’arborescence, une observation conventionnelle est indiquée dans l’ordre suivant : nom du point de visée arrière, nom du Point d’instrument et nom du point de visée avant.


Fenêtre Propriétés - Tabulation Levé - Page Résumé (Observations conventionnelles)

La page Résumé fait une synthèse des détails d’observation pour l’observation conventionnelle sélectionnée dans la fenêtre Propriétés.

1.	Sélectionnez l’observation dans la Vue de levé.

2.	Pour y accéder à partir de la fenêtre Propriétés, cliquez sur le bouton de menu Résumé [image].

Point d'instrument

Affiche le nom du point utilisé pour la configuration d’instrument à partir de laquelle l’observation a été effectuée.

Hauteur d'instrument

Affiche la hauteur de l’instrument.

Point visée arrière

Affiche le nom du point utilisé comme visée arrière.

Point visée avant

Affiche le nom du point de destination vers lequel l’observation est effectuée.

Etat

Affiche l’état de l’observation sélectionnée. L’état d’une observation est pris en compte par le recalcul, ou lorsqu’un calcul de levé est effectué.

Pour modifier ce champ, cliquez sur la flèche et sélectionnez l’option de la liste déroulante que vous souhaitez appliquer à l’observation.

Désactivez une observation si vous ne souhaitez pas qu’elle soit utilisée pour déduire la position du point.

Si l’observation est une vérification d’un point de visée arrière, activez-la comme vérification. Cela permet de garantir qu’une fermeture sera rapportée et que l’observation ne sera pas utilisée pour déterminer la coordonnée.

Le Rapport de recalcul indique les observations activées comme vérifications mais pas les observations désactivées.

Utiliser dans Ajustement de réseau

Si vous ne souhaitez pas que l’observation sélectionnée soit utilisée dans un ajustement de réseau (si le module Ajustement de réseau est installé sur votre ordinateur), mais qu’elle soit quand même utilisée dans un recalcul, désélectionnez cette case à cocher.


Fenêtre Propriétés - Tabulation Levé - Page Observations (Observations conventionnelles)

Utilisez la page Observations pour afficher les composants d’observation entre les configurations d’instrument et de destination de l’observation sélectionnée.

1.	Sélectionnez l’observation dans la Vue de levé.

2.	Pour y accéder à partir de la fenêtre Propriétés, cliquez sur le bouton de menu Observations [image].

Azimut visée arrière

Affiche l’azimut calculé et la qualité entre le point d’instrument et le point de visée arrière.

NOTE : Si le point de visée arrière comporte des coordonnées, celles-ci sont utilisées pour déterminer l’azimut vers le point de visée arrière. S’il existe également un azimut saisi au clavier, la qualité de cet azimut saisi au clavier est comparée à celle de l’azimut calculé à l’aide des coordonnées du point de visée arrière, et l’azimut de la plus haute qualité est utilisé. Si la qualité de l’azimut calculé et celle de l’azimut saisi au clavier sont les mêmes, alors l’azimut calculé est utilisé.

Angle horizontal, Angle vertical, Distance inclinée

Affiche les valeurs observées des composants d’observation lorsque l’option d’affichage est configurée sur AH,AV,DI.

Azimut, Distance horizontale, Delta élévation

Affiche les valeurs des composants d’observation réduits au système de coordonnées de grille lorsque l’option d’affichage est configurée sur AZ,DH,DV.

NOTE : Lors de l’importation, les corrections atmosphériques ainsi que les corrections de réfraction et de courbure sont appliquées à l’observation. L’observation horizontale est réduite à un angle du point de visée arrière. L’angle vertical est corrigé pour éliminer la courbure et la réfraction. La correction PPM est appliquée à la distance inclinée.

Type d'observation

Affiche la face (une, deux) de l’instrument utilisé lorsque l’observation a été effectuée ou si la moyenne d’observations de face une et de face deux a été calculée. Une observation dont la moyenne a été calculée correspond à la moyenne d’une paire face un/face deux, observée entre le point de visée arrière et le point observé avec la même configuration.

L’observation peut également être dérivée d’un angle tourné moyen du Trimble Survey Controller (par exemple lorsqu’une série d’angles est observée).

Afficher détails de sol au sol

Sélectionnez cette case à cocher pour afficher les observations réduites aux hauteurs au sol des marques qui sont occupées par le point d’instrument et le point de visée avant.


Fenêtre Propriétés - Tabulation Levé - Observations de niveau

Vous pouvez afficher des informations sur toute observation de niveau mesurée sur le terrain. Pour afficher les informations relatives au levé d’une observation de niveau sélectionnée, utilisez la tabulation Levé de la fenêtre Propriétés.

Les informations de la tabulation Levé sont organisées en pages. Pour accéder à une page, cliquez sur le bouton de menu approprié comme indiqué ci-dessous :

Résumé
Statistiques d’observation
NOTE : Si vous avez l’intention d’effectuer un ajustement de réseau à l’aide des données d’observation de niveau, des erreurs standard doivent obligatoirement être définies pour les observations de niveau. Des erreurs standard sont calculées dans les cas suivants :

  Si vous importez des données de niveau numérique, l’erreur standard est calculée à l’aide du nombre de tours et de l’erreur de lecture de la mire du dialogue Editer styles d’ajustement (tabulation Contrôles terrestres).

  Lorsque vous importez des données de niveau dans le Format Trimble Data Exchange. Si vous incluez le nombre de tours dans le Format Trimble Data Exchange, l’Erreur standard de delta élévation est calculée sur la base du nombre de tours. Si vous incluez la valeur de l’Erreur standard de delta élévation dans le Format Trimble Data Exchange, cette valeur est utilisée.

Si aucune erreur standard n’est définie ou calculée, les observations de niveau ne sont pas disponibles pour l’ajustement.

Vous pouvez entrer l’erreur standard pour chaque observation dans la fenêtre Propriétés.


Fenêtre Propriétés - Tabulation Levé - Page Résumé (Observations de niveau)

La page Résumé de la fenêtre Propriétés fait une synthèse des détails d’observation d’une observation de niveau sélectionnée.

1.	Sélectionnez l’observation de niveau.

2.	Pour accéder à la page à partir de la fenêtre Propriétés, cliquez sur le bouton de menu Résumé [image].

De point

Affiche le nom du point à partir duquel la série de niveaux a commencé.

Vers point

Affiche le nom du point sur lequel la série de niveaux s’est terminée.

Delta élévation

Affiche le changement d’élévation entre le point « de » et le point « vers ».

Etat

Sélectionnez dans cette liste l’état que vous souhaitez que l’observation ait. Cet état est pris en compte lors des recalculs ou des calculs de levé.

Si l’observation est une vérification d’un point, activez-la comme vérification. Cela permet de garantir qu’une fermeture est rapportée et que l’observation ne sera pas utilisée pour déterminer l’élévation. Si vous ne souhaitez pas que l’observation soit utilisée pour déterminer la position du point, désactivez-la.

Le Rapport de recalcul indique les observations activées comme vérifications mais pas les observations désactivées.


Fenêtre Propriétés - Tabulation Levé - Page Statistiques d'observation (Observations de niveau)

La page Statistiques d’observation permet d’afficher les statistiques et indicateurs de qualité appropriés.

Pour y accéder à partir de la fenêtre Propriétés:

  Dans la tabulation Levé, cliquez sur le bouton de menu Statistiques d’observation [image].

Erreur standard (Delta élév)

Cette valeur correspond à l’Erreur standard de delta élévation L’erreur est calculée lorsque vous :

  importez des fichiers de niveaux numériques (l’Erreur standard de delta élévation est calculée sur la base du nombre de tours et de l’erreur de lecture du jalon).

  importez des données de niveau dans le Format Trimble Data Exchange. Si vous incluez le nombre de tours dans le Format Trimble Data Exchange, l’Erreur standard de delta élévation est calculée sur la base du nombre de tours. Si vous incluez la valeur de l’Erreur standard de delta élévation dans le Format Trimble Data Exchange, cette valeur est utilisée.

Vous pouvez également saisir votre propre erreur standard.

Nombre de tours

Il s’agit du nombre de configurations d’instrument prises pour déterminer la delta élévation entre les deux points.

Longueur de ligne

Il s’agit de la longueur de la boucle de niveau. Elle est calculée en additionnant les distances de visée arrière et de visée avant.


Fenêtre Propriétés - Tabulation Levé - Observations par télémètre à laser

Vous pouvez afficher des informations complètes sur toute observation terrestre mesurée sur le terrain à l’aide d’un télémètre à laser.

Utilisez la tabulation Levé de la fenêtre Propriétés pour afficher et éditer les informations relatives au levé de l’observation par télémètre à laser sélectionnée. Les informations sur le levé sont organisées en pages.

Vous pouvez accéder aux pages suivantes en cliquant sur le bouton de menu approprié :

Résumé
Observations


Fenêtre Propriétés - Tabulation Levé - Page Résumé (Observations par télémètre à laser)

La page Résumé fait une synthèse des détails d’observation pour l’observation par télémètre à laser sélectionnée dans la fenêtre Propriétés.

Pour y accéder à partir de la fenêtre Propriétés :

  Dans la tabulation Levé, cliquez sur le bouton de menu Résumé [image].

Point d'instrument

Indique le point à partir duquel l’observation par télémètre à laser a été effectuée.

Hauteur d'instrument

Indique la hauteur de l’instrument.

Point de destination

Indique le point vers lequel l’observation par télémètre à laser est effectuée.

Hauteur de destination

Indique la hauteur de la destination.

Etat

Affiche l’état de l’observation sélectionnée. L’état d’une observation est pris en compte par le recalcul, ou lorsqu’un calcul de levé est effectué. Pour modifier ce champ, cliquez sur la flèche et sélectionnez l’option de la liste déroulante que vous souhaitez appliquer à l’observation.

Désactivez une observation si vous ne souhaitez pas qu’elle soit utilisée pour déduire la position du point.

Activez une observation comme vérification si vous souhaitez que l’observation soit utilisée uniquement s’il n’y a aucune autre observation ou coordonnée qui puisse être utilisée pour déduire la position du point.

Le Rapport de recalcul indique les observations activées comme vérifications mais pas les observations désactivées.


Fenêtre Propriétés - Tabulation Levé - Page Observations (Observations par télémètre à laser)

Utilisez la page Observations pour afficher les composants d’observation et pour éditer la déclinaison magnétique entre les configurations d’instrument et de destination de l’observation sélectionnée.

Pour y accéder à partir de la fenêtre Propriétés:

  Dans la tabulation Levé, cliquez sur le bouton de menu Observations [image].

Azimut magnétique

Indique l’azimut magnétique observé de l’observation.

Déclinaison

Indique la déclinaison magnétique entre le nord magnétique et le nord vrai. Cette valeur est requise pour calculer l’azimut vrai. Utilisez le clavier pour entrer une valeur afin de modifier ce champ.

Angle vertical

Indique l’angle vertical mesuré lorsque l’observation a été effectuée. Si cette valeur est nulle, il est supposé que la distance est horizontale.

Distance inclinée

Indique la distance inclinée de l’observation.


Fenêtre Propriétés - Tabulation Note

 Vue d'ensemble

Les enregistrements de notes sont des entités non graphiques de la base de données. C’est à dire qu’elles n’apparaissent pas ni ne peuvent être sélectionnées dans la fenêtre graphique.

Utilisez la tabulation Note de la fenêtre Propriétés pour afficher et éditer les enregistrements de notes si vous avez importé une étude du Trimble Survey Controller, qui contient des notes saisies par l’utilisateur.

Pour y accéder,

1.	Effectuez l'une des opérations suivantes :

  Sélectionnez Groupes de sélection. Choisissez le fichier .dc qui contient les notes que vous souhaitez afficher.

  Choisissez Sélectionner / Par interrogation / Notes.

2.	Ouvrez la fenêtre Propriétés.

La fenêtre Propriétés affiche toutes les entités de note avec une icône de crayon [image].


Fenêtre Propriétés - Tabulation CAD (Points)

Utilisez la tabulation CAD de la fenêtre Propriétés pour afficher et éditer les informations relatives au CAD d’un point sélectionné.

Couche

Affiche la couche à laquelle le point sélectionné est assigné. Sélectionnez une couche dans la liste et utilisez le clavier pour saisir un nom de nouvelle couche afin de modifier ce champ.

Style de point

Affichez ou sélectionnez le style de point dans la liste.

Code de caractéristiques

Affichez ou entrez le code de caractéristiques du point sélectionné.

Description

Affichez ou entrez la Description du point sélectionné.


Fenêtre Propriétés - Tabulation Ligne

Utilisez la tabulation Ligne pour afficher et éditer les champs de l’entité de ligne sélectionnée.

Pour y accéder à partir de la Vue en plan, effectuez l'une des opérations suivantes :

  Double-cliquez sur l’entité de ligne.

  Sélectionnez l’entité de ligne dans la fenêtre graphique et ouvrez la fenêtre Propriétés.

De

Affiche le nom du point « de » de la ligne sélectionnée. Utilisez le clavier pour entrer le nom du point afin de modifier ce champ.

Vers

Affiche le nom du point « vers » de la ligne sélectionnée. Utilisez le clavier pour entrer le nom du point afin de modifier ce champ.

Azimut

Affiche l’azimut entre les points « de » et « vers ».

Distance

Affiche la distance entre les points « de » et « vers ».

Couche

Affiche la couche à laquelle la ligne sélectionnée est assignée. Sélectionnez une couche dans la liste et utilisez le clavier pour saisir un nom de nouvelle couche afin de modifier ce champ.

Style de ligne

Affiche le style de ligne de la ligne sélectionnée. Sélectionnez un style de ligne dans la liste pour modifier ce champ.


Fenêtre Propriétés - Tabulation Courbe

Utilisez la tabulation Courbe pour afficher et éditer les champs de l’entité de courbe sélectionnée.

Pour y accéder à partir de la Vue en plan, effectuez l'une des opérations suivantes :

  Double-cliquez sur l’entité de courbe.

  Sélectionnez l’entité de courbe dans la fenêtre graphique et ouvrez la fenêtre Propriétés.

De

Affiche le nom du point « de » de la courbe sélectionnée. Utilisez le clavier pour entrer le nom du point afin de modifier ce champ.

Vers

Affiche le nom du point « vers » de la courbe sélectionnée. Utilisez le clavier pour entrer le nom du point afin de modifier ce champ.

Tangente d’entrée, Tangente de sortie

Affiche les tangentes de la courbe sélectionnée. Utilisez le clavier pour entrer une valeur afin de modifier ces champs.

Coefficient de lissage

Définit le degré de courbure de la courbe sélectionnée. Entrez une valeur entre 0 et 1 pour modifier ce champ.

Couche

Affiche la couche à laquelle l’entité de courbe sélectionnée est assignée. Sélectionnez une couche dans la liste et utilisez le clavier pour saisir un nom de nouvelle couche afin de modifier ce champ.

Style de ligne

Affiche le style de ligne de la courbe sélectionnée. Sélectionnez un style de ligne dans la liste pour modifier ce champ.


Fenêtre Propriétés - Tabulation Arc

Utilisez la tabulation Arc pour afficher et éditer les champs de l’arc sélectionné.

Pour y accéder à partir de la Vue en plan, effectuez l'une des opérations suivantes :

  Double-cliquez sur l’entité d’arc.

  Sélectionnez l’entité d’arc dans la fenêtre graphique et ouvrez la fenêtre Propriétés.

De

Affiche le nom du point « de » de l’arc sélectionné. Utilisez le clavier pour entrer le nom du point afin de modifier ce champ.

Vers

Affiche le nom du point « vers » de l’arc sélectionné. Utilisez le clavier pour entrer le nom du point afin de modifier ce champ.

Centre

Affiche le nom du point « central » de l’arc sélectionné. Utilisez le clavier pour entrer le nom du point afin de modifier ce champ.

Nord, Est

Affiche les coordonnées au point « central » de l’arc. Utilisez le clavier pour entrer des valeurs afin de modifier ces champs.

NOTE : Si le point central a été défini pour l’arc, les coordonnées ne sont pas utilisées. Le point central est utilisé.

Couche

Affiche la couche à laquelle l’entité d’arc sélectionnée est assignée. Sélectionnez une couche dans la liste et utilisez le clavier pour saisir un nom de nouvelle couche afin de modifier ce champ.

Style de ligne

Affiche le style de ligne de l’arc sélectionné. Sélectionnez un style de ligne dans la liste pour modifier ce champ.

L'arc est inférieur à 180 degrés.

Cette case à cocher est sélectionnée par défaut. Désélectionnez cette case à cocher pour que l’angle de l’arc sélectionné soit supérieur à 180 degrés.


Fenêtre Propriétés - Tabulation Annotation de point

Utilisez la tabulation Annotation de point pour afficher et éditer les champs de l’entité d’annotation sélectionnée assignée à un point.

Pour y accéder à partir de la Vue en plan, effectuez l'une des opérations suivantes :

  Double-cliquez sur l’entité d’annotation.

  Sélectionnez l’entité d’annotation dans la fenêtre graphique et ouvrez la fenêtre Propriétés.

Point

Affiche le nom du point auquel l’annotation sélectionnée est assignée.

Expression de texte

Affiche l’expression de texte de l’entité d’annotation sélectionnée. L’expression de texte peut être une définition de texte ou un code de champ inséré dans un modèle d’annotation. Utilisez le clavier pour entrer une valeur afin de modifier ce champ.

Est, Nord

Affiche les coordonnées relatives à la position d’ancrage par défaut de la « ligne de base gauche » de l’entité d’annotation sélectionnée. Utilisez le clavier pour entrer des valeurs afin de modifier ces champs.

Orientation

Affiche l’orientation de l’entité d’annotation. Utilisez le clavier pour entrer une valeur afin de modifier ce champ.

Couche

Affiche la couche à laquelle l’entité d’annotation sélectionnée est assignée. Sélectionnez une couche dans la liste et utilisez le clavier pour saisir un nom de nouvelle couche afin de modifier ce champ.

Style de texte

Affiche le style de texte de l’entité d’annotation sélectionnée. Sélectionnez un style de texte dans la liste pour modifier ce champ.


Fenêtre Propriétés - Tabulation Annotation de ligne

Utilisez la tabulation Annotation de ligne pour afficher et éditer les champs de l’entité d’annotation sélectionnée assignée à une entité de ligne.

Pour y accéder à partir de la Vue en plan, effectuez l'une des opérations suivantes :

  Double-cliquez sur l’entité d’annotation.

  Sélectionnez l’entité d’annotation dans la fenêtre graphique et ouvrez la fenêtre Propriétés.

Ligne

Affiche les noms des points utilisés pour définir la ligne à laquelle l’entité d’annotation sélectionnée est assignée.

Expression de texte

Affiche l’expression de texte de l’entité d’annotation sélectionnée. L’expression de texte peut être une définition de texte ou un code de champ inséré dans un modèle d’annotation. Utilisez le clavier pour entrer une valeur afin de modifier ce champ.

Est, Nord

Affiche les coordonnées relatives à la position d’ancrage par défaut de la « ligne de base gauche » de l’entité d’annotation sélectionnée. Utilisez le clavier pour entrer des valeurs afin de modifier ces champs.

Orientation

Affiche l’orientation de l’entité d’annotation. Utilisez le clavier pour entrer une valeur afin de modifier ce champ.

Couche

Affiche la couche à laquelle l’entité d’annotation sélectionnée est assignée. Sélectionnez une couche dans la liste et utilisez le clavier pour saisir un nom de nouvelle couche afin de modifier ce champ.

Style de texte

Affiche le style de texte de l’entité d’annotation sélectionnée. Sélectionnez un style de texte dans la liste pour modifier ce champ.


Fenêtre Propriétés - Tabulation Annotation d'arc

Utilisez la tabulation Annotation d’arc pour afficher et éditer les champs de l’entité d’annotation sélectionnée assignée à une entité d’arc.

Pour y accéder à partir de la Vue en plan, effectuez l'une des opérations suivantes :

  Double-cliquez sur l’entité d’annotation.

  Sélectionnez l’entité d’annotation dans la fenêtre graphique et ouvrez la fenêtre Propriétés.

Arc

Affiche les noms des points utilisés pour définir l’arc auquel l’entité d’annotation sélectionnée est assignée.

Expression de texte

Affiche l’expression de texte de l’entité d’annotation sélectionnée. L’expression de texte peut être une définition de texte ou un code de champ inséré dans un modèle d’annotation. Utilisez le clavier pour entrer une valeur afin de modifier ce champ.

Est, Nord

Affiche les coordonnées relatives à la position d’ancrage par défaut de la « ligne de base gauche » de l’entité d’annotation sélectionnée. Utilisez le clavier pour entrer des valeurs afin de modifier ces champs.

Orientation

Affiche l’orientation de l’entité d’annotation. Utilisez le clavier pour entrer une valeur afin de modifier ce champ.

Couche

Affiche la couche à laquelle l’entité d’annotation sélectionnée est assignée. Sélectionnez une couche dans la liste et utilisez le clavier pour saisir un nom de nouvelle couche afin de modifier ce champ.

Style de texte

Affiche le style de texte de l’entité d’annotation sélectionnée. Sélectionnez un style de texte dans la liste pour modifier ce champ.


Fenêtre Propriétés - Tabulation Texte

Utilisez la tabulation Texte pour afficher et éditer les champs de l’entité de texte sélectionnée.

Pour y accéder à partir de la Vue en plan, effectuez l'une des opérations suivantes :

  Double-cliquez sur l’entité de texte.

  Sélectionnez l’entité de texte dans la fenêtre graphique et ouvrez la fenêtre Propriétés.

Expression de texte

Affiche l’expression de texte de l’entité de texte sélectionnée. L’expression de texte peut être une définition de texte ou un code de caractéristiques inséré. Utilisez le clavier pour entrer une valeur afin de modifier ce champ.

Est, Nord

Affiche les coordonnées relatives au type d’ancrage spécifié pour l’entité de texte sélectionnée. Utilisez le clavier pour entrer des valeurs afin de modifier ces champs.

Orientation

Affiche l’orientation de l’entité de texte. Utilisez le clavier pour entrer une valeur afin de modifier ce champ.

Couche

Affiche la couche à laquelle l’entité de texte sélectionnée est assignée. Sélectionnez une couche dans la liste et utilisez le clavier pour saisir un nom de nouvelle couche afin de modifier ce champ.

Style de texte

Affiche le style de texte de l’entité de texte sélectionnée. Sélectionnez un style de texte dans la liste pour modifier ce champ.


Fenêtre Propriétés - Tabulation Attributs

Utilisez la tabulation Attributs de la fenêtre Propriétés pour gérer les attributs de votre projet. La fenêtre Propriétés vous permet d’afficher, d’éditer, d’ajouter et de supprimer des attributs dans Trimble Geomatics Office.

Pour y accéder à partir de la fenêtre Propriétés, sélectionnez la tabulation Attributs.

NOTE : Vous devez configurer le projet pour des attributs pour que la tabulation Attributs soit disponible.

Outil Sélection des attributs

L’outil Sélection des attributs affiche l’attribut courant sélectionné pour le point. Si plus d’un attribut est associé au point, cliquez sur la flèche et sélectionnez l’attribut de la liste déroulante que vous voulez afficher.

Nom d'attribut

Affiche les noms de toute propriété d’attribut sélectionnée associée au point sélectionné.

Valeur

Affiche les valeurs de toute propriété d’attribut sélectionnée associée au point sélectionné. Pour éditer une valeur d’attribut, cliquez sur le champ approprié et sélectionnez une valeur dans la liste, ou utilisez le clavier pour saisir une valeur dans le champ si la liste de menu n’est pas disponible.


Fenêtre Propriétés - Tabulation Général

La tabulation Général de la fenêtre Propriétés s’affiche lorsque vous ouvrez la fenêtre Propriétés sans qu’aucune entité ne soit sélectionnée, ou lorsque plus d’une entité est sélectionnée dans la liste de sélection de la fenêtre Propriétés.

La tabulation Général affiche le message « Aucune propriété disponible ».

Pour afficher les propriétés d’une entité, sélectionnez une entité dans la fenêtre graphique ou dans la liste de sélection de la fenêtre Propriétés.


Fenêtre Propriétés - Tabulation Levé - Observations réduites

Vous pouvez afficher des informations complètes sur les observations terrestres réduites, telles que les déports, qui ont été mesurées sur le terrain à l’aide du Trimble Survey Controller.

Utilisez la tabulation Levé de la fenêtre Propriétés pour afficher et éditer les informations relatives au levé d’une observation sélectionnée. Les informations sur le levé sont organisées en pages.

Pour accéder à une page de la tabulation Levé, cliquez sur le bouton de menu. Il est possible d’accéder aux pages suivantes :

Résumé
Observations


Fenêtre Propriétés - Tabulation Levé - Page Résumé (Observations réduites)

La page Résumé fait une synthèse des détails de l’observation réduite sélectionnée dans la fenêtre Propriétés.

Pour y accéder à partir de la fenêtre Propriétés:

  Dans la tabulation Levé, cliquez sur le bouton de la page Résumé [image].

De

Affiche le nom du point à partir duquel la mesure a été effectuée.

Vers

Affiche le nom du point vers lequel la mesure a été effectuée.

Etat

Affiche l’état de l’observation de déport sélectionnée. L’état d’une observation est pris en compte par le recalcul, ou lorsqu’un calcul de levé est effectué. Pour modifier ce champ, cliquez sur la flèche et sélectionnez l’option de la liste déroulante que vous souhaitez appliquer à l’observation.


Fenêtre Propriétés - Tabulation Levé - Page Observations (Observations réduites)

Utilisez la page Observations pour les composants d’observation de l’observation réduite.

Pour y accéder à partir de la fenêtre Propriétés:

  Dans la tabulation Levé, cliquez sur le bouton de menu Observations [image].

Azimut

Indique l’azimut observé de l’observation.

Distance horizontale

Indique la distance horizontale mesurée de l’observation.

Distance verticale

Indique la distance verticale mesurée de l’observation.


Coordonnées - Vue d'ensemble

Plus amples informations

 Voir aussi

Lorsque vous saisissez des coordonnées connues pour un point dans un projet Trimble Geomatics Office, une coordonnée est enregistrée avec les autres données de levé du point. Chaque point peut comporter au maximum une coordonnée WGS-84 entrée par l’utilisateur et une coordonnée de grille/locale entrée par l’utilisateur.

Si vous éditez une coordonnée existante entrée par l’utilisateur, l’ancienne coordonnée est remplacée par la nouvelle.

Vous pouvez entrer des coordonnées pour les points existant à l’aide :

  De la fenêtre Propriétés / bouton Ajouter nouvelle coordonnée

  Du dialogue Points (si le module Ajustement de réseau est installé sur votre ordinateur)

  De l’importation d’un fichier comprenant des coordonnées pour des points qui existent déjà dans la base de données.

Vous pouvez entrer des coordonnées pour de nouveaux points avec des coordonnées saisies au clavier à l’aide :

  Du dialogue Insérer points

  De l’importation d’un fichier comprenant des coordonnées pour des points qui n’existent pas déjà dans la base de données.

Toutes les coordonnées entrées dans Trimble Geomatics Office sont ajoutées comme coordonnées sous-jacentes du point et peuvent être affichées dans des pages Coordonnées séparées de la fenêtre Propriétés.

NOTE : Pour afficher la position d’un point, utilisez la page Résumé. La coordonnée entrée est utilisée pour positionner le point s’il a la même qualité (ou une qualité supérieure) que celle des observations vers le point.


Fenêtre Propriétés - Tabulation Implantation - Vue d'ensemble

La tabulation Implantation affiche des informations sur des points qui ont été implantés sur le terrain à l’aide du Trimble Survey Controller. Les pages suivantes sont disponibles à partir de la tabulation Implantation :

  Delta implantés

  Détails linéaires

  Point d’entrée en terre

  Eléments géométriques

  MNT


Fenêtre Propriétés - Tabulation Implantation - Page Delta implantés

Utilisez la page Delta implantés pour afficher la différence entre la position du point de projet et celle du point implanté.

Pour y accéder à partir de la fenêtre Propriétés:

  Dans la tabulation Levé, cliquez sur le bouton de menu Delta implantés [image].

Méthode

Ce champ indique l’élément implanté. Par exemple, un point, une ligne, un arc ou une route.

Si l’option Polaire est sélectionnée, les champs suivants s’affichent :

Azimut

Ce champ indique l’azimut du point de projet au point implanté.

Longueur

Ce champ indique la distance entre le point de projet et le point implanté.

Si l’option Delta de grille est sélectionnée, les champs suivants s’affichent :

Delta nord

Ce champ indique la différence de nord entre le point de projet et le point implanté.

Delta est

Ce champ indique la différence d’est entre le point de projet et le point implanté.

Si l’option Station et déport est sélectionnée, les champs suivants s’affichent :

Station delta

Ce champ indique la différence de distance (parallèlement à la ligne centrale de projet) entre le point de projet et le point implanté.

Déport delta

Ce champ indique la différence de distance (à la perpendiculaire de la ligne centrale de projet) entre le point de projet et le point implanté.

Delta élévation

Ce champ indique la différence d’élévation entre le point de projet (au point à implanter) et le point implanté.

Déblai/Remblai

Ce champ indique la différence d’élévation entre le point de projet (au point qui a effectivement été implanté) et le point implanté.

Les valeurs de delta élévation et de déblai/remblai sont les mêmes à moins que :

  La surface de projet ne soit pas un plan horizontal, et

  Il y ait une erreur significative dans l’erreur horizontale lors l’implantation du point

La figure suivante illustre la différence entre la delta élévation et le déblai/remblai lorsque la surface de projet n’est pas un plan horizontal.

Si le point implanté était un point d’entrée en terre, alors les delta reportés sont de la position implantée au point d’entrée en terre calculé.


Fenêtre Propriétés - Tabulation Implantation - Détails linéaires

Utilisez la page Détails linéaires pour afficher les détails d’implantation d’une caractéristique linéaire (par exemple une ligne, un arc ou une route).

Pour y accéder à partir de la fenêtre Propriétés:

  Dans la tabulation Levé, cliquez sur le bouton de menu Détails linéaires [image].

Nom

Ce champ indique le nom de la caractéristique linéaire à partir de laquelle le point est implanté.

Station

Ce champ indique la station de projet du point sélectionné pour implantation.

Déport horizontal

Ce champ indique le déport horizontal du point de projet au point implanté. Pour une route, ce champ indique le déport horizontal de construction.

Déport vertical

Ce champ indique le déport vertical du point de projet au point implanté. Pour une route, ce champ indique le déport vertical de construction.

Si une route a été implantée, les champs suivants sont également affichés :

Déport de projet

Ce champ indique le déport de projet du point à implanter.

Elévation de projet

Ce champ indique l’élévation de projet du point à implanter.

Nom de caract.

Ce champ indique le nom de caractéristique.

Méthode de déport horz

Ce champ indique comment le déport horizontal a été appliqué. Le déport horizontal peut être appliqué :

  horizontalement

  en suivant la pente de l’élément précédent du modèle.


Fenêtre Propriétés - Tabulation Implantation - Point d'entrée en terre

Utilisez la page Point d’entrée en terre pour afficher les détails d’un point d’entrée en terre. Cette option est disponible uniquement lorsque le point implanté était un point d’entrée sur une route.

Pour y accéder à partir de la fenêtre Propriétés:

  Dans la tabulation Levé, cliquez sur le bouton de menu Point d’entrée en terre [image].

Le point auquel une surface de projet et la surface naturelle se croisent est connu comme le point d’entrée en terre. Le diagramme suivant montre la coupe transversale typique d’une route :

Où :

A = Distance jusqu’à la ligne centrale

B = Distance horizontale du point d’entrée en terre au point de carrefour

C = Distance verticale du point d’entrée en terre au point de carrefour

D = Talus

E = Distance inclinée jusqu’au point de carrefour

F = Déport horizontal de construction

G = Largeur fossé en déblai

H = Point de carrefour

J = Point d’entrée en terre

Talus

Ce champ indique la valeur du talus.

Largeur fossé en déblai

Ce champ indique la valeur de la largeur du fossé en déblai. Elle correspond à « G » dans le diagramme ci-dessus.

Type de talus

Ce champ indique le type de talus (par exemple déblai ou remblai).

Distance horz à la ligne centrale

Ce champ indique la distance horizontale du point d’entrée en terre implanté à la ligne centrale.

Elévation

Ce champ indique l’élévation implantée du point d’entrée en terre.

Distance horizontale au carrefour

Ce champ indique la distance horizontale du point d’entrée en terre au carrefour.

Distance verticale au carrefour

Ce champ indique la distance verticale du point d’entrée en terre au carrefour.

Distance inclinée au carrefour

Ce champ indique la distance inclinée du point d’entrée en terre au carrefour.

Dist inclinée + déport construct

Ce champ indique la distance inclinée du point implanté au point de carrefour. Dans le diagramme ci-dessus, elle correspond à la distance inclinée du point de carrefour « H » à un point défini. Ce point est défini en projetant le talus jusqu’à ce qu’il croise la position implantée. La valeur est nulle (?) si :

  aucun déport horizontal de construction n’est appliqué

  le déport horizontal de construction est appliqué horizontalement.


Fenêtre Propriétés - Tabulation Implantation - Eléments géométriques

Utilisez la page Eléments géométriques pour afficher les distances du point d’entrée en terre aux points sur le modèle.

Pour y accéder à partir de la fenêtre Propriétés:

  Dans la tabulation Levé, cliquez sur le bouton de menu Eléments géométriques [image].

Nom

Ce champ indique le nom du point sur le modèle.

Distance horizontale du point d'entrée en terre

Ce champ indique la distance horizontale du point d’entrée en terre au point sur le modèle.

Distance verticale du point d'entrée en terre

Ce champ indique la distance verticale du point d’entrée en terre au point sur le modèle.


Fenêtre Propriétés - Tabulation Implantation - MNT

Utilisez la page MNT pour afficher les détails sur les points MNT implantés.

Pour y accéder à partir de la fenêtre Propriétés:

  Dans la tabulation Levé, cliquez sur le bouton de menu MNT [image].

Nom de MNT

Ce champ indique le nom du MNT.

Déport vertical

Ce champ indique le déport vertical spécifié entre le MNT et la surface à implanter.

Déblai/Remblai

Ce champ indique la différence d’élévation entre le MNT (y compris le déport vertical) et le point implanté.


Fenêtre Propriétés - Tabulation Levé - Observations d'azimut

Utilisez la tabulation Levé de la fenêtre Propriétés pour afficher et éditer les observations d’azimut. Les informations sur le levé sont organisées en pages.

Vous pouvez accéder à une page en cliquant sur le bouton de menu approprié comme indiqué ci-dessous :

Résumé
Avertissement
Statistiques d’observation


Fenêtre Propriétés - Tabulation Levé - Page Résumé (Observations d'azimut)

La page Résumé fait une synthèse des détails d’observation de l’observation d’azimut sélectionnée dans la fenêtre Propriétés.

1.	Sélectionnez l’observation d’azimut dans la Vue de levé.

2.	Pour y accéder à partir de la fenêtre Propriétés, cliquez sur le bouton de menu Résumé [image].

NOTE : Les observations d’azimut sont précédées du préfixe AZ.

De point

Affiche le nom du point de référence à partir duquel l’observation est effectuée.

Vers point

Affiche le nom du point vers lequel l’observation d’azimut est effectuée.

Azimut

Ce champ affiche l’azimut entre les points « de » et « vers ». Un azimut de grille ou un azimut avant de section normale peut être affiché. Ce champ ne peut être édité que lorsque les champs Type d’azimut et Affichage d’azimut sont les mêmes.

Qualité

Ce champ indique la qualité de l’observation d’azimut.

Type d'azimut

Ce groupe indique si l’azimut est enregistré dans la base de données comme azimut de grille ou comme azimut avant de section normale. Vous ne pouvez pas éditer ce champ.

NOTE : Si le point « vers » ne comporte pas d’azimut, la conversion entre l’Azimut de grille et l’Azimut avant de section normale ne peut pas être calculée. Dans ce cas, l’Azimut s’affiche comme Nul (?).

Affichage d’azimut

Utilisez ces options d’affichage pour choisir comment l’azimut est affiché dans la fenêtre Propriétés.

  Grille − sélectionnez cette option pour afficher l’Azimut de grille

  Avant section normale − sélectionnez cette option pour afficher l’Azimut avant de section normale.

NOTE : Pour que vous puissiez éditer le champ Azimut, les informations du champ Type d’azimut et de ces options d’affichage doivent être identiques.

Etat

Sélectionnez dans cette liste l’état que vous souhaitez pour l’observation. Cet état est pris en compte lors des recalculs ou des calculs de levé.

Si vous souhaitez qu’une observation soit utilisée pour déterminer la position d’un point lorsqu’il n’y a aucune autre observation ou coordonnée disponible, activez cette observation comme vérification. Si vous ne souhaitez pas que l’observation soit utilisée pour déterminer la position du point (par exemple si l’observation provoque une erreur de fermeture), désactivez-la.

Le Rapport de recalcul indique les observations activées comme vérifications mais pas les observations désactivées.

Utiliser dans Ajustement de réseau

Désélectionnez cette case à cocher si vous ne souhaitez pas que l’observation sélectionnée soit utilisée dans l’ajustement de réseau (si le module Ajustement de réseau est installé sur votre ordinateur), mais qu’elle soit utilisée lorsque la base de données fait l’objet d’un recalcul.


Fenêtre Propriétés - Tabulation Levé - Page Statistiques d'observation (Observations d'azimut)

La page Statistiques d’observation affiche les statistiques de la solution appropriée ainsi que les indicateurs de qualité en fonction du type d’observation sélectionné. Dans le cas d’une observation d’azimut, l’erreur standard est indiquée. Utilisez cette page pour vérifier la qualité de la solution.

Pour y accéder à partir de la fenêtre Propriétés :

  Dans la tabulation Levé, cliquez sur le bouton de menu Statistiques d’observation [image].

Erreur standard

Utilisez ce champ pour entrer l’erreur standard de l’observation d’azimut.

NOTE : Une erreur standard de 0°00'00” fixe l’azimut en ajustement.


Coordonnées dans la fenêtre Propriétés

Trimble Geomatics Office vous permet d’ajouter des coordonnées à un point de la fenêtre Propriétés à l’aide du bouton Ajouter coordonnée [image]. Lorsque vous cliquez sur ce bouton, un menu de raccourcis s’affiche. A partir de ce menu, sélectionnez le type de coordonnées que vous devez ajouter, c’est à dire WGS-84, Grille ou Local. Les coordonnées enregistrées pour un point, ainsi que la source de la coordonnée, sont affichées dans la liste arborescente gauche de la fenêtre Propriétés.

Le nombre de coordonnées saisies pouvant être associées à un point est limité. Vous pouvez saisir seulement une coordonnée WGS-84 et une coordonnée de grille/locale pour chaque point. Si vous avez déjà ajouté une coordonnée WGS-84, l’option coordonnée WGS-84 n’est plus disponible.

Ajoutez une coordonnée uniquement lorsque vous êtes sûr que vous voulez utiliser cette coordonnée plutôt que des observations, car les recalculs utilisent les coordonnées saisies au clavier plutôt que les observations de la même qualité. Cela permet de garantir qu’un point de projet est utilisé plutôt qu’une observation implantée.

NOTE : Un préfixe est ajouté au nom de la coordonnée saisie pour indiquer si la coordonnée est une coordonnée de grille (G), locale (L) ou WGS-84 (W).

Des coordonnées supplémentaires peuvent être générées par Trimble Geomatics Office et enregistrées pour un point. Ces coordonnées peuvent inclure :

  des coordonnées dérivées d’un ajustement

  des coordonnées WGS-84 ajustées enregistrées pour la calibration

Lorsque vous ajoutez une coordonnée, les composants de coordonnées s’affichent comme nuls (?). Vous pouvez utiliser le bouton Insérer valeur courante [image] pour remplir chaque champ de composant de coordonnées avec la valeur et la qualité courantes du composant de coordonnées telles qu’elles sont indiquées dans la page Résumé. Ce bouton est disponible uniquement si vous avez sélectionné le champ du composant de coordonnées que vous souhaitez remplir.

Edition des coordonnées

Vous pouvez éditer des coordonnées saisies. Cependant, vous ne pouvez pas éditer de coordonnées supplémentaires, telles que les coordonnées d’ajustement et de calibration que le logiciel crée.

Vous ne pouvez éditer les composants de coordonnées que si l’affichage courant est identique au paramètre Enregistrer sous type de coordonnées. Lorsque des composants de coordonnées sont édités, la source est modifiée et passe à saisie au bureau, ou sources multiples si seulement certains des composants sont édités.

Coordonnées de sources multiples

La source d’une coordonnée est indiquée dans la fenêtre Propriétés. Si une coordonnée a été dérivée de deux sources ou plus, elle est affichée comme coordonnée de sources multiples. Par exemple, si la coordonnée horizontale d’un point est dérivée d’un fichier importé et que la hauteur a été entrée dans Trimble Geomatics Office, le point est affiché comme comportant des coordonnées dérivées de sources multiples.

ASTUCE : Vous pouvez utiliser l’Info-bulle de la page de coordonnées pour examiner les sources de chacun des composants de coordonnées.

Déplacement des coordonnées

Lorsqu’une coordonnée est déplacée ou transformée dans Trimble Geomatics Office, toutes les observations sont désactivées, vous devez donc les réactiver si vous souhaitez calculer les coordonnées une nouvelle fois. Pour cela, utilisez le dialogue Edition multiple.

Suppression des coordonnées

Pour supprimer une coordonnée, utilisez le bouton Supprimer [image].

NOTE : Les coordonnées fixées pour ajustement ne peuvent pas être éditées.

Qualité des coordonnées

Une qualité de coordonnées peut être assignée à chaque coordonnée de la fenêtre Propriétés. Les qualités de coordonnées suivantes peuvent être éditées :

  Qualité Contrôle

  Qualité Levé

  Qualité Mappage

  Qualité Inconnu

Vous ne pouvez pas modifier la qualité d’un point de contrôle fixe.


Fenêtre Propriétés - Tabulation Levé - Page Evénements

Utilisez la tabulation Levé de la fenêtre Propriétés pour afficher des données d’événement.

Pour y accéder à partir de la fenêtre Propriétés :

  Dans la tabulation Levé, cliquez sur le bouton de menu Evénements [image].

ID événement de récepteur

Ce champ indique l’ID de l’événement.

Méthode d'interpolation

Ce champ indique le type de méthode d’interpolation, à savoir linéaire ou quadratique.

Points entrés

Ce champ indique le nombre de points utilisés pour l’adaptation à l’arc d’interpolation.

ID port de récepteur

Ce champ indique l’ID du port de récepteur.

Type de solution

Ce champ indique le type de solution.

RMS

Indique la valeur RMS.

Etat

Sélectionnez dans cette liste l’état que vous souhaitez pour l’observation. Cet état est pris en compte lors des recalculs ou des calculs de levé.


Fenêtre Propriétés - Tabulation Station et Déport

La tabulation Station et Déport affiche les informations concernant les points mesurés utilisant un programme Station et Deport dans un périphérique Trimble.

Dans la fenêtre Propriétés:

1.	Sélectionnez la tabulation Station et Déport.

2.	Cliquez sur le bouton Station/Déport.

Nom

Ce champ contient le nom de la caractéristique linéaire.

Station

Ce champ contient la station sur la caractéristique linéaire pour le point mesuré.

NOTE: Pour l’instrument Trimble 3300, s’il y a un déport de construction utilisé de la ligne de référence, le déport est présenté dans le champ de nom à côté du nom de point de début et de fin de la ligne.

Déport

Ce champ contient le déport de la caractéristique linéaire pour le point mesuré.

Elévation delta

Ce champ contient la différence en élévation entre la station sur la caractéristique linéaire et le point mesuré.


    Calibration du site GPS - Vue d'ensemble

Plus amples informations

Comment ...	Dialogue

Vue d'ensemble

Le procédé de calibration du site GPS établit le rapport entre les données WGS-84 collectées par les récepteurs GPS et les positions de contrôle locales (exprimées sous forme d'une grille de carte locale avec des élévations au-dessus du niveau de la mer).  Ce rapport est défini par une série de transformations mathématiques. La fonctionnalité de la calibration du site GPS vous permet d'apparier les points de contrôle locaux et GPS à utiliser dans la calibration. (Les coordonnées GPS doivent être dérivées des points et des observation GPS et les points de grille doivent être dérivés des points de grille et des observations terrestres). Ensuite Trimble Geomatics Office les calcule et applique des transformations mathématiques utilisant les moindres carrées. Vous pouvez récalibrer à tout moment.

Les transformations mathématiques appliquées afin de convertir les positions WGS-84 aux coordonnées de grille sont:

1.	Une transformation de datum pour convertir les coordonnées WGS-84 de latitude, de longitude et de hauteur ellipsoïdale aux coordonnées de latitude, de longitude et de hauteur ellipsoïdale relatives à l'ellipsoïde de la grille de carte locale.

2.	Une projection de carte pour convertir les coordonnées locales de l'ellipsoïde, de latitude et de longitude aux coordonnées du Nord et de l'Est de grille de carte locale (la valeur de la hauteur n'est pas altérée dans ce procédé).

3.	Un modèle de géoïde à la hauteur WGS-84 pour obtenir une élévation approximative au-dessus du niveau de la mer.

4.	Un ajustement horizontal des coordonnées de grille transformées pour mieux s'adapter aux données de contrôle locales. Cet ajustement tient compte de toute variation locale dans le système de projection qu'il n'est pas possible de satisfaire dans la transformation de datum en général.

5.	Un ajustement de hauteur pour convertir les hauteurs au-dessus de l'ellipsoïde local ou les élévations dérivées du géoïde aux élévations de contrôle locales au-dessus du niveau de la mer.

Pour changer les paramètres de calibration (qui définissent les résidus les plus hauts permis), utilisez le registre de l'ordinateur.

Les ajustements horizontal et vertical ci-dessus sont stockés comme partie de la définition du système de coordonnées pour le projet. Tous les points GPS dans la base de données sont mis à jour utilisant les paramètres de calibration avec le résultat des valeurs de coordonnées de grille locale plus précises.

Pour enregistrer la nouvelle définition du système de coordonnées (y compris les paramètres de calibration) utilisez un site à utiliser dans des projets au futur dans la même zone, cliquez sur Enregistrer sous site du dialogue Calibration.

Si vous enregistrez une calibration sous forme d'un site avec l'intention d'utiliser le site dans un autre projet, vérifiez que la surface du projet soit entourée complètement par les points utilisés dans la calibration. Par exemple, dans le diagramme suivant, il est valide d'utiliser la définition du site pour le projet B, mais pas pour le projet C.

Si vous utilisez la touche [image] dans Trimble Survey Controller pour commencer une base cinématiques en temps réel (RTK), et transférez le fichier Survey Controller (.dc) à Trimble Geomatics Office, la position de base (et donc tous les points du mobile à partir de la base) sont d'une qualité inconnue (pour tous les composants, le horizontal, la hauteur et l'élévation).

Si vous êtes sûr que la calibration du site GPS a amélioré la qualité des points, utilisez la fenêtre Propriétés pour changer la qualité du point de base.

NOTE: Par défaut, si vous effectuez une calibration du site GPS utilisant un point de station de base RTK avec des coordonnées WGS-84, les coordonnées de grille pour le points seront désactivées. Cependant, vous pouvez utiliser la position de grille pour une calibration de site GPS. Si vous choisissez de le faire, vous devez activer la coordonnée de grille (vérifiez qu'elle a une qualité plus haute que la coordonnée WGS-84), explosez les points par type de données et puis calibrez à nouveau. Si vous ne l'effectuez pas et les vecteurs GPS proviennent de la position de grille:

  aucun point GPS dérivé de la base ne peut être utilisé dans la calibration

  il se peut qu'une erreur, égale à l'erreur résiduelle de calibration est introduite.

NOTE: Si vous avez effectué une calibration du site GPS et vous sélectionnez la case à cocher Utiliser coordonnées au sol dans le dialogue Paramètres du site local, la calibration que vous avez défini dans le projet devient non valide. Un message d'erreur vous avertit que de continuer avec l'opération supprimera la calibration.

Utilisation des modèles de géoïde

Si le système de coordonnées du projet définit l'utilisation d'un modèle de géoïde pour la détermination des élévations de points, alors les élévations sont déterminées directement à partir des hauteurs WGS-84 par l'interpolation sur la grille de modèle de géoïde. Cependant, il est toujours possible d'appliquer un ajustement de hauteur sur les élévations produites par un modèle de géoïde afin de tenir en compte des petites variations locales qu'un modèle de géoïde à grande échelle ne peut pas tenir en compte. Les élévations déterminées à partir d'une calibration verticale sont assignées une qualité de levé.

Règles pour assurer une calibration utile

  Les coordonnées WGS-84 doivent être assez correctes.

Il est possible de générer des points GPS autonomes. Cependant, vous ne devriez pas utiliser plus d'un point WGS-84 autonome dans une calibration, par exemple vérifiez que seulement une seule station de base a été installée utilisant la touche [image] dans Trimble Survey Controller. D'autres stations de base devraient être installées sur des positions mesurées par un vecteur GPS par rapport à la station de base autonome. Ainsi le rapport entre les deux est défini et vous permet d'effectuer une calibration avec des points utilisés à partir de l'une ou l'autre station de base correctement. C'est décrit ci-dessous:

La figure suivante ne montre aucun rapport entre les stations de base:

Les points de Base 1 et de Base 2 ne devraient être utilisés pour la calibration.

Dans la figure suivante, un rapport est défini entre les stations de base par l'occupation de la deuxième station de base en utilisant la première station de base:

Les points de Base 1 et de Base 2 peuvent être utilisés pour une calibration.

  Les meilleures coordonnées possibles devraient être utilisées pour la station de base initiale dans un levé.

La précision des vecteurs GPS (en temps réel ou post traités) est influencée par la précision de la coordonnée de base. Une erreur de jusqu'à une partie par million (1 ppm) peut être introduite pour chaque dix mètres d'erreur dans les coordonnées de base. Par exemple, si votre point de référence WGS-84 primaire a une erreur de cent mètres et votre ligne de base est d'une longueur de deux kilomètres, il se peut qu'il y aura une erreur supplémentaire inutile de deux centimètres dans vos vecteurs GPS.


Modification des paramètres de calibration à l'aide du registre

 Vue d'ensemble et plus amples informations

Lors de l'installation de Trimble Geomatics Office, un paramètre du registre est ajouté à votre ordinateur qui contrôle:

  le nombre d'itérations effectuées au calcul par les moindres carrés des paramètres de calibration

  l'erreur maximum permise entre les positions de grille connues et calibrées des points de calibration

En utilisant les entrées, vous pouvez contrôler à quel moment un avertissement se présente. L'avertissement indique que le calcul n'a pas satisfait le critère d'erreur maximum spécifié.

Entrées du registre

Les entrées (Erreur maximum et Compte d'itérations maximum) se trouve sous la clé de registre:

HKEY_CURRENT_USER\Software\Trimble\Trimble Geomatics Office\1.6\Calibration

Erreur maximum

Pour changer l'Erreur maximum, utilisez cette entrée. La valeur est l'erreur (résiduelle) polaire horizontale maximum permise entre toute position de point de calibration calculée et sa position de grille connue associée.

Lorsque cette erreur spécifiée est dépassée un message d'erreur s'affiche indiquant que le calcul de calibration a dépassé le résidu horizontal maximum défini.

Les résultats de calibration peuvent dépasser l'erreur maximum spécifiée dans toutes les situation suivantes:

  il y avait une erreur dans l'entrée d'un des points de contrôle de grille

  un point de contrôle était identifié incorrectement ou utilisé incorrectement pendant le levé

  le calcul de calibration n'a pas convergé dans le nombre d'itérations spécifié.

La plus petite la valeur dans le champ Erreur maximum, la plus grande la possibilité de recevoir le message indiquant que l'erreur maximum a été dépassée. C'est à cause du fait que des critères plus serrés pour faire correspondre les positions calculées aux positions connues ont été définis.

La valeur par défaut pour ce champ (toujours définie en mètres) est 0,1 m.

Compte d'itérations maximum

Pour changer le nombre maximum d'itérations, utilisez cette entrée. Elle fournit de contrôle sur le nombre maximum d'itérations à effectuer sur le calcul par les moindres carrées des paramètres de calibration. La configuration de cette valeur sur un nombre plus élevé aura l'effet que les calculs effectueront plus itérations, à moins que le calcul détecte que les paramètre calculés ne changent plus avec chaque itération.

L'intervalle valide pour cette valeur est de 5 à 5000. Si vous configurez une valeur inférieure à 5, la valeur minimum de 5 est utilisée. Si vous configurez une valeur supérieure à 5000, la valeur maximum de 5000 itérations est utilisée. Le nombre d'itérations par défaut est 10.


Commande Calibration du site GPS

Utilisez cette commande pour effectuer une calibration du site GPS.

Pour y accéder, sélectionnez Levé / Calibration du site GPS.

Le dialogue Calibration du site GPS s'affiche.

NOTE: Avant que le dialogue Calibration du site GPS s'affiche, si vous avez des paramètres de transformation d'ajustement de réseau définis, ils seront écartés lorsque vous effectuez une calibration du site GPS. Si vous avez des points avec le même nom qui ont été fusionnés, un message vous demande si vous voulez exploser les points par type de données.


Dialogue Calibration du site GPS

 Vue d'ensemble et plus amples informations

Utilisez ce dialogue pour effectuer une calibration du site GPS.

Pour y accéder, sélectionnez Levé / Calibration du site GPS.

Pour effectuer une calibration du site GPS:

1.	Spécifiez quels composants de calibration doivent être calculés.

2.	Définissez la liste des paires de coordonnées. Une paire est une position dérivée de la grille et une position dérivée de GPS pour le même point physique.

3.	Appliquez la calibration courante à la base de données.

Composants de calibration

Le groupe Composants de calibration contient les cases à cocher et les champs suivants:

Transformation de datum

Normalement il y aura une transformation de datum prédéfinie disponible comme partie de la définition du système de coordonnées. Cette transformation de datum peut être une transformation à trois ou à sept paramètres, une définition de régression multiple ou basée sur des fichiers de grille de datum (par exemple, NADCON). Dans ces cas, il est peu probable que vous voudrez calculer une transformation de datum différente, donc laissez la case à cocher Transformation de datum vide. Cependant, s'il n'y a aucunes informations de transformation de datum disponible ou si vous doutez les paramètres disponibles, alors calculez une transformation de datum.

Pour calculer une transformation de datum entre l'ellipsoïde WGS-84 et l'ellipsoïde local utilisé comme partie de la définition du système de coordonnées pour le projet, sélectionnez cette case à cocher. Les options Transformation à trois paramètres et Transformation à sept paramètres sont disponibles. Sélectionnez quel type de transformation de datum à calculer.

NOTE: Cette option n'est pas disponible sur les projets qui ont un système de coordonnées de projection (Mercator transversal) par défaut car la projection par défaut est basées directement sur l'ellipsoïde WGS-84 signifiant qu'aucune transformation de datum n'est requise.

Si une transformation de datum est calculée et la calibration est appliquée à la base de données, le champ Datum dans la tabulation Système de coordonnées dans le dialogue Propriétés du projet affiche 'Calibration'.

Mettre à jour origine de projection par défaut

Cette option n'est active que si la définition du système de coordonnées courant utilise une projection par défaut. Pour mettre à jour l'origine de projection utilisant le premier point de contrôle à deux dimensions dans la Liste points, utilisez cette option. Etant donné que la projection par défaut n'a pas une origine de latitude et de longitude d'une constante nord et une constante est définie, la mise à jour de l'origine assurera que l'origine du projet est locale (c'est à dire, près des données de levé dans votre projet).

Si vous effectuez une calibration GPS sur un projet avec la projection par défaut déjà définie, alors vous devez vous assurer que cette case à cocher est effacée afin que la définition de projection par défaut n'est pas modifiée.

Ajustement horizontal

Pour calculer un ajustement horizontal, sélectionnez cette option. La case à cocher Fixer facteur d'échelle à 1 devient disponible. Si vous ne voulez pas que vos distances soient mises à l'échelle, utilisez cette option pour fixer le facteur d'échelle à 1. L'ajustement horizontal est une transformation de plan consistant en des translations dans les directions nord/sud et est/ouest, une rotation autour d'une origine définie et un facteur d'échelle. Les paramètres de transformation sont calculé utilisant les méthodes par les moindres carrées pour trouver la transformation que fournit des paramètres d'ajustement que, quand appliqués aux positions GPS, s'adaptent mieux aux coordonnées de grille de contrôle.

L'ajustement horizontal réduit toute erreur résiduelle entre les coordonnées de contrôle et les coordonnées de grille calculées des position GPS.

ASTUCE: En général il est prudent de calculer un ajustement horizontal sans le facteur d'échelle fixé à 1 initialement afin de vérifier si la facteur d'échelle calculé est près de 1. Si le facteur d'échelle calculé n'est pas près de 1, ce peut indiquer un problème dans les paires de points de calibration sélectionnées.

Ajustement vertical

Pour calculer un ajustement vertical, sélectionnez cette option. Il est un ajustement de plan incliné consistant en un déplacement vertical à une origine définie et des inclinaisons dans les directions nord et est. Les paramètres pour cet ajustement sont calculés utilisant les méthodes par les moindres carrés pour trouver un plan d'ajustement que s'adapte mieux aux élévations dérivées des hauteurs GPS avec des élévations de point de contrôle. Ce nécessite trois paires de points de calibration à trois dimensions. Avec une seule paire de points de calibration à trois dimensions, seulement un paramètre du déplacement vertical peut être calculé. S'il y a deux paires de points de calibration à trois dimensions disponibles, le système définit un plan de correction que s'adapte exactement à ces paires.

Si le projet utilise un modèle de géoïde, l'ajustement vertical est calculé et appliqué sur les corrections de modèle de géoïde.

Modèle géoïde

Le champ Modèle géoïde affiche le nom du modèle de géoïde sélectionné comme partie de la définition du système de coordonnées. Lorsqu'il n'y a aucun modèle de géoïde, "Aucun" s'affiche. C'est un champ en affichage seul. Le paramètre Modèle géoïde ne peut être changé qu'en utilisant la tabulation Système de coordonnées dans le dialogue Propriétés du projet.

Liste points

Cliquez sur Liste points pour sélectionner les paires de points de calibration à utiliser dans les calculs de calibration. Le dialogue Calibration du site GPS est masqué et le dialogue Liste points s'affiche.

Résumé du calcul

Les champs dans ce groupe présentent un résumé du dernier calcul de calibration effectué. Utilisez le résumé pour confirmer que la calibration calculée est valide sans la nécessité d'étudier le rapport de calcul.

Facteur d'échelle ajustement horizontal

Le champ Facteur d'échelle ajustement horizontal présente le facteur d'échelle calculé pour l'ajustement horizontal. Si vous avez sélectionné la case à cocher Fixer facteur d'échelle à 1, celle-ci affichera 1. Utilisez ce champ pour confirmer que le facteur d'échelle calculé pour l'ajustement horizontal est près de 1. Si ce n'est pas le cas, il indiquerait qu'il y a un problème avec une ou plusieurs des paires de points de calibration.

Inclinaison max ajustement vertical

Le champ Inclinaison max ajustement vertical présente l'inclinaison maximum pour l'ajustement de hauteur calculé basée sur les valeurs de Pente nord et Pente est calculées.

Résidu max horizontal et  Résidu max vertical

Les champs Résidu max horizontal et Résidu max vertical affichent les résidus maximum pour le dernier calcul effectué.

OK

Cliquez sur OK pour appliquer la dernière calibration calculée à la base de données.

Un recalcul est effectué afin de vérifier que tous les points GPS sont mis à jour selon la calibration GPS. Les détails du système de coordonnées sont mis à jour avec les paramètres de calibration.

Pour afficher les détails de calibration:

1.	Sélectionnez Fichier / Propriétés du projet.

2.	Sélectionnez la tabulation Système de coordonnées.

3.	Cliquez sur Détails.

4.	Sélectionnez la tabulation Ajustement horizontal ou la tabulation Ajustement vertical.

Annuler

Cliquez sur Annuler pour quitter le dialogue Calibration du site GPS sans appliquer la calibration calculée.

Calculer

Cliquez sur Calculer pour calculer les paramètres de calibration GPS après avoir modifier les paramètres des composants de calibration. Une fois que le calcul a été effectué les résultats sont résumés dans les champs de Résumé de calcul.

Enregistrer sous site

Cliquez sur Enregistrer sous site pour enregistrer la calibration courante ainsi que les détails du système de coordonnées dans la base de données de système de coordonnées sous forme d'un définition de site. Le dialogue Enregistrer sous site s'affiche. Saisissez le nom pour la définition du site enregistrée et cliquez sur OK. Vous pouvez utiliser cette définition du site comme la définition du système de coordonnées pour les projets au futur.

Système de coord

Cliquez sur Système de coord pour afficher les détails du système de coordonnées courant du projet. Le même dialogue s'affiche comme le dialogue Détails dans la tabulation Fichier / Propriétés du projet / Système de coordonnées.

Rapport

Cliquez sur Rapport pour afficher un rapport détaillé du calcul de calibration. Ce rapport affiche tous les paramètres calculées ainsi qu'une liste des coordonnées de points de contrôles calculées comparées à leurs positions connues et les valeurs résiduelles individuelles.


Calibration du site GPS - Dialogue Liste points

Utilisez ce dialogue pour sélectionner les paires de points que vous voulez utiliser pour calculer les paramètres du site GPS.

Pour y accéder:

1.	Sélectionnez Levé / Calibration du site GPS. Le dialogue Calibration du site s'affiche.

2.	Cliquez sur Liste points.

Le dialogue Calibration du site GPS – Liste points s'affiche.

Ce dialogue présente une liste défilante des groupements de paires de points de calibration. Il n'y a aucune limite spécifique au nombre de paires pouvant être définies pour calculer une calibration. L'ajout d'un grand nombre de paires n'aura pas toujours l'effet d'améliorer les résultats de calibration mais il fournira des vérifications supplémentaires sur la validité des paramètres calculés.

Chaque paire de points de calibration consiste en un point GPS (un point avec une position GPS ou, une position dérivée d'un vecteur GPS), et un point de grille (en général un point de contrôle, un point ajusté ou un point dérivé des observations terrestres) que ne peut pas être un point dérivé de GPS.

NOTE: Un point  GPS doit venir d'un vecteur GPS provenant d'une coordonnées WGS-84. Vous ne pouvez pas utiliser un vecteur GPS provenant d'une coordonnée de grille pour la définition d'un point GPS.

Lors de la sélection ou la spécification des points pour l'inclusion dans la liste de points, le système vérifie que chaque point GPS a une dérivation GPS et que chaque point de grille n'a pas une dérivation GPS.

Les positions des paires de points de calibration sont soulignées dans la fenêtre de graphiques avec un symbole de boîte sur le point GPS et un symbole de croix sur le point de contrôle et une ligne liant les deux positions. Vous devez effectuer un zoom avant sur l'une des paires de points de calibration pour voir ce détail. Les symboles et la ligne de liaison de la paire de points de calibration courante est affichées par des lignes pointillées.

En général, vous sélectionnez les points des paires de points de calibration à l'aide du remplissage de champ pour choisir les points appropriés de la fenêtre de graphiques. Le remplissage de champ ne permet que la sélection des points dérivés de GPS comme un point GPS et des points sans une dérivation GPS comme un point de grille. Vous pouvez également saisir les noms des points à utiliser dans le champs Point GPS et Point de grille.

Si un point GPS a une élévation de grille utilisée (et que n'est pas dérivée de la hauteur GPS et le modèle de géoïde), vous pouvez utiliser le même point dans le point GPS et le point de grille dans le calcul d'un ajustement vertical. Si vous le faites, le champ Type est configuré sur Vertical.

NOTE: Si vous avez deux points séparés (un point GPS et un point de grille) avec le même nom, lorsque le point GPS est sélectionné, le point de grille avec le même nom est sélectionné automatiquement. Si les coordonnées GPS et de grille sont enregistrées sous le même point, vous devrez exploser les points par type.

Si vous n'avez pas des points de contrôle de grille ou des points GPS appropriés dans la base de données, vous pouvez spécifier les valeurs de coordonnées appropriées de ces points dans les champs fournis. En ce cas, vous ne devez pas spécifier le nom du point pour le point car Trimble Geomatics Office attendra un point avec ce nom dans la base de données. Les positions entrées de cette façon ne sont pas stockées dans la base de données.

Pour spécifier les valeurs de coordonnées (point GPS ou de grille) pour la calibration:

1.	Cliquez sur l'icône (+) à côté du Point GPS ou Point de grille dans les champs Nom.

L'affichage d'arborescence se développe pour afficher les champs de coordonnées.

2.	Utilisez le remplissage de champ ou saisissez une valeur à l'aide du clavier pour compléter les champs de coordonnées.

Si vous ne pouvez pas sélectionner un point GPS pour l'utilisation dans une calibration du site GPS, référez-vous à la rubrique Problèmes dans la sélection des points pour une calibration du site GPS.

Type

Pour spécifier comment il faut utiliser la paire de points de calibration dans le calcul de calibration, utilisez ce champ. Trimble Geomatics Office remplit ce champ automatiquement avec le type approprié lorsque vous avez sélectionné une paire de coordonnées. Vous pouvez changer le type, à condition que l'option sélectionné est valide. Par exemple, si le point de contrôle n'a pas une élévation, vous ne pouvez pas configurer le champ Type pour la paire Horz et Vert.

Les options disponibles sont:

  Horz et Vert -	spécifie que la paire de points de calibration est appropriée à utiliser dans la détermination des ajustements horizontal ainsi que vertical.

  Horizontal -    spécifie que la paire de points de calibration n'est appropriée qu'à utiliser dans la détermination d'un ajustement horizontal. En effet, la valeur de l'élévation du point de grille n'est pas fiable.

  Vertical -        spécifie que la paire de points de calibration n'est appropriée qu'à utiliser dans la détermination d'un ajustement vertical. En effet, les valeurs du Nord et de l'Est du point de grille ne sont pas fiables.

  (Ignoré) -       spécifie que la paire de points de calibration ne doit pas être utilisée dans le calcul de tout paramètre de transformation. Cette option sert lorsque vous avez trouvé un problème dans votre calcul de calibration et vous essayez de localiser une paire de points de calibration suspecte. Vous pouvez utiliser ce paramètre pour supprimer une paire de points de calibration temporairement du calcul afin de voir si les résultats sont améliorés.

  (Non valide)   indique que la paire de points sélectionné n'est pas une paire de points valide.

Statistiques

Les champs dans ce groupe présentent les résultats de la calibration basés sur les valeurs que vous avez ajouté à la Liste points. Utilisez le résumé pour vérifier les statistiques de calibration et identifier des paires de points mauvaises avant d'accepter la Liste points.

Facteur d'échelle ajustement horizontal

Le champ Facteur d'échelle ajustement horizontal affiche le facteur d'échelle calculé pour l'ajustement horizontal. Si vous avez sélectionné la case à cocher Fixer facteur d'échelle à 1, cette case affichera 'un'. Utilisez ce champ pour confirmer que le facteur d'échelle calculé est près d'un. S'il n'est pas près d'un, il indique qu'il y a un problème avec une ou plus des paires de points de calibration.

Inclinaison max ajustement vertical

Le champ Inclinaison max ajustement vertical présente l'inclinaison maximum pour l'ajustement de hauteur calculé, basé sur les valeurs de pente nord et de pente est calculées.

Résidu max horizontal et Résidu max vertical

Les champs Résidu max horizontal et Résidu max vertical affichent les résidus maximum.

Insérer

Cliquez sur Insérer pour insérer un enregistrement de paire de points de calibration vide. Ce sert si vous utilisez la Projection par défaut, et vous voulez mettre à jour l'origine de la projection. La première paire de points est utilisée pour mettre à jour l'origine, donc utilisez ce bouton pour insérer une nouvelle paire de points de calibration vide au début de la liste.

Supprimer

Cliquez sur Supprimer pour supprimer une paire de points de calibration de la liste.


Problèmes dans la sélection des points pour une Calibration du site GPS

Si vous utilisez le  dialogue Calibration du site GPS – Liste points pour sélectionner les points pour une calibration du site GPS, et le système ne vous permet pas de sélectionner un point pour le point GPS ou le point de grille de la paire de points de calibration:

Un point GPS ne peut pas être sélectionné parce que le point est dérivé d'une grille

Le point est le résultat d'un vecteur dans lequel le 'point de' du vecteur a été dérivé d'une position de grille, Ce peut arriver si vous avez entré au clavier  une position de grille pour une station de base après un levé cinématique en temps réel (RTK). Tout vecteur provenant de la station de base est appliqué. Cependant, la position pour tous ces points est dérivée en utilisant une combinaison des valeurs WGS-84 (le vecteur) et les valeurs de grille (la position WGS-84 de base). Vous ne pouvez pas utiliser tout point ayant les vecteurs appliqués de la base. Ce fait est illustré dans les figures suivantes:

  Point 1 ne peut pas être utilisé comme un point GPS dans une calibration du site GPS.

La position de grille de base est convertie à WGS-84 et puis le vecteur est appliqué à la position de point 1.

 Point 1 peut être utilisé comme un point GPS dans une calibration du site GPS.

Ce vecteur est appliqué à la position WGS-84 de base pour déterminer la position du point 1.

Pour le fixer, utilisez la fenêtre Propriétés pour éditer les coordonnées du point et saisir une position WGS-84. Celle-ci sera utilisée plutôt que la position de grille, et tous les vecteurs provenant de la station de base seront disponibles comme les points GPS dans une calibration du site GPS.

De la même façon qu'un point GPS ne peut pas être sélectionné du fait que le 'point de' est dérivé d'une position de grille, un point de grille ne peut pas être utilisé si le 'point de' est dérivé d'une position GPS. Un point de grille doit provenir d'une observation terrestre.


Transformation à trois paramètres

Utilisez une Calibration du site GPS pour définir une transformation de datum si vous n'avez pas un système de coordonnées défini et vous voulez créer vos propres paramètres de transformation de datum pour votre surface de projet.

L'algorithme de transformation à trois paramètres calcule les valeurs de translation le long des axes cartésiens X, Y, et Z à partir du centre de l'ellipsoïde WGS-84 au centre de l'ellipsoïde de la projection locale. Les positions de point de contrôle sont utilisées pour déterminer le positionnement spatial de l'ellipsoïde local et ainsi son rapport à l'ellipsoïde WGS-84.


Transformation à sept paramètres

Utilisez une Calibration du site GPS pour définir une transformation de datum si vous n'avez pas un système de coordonnées défini et vous voulez créer vos propres paramètres de transformation de datum pour votre surface de projet.

L'algorithme de transformation à sept paramètres calcule trois valeurs de translation, trois valeurs de rotation et un changement d'échelle entre l'ellipsoïde WGS-84 et l'ellipsoïde pour la projection locale. Les trois valeurs de translation sont calculées le long des axes cartésiens X, Y, et Z à partir du centre de l'ellipsoïde WGS-84 au centre de l'ellipsoïde de la projection locale comme dans la transformation à trois paramètres. Il calcule les rotations autour de chaque axe entre les deux ellipsoïdes et un facteur d'échelle tenant en compte toute différence de taille entre les deux ellipsoïdes (à cause des demi-grands axes ellipsoïdaux différents et les valeurs d'aplatissement).

Les positions de point de contrôle sont utilisées pour déterminer le positionnement spatial de l'ellipsoïde local et ainsi son rapport à l'ellipsoïde WGS-84. Il existe des corrélations étroites entre les valeurs de translation et de rotation calculées (c'est à dire, le résultat d'un très petit angle de rotation appliqué sur un grand rayon tel que le rayon de la terre est très semblable à une translation). Ce signifie que c'est facile d'arriver aux grandes valeurs calculées de compensation qui ne sont pas précises si les points de contrôle ne couvrent pas une surface assez grande. Normalement, toute valeur de rotation calculé ne devrait pas dépasser 10° de l'arc.


Observation GPS - Vue d'ensemble

 Voir aussi

Une observation GPS, ou ligne de base, est un vecteur entre deux points observés utilisant GPS. Les deux points utilisés pour déterminer une ligne de base dépendent de la technique de levé utilisée. Par exemple, dans un levé cinématique en temps réel (RTK) ou un levé cinématique post traité (PPK), une ligne de base existe de la station de base à chaque point mobile.

Cependant, avec les levés statique et statique rapide, il n'y a pas de station de base. Avec ces techniques de levé, un vecteur GPS existe entre les points qui ont été occupés au même temps, comme indiqué dans la table suivante:

Pour afficher le type de ligne de base, sélectionnez-la et ouvrez la fenêtre Propriétés.

Vecteurs statique, statique rapide, et cinématique post traité (PPK)

Les observations statique, statique rapide et PPK qui n'ont pas été post traitées, sont des lignes de base non traitées. Lorsque les lignes de base ont été post traitées, elles deviennent des lignes de base post traitées.

Lorsque vous supprimez une ligne de base post traitée, la ligne de base retournera à une ligne de base potentielle. Vous pouvez supprimer la ligne de base par la suppression des points. Sinon,  désactivez la ligne de base si vous ne voulez pas l'utiliser dans des calculs de levé ultérieurs, mais vous voulez conserver les points.

Vecteurs continus PPK

Lorsque les vecteurs continus PPK ont été post traités, ils sont affichés dans la même couleur que les lignes de base PPK, mais la méthode Champ s'affiche comme 'Continu' dans la fenêtre Propriétés.

Vecteurs RTK

Les vecteurs RTK ne peuvent pas être les lignes de base potentielles, ils sont affichés comme 'Lignes de base RTK' dans la fenêtre Propriétés. Les vecteurs continu RTK sont affichés dans la même couleur que les ligne de base RTK, mais la méthode Champ s'affiche comme 'Continu' dans la fenêtre Propriétés.

Vecteurs de vérification

Si l'état d'une ligne de base est configuré sur 'Vérifier', la ligne de base s'affichera avec des lignes à traits.

Vecteurs désactivés

Si l'état d'une ligne de base est configuré sur 'Désactivé', la ligne de base s'affichera avec la couleur appropriée pour les lignes désactivées.


Occupations GPS

Un enregistrement d'Occupation GPS est créé chaque fois qu'un point est observé utilisant GPS.

Pour afficher les enregistrements d'Occupation GPS utilisant la fenêtre Propriétés:

1.	Sélectionnez l'une des choses suivantes:

  un point observé utilisant GPS

  la ligne de base (dans Vue de levé)

2.	Cliquez sur l'icône [image] pour afficher l'enregistrement d'Occupation GPS.

Si vous sélectionnez un point observé utilisant GPS dans la fenêtre Propriétés, vous pouvez afficher tous les enregistrements d'Occupation GPS pour ce point. Si vous sélectionnez une ligne de base utilisant la fenêtre Propriétés, vous pouvez afficher l'occupation GPS qui a produite la ligne de base.

Un enregistrement d'occupation GPS comprend:

  Le nom du point.

  La durée (en heures) de l'occupation et l'heure de début de l'occupation.

  Les détails d'antenne GPS y compris la hauteur d'antenne, le type et la méthode.

Lors de l'importation des données l'Heure de début et l'Heure de fin sont converties de l'heure GPS à UTC (Temps universel coordonné – en effet Greenwich Mean Time). Ces temps sont alors affichés comme les heures locales utilisant le Fuseau horaire courant configuré dans Microsoft Windows.


Constante prisme

La constante prisme est une correction à la distance inclinée mesurée avec une station totale.  Elle est appliquée pendant le recalcul. Elle est nécessaire parce que la surface du réflecteur de prisme n'est pas directement au-dessus du point de levé mesuré. En général, la constante est une valeur négative.

Où:

	est la distance inclinée correcte
	est la distance inclinée mesurée
	est la constante de prisme

La valeur de la constante de prisme est stockée avec les informations de configuration de destination .

Pour afficher et éditer une valeur de constante de prisme:

1.	Sélectionnez un point de configuration de destination ou une observation conventionnelle.

2.	Ouvrez la fenêtre Propriétés.

3.	Cliquez sur le bouton [image].

NOTE: Dans le logiciel 3300/3600/5600 Elta et 3600/5600 GDM, la constante de prisme est appliquée dans l’instrument, mais elle n’est pas stockée dans le fichier. La constante de prisme s’affiche comme ‘?’.

Si une constante de prisme incorrecte a été configuré dans l’instrument et vous voulez corriger la valeur dans Trimble Geomatics, dans le champ de constante de prisem de la fenêtre de propriétés, entrez la différence entre la constante de prisme incorrecte et la constante de prisme correcte. Par exemple, si vous avez entré une constante de prisme de “20”, et vous devriez avoir entré “5”, entrez “–15”.

ASTUCE: Vous pouvez utiliser le dialogue Edition multiple pour l'édition des observations multiples.


Corrections pour l'excentricité de montage d'équipement de mesure de distance électronique (EDM)

La plupart des stations totales modernes n'ont pas besoin d'une correction pour l'excentricité de montage, car la raie de mesure EDM (mesure de distance électronique) est identique à la ligne de visée du télescope.  Cette correction n'est pas requise si un EDM monté à télescope est utilisé avec un réflecteur inclinant et le réflecteur/destination est conçu à correspondre à la configuration de l'EDM/télescope.

Cependant, dans certains cas il peut être nécessaire de calculer une correction à la distance inclinée mesurée afin de trouver la distance inclinée vraie que vous auriez observé si la raie de mesure EDM était identique avec la ligne de visée du télescope. La correction pour l'excentricité de montage dépend du type de système de montage, et elle est appliquée automatiquement par Trimble Geomatics Office utilisant les formules appropriées indiquées ci-dessous.

Les valeurs de déport appropriées sont stockées dans l'enregistrement d'instrument dans le fichier de carnet de terrain.

1. EDM monté standard avec un réflecteur non inclinant.

Où:

	est la distance inclinée correcte
	est le déport EDM (à l'instrument)
	est le déport du réflecteur (à la destination)
	est l'angle vertical de zénith (observé)
	est la distance inclinée mesurée (après l'application de la correction de constante de prisme)

2. EDM monté à télescope avec un réflecteur non inclinant.

Où:

	est la distance inclinée corrigée
	est le déport EDM (à l'instrument)
	est le déport du réflecteur (à la destination)
	est l'angle vertical de zénith (observé)
	est la distance inclinée mesurée
Cette correction est appliquée pendant l'importation d'un fichier de périphérique.


Correction de température et pression

Les corrections de pression et température sont faites aux distances mesurées électroniquement pour corriger des variations dans la longueur mesurée à cause de température et de pression. Vous pouvez saisir des valeurs appropriées lors du travail sur le terrain ou éditer le fichier de périphérique avant son importation.

Les mesures de distance électroniques effectuées par les EDM sont basées sur la vitesse de la lumière à travers l'air. La température et la pression ont un effet sur cette vitesse.

Cette correction atmosphérique n'est effectuée aux mesures de distance que si l'EDM n'a pas déjà appliqué une correction appropriée.

Où:

	est la distance inclinée corrigée
	est la distance inclinée mesurée
	est un index réfractif de groupe pré-configuré pour les fréquences de modulation et porteuse de l'instrument EDM.
	est un constante pour la longueur d'onde porteuse (mm Hg) de l'instrument EDM
	est la pression atmosphérique en mm de mercure
	est la température d'air sec en degrés Celsius

NOTE: Trimble vous recommande d'enregistrer la pression et la température dans le Trimble Survey Controller. Cette correction est appliquée aux distances inclinées lors de l’importation d’un fichier .dc.

NOTE:  Dans le logiciel 3600/5600 Zeiss Elta, 3600/5600 GDM, et TDS Survey Pro, les corrections de pression et de température sont appliquées dans l’instrument. Donc, les observations sont déjà corrigées avant d’être importées dans Trimble Geomatics Office.


Collimation verticale

La correction de collimation verticale est une correction angulaire appliquée directement à l'angle vertical.

La collimation verticale d'un instrument est établie par l'observation soigneuse d'un objet utilisant les deux faces de l'instrument. La valeur de correction établie indique l'erreur constante à cause du positionnement légèrement incorrect des réticules dans le télescope.

Lorsque l'erreur de collimation dans un instrument a été établie, un enregistrement de collimation sera stocké dans le fichier de carnet de terrain et la collimation verticale est appliquée à tous les angles verticaux suivants:

Face 1:
Face 2:
Où:

	est l'angle vertical corrigé
	est l'angle vertical observé (angle de zénith)
	est la correction de collimation verticale. Cette correction est appliquée lors de l'importation d'un fichier de carnet de terrain.
NOTE: Trimble Survey Controller n’a aucun programme de collimation. Aucune correction de collimation n’est effectuée lors de l’importation.

NOTE: Dans le logiciel 3300/3600/5600 Elta, 3600/5600 GDM, et TDS Survey Pro, la collimation verticale est appliquée avant son importation Trimble Geomatics Office.


Courbure et réfraction

La correction de courbure et réfraction est faite à l'angle vertical observé pour tenir en compte l'effet de la courbure de la terre et la réfraction verticale atmosphérique des raies spectrales.

La correction de courbure et réfraction est appliquée lors de l'importation d'un fichier de carnet de terrain.

Où:

	est l'angle vertical de zénith corrigé
	est l'angle vertical de zénith observé
	est la distance inclinée corrigée
	est le coefficient de réfraction – en général 0,142 ou 0,20 selon la valeur dans l'enregistrement d'Etude dans le fichier de carnet de terrain.
	est le rayon de la terre
NOTES:

•	Ce format retourne une valeur en radians.

 La correction de courbure et réfraction est appliquée lors de l'importation d'un fichier Trimble Survey Controller.

  Dans le logiciel 3300/3600/5600 Elta, la courbure et la réfraction sont appliquées dans l’instrument. Donc, les observations sont déjà corrigées avant leur importation dans Trimble Geomatics Office.

  Dans le logiciel 3600/5600GDM, la courbure et la réfraction sont appliquées aux coordonnées calculées. Les observations brutes sont importées et corrigées dans Trimble Geomatics Office.

•	Dans le logiciel TDS Survey Pro, la correction de courbure et réfraction est appliquée lors de l’importation dans Trimble Geomatics Office.


Collimation horizontale

NOTE: Trimble Survey Controller n'a pas un programme de collimation. Aucune correction de collimation n'est effectuée lors de l'importation.

S'il y a un enregistrement de collimation rencontré, Trimble Geomatics Office applique la collimation horizontale selon la formule suivante:

Face 1:
Face 2:
Où:

	est l'angle horizontal corrigé
	est l'angle horizontal observé
	est la correction de collimation horizontale

Cette correction de collimation, si nécessaire, est appliquée lors de l'importation d'un fichier de carnet de terrain.

NOTE: Dans le logiciel 3300/3600/5600 Elta, 3600/5600 GDM, et TDS Survey Pro, la collimation horizontale est appliquée avant son importation dans Trimble Geomatics Office.


Orientation horizontale

 Voir aussi

Les angles horizontaux terrestres sont orientés lors du recalcul en utilisant l'azimut à la visée arrière. L'angle horizontal est ajouté à l'azimut et le résultat est l'azimut au point de visée avant. Ainsi toutes les observations de la configuration sont orientées.

L'azimut au point de visée arrière est déterminé utilisant les règles suivantes (seulement pour les observations conventionnelles):

  Si seulement la position du point d'instrument et des points de visée arrière sont connues, (c'est à dire, il n'y a aucun azimut entré au clavier) le recalcul utilise l'azimut de visée arrière déterminé de l'inverse entre les deux points.

  Si seulement un azimut entré au clavier est disponible et le point "vers" n'a pas une coordonnée, le recalcul l'utilise comme l'azimut de visée arrière.

  Si les positions du point d'instrument et du point de visée arrière sont connues, et un azimut entré au clavier est disponible, le recalcul détermine la qualité de chaque azimut et utilise la hiérarchie dans la table ci-dessous pour déterminer quel azimut à utiliser comme l'azimut de visée arrière.

Par exemple, si une paire de points de visée arrière et d'instrument de qualité de levé existe, ainsi qu'un azimut entré au clavier de qualité de contrôle, l'azimut entré au clavier est utilisé pour l'azimut de visée arrière.

NOTE: Pour calculer la qualité d'un azimut de visée arrière, la qualité horizontale du point d'instrument et du point de visée arrière sont étudiées et la qualité la plus basse est prise. Par exemple, si le point d'instrument a une qualité de contrôle et le point de visée arrière a une qualité de mappage, l'azimut de visée arrière aura la qualité de mappage.

Type	Qualité
Dérivé des coordonnées	Contrôle
Azimut entré au clavier	Contrôle
Dérive des coordonnées	Levé
Azimut entré au clavier	Levé
Dérivé des coordonnées	Mappage
Azimut entré au clavier	Mappage

NOTE: Si vous voulez qu'un recalcul utilise l'azimut entré au clavier plutôt que l'azimut déterminé entre l'instrument et le point de visée arrière, augmentez la qualité de la position de l'azimut entré au clavier afin qu'elle est plus haute dans la hiérarchie que l'azimut déterminé entre l'instrument et le point de visée arrière.

ASTUCE: Vous pouvez éditer la qualité d'un azimut entré au clavier.


Angles horizontaux

 Voir aussi

Pour les observation terrestres, Trimble Geomatics Office stocke des angles horizontaux non orientés entre deux points. Les angles sont orienté lors d'un recalcul.

Pendant l'importation, l'angle horizontal est déterminé par:

Où:

	est l'angle horizontal stocké dans Trimble Geomatics Office
	est l'observation horizontale au point de visée avant
	est l'observation horizontale au point de visée arrière
La valeur BSObs est stockée dans l'enregistrement BKB dans le fichier de carnet de terrain. (L'enregistrement BKB est le type d'enregistrement dans un fichier .dc assigné à l'enregistrement de visée arrière.)

Lorsqu'il n'y a aucun enregistrement de gisement arrière, Trimble Geomatics Office traite le premier point observé comme le point de gisement arrière.


Réduction de pente

Vue d'ensemble

Pour chaque observation terrestre, Trimble Geomatics Office réduit la distance inclinée à ses composants horizontaux et verticaux pendant un recalcul.

Les deux formules suivantes sont utilisées:

Où:

	est la distance horizontale à l'élévation du point de source
	est la différence de hauteur entre les points de source et de destination
	est la distance inclinée (après les corrections d'observation)
	est l'angle vertical (zénith)
Référez-vous au diagramme de réduction. Ce diagramme éclaircit les valeurs utilisées dans les deux formules ci-dessus.


Correction du niveau de l'ellipsoïde

La correction du niveau de la mer (ou correction du niveau de l'ellipsoïde) est appliquée à la distance horizontale calculée. Pour vérifier si les corrections du niveau de la mer sont appliquées, sélectionnez Fichier / Propriétés du projet / tabulation Recalculer.

Cette correction réduit une distance horizontale à la distance que aurait été mesurée au niveau de l'ellipsoïde ( pas au niveau du géoïde, étant le niveau de la mer). Cette différence n'est pas considérable avec la réduction.

Si la case à cocher de corrections du niveau de l'ellipsoïde est sélectionnées, les observations terrestres sont appliquées le long de l'ellipsoïde local. Si la case à cocher de corrections du niveau de l'ellipsoïde est effacée, les observations terrestres sont appliquées au sol.

La formule de correction du niveau de l'ellipsoïde est:

Où:

	est le rayon de la terre
Référez-vous au diagramme de réduction. Ce diagramme éclaircit les valeurs utilisées dans la formule ci-dessus.

NOTE: La correction de niveau d’ellipsoïde n’est pas appliquée dans une étude à facteur d’échelle seul. Lorsque vous importez un fichier à facteur d’échelle seul, la correction de niveau d’ellipsoïde dans Trimble Geomatics Office est désactivée automatiquement à moins que vous n’ayez sélectionné la case à cocher Utiliser coordonnées au sol dans le dialogue Paramètres du site local.


Rayon de la terre

Dans Trimble Geomatics Office, le rayon de la terre est calculé utilisant la latitude du projet et l'ellipsoïde du projet. Si une latitude du projet n'est pas définie pour votre projet, une latitude du projet par défaut de 30 degrés est utilisée. Par exemple, sur l'ellipsoïde WGS-84, avec une latitude du projet de 30 degrés, la valeur du rayon de la terre est 6383.480,917.

NOTE: Définissez la latitude du projet avant l'importation des observations terrestres. Si vous définissez la latitude du projet après l'importation des données terrestres, la correction ellipsoïdale et la correction de courbure et réfaction, ne sont pas réappliquées utilisant le rayon de la terre nouvellement calculé.

Pour définir la latitude du projet:

1.	Sélectionnez Fichier / Propriétés du projet. Le dialogue Propriétés du projet s'affiche.

2.	Sélectionnez la tabulation Système de coordonnées.

3.	Dans le groupe Paramètres du site local, cliquez sur Changer.

4.	Dans le groupe Emplacement du projet, saisissez la latitude du projet.


Observation conventionnelles - Vue d'ensemble

Trimble Geomatics Office prend en charge les points observés à l'aide des instruments de levé conventionnels. Ces points sont dénommés des observations conventionnelles.

Dans Trimble Geomatics Office vous pouvez:

  importer des observations conventionnelles.

  importer des observations de télémètre laser

  importer des fichiers de niveau numérique


Réduction des observation conventionnelles

L'opération de Recalculer de Trimble Geomatics Office utilise les mesures de distance et les angles observés pour calculer les coordonnées des points observés conventionnellement.

Importation

Lors de l'importation d'un fichier, les corrections suivantes sont appliquées:

  Correction de pression et de température, appliquée à la distance inclinée.

  Courbure et réfraction, appliquée à l'angle vertical.

  Angle horizontal, réduit les observations horizontales aux angles horizontaux du point de visée arrière. Les lecteurs à deux faces sont stockées comme des lectures de face simple.

  Corrections pour l'excentricité de montage d'équipement de mesure de distance électronique (EDM).

  Collimation horizontale (pour des fichiers .sdr seuls).

  Collimation verticale (pour des fichiers .sdr seuls).

Recalcul

Lors d'un recalcul, les corrections suivantes sont appliquées:

  Correction d'ellipsoïde

  Orientation horizontale

  Constante prisme


Diagramme de réduction

Dans ces corrections, le diagramme suivant éclaircit les valeurs utilisées dans les formules:

Où:

	est l'angle vertical (zénith)
	est la distance inclinées (après les corrections d'observation)
	est l'élévation du point de source au-dessus de l'ellipsoïde
	est la différence de hauteur entre les points de source et de destination
	est la distance horizontale à l'élévation du point de source
	est la corde au niveau de la mer (ou niveau d'ellipsoïde)
	est la distance au niveau de la mer (ellipsoïde)


Observations conventionnelles - Observations à deux faces

Les observations de Face 2 sont converties et stockées dans la base de données sous forme d'observation à face simple. Pour sélectionner une observation, utilisez la fenêtre Propriétés. La face (une ou deux) est affichée.

S'il y a des observation de Face 1 et Face 2 vers et depuis les mêmes points, on fait la moyenne  des observations de Face 1 et Face 2 et elles sont importées sous forme d'une seule observation. Le type d'observation est indiqué comme moyen.


Observations conventionnelles - Cheminement

A l'aide du logiciel Trimble Survey Controller, TDS Survey Pro, ou 3600/5600 Elta, vous pouvez calculer une erreur de fermeture de cheminement pour vos observations conventionnelles et ajuster le cheminement.

Si vous avez ajusté un cheminement et vous voulez modifier les valeurs ajustées, effectuez les étapes suivantes:

1.	Sélectionnez le point pour lequel vous voulez modifier les coordonnées.

2.	Ouvrez la fenêtre Propriétés et développez le point en utilisant la structure d'arborescence

3.	Vérifiez s'il y a déjà une coordonnée locale/de grille pour le point. S'il n'y en a pas, ajoutez une coordonnées à l'aide de l'outil Ajouter coordonnée [image].

4.	Cliquez dans les champs de coordonnées et saisissez les valeurs de coordonnées connues.

5.	Dans les Contrôles de sélection de qualité à côté des champs de coordonnées, sélectionnez l'option Qualité de la liste déroulante. La qualité doit être de levé ou plus haute afin d'être utilisée plutôt que l'observation.

6.	Cliquez sur OK.

7.	Appuyez sur F4 pour effectuer un recalcul des données de levé.

Un recalcul détermine toute erreur de fermeture entre les coordonnées connues et les coordonnées observées.

Les erreurs de fermeture sont affichées dans le Rapport de recalcul. Si vous ne voulez pas voir ces erreurs de fermeture, désactivez l'observation.

Pour afficher les erreurs de fermeture, effectuez l'une des choses suivantes:

  Sélectionnez Rapports / Rapport de recalcul.

  Cliquez sur le bouton de la page d'Avertissement [image] dans la fenêtre Propriétés et cliquez sur l'outil de dérivation de point [image].

  Cliquez sur le bouton de la page Résumé [image] dans la fenêtre Propriétés et cliquez sur l'outil de dérivation de point [image].


Observation de télémètre laser - Vue d'ensemble

Trimble Geomatics Office prend en charge les points observés à l'aide des télémètres laser avec Trimble Survey Controller de version 7.00 ou ultérieure. Pour plus amples informations concernant quels télémètres laser Trimble Survey Controller prend en charge, référez-vous au Manuel de référence Trimble Survey Controller.

La façon dont les observations de télémètre laser sont affichées dépend du paramètre apparence dans la fenêtre Propriétés.

Dans Trimble Geomatics Office vous pouvez:

  Importer les fichiers Survey Controller (*.dc) contenant des observations de télémètres laser. (Les points de télémètres laser observés sont importés dans Trimble Geomatics Office de la même façon que les points conventionnels.)

  Afficher des observations de télémètre laser dans la fenêtre Propriétés.

  Sélectionner et rapporter des observations de télémètre laser.


    Observations de télémètres laser - Affichage

Les observations de télémètre laser sont affichées dans la Vue de levé. La couleur sous laquelle elles sont affichées dépend du groupe d'arrière plan configuré dans la tabulation Affichage / Options/ Légende de levé.

ASTUCE: Choisissez Sélectionner / Par interrogation et utilisez l'option Points de télémètre laser pour sélectionner tous les points de télémètre laser dans la base de données. Les points de télémètre laser ont une observation de télémètre laser.

Pour afficher un point observé à l'aide d'un télémètre laser:

1.	Dans la Vue de levé, ouvrez la fenêtre Propriétés.

2.	Sélectionnez une observation de télémètre laser.

Utilisez les boutons de menu pour afficher les détails d'observation de télémètre laser suivants:


Paramètres de transformation d'ajustement et de calibration

 Comment ...

Lorsqu'une calibration du site GPS est effectué, des paramètres de calibration sont calculés établissant le rapport entre les points WGS-84 (points GPS) et les positions de grille locale dans la base de données.

Lorsqu'un ajustement de réseau contraint est effectué (si vous avez le module Ajustement de réseau installé), les paramètres de transformation sont calculés. Ces paramètres établissent le rapport entre les observations et le contrôle fixe.

Trimble Geomatics Office ne permet que des paramètres d'ajustement vertical ou horizontal dérivés d'un calibration du site GPS ou des paramètres d'ajustement dérivés d'un ajustement. Si vous avez effectué un ajustement et vous avez calculé les paramètres de transformation d'ajustement, et puis vous essayez d'effectuer une calibration du site GPS, vous serez averti que les paramètres seront supprimés si vous continuez avec la calibration.

NOTE: Vous ne pouvez pas transféré les paramètres de transformation d'ajustement dérivés d'un ajustement vers le Trimble Survey Controller. Pour ce faire, vous devez effectuer une calibration du site GPS utilisant des coordonnées de calibration enregistrées (sous forme des points WGS-84) dérivées d'un ajustement sur le datum WGS et des coordonnées ajustées (sous forme des points de grille) dérivées d'un ajustement de réseau contraint sur le datum local.


Références

Livres

Barry, B. Austin. 1991. Errors in Practical Measurement in Surveying, Engineering, and Technology. Edité par M. D. Morris. ISBN 0-910845-47-6. Rancho Cordova: Landmark Enterprises.

Hofmann-Wellenhof, B., Herbert Lichtenegger, et James Collins. 1993. Global Positioning System Theory and Practice. Second Edition. ISBN: 3-211-82477-4 et ISBN 0-387-82477-4. New York: Springer-Verlag Wien.

Leick, Alfred. 1995. GPS Satellite Surveying. Deuxième édition. ISBN: 0-471-30626-6. New York: John Wiley and Sons, Inc.

Pope, Allen J. 1976. The Statistics of Residuals and the Detection of Outliers. NOAA Technical Report NOS65 NGS1, Rockville, MD.

Seeber, Günter. 1993. Satellite Geodesy, Foundations, Methods, and Applications. ISBN: 3-11-012753-9. Berlin et New York: Walter deGruyter.

U.S. National Geodetic Survey (NGS). 1986. Geodetic Glossary. Silver Spring, MD: NGS Publications Information Center.

Van Sickle, Jan. 1996. GPS for Land Surveyors. ISBN 1-575-04041-7. Ann Arbor: Ann Arbor Press, Inc.

Wolf, Paul R. et Charles D. Ghilani. 1997. Adjustment Computations: Statistics and Least Squares in Surveying and GIS. ISBN: 0471-16833-5. New York: John Wiley and Sons, Inc.

Autres ressources - Publications / Journaux dans les Etats-Unis

Les publications suivantes sont une source utile des informations concernant le génie civil et d'autres aspects de la topographie:

ACSM Bulletin. ISSN: 0747-9417. Publié tous les deux mois par l'American Congress on Surveying and Mapping, 5410 Grosvenor Land, Suite 100, Bethesda, MD 20814-2144 Etats-Unis.

GPS World: News and Applications of the Global Positioning System. ISSN: 1048-5104 Publié chaque mois par Advanstar Communications Inc. Subscription offices: P.O. Box 6148, Duluth, MN 55806-6148 Etats-Unis.

P.O.B. - Point of Beginning. ISSN: 0739-3865. Publié chaque mois par Business News Publishing Company, 755 W. Big Beaver Road, Ste. 1000, Troy, MI 48084 Etats-Unis.

Professional Surveyor. ISSN: 0278-1425. Publié dix fois chaque année par Professional Surveyors Publishing Company, Inc., 1713-J Rosemont Avenue, Frederick, MD 21702-4170 Etats-Unis.


Le Navigateur de zoom

Le Navigateur de zoom s'affiche sous forme d'une petite fenêtre en bas de la Barre du projet. Il reflète les données affichées dans la fenêtre de graphiques principale de Trimble Geomatics Office. Lorsque vous utilisez tout outil de zoom dans la fenêtre principale, la boîte dans le Navigateur de zoom change en conséquence afin de refléter ce changement. L'étendue de votre projet est toujours affichée.

Afficher (ou masquer) le Navigateur de zoom

Pour masquer (ou afficher) le Navigateur de zoom dans la barre Trimble Geomatics Office:

1.	Dans la Barre du projet, cliquez avec le bouton droit de la souris.

Le menu de raccourcis s'affiche.

2.	Sélectionnez Navigateur de zoom.

Utilisation du Navigateur de zoom

Pour utiliser le Navigateur de zoom:

  Cliquez sur une zone du projet dans le Navigateur de zoom.

NOTE: Cette zone devient le centre de la fenêtre de graphiques principale.

Pour utiliser le Navigateur de zoom pour faire un zoom:

1.	Dans le Navigateur de zoom, faites glisser-déplacer une boîte autour de la zone d'intérêt.

La fenêtre principale de graphiques affiche les mêmes données.

2.	Lorsque vous pointez au milieu de la boîte dans le Navigateur de zoom, le pointeur devient un [image]. Ce veut dire que vous pouvez faire glisser-déplacer la boîte sur la zone du projet sans modifier le zoom. L'affichage du projet dans la fenêtre de graphiques principale reflète ces mouvements.

  Pour faire un zoom à l'étendue du projet, double-cliquez dans le Navigateur de zoom.

NOTE: Si vous avez une souris à roulette, vous pouvez utiliser la roulette pour modifier la taille de la boîte du Navigateur de zoom.


Barre du projet - Vue d'ensemble

La Barre du projet consiste en des groupes nommés listant les raccourcis aux tâches les plus couramment utilisées dans Trimble Geomatics Office. Chaque raccourci est disponible dans un menu.

Pour accéder à une tâche rapidement:

1.	Cliquez sur un raccourci dans la Barre du projet.

2.	Cliquez sur un groupe pour vous déplacer à un groupe de raccourcis différent.

Pour modifier la taille des icônes de raccourci:

1.	Cliquez avec le bouton droit de la souris dans la zone de la Barre du projet.

Un menu de raccourcis s'affiche.

2.	Sélectionnez l'option Grandes icônes ou Petites icônes.

Pour masquer (ou afficher) la Barre du projet, effectuez l'une des choses suivantes:

 	Cliquez avec le bouton droit de la souris dans la zone de la Barre du projet.

Un menu de raccourcis s'affiche. Sélectionnez Masquer.

  	Sélectionnez Affichage / Barre du projet

Dans la barre d'outils, cliquez sur l'icône Barre du projet

Lorsqu'il n'y a aucun projet ouvert, les groupes Projets et Utilitaires sont disponibles. Lorsqu'un projet est ouvert, la liste de groupes entière (sauf le groupe Utilitaires) devient disponible:

  Le groupe Levé de Trimble

  Le groupe  Projets

  Le groupe Importer

  Le groupe Traiter (lorsque vous avez le module Trimble WAVE Baseline Processing (Traitement de lignes de base WAVE de Trimble) installé)

  Le groupe Ajustement (lorsque vous avez le module Trimble Network Adjustment (Ajustement de réseau de Trimble)

  Le groupe Exporter

  Le groupe CAD

  Le groupe Utilitaires


Barre du projet - le groupe Levé de Trimble

Le groupe Levé de Trimble contient des raccourcis qui fournissent d'accès rapide aux tâches les plus couramment utilisées dans la Vue de levé. Ces tâches sont listées dans l'ordre dont elles sont utilisées en général.


Barre du projet - le groupe Projets

Le groupe Projets contient des raccourcis qui fournissent d'accès rapide aux commandes du projet.


Barre du projet - le groupe Importer

Le groupe Importer contient des raccourcis qui fournissent d'accès rapide pour l'importation à partir des sources de données de levé couramment utilisées.


Barre du projet - le groupe Traiter

Le groupe Traiter contient des raccourcis qui fournissent d'accès rapide aux tâches de traitement de lignes de base couramment utilisées.

NOTE: Le module Traitement de lignes de base doit être installé.


Barre du projet - le groupe Ajustement

Le groupe Ajustement contient des raccourcis qui fournissent d'accès rapide aux tâches d'ajustement couramment utilisées.

NOTE: Le module Ajustement de réseau doit être installé.


Barre du projet - le groupe Exporter

Le groupe Exporter contient des raccourcis qui fournissent d'accès rapide pour l'exportation des données dans votre projet aux formats de données couramment utilisés.


Barre du projet - le groupe CAD

Le groupe CAD contient des raccourcis qui fournissent d'accès rapide aux tâches couramment utilisées dans la Vue en plan.


Barre du projet - le groupe Utilitaires

Le groupe Utilitaires contient des raccourcis qui fournissent d'accès rapide aux utilitaires couramment utilisés compris dans le système Trimble Geomatics Office.


Attributs - Vue d'ensemble
Attributs − Vue d'ensemble

Plus amples informations

Comment ...	Dialogue	Voir aussi
Vue d'ensemble

Un attribut est un renseignement descriptif à propos d'un code de caractéristiques. Par exemple, avec le code de caractéristiques 'arbre' vous pouvez avoir des attributs d'espèce, de hauteur, de circonférence et de condition. Chaque code de caractéristiques 'arbre' collecté sur le terrain aura ses propres valeurs pour ces attributs.

Trimble Geomatics Office prend en charge la gestion des informations d'attribut avec les points.  A l’aide de Trimble Geomatics Office vous pouvez:

  utiliser l'utilitaire Trimble Feature and Attribute Editor™ (Editeur d'attributs et caractéristiques de Trimble) pour créer des bibliothèques d'attributs et caractéristiques contenant les définitions d'attributs

   importer les fichiers Survey Controller (*.dc) contenant des attributs

  afficher et éditer les informations d'attributs

  ajouter les attributs

  exporter les donnée vers un format GIS (Système d'information géographique)

  produire un rapport des attributs

Utilisez l'utilitaire Feature and Attribute Editor pour définir un groupe d'attributs pour un code de caractéristiques. Pour chaque attribut, il faut définir un nom et un type d'attribut. Le type d'attribut détermine le format des informations d'attribut. Par exemple, si l'attribut pour 'arbre' est la hauteur, le type d'attribut est 'numérique'.

Vous pouvez également utiliser des fichiers de Dictionnaire d'attributs (*.ddf) pour collecter les informations d'attribut. Pour créer les fichiers .ddf, utilisez l'utilitaire Data Dictionary Editor™ de Trimble (Editeur du dictionnaire d'attributs de Trimble). Il fait partie du logiciel GPS Pathfinder Office.

Ordre de travail

L'ordre de travail pour la gestion des informations d'attribut à l'aide de Trimble Geomatics Office est:

1.	Créez un fichier de Bibliothèque d'attributs et caractéristiques (*.fcl) dans l'Editeur d'attributs et caractéristiques.

2.	Dans Trimble Geomatics Office, configurez un projet pour les attributs en sélectionnant le fichier .fcl ou .ddf.

3.   Transférez le .fcl ou .ddf au Trimble Survey Controller (version 7.00 ou ultérieure).

4.	Collectez les données sur le terrain.

5.	Transférez les données du terrain à Trimble Geomatics Office.

6.	Exportez les données sous forme d'un format GIS ou ascii personnalisé.


Bibliothèque d'attributs et caractéristiques - Vue d'ensemble

Plus amples informations

 Comment ...

Vue d'ensemble

Un fichier de Bibliothèque d'attributs et caractéristiques (*.fcl) est un fichier texte contenant les définitions des codes de caractéristiques, des attributs, des styles CAD et  des codes de contrôle. Pour éditer les bibliothèques d'attributs et caractéristiques, utilisez l'utilitaire Feature and Attribute Editor™de Trimble.

Les bibliothèques dattributs et caractéristiques sont un groupe d'instructions décrivant les opérations à réaliser lors du traitement des codes de caractéristiques. Par exemple:

  l'allocation d'un style de point spécifique à un point

  la création des lignes entre les points

  la définition d'attributs à collecter sur le terrain

  l'annotation des points et des lignes

Pour plus amples informations concernant la création des bibliothèques d'attributs et caractéristiques, référez-vous à l'aide en ligne de l'Editeur d'attributs et caractéristiques de Trimble.

Utilisez un fichier de Bibliothèque d'attributs et caractéristiques (*.fcl) pour:

  traiter les codes de caractéristiques

  définir les attributs pour la collecte de données sur le terrain.

Traitement de codes de caractéristiques

Pour plus amples informations concernant l'utilisation d'une bibliothèque d'attributs et caractéristiques pour le traitement des codes de caractéristiques, référez-vous à la rubrique Codes de caractéristiques – Vue d'ensemble.

Attribut

Pour configurer un projet pour les attributs, utilisez le dialogue Propriétés du projet. Vous pouvez sélectionner un fichier de Bibliothèque d'attributs et caractéristiques (*.fcl) ou un fichier de Dictionnaire d'attributs (*ddf).


Configuration d'un projet pour des attributs

 Vue d'ensemble et plus amples informations

Afin d'utiliser des attributs dans Trimble Geomatics Office, vous devez configurer le projet pour des attributs. Pour ce faire, sélectionnez le fichier Bibliothèque d'attributs et caractéristiques (*.fcl) ou le fichier de Dictionnaire d'attributs (*.ddf) que vous voulez utiliser.

ASTUCE: Pour plus amples informations à propos la sélection d'un fichier .fcl ou .ddf, référez-vous à la rubrique Configurer un projet pour les attributs.

Trimble Geomatics Office crée une table pour chaque code de caractéristiques ayant des attributs. Le fichier de bibliothèque n'est pas stocké dans le projet. Ce signifie que le projet n'est pas mis à jour s'il y a une modification à la bibliothèque.

Changement de la bibliothèque d'attributs et caractéristiques du projet

Pour changer la Bibliothèque d'attributs et caractéristiques du projet, sélectionnez-le comme décrit ci-dessus. Cependant, Trimble Geomatics Office ne vous permet de ce faire après:

  l'importation d'un fichier Survey Controller (*.dc) contenant des attributs

  l'ajout d'un enregistrement d'attributs à Trimble Geomatics Office

C'est à cause du fait que les tables de caractéristiques existantes sont supprimées, et des nouvelles tables sont créées utilisant la bibliothèque sélectionnée.

Lorsqu'il est nécessaire de changer la bibliothèque d'attributs et caractéristiques d'un projet, vous pouvez supprimer tous les points avec attributs, rouvrez le projet, et puis sélectionnez une nouvelle bibliothèque d'attributs et caractéristiques.

Trimble Survey Controller

Assurez-vous que la bibliothèque d'attributs et caractéristiques (ou fichier de dictionnaire d'attributs) utilisée dans Trimble Geomatics Office soit la même que celle utilisée sur le terrain avec  Trimble Survey Controller. Si vous utilisez des bibliothèques diverses, il se peut que certaines données d'attributs ne seront pas importées.

Pour plus amples informations à propos de l'importation des données avec attributs lorsque les bibliothèques sont différentes, référez-vous à l'Importation d'attributs avec des définitions différentes.


Fichiers de dictionnaire d'attributs - Vue d'ensemble
Fichiers de dictionnaire d'attributs − Vue d'ensemble

Plus amples informations

 Voir aussi

Vue d'ensemble

Un fichier de Dictionnaire d'attributs est un fichier définissant les caractéristiques et les attributs associés des caractéristiques.

Il peut être utilisé comme une alternative à un fichier de Bibliothèque d'attributs et caractéristiques.

Si vous avez l'utilitaire Data Dictionary Editor  (faisant partie de  GPS Pathfinder Office), vous pouvez l'utiliser pour créer des fichiers de Dictionnaire d'attributs (*.ddf). Un fichier .ddf contient les définitions d'attributs et caractéristiques.

Utilisation d'un fichier .ddf pour configurer un projet pour des attributs

Vous pouvez utiliser un fichier .ddf pour configurer un projet Trimble Geomatics Office pour des attributs. Si vous le faites, toute caractéristique de ligne ou de surface dans le fichier .ddf file est traitée comme une caractéristique de point. C'est du fait que Trimble Geomatics Office et Trimble Survey Controller (version 7.00 ou ultérieure) ne prennent en charge que les caractéristiques de point.

Utilisation d'un fichier .ddf dans le Feature and Attribute Editor

NOTE: La bibliothèque de caractéristiques utilisée pour la configuration d'un projet pour des attributs ne doit pas être la même que la bibliothèque d'attributs et caractéristiques utilisée pour le traitement des codes de caractéristiques. Cependant lorsque vous collectez des données d'attributs sur le terrain en utilisant un fichier de Bibliothèque d'attributs et caractéristiques, il faut utiliser la même bibliothèque pour le traitement des codes de caractéristiques.

Vous pouvez importer les fichiers .ddf aux fichiers de Bibliothèque d'attributs et caractéristiques (*.fcl) en utilisant l'utilitaire Feature and Attribute Editor. Vous pouvez également exporter un fichier .fcl file sous forme d'un fichier .ddf. Ces tâches servent si vous voulez utiliser les fichiers .ddf avec l'utilitaire Data Dictionary Editor (Editeur du dictionnaire d'attributs).

Utilisation d'un fichier .ddf dans le Trimble Survey Controller

Vous pouvez utiliser un fichier .ddf sur le terrain comme la bibliothèque définissant les attributs. Pour ce faire,  transférez un fichier .ddf au Trimble Survey Controller.


Importation d'attributs avec des définitions diverses

 Vue d'ensemble et plus amples informations

Lorsque vous importez un fichier (*.dc) Survey Controller contenant des attributs, les définitions d'attributs dans le fichier .dc sont comparées aux définitions d'attributs dans le projet. Lorsqu'elles diffèrent de toute façon, un dialogue s'affiche et vous demande si vous voulez:

  importer les informations d'attributs communes à toutes les deux bibliothèques d'attributs et caractéristiques ou tous les deux fichiers de dictionnaire d'attributs.

Lorsque vous sélectionnez cette option et que vous avez changé toute définition d'attribut dans le fichier de Bibliothèque d'attributs et caractéristiques du projet après avoir transférer le fichier au Trimble Survey Controller, vous perdrez les informations d'attributs pour le code de caractéristiques.  Lorsque vous n'ajoutez que des nouveaux codes de caractéristiques, aucunes données ne sont perdues.

  insérer des points avec les codes de caractéristiques mais sans les informations d'attributs.

  annuler l'importation

Pour éviter l'affichage de ce message, assurez-vous que le fichier utilisé pour configurer un projet pour des attributs soit le même que le fichier utilisé dans le Trimble Survey Controller.


Affichage/Edition des attributs dans Trimble Geomatics Office

 Vue d'ensemble et plus amples informations

Pour afficher et éditer les attributs d'un point dans Trimble Geomatics Office, utilisez la fenêtre Propriétés. Lorsque vous avez configuré un projet pour les attributs dans le dialogue Propriétés du projet, la tabulation Attribut s'affiche dans la fenêtre Propriétés. Lorsqu'il y a plus d'une caractéristique ayant attributs pour le point sélectionné, utilisez la liste déroulante pour sélectionner la caractéristique dont vous voulez afficher des attributs. Pour éditer une valeur d'attribut, cliquez sur le champ approprié dans la colonne Valeur.

Pour les attributs de menu:

  Sélectionnez une valeur de la liste de valeurs d'attribut disponibles.

Pour tout autre type d'attribut:

  Entrez la nouvelle valeur de l'attribut à l'aide du clavier. Vous ne pouvez pas utiliser le remplissage du champ pour les attributs.

NOTE: Lorsque vous éditez les informations d'attributs, Trimble Geomatics Office vérifie la définition de bibliothèque d'attributs et caractéristiques pour assurer que la nouvelle valeur soit valide.


Copie/coupure/collage des points ayant des attributs entre les projets

 Voir aussi

Pour copier, ou couper et coller des points ayant des attributs, les bibliothèques d'attributs et caractéristiques utilisées pour configurer les deux projets devraient être les mêmes.

Si les bibliothèques d'attributs et caractéristiques sont différentes, mais elles ont des définitions d'attributs communes, vous en êtes averti et vous avez l'option d'annuler le collage, de coller seulement des points (sans attributs), ou de coller des points ayant des attributs communs à toutes les deux bibliothèques. Trimble Geomatics Office enregistre les attributs qui n'ont pas été collés dans le rapport Import.rpt.

Vous ne pouvez pas sélectionner une bibliothèque différente pour la configuration d'un projet pour des attributs lorsque le projet contient déjà des points ayant des attributs. C'est à cause du fait que les définitions d'attributs ont été déjà configurées. Cependant, vous pouvez créer un nouveau projet, définir les nouvelles définitions d'attributs dans le Bibliothèque d'attributs et caractéristiques (ou fichier de Dictionnaire d'attributs), puis copier les points ayant des attributs du projet existant au nouveau projet.

NOTE: La copie ou la coupure et le collage ne fonctionne que si l'une des suivants est vrai:

  Tous les points copiés ou coupés ont des élévations.

  Le projet a une élévation par défaut spécifiée. Ce défaut est utilisé pour tout point avec une élévation nulle.

C'est à cause du fait que les coordonnées des points copiés/coupés sont réduites à l'ellipsoïde WGS-84. Le système de coordonnées du projet dans lequel vous collez sera appliqué aux points collés. Si le projet dans lequel vous collez possède un système de coordonnées différent, les coordonnées correctes s'afficheront.


Différences entre les fichiers .fcl et .ddf

Vous pouvez utiliser un fichier de Bibliothèque d'attributs et caractéristiques (*.fcl) ou un fichier de Dictionnaire d'attributs (*.ddf) dans votre projet Trimble Geomatics Office. Pour sélectionner un fichier pour le projet courant, sélectionnez Fichier / Propriétés du projet tabulation Général. Les différences entre les deux fichiers sont décrites ci-dessous.

Fichiers .fcl

Un fichier .fcl peut contenir des paramètres CAD (par exemples les styles de point, les styles de ligne, les styles d'annotation et les codes de contrôle) pour la codification ainsi que les définitions d'attributs. Lorsque vous sélectionner un fichier .fcl pour configurer un projet pour des attributs, Trimble Geomatics Office utilise les définitions d'attributs pour créer les tables requises.

Le même procédé se produit lorsque vous décidez d'exporter un fichier .fcl ayant des attributs au Trimble Survey Controller (version 7.00 ou ultérieure).

Si vous voulez utiliser des codes de caractéristiques et collecter des attributs, utilisez les fichiers .fcl. Si vous avez déjà un fichier .ddf mais vous voulez avoir des codes de caractéristiques, vous pouvez importer le fichier .ddf vers un fichier .fcl dans l'utilitaire Feature and Attribute Editor et puis utiliser le fichier .fcl.

Pour plus amples informations, référez-vous à l'aide de l'utilitaire Feature and Attribute Editor.

Fichiers .ddf

Un fichier .ddf ne peut pas contenir des paramètres CAD pour la codification. Il ne contient que les définitions d'attributs et caractéristiques. Lorsque vous traitez les codes de caractéristiques d'un fichier .ddf, aucun style de point, le ligne ou d'annotation ne sera assigné.

Vous pouvez créer un fichier .ddf de l'utilitaire Data Dictionary Editor (si vous disposez de GPS Pathfinder Office) ou de l'utilitaire Feature and Attribute Editor dans Trimble Geomatics Office. (Sélectionnez Fichier / Exporter dans l'Editeur d'attributs et caractéristiques pour exporter un fichier .ddf.)


Transfert de données - Vue d'ensemble

L'utilitaire Data Transfer de Trimble est le logiciel utilisé pour le transfert des données entre Trimble Geomatics Office et un périphérique. Un périphérique est un Survey Controller, un récepteur GPS de Trimble, une station totale conventionnelle, un carnet de terrain de format SDR, un niveau numérique, une unité de contrôle ou une carte PC. Avant de pouvoir transférer des données à l'aide de l'utilitaire Data Transfer, vous devez configurer le périphérique requis.

Si vous avez un périphérique configuré, pour accéder à l'utilitaire Data Transfer  à partir de Trimble Geomatics Office, effectuez le suivant:

1.	Sélectionnez Fichier / Importer ou Fichier / Exporter—selon ce que vous voulez faire. 	Le dialogue approprié s'affiche.

2.	Sélectionnez l'option requise.

Pour ouvrir l'utilitaire Data Transfer, sélectionnez une option que requiert une connexion à un périphérique.

Le dialogue Enregistrer sous ou Ouvrir s'affiche, selon ce que vous avez choisi.

3.	Sélectionnez le périphérique requis à partir duquel ou vers lequel vous voulez transférer des données.

Accéder à l'utilitaire Data Transfer du dehors d'un projet

Si vous n'avez pas un projet ouvert, vous pouvez transférer les fichiers entre votre ordinateur et un périphérique.

Pour ce faire:

  Dans Trimble Geomatics Office, sélectionnez Utilitaires / Transfert de données.

Accéder à l'utilitaire Data Transfer du dehors de Trimble Geomatics Office

Vous pouvez transférer les fichiers de votre ordinateur à un périphérique du dehors du logiciel Trimble Geomatics Office.

Pour ce faire:

  	Sélectionnez Démarrer / Programmes / Transfert de données.


Périphériques - Vue d'ensemble

Pour transférer les données entre un projet Trimble Geomatics Office et un périphérique ; il faut configurer le périphérique correctement.

Lorsque vous configurez un périphérique, il sera toujours disponible lors de l'importation ou de l'exportation. Pour plus amples informations, référez-vous à l'Aide de Data Transfer.


Dialogue Entrée DAT

Utilisez ce dialogue pour vérifier et corriger les valeurs dans le(s) fichier(s) GPS avant l'importation des observations GPS dans un projet Trimble Geomatics Office.

NOTE: Toute modification effectuée dans le dialogue n'aura un effet que sur les données importées. Les modifications n'auront pas un effet sur le fichier de données GPS  (*.dat).

Le dialogue s'affiche lorsque vous importez les:

  fichiers de données GPS (*.dat)

  fichiers RINEX  (*.XXo, *.XXn, *.obs, *.nav)

  fichiers DC avec des fichiers .dat associés

Utiliser

Effacez cette case à cocher lorsque vous ne voulez pas que les données soient transférées au projet.

Nom

Cette case affiche le nom du point où les données ont été observées. Lorsqu'il n'y a aucun nom assigné au point dans le fichier, l'ID de station est utilisée. Utilisez le clavier pour saisir un nom pour changer ce champ.

Nom du fichier

Cette case affiche la source de fichier des données.

Heure de début

Cette case affiche l'heure de début et la date de l'occupation.

Heure d'arrêt

Cette case affiche l'heure d'arrêt et la date de l'occupation.

Type de récepteur

Cette case liste le type de récepteur GPS utilisé pour la collecte des données.

N. de série récepteur

Cette case affiche le numéro de série du récepteur. Vous pouvez éditer ce numéro.

Hauteur d'antenne

Cette case affiche la hauteur d'antenne entrée pour l'occupation. Si la hauteur a été entrée incorrectement lors de l'occupation, utilisez le clavier pour saisir une valeur afin de changer ce champ.

Type d'antenne

Cette case affiche le type d'antenne connecté au récepteur. Si l'antenne a été entrée incorrectement lors de l'occupation, sélectionnez un type d'antenne de la liste pour changer ce champ.

Mesuré à

Cette case affiche jusqu'à où la hauteur d'antenne a été mesurée. Si la méthode a été entrée incorrectement lors de l'occupation, sélectionnez une option de la liste pour changer ce champ.

Code de caracts

Cette case affiche la valeur du code de caractéristiques assignée au point. Utilisez le clavier pour saisir une nouvelle valeur pour changer ce champ.

Remettre

Cliquez sur Remettre pour restaurer les paramètres d'origine.

ASTUCE: Pour appliquer les mêmes modifications à plus d'un point, enfoncez Ctrl et sélectionnez les champs requis sous la colonne que vous voulez modifier. Appuyer sur Maj si vous voulez sélectionner des lignes en séquence. Saisissez ou sélectionnez une valeur dans un champ sélectionné que vous voulez appliquer à tous les champs dans la sélection.


Dialogue Importation de niveau numérique

 Voir aussi

Utilisez le dialogue Importation de niveau numérique pour:

  Afficher les données d'un Fichier de niveaux numériques

  Spécifier entre quels points vous voulez calculer des élévations delta.

  Valider l'élévation du point de début

  Spécifier les erreurs de fermeture sur des points connus.

  Valider les noms de points

Le dialogue Importation de niveau numérique s'affiche lorsque vous importer des Fichiers de niveaux numériques (*.dat, *.raw). Pour ce faire:

1.	Sélectionnez Fichier / Importer. Le dialogue Importer s'affiche.

2.	Dans la tabulation Levé, sélectionnez Fichiers de niveaux numériques (*.dat, *.raw). Le dialogue Ouvrir s'affiche.

3.	Sélectionnez le Fichier niveau numérique.

4.	Le dialogue Importation de niveau numérique s'affiche.

Toute modification que vous effectuez dans le dialogue n'aura un effet que sur les données importées. Les modifications n'auront pas un effet sur le Fichier de niveaux numérique (*.dat, *.raw).

NOTE: Lorsque vous importez les données de niveaux numériques, vous pouvez spécifiez comment les points en double sont traités. Par défaut Trimble Geomatics Office fusionne les points avec le même nom lors de l'importation, donc des points en double ne sont pas créés. Dans le dialogue Options de points en double, vous pouvez utiliser l'option Fusionner points en double dans la tolérance pour spécifier la tolérance verticale avec laquelle il faut fusionner les points en double. La tolérance horizontale n'est pas utilisée lorsque vous importez des données de niveaux numériques.

NOTE: Vous pouvez importer les données directement d’un périphérique de niveau numérique DiNi. Lorsque vous les faites, le dialogue Importation de niveau numérique s’affichera.

ASTUCE: Lors de la collecte de données d'importation de niveau numérique, stockez les points que vous voulez importez dans Trimble Geomatics Office avec un préfixe ou suffixe unique. Par exemple, stockez tous les points que vous voulez utiliser comme des points de station avec le préfixe, STN. Ce vous permet de sélectionner des points facilement à l'aide du bouton Sélectionner par filtre dans le dialogue Importation de niveau numérique ou des points que vous ne voulez pas importer.

Point de début

Le nom du point de début. Le point de début dans une série de niveaux est indiqué par l'icône de point de début.

Elévation

L'élévation du point de début. Vous pouvez éditer l'élévation d'un point de début lorsque le point est sélectionné dans la table. La sélection d'une autre série de niveaux dans la table affiche le point de début de cette série dans le groupe d'élévation. La fin d'une série de niveaux et le début d'une autre série est indiquée par une ligne épaisse noire horizontale  dans la table.

Qualité d'élévation

La qualité du champ d'élévation.  Pour modifier la qualité d'élévation, cliquez sur le bouton Qualité.

Les données dans la table Niveaux numériques sont affichées dans l'ordre dont les observations ont été effectuées. Vous pouvez visualiser les colonnes suivantes:

Colonne	Qui indique
Points de station	Le nom du point et si le point a été sélectionné comme un point de station pour l'importation dans Trimble Geomatics Office.  Lorsque vous importez un fichier, toutes les observations sont sélectionnées par défaut. Le nom du point de station peut être édité.
VA	L'observation de niveau à la visée arrière.
VI	L'observation de niveau à une visée intermédiaire. Vous ne pouvez pas sélectionner la visée intermédiaire jusqu'à ce que la visée arrière correspondante est sélectionnée.
Vav	L'observation de niveau à la visée avant.
Elévation	L'élévation du point. Une icône est utilisée pour indiquer quel point a été utilisé comme le point de début du calcul de l'élévation. Ce correspond au point de début indiqué dans le groupe Elévation.  L'élévation du point de début dans chaque boucle peut être éditée du groupe Elévation. Si un point n'a pas d'élévation, un point de début n'est pas sélectionné pour calculer l'élévation du point.
Distance	La distance à la visée arrière, la visée avant ou la visée intermédiaire.
Description	La description du point dont la prise de niveau est en cours.
Deltas implantés	La différence de l'élévation entre un point implanté et la hauteur du projet pour la position du point.

Pour sélectionner des points dans la table, effectuez l'une des choses suivantes:

  Sélectionnez la case à cocher dans la colonne Point de station

  Cliquez sur l'en-tête de la colonne Point de station. Vous pouvez sélectionner tous les points dans la table en cliquant sur le début de la colonne et puis sélectionnant le premier point dans la colonne.

  Cliquez sur Sélectionner par filtre.

Une fois que vous spécifiez une sélection, les règles suivantes s'appliquent:

  Vous ne pouvez pas sélectionner une visée intermédiaire jusqu'à ce que la visée arrière précédente est sélectionnée.

  S'il y a plus d'une visée avec le même nom, toutes les autres observations (sauf les visées intermédiaires) avec le même nom seront sélectionnées.

Lorsque vous cliquez sur Sélectionner par filtre, le dialogue Définir stations de niveaux s'affiche.

NOTE: Les Niveaux numériques et Formats de stockage suivants sont pris en charge:

NOTE: Trimble Geomatics Office ne prend pas en charge des enregistrements de mesure Leica qui sont enregistrés au dehors de la série de niveaux, c'est à dire, 330 enregistrements.


Dialogue Définir stations de niveaux

Plus amples informations

 Voir aussi

Le dialogue Définir stations de niveau s'affiche lorsque vous cliquez sur Sélectionner par filtre dans le dialogue Importation de niveau numérique.  Utilisez cette option de filtre pour sélectionner ou désélectionner des points de station. Par défaut, tous les points de station sont sélectionnés dans le dialogue Importation de niveau numérique.

ASTUCE: Lors de la collecte de données d'importation de niveau numérique, stockez les points que vous voulez importer dans Trimble Geomatics Office avec un préfixe ou suffixe unique. Par exemple, stockez tous les points que vous voulez utiliser comme des points de station avec le préfixe, STN. Ce vous permet de sélectionner des points facilement à l'aide du bouton Sélectionner par filtre dans le dialogue Importation de niveau numérique .

Les options sont:

Sélectionner points avec

Utilisez cette option pour sélectionner des points de station dans la colonne Points de station du dialogue Importation de niveau numérique. Par exemple, vous pouvez sélectionner des points de station avec un préfixe ou suffixe spécifique.

NOTE:  Tout point qui n'a pas un préfixe ou suffixe particulier est effacé.

Désélectionner points avec

Utilisez cette option pour effacer, pour des points spécifique, la case a cocher dans la colonne Points de station du dialogue Importation de niveau numérique.

NOTE: Tout point qui n'a pas un préfixe ou un suffixe spécifique est coché.

Nom

Utilisez ce champ pour sélectionner des points par nom. Vous pouvez sélectionner les noms de points utilisant des noms génériques de ?, *, ou #.

Description

Utilisez ce champ pour sélectionner des points selon leur description. De nouveau, vous pouvez sélectionner la description utilisant des noms génériques.

Exemple:

Lorsque vous ne voulez pas la sélection des points avec le préfixe "VI" dans la colonne Points de station du dialogue Importation de niveau numérique:

1.	Dans le dialogue Définir stations de niveau, sélectionner l'option Désélectionner points avec.

2.	Sélectionnez l'option Nom et saisissez VI*.


L'utilisation des observations de niveau dans un ajustement de réseau

Si vous voulez effectuer un ajustement de niveau utilisant des données de niveau, vous devez avoir les erreurs standard d'observation de niveau définies. Le logiciel calcule les erreurs standards des circonstances suivantes:

  Si vous importez des données de niveau numérique, l'erreur standard (élévation delta) est calculée en utilisant le nombre de tours.

  Utilisant le format Trimble Data Exchange, l'erreur standard est calculée si vous comprenez le nombre de tours dans le format d'importation. Sinon, l'erreur standard (élévation delta) peut être utilisée directement du fichier.

Lorsqu'une erreur standard n'est pas définie, les observations de niveau ne sont pas marquées pour un ajustement.

NOTE: Vous pouvez entrer l'erreur standard ou le nombre de tours pour chaque élévation delta dans la fenêtre Propriétés.


Dialogue Echec d'exportation SIG

Ce dialogue s'affiche lors de l'échec de l'exportation SIG.

Quelques raisons possibles sont indiquées dans la table suivante:


Dialogue Sélectionner modèle de géoïde - Résolution des problèmes

Lors de l'importation d'un fichier de périphérique, si le nom du modèle de géoïde référencé dans le fichier ne correspond à aucun nom de géoïde dans la base de données de système de coordonnées (Current.csd), ce dialogue s'affiche. Choisissez le modèle de géoïde que vous voulez utiliser.

Ce dialogue s'est affiché à cause d'une des raisons suivantes:


Imprimante manquante

Pour imprimer ou tracer à partir de Trimble Geomatics Office, vous devez avoir une imprimante installée sur votre ordinateur.

Pour installer une imprimante:

1.	Cliquez sur Démarrer.

2.	Sélectionnez Paramètres / Imprimantes. La fenêtre Imprimantes s'affiche.

3.	Double-cliquez sur  Ajout d'imprimante.

Suivez les instructions d'installation pour ajouter votre imprimante à vos paramètres Microsoft Windows.

Après l'installation d'une imprimante, vous pouvez l'utiliser pour imprimer du logiciel Trimble Geomatics Office.


Sélectionner un modèle de grille de datum

Si le nom du Fichier de grille de datum combiné référencé dans le fichier .dc ne correspond à aucun Fichier de grille de datum dans la base de données de système de coordonnées, ce dialogue s'affiche.

Utilisez ce dialogue pour choisir le fichier de grille de datum combiné le plus approprié à utiliser dans le système.

Ce dialogue s'est affiché à cause d'une des raisons suivantes:


Message et dialogues lors de l'importation d'un fichier Survey Controller (*.dc)

Lorsque vous importer un fichier Survey Controller (*.dc) au logiciel Trimble Geomatics Office, le logiciel effectue un nombre de vérifications sur le fichier et vous avertit si le fichier .dc et le système de coordonnées du projet ne sont pas identiques.

La liste suivante fournit les messages et dialogues possibles qui s'affichent lorsqu'un fichier .dc est importé:

 Une ou plusieurs erreurs générées pendant l'importation du fichier

 Le fichier d'attributs et caractéristiques défini pour ce  projet est différent

 Un fichier d'attributs et caractéristiques n'a pas été défini pour le projet

  Impossible d'importer le fichier

  Dialogue Définition de projection pardéfaut

  Dialogue points en double

  Dialogue Système de coordonnées du projet

  Dialogue Sélectionner grille de datum

  Dialogue Sélectionner modèle de géoïde


Message - Une ou plusieurs erreurs générées pendant l'importation du fichier

Pendant l'importation du fichier, le logiciel a trouvé une ou plusieurs erreurs.

Effectuez l'une de choses suivantes:

  Pour annuler le message et ignorer les messages d'erreur, cliquez sur Non.

  Pour afficher les erreurs cliquez sur Oui.

Un rapport d'importation s'affiche dans l'afficheur HTML par défaut.

Cliquez sur la liaison Messages d'erreur et défilez jusqu'à la section Erreurs du rapport d'importation pour étudier les erreurs.

Si nécessaire, corrigez les erreurs et importez le fichier à nouveau.


Message Le fichier d'attributs et caractéristiques défini pour ce  projet est différent

Lors de l'importation d'un fichier Survey Controller (*.dc), la bibliothèque d'attributs et caractéristiques ou le fichier de dictionnaire d'attributs utilisé pour la configuration du projet pour des attributs a été différent de la bibliothèque ou du fichier référencé dans le fichier .dc.

Effectuez l'une de choses suivantes:

  Cliquez sur Oui pour importer les attributs communs au fichier .dc et au projet.

  Cliquez sur Non pour importer seuls les points du fichier .dc.

Afin d'éviter l'affichage de ce message à nouveau, assurez-vous que la bibliothèque utilisée soit la même que la bibliothèque utilisée pour la collecte des informations d'attribut sur le terrain.


Message - Un fichier d'attributs et caractéristiques n'a pas été défini pour le projet

Lors de l'importation d'un fichier Survey Controller (*.dc) contenant des informations d'attribut, Trimble Geomatics Office n'a pas pu trouver une bibliothèque d'attributs et caractéristiques ou un fichier de dictionnaire d'attributs pour le projet.

Effectuez l'une des choses suivantes:

  Cliquez sur OK pour n'importer que les données de levé et ignorer les informations d'attribut du fichier .dc.

  Cliquez sur Annuler pour configurer le projet des attributs dans le dialogue Propriétés du projet / tabulation Caractéristiques avec la même bibliothèque référencée par le fichier .dc, puis importez le fichier .dc à nouveau.


Impossible d'importer le fichier

Lors de l'importation d'un fichier, une erreur a été rencontrée avec le résultat que le fichier n'a pas été importé. Le message suivant s'affiche:

Impossible d'importer le fichier <nom du fichier>.

Vérifiez si le fichier est ouvert dans une autre application (par exemple, un fichier Survey Controller (*.dc) ouvert dans l'utilitaire DC File Editor). Autrement, il se peut que le fichier est endommagé.


    Codes de caractéristiques - Vue d'ensemble

Plus amples informations

Comment ...	Dialogue	Voir aussi
Vue d'ensemble

Les codes de caractéristiques sont une méthode d'enregistrer les points qu'il faut réunir pour une forme de traitement. Par exemple, vous pouvez coder chaque point collecté le long d'une ligne de barrière comme "BARRIERE" afin que, au retour au bureau, tous les points avec une code de caractéristiques BARRIERE peuvent être réunis par une style de ligne spécifique.

Pour travailler avec les codes de caractéristiques, vous devez:

  créer un fichier Bibliothèque d'attributs et caractéristiques (*.fcl)

  transférer la bibliothèque au Trimble Survey Controller

  transférer la bibliothèque au TDS Survey Pro

  collecter les données utilisant les codes de caractéristiques dans la bibliothèque

  transférer un fichier Survey Controller (*.dc) à Trimble Geomatics Office

  traiter les codes de caractéristiques

Une bibliothèque d'attributs et caractéristiques est un groupe d'instructions.

Pour créer des nouvelles bibliothèques, utilisez l'utilitaire Feature and Attribute Editor Editeur. Vous pouvez créer autant de bibliothèques que nécessaire. Lorsque les codes de caractéristiques sont traités, les instructions dans la bibliothèque informent Trimble Geomatics Office qu'est ce qu'il faut faire avec les données dans la sélection. Lorsque la valeur dans un code de caractéristiques correspond à une condition spécifiée dans la bibliothèque d'attributs et caractéristiques, les actions associées sont effectuées.

Pour plus amples informations à propos des bibliothèques d'attributs et caractéristiques, référez-vous à l'aide de l'utilitaire Feature and Attribute Editor (Editeur de caractéristiques et attributs de Trimble).


    Codes de caractéristiques - Ordre de traitement

 Vue d'ensemble et plus amples informations

Tous les codes de caractéristiques dans une sélection courante sont traités, mais la méthode de sélection que vous utilisez détermine l'ordre de traitement.

Groupes de sélection

Lors de l'importation d'un fichier Survey Controller (*.dc), un groupe de sélection est créé. Il a le même nom que le fichier .dc, et il enregistre l'ordre dont les points ont été collectés sur le terrain. Si vous utilisez cette groupe de sélection, les codes de caractéristiques sont traités selon l'ordre dont ils ont été collectés.

Autres méthodes de sélection

Vous pouvez également effectuer l'une des choses suivantes:

  Choisissez une commande du menu Sélectionner.

  Cliquez sur un point.

Avec ces méthodes, il se peut que le traitement ne sera pas effectué selon l'ordre dont les données ont été collectés.


    Codes de caractéristiques - Erreurs fréquentes

 Vue d'ensemble et plus amples informations

Les erreurs suivantes typiques peuvent se produire lors de l'utilisation des codes de caractéristiques.

Liaison incorrecte à cause de l'utilisation d'un groupe de sélection incorrect

Une liaison incorrecte peut se produire lorsque vous utilisez une sélection graphique (ou la commande Sélectionner / Tout ) pour sélectionner des points pour la codification au lieu de l'utilisation d'un groupe de sélection créé par le système pendant l'importation des données.

NOTE: Toutes les entités importées d'un fichier sont enregistrées dans un groupe de sélection. Dans le cas des fichiers Survey Controller (*.dc) le groupe de sélection est nommé selon le nom de l'étude interne du fichier. C'est à dire, le groupe de sélection créé conserve l'ordre des points importés. Dans le cas des fichiers .dc ce signifie que le groupe de sélection conserve l'ordre dans lequel les travaux de levé ont été réalisés. En général c'est nécessaire pour le traitement des codes de caractéristiques.

Mettant les codes de contrôle à côté du code de point incorrect

Par exemple: TROTT1 RB DEB

En ce cas le point est une observation d'un réverbère qui se trouve sur la ligne du trottoir. L'observation du trottoir est le début d'une nouvelle ligne de trottoir. Le réverbère est un point à tracer avec le symbole approprié. La commande DEB (Début) est appliqué au code de point qu'il suit (c'est à dire, le réverbère). Ce veut dire que la ligne du dernier point nommé TROTT1 est tracée au point courant, qui n'est pas intention du codage.

Le codage devrait être: TROTT1 DEB RB ou RB TROTT1 DEB

Manquement d'utiliser la commande DEB (Début)

Ce produira un résultat des lignes continues entre les points avec le même code, tracées dans les positions incorrectes.


Codes de champ - Vue d'ensemble
Codes de champ − Vue d'ensemble

Plus amples informations

 Comment ....

Vue d’ensemble

Les codes de champ sont des instructions spéciales qui ordonnent à Trimble Geomatics Office d’insérer des informations dans des champs. Les codes de champ sont aussi utilisés pour traiter le transfert de données de la base de données à l’importation/exportation ASCII ou les opérations de rapport personnalisées. Les codes de champ sont groupés sous plusieurs en-têtes. Pour une liste de chaque en-tête et liaisons aux détails de chaque code de champ, voir Détails de code de champ.

Trimble Geomatics Office utilise beaucoup les codes de champ. Par exemple, les codes de champ sont utilisés dans :

  le nom du projet et la date dans les en-têtes du projet lors de la création d’un nouveau projet

  annotations

  labels

  Etiquetage de point personnalisé

  importation/exportation ASCII

 définitions de rapport

Les codes de champ sont toujours contenus entre des paires de crochets. Tout caractère entre des paires de ‘[]’ est traité comme une expression possible de code de champ, et est remplacé par la valeur correspondante du code de champ, s’il y en a une.

Par exemple, vous pouvez instruire Trimble Geomatics Office de faire un étiquetage personnalisé de chaque point dans la base de données avec son nom en utilisant le code de champ ‘[Nom]’. Vous pouvez étendre ces labels pour inclure l’élévation en ajoutant le code de champ ‘[Elévation]’.

Par exemple, avec la configuration appropriée [$Zone de système de coordonnées] pourrait afficher “Zone Californie 3 0403” et [Code de caractéristiques] pourrait afficher “ARBRE”. D’autres informations, telles que la date et l’heure courantes du système d’ordinateur peuvent être sorties en utilisant les codes de champ [Date] et [Heure].

Comme autre exemple, le code de champ Nom=[Nom] Elv=[Elévation] pourrait produire le type de rapport suivant :

Nom=3397 Elv=18.762

Nom=3396 Elv=18.702

Nom=3432 Elv=?

Nom=3436 Elv=16.372

Pour accéder aux codes de champ, cliquez avec le bouton droit sur le champ auquel vous voulez ajouter le code de champ. Si les codes de champ sont disponibles, l’option Champs apparaît dans le menu de raccourcis.

Syntaxe

La syntaxe d’un code de champ comprend trois éléments :

[nom de contexte:options]

Utilisez le composant de contexte pour éliminer toute ambiguïté parmi les noms de codes de champ. Par exemple, le code de champ [Nom] affiche toujours le nom du point pour le point courant de la base de données. Si vous voulez afficher le nom du projet en utilisant un code de champ, celui-ci doit inclure le contexte $Projet (ça veut dire que le [$Nom de Projet] est requis). Autrement, le nom du point courant est retourné.

Le composant de contexte est facultatif. S’il est omis, le logiciel Trimble Geomatics Office suppose le contexte le plus logique.

Les options qui peuvent être incluses avec un code de champ sont normalement des modificateurs pour le code de champ. Ce composant est aussi facultatif.


Détails de code de champ

 Vue d'ensemble et plus amples informations

Les codes de champ sont groupés sous plusieurs en-têtes.


Codes de champ de base de données

 Vue d'ensemble et plus amples informations

Les codes de champ de base de données fournissent une sortie directe des valeurs stockées et calculées de la base de données. Vous avez l’option de pouvoir préfixer un code de champ de base de données avec le nom de la table dont les données doivent venir.

Par exemple, [Est du point] extrait la valeur Est de la table de points. Spécifier simplement [Est] aussi extrait la valeur Est de la table de point à condition que cette table soit la seule table avec un champ Est. Si toute ambiguïté est possible, le nom de la table devrait être spécifié.

Les codes de champ disponibles sont sensibles au contexte. Les noms valables dépendent du contexte mais, en général, les codes de champ associés avec les tables de bases de données sont :

Codes de champ de base de données

La valeur de champ d’une base de données est prise directement de la base de données (par exemple, Est).

Codes de champ calculés

La table suivante montre les codes de champ calculés qui sont valides :

$Centre{<separator>}

Pour un arc, il y a un code de champ spécial $Centre qui affiche soit le nom du point central (si un point de base de données était utilisé) ou les coordonnées du point central.

Vous pouvez spécifier un caractère séparateur pour l’affichage entre les valeurs de coordonnées. Si le caractère séparateur est omis, le caractère ‘séparateur de liste’ du panneau de contrôle de Microsoft Windows / paramètres régionaux est utilisé.

Première ordonnée

La valeur de la première ordonnée (Est ou Nord) selon l’ordre de sortie.

Deuxième ordonnée

La valeur de la deuxième ordonnée (Nord ou Est) selon l’ordre de sortie.

Nom de la première ordonnée

Le nom de la première ordonnée (Est ou Nord) selon l’ordre de sortie.

Nom de la deuxième ordonnée

Le nom de la deuxième ordonnée (Est ou Nord) selon l’ordre de sortie.


Codes de champ du système
Codes de champ de système

 Vue d'ensemble et plus amples informations

Les codes de champ de système donnent accès à un nombre de détails valides pour tout le système, et fournissent un nombre de fonctions spéciales. Utilisez les codes de champ de système pour ajouter des informations supplémentaires aux rapports.

NOTE: Chacun de ces codes de champ de système peut être préfixé avec $Système. Par exemple, [$Heure.Système] montre l’heure courante de l’ordinateur au moment où le code de champ est utilisé.

Bien que ces codes de champ n’aient pas besoin d’être préfixés avec $Système, utilisez ce préfixe pour éviter toute ambiguïté. Par exemple, [Heure] affiche la même valeur que [$Heure.Système].

Heure

Affiche l’heure courante pour l’ordinateur au moment où le code de champ est utilisé.

Date

Affiche la date courante du système pour l’ordinateur au moment où le code de champ est utilisé.

Résumé

Affiche le résumé de l’’entité courante’.

NOTE: Le code de champ résumé utilisé pour un point est défini dans le fichier options.ini associé au projet. Généralement, il utilise le nom et les codes de champ de caractéristiques sous la forme [Nom] : [Code de caractéristiques]

Espace:<compter>

Affiche un ou plusieurs espaces.

Par exemple, le code de champ [$Système.espace:3] fournit trois espaces.

Répéter<chaîne>:<compter>

Affiche une partie de texte répétée. Le texte que vous voulez avoir répété doit être inclus entre parenthèses (< >).

Par exemple, le code de champ [$Système.Répéter<->:80] affiche quatre-vingts caractères moins. Pour de plus amples informations, voir Ecran entrée de paramètres.

Colonne:<nombre>

Affiche les espaces de la position courante sur une ligne jusqu’à la colonne spécifiée.

Par exemple, le code de champ [$Système.colonne:40] insère des espaces jusqu’à ce que la quarantième colonne dans la ligne courante soit atteinte.

Remplacer<trouver><remplacer>

Remplace toutes les occurrences du texte spécifié ‘trouver’ avec le texte ‘remplacer’. Le code de champ Remplacer<trouver><remplacer> doit suivre les champs que trouver et remplacer doit actionner. Si le code de champ Remplacer apparaît avant le champ que vous voulez trouver ou remplacer, le champ ne sera pas changé.

Il est facultatif de spécifier le texte ‘remplacer’. S’il n’est pas spécifié, le texte ‘trouver’ sera supprimé de la ligne.

Par exemple, le code de champ [$Système.remplacer<°><degrés>] trouve toutes les occurrences d’un symbole de degré dans la ligne courante et les remplace avec le mot ‘degrés’.

Caractère:code

Insère un caractère en spécifiant son code de caractère.

NOTE: La sortie du caractère varie selon la police utilisée.

Par exemple, le code de champ [$Système.caractère:169] donne © pour la plupart des polices.

Nom de l’utilisateur

Affiche le nom d’entrée du réseau de l’utilisateur.

Nom de l’ordinateur

Affiche l’identification du réseau. Vous pouvez changer cette identification dans votre système à partir du réseau / panneau de contrôle Microsoft Windows.

A propos de

Affiche le nom du produit et le numéro de version.

Par exemple, Trimble Geomatics Office, Version 1.00.

Nom du répertoire

Il y a un nombre de codes de champ différents que vous pouvez utiliser pour fournir les noms  spécifiques de dossiers reliés au système.

Les noms de répertoire disponibles sont :

Exportation ASCII

Deux codes de champ qui sont valides seulement pour l’exportation ASCII sont:

Compteur	Affiche le nombre de codes de champ utilisés. Il peut être utilisé pour identifier individuellement chaque valeur exportée.
Ligne	Affiche le numéro de la ligne dans un fichier d’exportation. Il est fourni pour permettre les définitions ASCII qui ont un compte de ligne pour chaque ligne dans le fichier.


Dialogue d'entrée de paramètre

AVERTISSEMENT : Trimble Navigation Limited ne prend en charge l’utilisation ou la modification de cette base de données que par l’utilisation du logiciel Trimble de marque déposée Trimble Geomatics Office. Trimble ne prend en charge ni autorise l’utilisation ou la modification, ni garantit la performance de cette base de données au moyen d’autres applications de logiciel. Trimble peut à tout moment modifier le format de cette base de données la rendant incompatible avec d’autres applications de logiciel.

Ce dialogue apparaît si vous devez entrer une ou des valeurs pour compléter une interrogation de sélection ou de code de champ. Les interrogations ‘sélecter’ et les codes de champ qui exigent des paramètres ont une ellipse (…). Le dialogue apparaît dans un nombre d’endroits dans Trimble Geomatics Office.

Nom

Affiche le(s) paramètre(s) à définir.

Valeur

Saisissez la(les) valeur(s) dont vous avez besoin.

Un dialogue d’entrée de paramètre apparaît quand un rapport de base de données qui a été basé sur les résultats d’une recherche de base de données est effectué.


Codes de champ de projet

 Vue d'ensemble et plus amples informations

Les codes de champ de projet permettent d’accéder à des informations spécifiques de projet. Utilisez ces codes de champ pour ajouter des informations supplémentaires aux rapports.

NOTE: Chacun de ces codes de champ de projet peut être préfixé avec $Projet. Par exemple, [$Projection du projet] affiche le nom du type de projection du système de coordonnées utilisé pour le projet.

Bien que ces codes de champ n’aient pas besoin d’être préfixés avec $Projet, utilisez ce préfixe pour éviter toute ambiguïté. Par exemple, [Nom] affiche la même valeur que [$Nom du projet].

Nom

Affiche le nom du projet que vous avez utilisé pour le projet.

Chemin

Affiche le nom du dossier dans lequel les projets sont stockés.

NOTE: Ce code de champ fournit la même sortie que $Système.Répertoire de projets

Projection

Affiche le nom du type de projection de système de coordonnées utilisé pour le projet. Par exemple, Mercator Transverse.

Nom auto

Affiche le nom du nom de point auto courant.

Par exemple, AUTO 0009 est retourné si le précédent nom du point auto attribué était AUTO 0008

Couche CAD

Affiche le nom de la couche CAD courante.

Unités:type

Affiche le nom d’unité approprié. Vous devez toujours fournir un type.

NOTE: Distance, élévation, coordonnée et angle sont disponibles.

Par exemple, le code de champ [$Unités du projet:distance] affiche l’unité de distance telle qu’elle est configurée pour le système. Vous pouvez inclure le nom ‘:prolixe’ pour obtenir une valeur complète du code de champ. Par exemple, pour obtenir ‘mètres’ au lieu de ‘m’ utilisez [$Unités du projet:distance:prolixe].

Type:distance

Affiche le type d’affichage pour la distance. Vous pouvez configurer ceci dans le fichier / propriétés du projet / unités et format.

Type:LLH

Affiche le type d’affichage pour la latitude, la longitude, et la hauteur (locale ou WGS-84).


Codes de champ d'en-têtes

 Vue d'ensemble et plus amples informations

Les codes de champ d’en-têtes donnent accès aux données enregistrées dans les champs d’en-têtes du projet dans la Tabulation de détails du projet du dialogue de propriétés du projet. Utilisez ces codes de champ pour ajouter des informations supplémentaires aux rapports.

NOTE: Chacun de ces codes de champ d’en-têtes peut être préfixé avec $En-têtes. Par exemple, [$Titre d’en-têtes] affiche le titre pour le projet courant.

Si un des en-têtes a changé, les codes de champ d’en-têtes disponibles changent pour refléter ce changement.

Bien que ces codes de champ n’aient pas besoin d’être préfixés avec $En-têtes, utilisez ce préfixe pour éviter toute ambiguïté.

Les codes de champ d’en-têtes standard sont :

Titre

Affiche le titre du projet courant.

Description

Affiche une description du projet.

Référence

Affiche un numéro de référence du projet de telle sorte que vous puissiez relier le projet au procédé d’estimation du prix utilisé par le bureau.

Géomètre de terrain

Cette sortie de code de champ pourrait être utile pour une référence ultérieure.

Opérateur d’ordinateur

Cette sortie de code de champ pourrait être utile pour une référence ultérieure.

Date

Affiche la date de création du projet. C’est effectué automatiquement en utilisant le code de champ de date dans le projet modèle.

Datum vertical

Affiche le datum vertical pour le projet.

Equipement

Affiche le(s) nom(s) de série de votre équipement.


Codes de champ du système de coordonnées
Codes de champ de système de coordonnées

 Vue d'ensemble et plus amples informations

Les codes de champ de système de coordonnées donnent accès aux détails spécifiques du système de coordonnées pour le projet. Utilisez ces codes de champ pour ajouter des informations supplémentaires aux rapports.

NOTE: Chacun de ces codes de champ de système de coordonnées peut être préfixé avec $Système de coordonnées. Par exemple, [$Système de coordonnées.Système] affiche le nom du système de coordonnées utilisé pour le projet.

Bien que ces codes de champ n’aient pas besoin d’être préfixés avec $Système de coordonnées, utilisez ce préfixe pour éviter toute ambiguïté.

NOTE: Les codes de champ $Système de coordonnées retournent ‘pas sélectionné’ quand il n’y a pas de configuration disponible.

Système

Affiche le nom du système de coordonnées qui est utilisé pour le projet (par exemple, plan d’état 1983 des Etats Unis).

Zone

Affiche le nom de la zone du système de coordonnées qui est utilisée pour le projet (par exemple, zone 3 0403 Californie).

Datum

Affiche le nom du datum sur lequel la projection du système de coordonnées est basée (par exemple, nord américain 1983 (Conus)).

Géoïde

Affiche le nom du modèle du géoïde qui est utilisé par le projet (par exemple, GEOID96 (EU continentaux).

Site

Affiche le nom de toute définition du site pour le projet.


Codes de champ de projection

 Vue d'ensemble et plus amples informations

Les codes de champ de projection donnent accès aux détails spécifiques de la projection utilisée dans le système de coordonnées du projet. Utilisez ces codes de champ pour ajouter des informations supplémentaires aux rapports.

NOTE: Chacun de ces codes de champ de projection peut être préfixé avec $Projection. Par exemple, [$Nom de projection] affiche le nom de la projection du système de coordonnées utilisée pour le projet.

Bien que ces codes de champ n’aient pas besoin d’être préfixés avec $Projection, utilisez ce préfixe pour éviter toute ambiguïté.

Nom

Affiche le nom du type de projection (par exemple, Mercator Transverse).

Les codes de champ de projection qui restent retournent les valeurs réelles du paramètre pour la projection utilisée par le projet.

NOTE: Pas tous les codes de champ sont valides pour tous les types de projection. Ceci signifie que pas tous les codes de champ sont valides dans chaque cas. La liste suivante des codes de champ donnent la liste complète de tous les paramètres de projection qui peuvent être retournés. Selon la projection utilisée par le projet, seuls les codes de champ appropriés retourneront des valeurs. Tous les codes de champ qui ne sont pas valides pour la projection retournent ‘N/A’ (ce qui veut dire Non applicable).

Chacun des codes de champ suivants affiche la valeur qui est détaillée :

Constante nord

La valeur de constante nord pour la projection.

Constante est

La valeur de constante est pour la projection.

Constante d’élévation

La valeur de hauteur d'ellipsoïde à l’origine d’une projection de plan.

Hauteur d’origine

La valeur d’élévation à l’origine d’une projection de plan.

Latitude origine

La latitude origine pour la projection.

Longitude origine

La longitude origine pour la projection.

Facteur d’échelle

Le facteur d’échelle central pour la projection (basé sur l’unité).

Azimut

La valeur oblique par azimut des projections Mercator oblique et R.S.O. avec une définition oblique par azimut.

Convergence

La valeur d’angle oblique pour les projections Mercator oblique et R.S.O. avec une définition de convergence oblique d’origine.

Parallèle 1

Le parallèle nord utilisé dans la définition du Parallèle de Lambert 2 et la projection conique de surface égale d’Albers.

Parallèle 2

Le parallèle sud utilisé dans la définition du Parallèle de Lambert 2 et la projection conique de surface égale d’Albers.

Latitude point 1, longitude point 1, latitude point 2, longitude point 2

Ces quatre codes de champ sont utilisés par les projections Mercator oblique et R.S.O. définies par deux points, avec ces champs étant les positions de point de définition.


Codes de champ de datum

 Vue d'ensemble et plus amples informations

Les codes de champ de datum donnent accès aux détails spécifiques du datum utilisé dans le système de coordonnées du projet. Utilisez ces codes de champ pour ajouter des informations supplémentaires aux rapports.

NOTE: Chacun de ces codes de champ de datum peut être préfixé avec $Datum. Par exemple, [$Méthode de datum] affiche le nom de la méthode de datum utilisée pour le projet.

Bien que ces codes de champ n’aient pas besoin d’être préfixés avec $Datum, utilisez ce préfixe pour éviter toute ambiguïté.

NOTE: Pas tous les codes de champ sont valides pour tous les types de datum. Ceci signifie que pas tous les codes de champ sont valides dans chaque cas. La liste suivante des codes de champ donnent la liste complète de tous les paramètres de datum qui peuvent être retournés. Selon le datum utilisé par le projet, seuls les codes de champ appropriés retourneront des valeurs. Tous les codes de champ qui ne sont pas valides pour le datum retournent ‘N/A’ (qui veut dire Non applicable).

Chacun des codes de champ suivants affiche la valeur détaillée :

Méthode

Le nom de la méthode de transformation de datum utilisée (par exemple, trois-paramètres).

Translation X

La valeur X de translation (disponible pour les transformations de datum de trois- et sept- paramètres).

Translation Y

La valeur Y de translation (disponible pour les transformations de datum de trois- et sept-paramètres).

Translation Z

La valeur Z de translation (disponible pour les transformations de datum de trois- et sept-paramètres).

Rotation X

La valeur X de rotation (disponible seulement pour les transformations de datum de sept-paramètres).

Rotation Y

La valeur Y de rotation (disponible seulement pour les transformations de datum de sept-paramètres).

Rotation Z

La valeur Z de rotation (disponible seulement pour les transformations de datum de sept-paramètres).

Facteur d’échelle

La valeur du facteur d’échelle de transformation de datum de sept-paramètres exprimée en parties par million (ppm).

Nom d’ellipsoïde

Le nom de l’ellipsoïde utilisé dans le projet.

Aplatissement de l’ellipsoïde

L’aplatissement de l’ellipsoïde.

Demi-grand axe

Le demi-grand axe de l’ellipsoïde.


Codes de champ de calibration

 Vue d'ensemble et plus amples informations

Les codes de champ de calibration donnent accès aux détails spécifiques de la calibration du site GPS utilisée dans le système de coordonnées du projet. Utilisez ces codes de champ pour ajouter des informations supplémentaires aux rapports.

NOTE: Chacun de ces codes de champ de calibration peut être préfixé avec $Calibration. Par exemple, [$Rotation de calibration] affiche la valeur de rotation pour la transformation de plan pour la calibration du site GPS.

Bien que ces codes de champ n’aient pas besoin d’être préfixés avec $Calibration, utilisez ce préfixe pour éviter toute ambiguïté.

NOTE: Les codes de champ de calibration sont séparés en deux groupes relatifs aux composants horizontaux et verticaux d’une calibration du site GPS. Tous les codes de champ dans l’un ou l’autre des groupes sont valides si ce composant a été calculé. Ceci signifie que les composants de calibration verticale sont tous valides si la calibration verticale est disponible dans le système de coordonnées. La liste suivante des codes de champ donnent la liste complète de tous les paramètres de calibration qui peuvent être retournés. Tous les codes de champ qui ne sont pas valides pour la calibration retournent ‘N/A’ (qui veut dire Non applicable).

Chacun des codes de champ suivants affiche la valeur détaillée :

Origine vert nord

La valeur Nord du point d’origine de transformation verticale.

Origine vert est

La valeur Est du point d’origine de transformation verticale.

Correction constante vert

Le déplacement vertical qui est appliqué au point d’origine de transformation.

Correction vert nord

La correction verticale par unité nord qui est appliquée dans la transformation.

Correction vert est

La correction verticale par unité est qui est appliquée dans la transformation.

Origine horz nord

La valeur Nord du point d’origine de transformation horizontale.

Origine horz est

La valeur Est du point d’origine de transformation horizontale.

Translation horz nord

La valeur de translation horizontale appliquée dans la direction nord.

Translation horz est

La valeur de translation horizontale appliquée dans la direction est.

Facteur d’échelle

Le facteur d’échelle (basé sur l’unité) appliqué dans la transformation horizontale.

Rotation

La rotation appliquée dans la transformation horizontale.


Codes de champ spéciaux

 Vue d'ensemble et plus amples informations

Ces codes de champ spéciaux donnent accès à un nombre de détails pour tout l’ordinateur et fournissent un nombre de fonctions spéciales. Utilisez ces codes de champ pour ajouter des informations supplémentaires aux rapports.

$Fichier.nom de fichier

Ce code de champ affiche le contenu du fichier spécifié.

NOTE: A cause d’un conflit avec le caractère deux-points utilisé dans certaines définitions de codes de champ, vous ne pouvez pas utiliser le caractère deux-points (:) pour définir un disque de l’ordinateur. Si vous avez besoin de spécifier un disque, utilisez <disque> au lieu de ‘disque:’. Par exemple, <C> spécifie le disque C:

Trimble Geomatics Office utilise certains codes de champ spéciaux pour retourner des répertoires standard.

$Parent (où le TGOffice.exe est situé)

$Système (le sous-répertoire du système sous $Parent où le TGOffice.exe est situé)

$Projet (le répertoire du courant projet)

Par exemple [$FICHIER.<c>\temp\données.txt] ou [$fichier.$système\monfichier.txt]

NOTE: Ici, vous pouvez aussi utiliser ‘:développer’.

$Variable.env

Affiche la variable d’environnement nommée.

$Touche.touche

Affiche la chaîne avec la touche donnée, prise d’un fichier .ini (ou de répertoire).

Le format du code de champ ‘nom’ est touche{,section}{;chemin}

Si la section est omise, ‘code de champ’ est utilisé, et si le chemin est omis, on suppose que c’est $Système.

Comme avec $Fichier, le chemin peut commencer avec un des suivants :

  $Parent

  $Système

  $Projet

NOTE: Si le chemin est seulement $Projet alors le fichier .ini du projet* est utilisé (Options.ini est d’abord examiné, puis tgo_v100.ini — ce sont des fichiers .ini associés à chaque projet) Si le chemin est seulement $Système alors la zone de Trimble Geomatics Office dans le répertoire est utilisée.

$Texte.texte

Affiche le texte que vous avez saisi. C’est utile pour les rapports.

Par exemple [$Texte.code de caractéristiques:20] affiche le label ‘code de caractéristiques’ placé à 20 espaces du point.


Codes de champ de modèle de plan

 Vue d'ensemble et plus amples informations

Dans la définition des modèles de plan il y a plusieurs codes de champ spéciaux fournis. Utilisez-les dans une définition de modèle de plan pour vous permettre d’afficher les détails spécifiques au report.

NOTE: Chacun de ces codes de champ de modèle de plan peut être préfixé avec $Report. Par exemple, [$Echelle de report] affiche l’échelle de report pour le report.

Bien que ces codes de champ n’aient pas besoin d’être préfixés avec $Report, utilisez ce préfixe pour éviter toute ambiguïté.

Echelle

Affiche l’échelle de report pour le report.

Nom de première ordonnée d'origine

Affiche le nom de première ordonnée d'origine. Par exemple, avec le fichier / propriétés du projet / unités et format / option d’ordre de coordonnées configuré à Nord en premier, [$Report.Nom de première ordonnée d’origine] affiche ‘Nord’.

Nom de deuxième ordonnée d'origine

Affiche le nom de deuxième ordonnée d'origine. Par exemple, avec le fichier / propriétés du projet / unités et format / option d’ordre de coordonnées configuré à Nord en premier, [$Report.Nom de deuxième ordonnée d’origine] affiche ‘Est’.

Première ordonnée d’origine

Affiche la valeur de la première ordonnée d’origine. Par exemple, avec le fichier / propriétés du projet / unités et format / option d’ordre de coordonnées configuré à Nord en premier, [$Report.Nom de première ordonnée d’origine] affiche la valeur Nord d’origine du report.

Deuxième ordonnée d’origine

Affiche la valeur de la deuxième ordonnée d’origine. Par exemple, avec le fichier / propriétés du projet / unités et format / option d’ordre de coordonnées configuré à Nord en premier, [$Report.Nom de deuxième ordonnée d’origine] affiche la valeur Est d’origine du report.

Origine N

Affiche la valeur Nord d’origine du report.

Origine E

Affiche la valeur Est d’origine du report.

Rotation

Affiche les valeurs de rotation spécifiées pour le report.

Modèle de plan

Affiche le nom du modèle de plan tel que spécifié dans la définition du modèle de plan.


Modificateurs

 Vue d'ensemble et plus amples informations

Les modificateurs de code de champ sont des commandes spéciales que vous pouvez utiliser pour modifier la sortie d’un code de champ.

Utilisez les modificateurs pour :

  obtenir la sortie spécifique requise du code de champ

  fournir des informations supplémentaires pour le formatage ou des paramètres supplémentaires lors de la sortie de données

Un nombre de modificateurs peut être utilisé pour modifier un seul code de champ. Quand un nombre de modificateurs est utilisé, ils sont généralement spécifiés comme ci-dessous :

	[Code de champ{:Elémentdeformat}{:Elémentdeformat}{:Elémentdeformat}{:Elémentdeformat}]
où Elémentdeformat est soit un nom de modificateur ou une mise en forme. Les accolades indiquent que ce sont des éléments facultatifs.

Quand le système spécifie un modificateur, il accepte soit les majuscules soit les minuscules ou un mélange des deux.

Les modificateurs possibles sont les suivants :

Valeur de largeur (numérique)

Spécifie la largeur de la sortie d’un code de champ. Si la valeur de largeur est négative, la sortie est alignée à gauche, sinon elle est alignée à droite.

Par exemple, si Code de caractéristiques = « exemple de code », alors [Code de caractéristiques:15] affiche «     exemple de code » (ce qui signifie qu’il est allongé jusqu’à 15 caractères en ajoutant quatre espaces devant).

Valeurs décimales de largeur (numérique)

Spécifie la largeur et le nombre de décimales pour la sortie d’un code de champ. Le modificateur a la forme L.D où L est la largeur et D le nombre de décimales.

Si la valeur de largeur est négative, le champ est aligné à gauche, sinon il est aligné à droite.

Si la valeur de largeur commence par un zéro, mais n’est pas zéro juste (par exemple, 08,3), alors on allonge avec un zéro au lieu d’un espace.

Par exemple, 	si Elévation = 10,3258

Alors	[Elévation:0,3] affiche 10,326

Et	[Elévation:07,2] affiche 0010,33

Unités

Suffixent les valeurs avec leurs unités. Par exemple, m pour mètres, pdi pour pieds internationaux, et pdl pour pieds américains de levé. Vous pouvez utiliser ce modificateur pour vous assurer que les unités sont affichées même si l’option Affiche unités dans le fichier / propriétés du projet / tabulation d’unités et formats est configuré à Non.

Aucune unité

Assure que les valeurs ne sont pas suffixées avec leurs unités. Vous pouvez utiliser ce modificateur pour vous assurer qu’aucune unité n’est affichée même si l’option Affiche unités dans le fichier / propriétés du projet / tabulation d’unités et formats est configuré à Oui.

Majuscule

Convertit tous les caractères alpha qui sont retournés d’un code de champ en majuscule.

Minuscule

Convertit tous les caractères alpha qui sont retournés d’un code de champ en minuscule.

Droite

Aligne à droite la sortie d’un code de champ dans la largeur de champ spécifiée. Si aucune largeur de champ n’est spécifiée, aucun alignement ne sera effectué.

Gauche

Aligne à gauche la sortie d’un code de champ dans la largeur de champ spécifiée. Si aucune largeur de champ n’est spécifiée, aucun alignement ne sera effectué.

Centre

Aligne la sortie d’un code de champ dans la largeur de champ spécifiée. Si aucune largeur de champ n’est spécifiée, aucun alignement ne sera effectué.

Par exemple, si Code de caractéristiques = "exemple"

Alors [Code de caractéristiques:11:centre] affiche "  exemple  ".

Ce qui veut dire qu’il est retourné dans ce cas-là avec deux espaces devant et deux derrière.

Tailler

Taille la sortie à la largeur donnée (devant/ derrière si la largeur est positive ou négative).

Par exemple, si Code de caractéristiques = exemple

Alors [Code de caractéristiques:4:tailler] affiche ‘exem’

Et [Code de caractéristiques:–4:tailler] affiche ‘mple’

Développer

Développe tous les codes de champ qui se trouvent dans le texte retourné par le code de champ auquel ‘développer’ s’applique.

Par exemple, le code de champ [$Fichier.$Système\texte.txt:développer], quand il est utilisé pour importer un fichier, développe tous les codes de champ valide qui se trouvent dans le fichier Texte.txt avant de l’afficher.

Standard

Convertit la sortie à une ponctuation numérique standard (virgule pour le symbole de décimale et pas de symbole de groupement de chiffres ; ce qui veut dire, pas de séparateur de mille).

NOTE: Ce modificateur s’applique seulement pour les définitions de rapport. Les routines d’importation peuvent recevoir des données numériques sous la forme standardisée (pas de symbole de groupement de chiffres et virgule pour la décimale) ou des valeurs utilisant le symbole de groupement de chiffres et le symbole de décimale configurées dans le système Microsoft Windows. Les routines d’exportation convertissent les sorties numériques à la forme standardisée.

Inverser

Inverse le sens des codes de champ appropriés. Par exemple, [Azimut:inverser] inverse l’azimut qui serait normalement retourné pour une ligne donnée.

Quadrant

Affiche un azimut comme un gisement quadrant. Par exemple, un azimut de 228°37'25" apparaît comme N81°22'35"W.

Cercle entier

Affiche un azimut comme un gisement de cercle entier. Par exemple, si votre format d’azimut est configuré au gisement quadrant nord 81 degrés 22 minutes et 35 secondes ouest, il apparaît comme 228°37'25".

Prolixe

Utilise les termes complets où c’est approprié. Par exemples, si les unités sont abrégées à ‘m’, utilisez le prolixe modificateur pour afficher le mot ‘mètres’.

Fractionner

Fractionne la chaîne à la décimale.

Par exemple, le modificateur [Elévation:fractionner] donne l’élévation sans la décimale.

NOTE: Ce modificateur est utile pour la position d’annotation du point Sur quand vous concevez les modèles d’annotation.

Ce modificateur remplace le ‘.’ avec un seul espace mais vous pouvez l’augmenter en utilisant :fractionner<3>, par exemple.

PartEnt

Enlève la partie fractionnaire d’une valeur (jusqu’à ‘.’).

PartFrac

Enlève la partie intégrale d’une valeur (à partir de ‘.’).

Impliqué

Traite l’importation et l’exportation de fichiers ASCII où la décimale n’est pas présente en valeurs numériques mais sa position est impliquée. Le modificateur est utilisé en conjonction avec la largeur standard et les modificateurs de décimale.

Par exemple, [Nord:10,3:impliqué] sortirait la valeur Nord de 4853297,692 comme 4853297692. La même définition de codes de champ permet l’interprétation correcte de valeurs avec des décimales impliquées pour l’importation.


Mises en forme

 Vue d'ensemble et plus amples informations

Les mises en forme vous laissent construire des codes de champ.

Par exemple, vous pouvez construire un code de champ pour une date utilisant des composants de mise en forme.

[Date:<{DDDD}, {MMM} {D}-{YYYY}!>]

Ceci donne dimanche 17 Oct 1999!

NOTE: Vous pouvez inclure tout texte dans le code de champ. Dans l’exemple ci-dessus, un point d’exclamation était inclus.

Les mises en forme sont toujours incluses entre parenthèses (< >) (pour des mises en forme complètes) ou entre des accolades ({ }) (pour les composants d’une mise en forme).

Les mises en formes ne s’appliquent qu’à certains types de données. Celles-ci sont :

  Mises en forme de données

  Mises en forme d’heure

  Mises en forme d’angles/gisements

Les dates et heures fournissent les composants d’une mise en forme alors que les angles/gisements fournissent les mises en forme compètes.


Mises en forme de date

 Vue d'ensemble et plus amples informations

Les mises en forme de date fournissent le contrôle sur l’affichage du composant de date pour un code de champ. Par exemple, le code de champ de date qui inclut le composant dd affiche la date avec un zéro devant.

NOTE: Des exemples de mises en forme de date utilisant des valeurs représentatives sont disponibles sous le menu de raccourcis champs / expert / plus / option de mise en forme. Ces exemples de mises en forme sont automatiquement convertis pour utiliser les codes de champ détaillés inscrits ici.

La liste complète des types disponibles de mise en forme de date est comme ci-dessous :


Mises en forme d'heure

 Vue d'ensemble et plus amples informations

Les mises en forme d’heure fournissent le contrôle sur l’affichage du composant d’heure pour un code de champ. Par exemple, le code de champ d’heure peut inclure le composant hh pour afficher l’heure avec un zéro devant (horloge à 12 heures).

NOTE: Des exemples de mises en forme d’heure utilisant des valeurs représentatives sont disponibles sous le menu de raccourcis champs / expert / plus / option de mise en forme. Ces exemples de mises en forme sont automatiquement convertis pour utiliser les codes de champ détaillés inscrits ici.

La liste complète des types disponibles de mise en forme d’heure est :

NOTE: Les composants de l’heure du jour de la mise en forme doivent être mis dans l’ordre, avec les heures avant les minutes avant les secondes et les millisecondes et t ou tt en dernier. Vous pouvez omettre n’importe lequel des éléments de formatage.

Les composants de la date et l’heure peuvent tous les deux être utilisés pour les codes de champ de date et d’heure ; les données passées ont toutes les informations. Le cas des composants est important pour distinguer entre les minutes (m) et les mois (M).

Exemples:

<DDDD, MMM D, YYYY!> donne ‘Jeudi 14 Oct 1999!’ – qui est la date courante.

<hh mm sstt!> donne ‘13 02 07!’ – qui est l’heure courante.


Mises en forme angles / coordonnées
Mises en forme d’angles / gisements

 Vue d'ensemble et plus amples informations

Les mises en forme d’angles/gisements fournissent le contrôle de l’affichage du composant angle/gisement pour un code de champ. Par exemple, le code de champ d’azimut peut inclure le composant <dm.mm> pour afficher l’azimut en degrés et minutes décimales.

NOTE: Les mises en forme d’angles/gisements sont disponibles au menu de raccourcis champs / expert / plus / option de mises en forme.

Les types disponibles de mises en forme d’angles/gisements sont les suivants :

Secondes décimales

Affiche les angles/gisements utilisant les degrés, les minutes, et les secondes décimales (quand les unités d’angle sont configurées en degrés). Cette mise en forme insère les caractères de disposition <dms.ss>. Par exemple, [Azimut:0,3:<dms.ss>] affiche une valeur d’azimut avec trois décimales pour les secondes.

Minutes décimales

Affiche les angles/gisements utilisant les degrés et les minutes décimales (quand les unités d’angle sont configurées en degrés). Cette mise en forme insère les caractères de disposition <dm.mm>. Par exemple, [Azimut:0,4:<dm.mm>] affiche une valeur d’azimut avec quatre décimales pour les minutes.

Degrés décimaux

Affiche les angles/gisements utilisant les degrés décimaux (quand les unités d’angle sont configurées en degrés). Cette mise en forme insère les caractères de disposition <d.dd>. Par exemple, [Azimut:0,6:<d.dd>] affiche une valeur d’azimut avec six décimales pour les degrés.

Degrés décimaux signés

Affiche les angles/gisements utilisant les degrés décimaux (quand les unités d’angle sont configurées en degrés) et inclut un caractère de signe devant pour la valeur. Cette mise en forme insère les caractères de disposition <+d.dd>. Par exemple, [Latitude<+d.dd>:0.8] affiche une valeur de latitude avec huit décimales pour les degrés y compris le signe de caractère devant (+ or –) pour la valeur.


Substitution de code de champ

 Vue d'ensemble et plus amples informations

La substitution de code de champ reconnaît certaines séquences spéciales de caractères qui ne sont pas des codes de champ. Celles-ci sont exposées ci-dessous :

Exemples:

Pour créer un format de sortie ASCII délimité par la tabulation, utiliser les codes de champ suivants :

[Nom]\t[Est]\t[Nord]\t[Elévation]\t[Code de caractéristiques]

Pour annoter une ligne avec l’azimut et la distance sur deux lignes, utiliser les codes de champ suivants :

[Azimut]|[Distance]


Codes de champ et importation ASCII

 Vue d'ensemble et plus amples informations

Les codes de champ permis dans un format d’importation sont :

  Seulement les champs sur la table de points.

  Pas de sous-enregistrements et pas de codes de champ de système.

(Une exception est [$Système.Répéter<trouver><remplacer>] qui est permis.)

Certains codes de champ calculés sont permis (par exemple, Latitude/Longitude).

$Sauter

Il y a un moyen spécial ($Sauter) pour aider à l’interprétation d’entrée ASCII :

Espace

Ce code de champ saute le nombre d’espaces donnés par la largeur, ou si la largeur n’est pas donnée, saute jusqu’à un espace non-vide. Vous devez vous assurer que vous placez un espace avant le ‘sauter’ afin que Trimble Geomatics Office sache où le champ précédent finit. Par exemple :

[Nom] [$Sauter.Espace][Nord] [$Sauter.Espace][Est] [$Sauter.Espace]

Chiffre

Ce code de champ saute un chiffre (‘0’...’9’) et les trois caractères suivants (‘+’, ‘–’, ’.’). S’il y a une largeur, il saute ce nombre de caractères appropriés ; sinon, il saute jusqu’au premier caractère qui n’est ni un chiffre, ni un des caractères ‘+’, ‘–‘ & ‘.’.

NOTE: Ceci ne marche pas avec des nuls ou des exponentiels.

Ligne

Ce code de champ saute le nombre de lignes données dans la largeur, ou une ligne.

Texte

Ce code de champ saute le nombre donné de caractères ou jusqu’à un caractère de non-impression (par exemple, ‘\n’).

Colonne

Ce code de champ saute jusqu’au nombre de colonne donné.

A:chaîne

Ce code de champ saute jusqu’au premier caractère après la chaîne donnée (par exemple, [$Sauter.à:ENTITES]).

NOTE: La chaîne ne peut pas inclure de nouvelles lignes.

Il y a certaines ‘fonctions de formatage’ spéciales.

Standard

Cette fonction de formatage convertit l’entrée d’une ponctuation standard : ‘.’ est changé au caractère décimal.

Sauter

Cette fonction de formatage lit le champ comme un champ normal mais ne le stocke pas (par exemple [Point.Horz Prec:10,3:sauter]).

= (Signe égal)

Cette fonction de formatage attribue la valeur du champ en utilisant le reste de la chaîne de formatage. Il n’examine pas le fichier (par exemple, [Code de caractéristiques =importé]). De tels champs doivent arriver en premier dans la chaîne du code de champ de définition (sinon, l’analyse grammaticale s’y perd). Par exemple ‘[Elévation:=0,0][Nom],[Nord],[Est]’

NOTE: Quand vous définissez un format d’importation ASCII, il est important soit de spécifier la largeur dans un code de champ (par exemple, [WGS Latitude:11]) ou de spécifier le caractère séparateur entre les codes de champ (par exemple, [WGS Latitude],[WGS Longitude],[WGS Hauteur] où les virgules indiquent la fin des ‘champs’). C’est parce que quand les routines d’importation essaient d’interpréter les valeurs d’un fichier ASCII, elles utilisent la définition de formatage pour calculer la portion de chaque fichier qui doit être lue dans un champ donné de base de données. La largeur ou les caractères séparateurs permettent de localiser les portions appropriées des lignes de fichier.

NOTE: Pour afficher les heures d’événement à la milliseconde près, ajoutez la mise en forme d’heure suivante au code de champ, <H:mm:ss.nnn>, où nnn est le nombre de secondes décimales. Par exemple, [Heure d’obs. de point:<H:mm:ss.nnn>]. Pour augmenter la précision temporelle, augmentez le nombre de secondes décimales. Par exemple, nnnn sorts les 4 premières secondes décimales.


Dialogue de paramètres de code de champ

Utilisez ce dialogue pour compléter le code de champ. Le titre du dialogue indique le contexte.

Nom

Cette colonne affiche le(s) paramètre(s) à définir.

Valeur

Saisissez la(les) valeur(s) dont vous avez besoin.


Transfert des fichiers au Trimble Survey Controller - Vue d'ensemble

Plus amples informations

Comment ...	Dialogue

Vue d'ensemble

Vous pouvez transférer des fichiers requis sur le terrain au Trimble Survey Controller. Pour ce faire, vous devez avoir une configuration de périphérique Survey Controller dans l'utilitaire Data Transfer.

Vous pouvez transférer les types de fichier suivants:

  Fichiers d'antenne

  Fichiers de grille de datum (*.cdg)

  Fichiers de carnet de terrain (*.dc)

  Fichiers de dictionnaire d'attribut (*.ddf)

  Points MNT

  Bibliothèques de caractéristiques et attributs (*.fcl)

  Fichiers de grille de géoïde (*.ggf)

  Fichiers de grille nationale UK (*.pgf)

  Fichiers CSV

  Fichiers de carte d’arrière-plan


Option Détails d'antenne au Survey Controller

Si vous avez édité des informations d'antenne dans un fichier d'antenne, utilisez cette option pour transférer ces informations au Trimble Survey Controller.

  Pour le Trimble Survey Controller 6.00 ou plus récente, les détails d'antenne sont stockés dans le fichier Antenna.ini.

 Pour le Trimble Survey Controller 5.03 ou précédente, les détails d'antenne sont stockés dans le fichier Antenna.dat.

NOTE: Pour exporter les détails d'antenne, le fichier d'antenne approprié doit se trouver dans le dossier Program Files\Common Files\Trimble\Config.


Option Fichiers Survey Controller (*.dc)

Les fichiers Survey Controller (*.dc) fournissent la liaison entre Trimble Geomatics Office et le Trimble Survey Controller. Ces fichiers sont créés lorsque vous importez une étude de l'étude Trimble Survey Controller ou lors de l'exportation d'un projet Trimble Geomatics Office à ce format de fichier.

A l'aide de Trimble Geomatics Office, vous pouvez:

  Importer un fichier Survey Controller (*.dc)

  Exporter un projet Trimble Geomatics Office à un fichier Survey Controller (*.dc) afin d'être transféré au Survey Controller plus tard.

Pour importer un fichier Survey Controller (*.dc):

1.	Sélectionnez Fichier / Importer. Le dialogue Importer s'affiche.

2.	Sélectionnez la tabulation Levé.

3.	Sélectionnez l'option Fichiers Survey Controller (*.dc). Le dialogue Ouvrir s'affiche.

4.	Sélectionnez le fichier que vous voulez importer d'un dossier et cliquez sur Ouvrir.

Pour exporter le projet Trimble Geomatics Office courant à un fichier Survey Controller (*.dc):

1.	Sélectionnez Fichier / Exporter. Le dialogue Exporter s'affiche.

2.	Sélectionnez la tabulation Levé.

3.	Sélectionnez l'option Fichiers Survey Controller (*.dc).

4.	Cliquez sur Options pour ouvrir le dialogue Options carnet de terrain et spécifiez le nom interne du fichier et la version du fichier Survey Controller.

5.	Cliquez sur OK pour accepter les paramètres et retourner au dialogue Exporter.

6.	Cliquez sur OK. Le dialogue Enregistrer sous s'affiche.

7.	Sélectionnez le dossier dans lequel vous voulez enregistrer le fichier.

8.	Par défaut, le nom du fichier utilise le nom du projet. Saisissez un nom dans le champ Nom du fichier pour le changer et cliquez sur Enregistrer.


Option Survey Controller

A l'aide de Trimble Geomatics Office vous pouvez transférer les données directement entre un projet Trimble Geomatics Office et le Trimble Survey Controller. Lors de l'importation d'une étude du Trimble Survey Controller elle est convertie au format d'un fichier Survey Controller (*.dc)  utilisé par Trimble Geomatics Office.

Utilisez Trimble Geomatics Office pour:

  importer une étude du Trimble Survey Controller

  exporter un projet Trimble Geomatics Office au Trimble Survey Controller

Pour accéder à l'option, effectuez l'une des choses suivantes:

  Pour importer une étude Survey Controller à un projet Trimble Geomatics Office:

1.	Sélectionnez Fichier / Importer. Le dialogue Importer s'affiche.

2.	Sélectionnez la tabulation Levé.

3.	Sélectionnez l'option Périphérique de levé et cliquez sur OK. Le dialogue Transfert de données Ouvrir s'affiche.

  Pour exporter un projet Trimble Geomatics Office au Survey Controller:

1.	Sélectionnez Fichier / Exporter. Le dialogue Exporter s'affiche.

2.	Sélectionnez la tabulation Levé.

3.	Sélectionnez l'option Périphérique de levé.

4.	Cliquez sur Options pour ouvrir le dialogue Options carnet de terrain pour spécifier le nom interne du fichier et la version du fichier Survey Controller.

5.	Cliquez sur OK pour accepter les paramètres et retourner au dialogue Exporter.

6.	Cliquez sur OK. Le dialogue Enregistrer sous s'affiche.

7. 	Sélectionnez un périphérique de survey controller.

8.	Cliquez sur Ouvrir. L'ordinateur connecte au Trimble Survey Controller.

9.	Sélectionnez l'option Mémoire principale ou Carte de données de levé (si disponible).

10.	Cliquez sur Ouvrir. Le logiciel lit le dossier que vous avez sélectionné, et affiche toutes les études Trimble Survey Controller dans le dossier.

11.	Dans le champ Nom du fichier, saisissez un nom pour le nouveau fichier .dc.

12.	Cliquez sur Enregistrer pour créer le fichier .dc et le transférer au Trimble Survey Controller.

NOTE: Si vous transférez des données vers ou depuis le Trimble Survey Controller de version 7.60, utilisez le format du fichier de version 7.50 dc.


Dialogue Options Carnet de terrain

Utilisez ce dialogue pour spécifier le type de carnet de terrain auquel vous exporter des données.

Pour y accéder:

1.	Sélectionnez Fichier / Exporter. Le dialogue Exporter s'affiche.

2.	Sélectionnez la tabulation Levé.

3.	Sélectionnez l'une des options suivantes:

  Points MNT au fichier

  Points MNT au Survey Controller

  Bibliothèque de caractéristiques et attributs au Survey Controller

  Fichiers SDR

  Survey Controller

  Fichiers Survey Controller

4.	Cliquez sur Options.

NOTE: Les champs disponibles dépendent de l'option sélectionnée.

Version

Sélectionnez la version relative au format de données.

Etude / Nom du fichier

Affiche le nom de l'étude ou le type de fichier à exporter. Lors de l'exportation d'une étude, le nom du projet Trimble Geomatics Office est saisi par défaut. Utilisez le clavier pour saisir un nouveau nom pour le fichier.

Sortir caractéristiques et attributs

Sélectionnez cette case à cocher pour transférer la bibliothèque de caractéristiques et attributs utilisée pour le projet ainsi que toute information de caractéristiques et attributs assignée aux points  au projet que vous exportez. Cette option n'est disponible que pour un Trimble Survey Controller 7.00 ou plus récent.

Exporter lignes et arcs

Sélectionnez cette case à cocher pour exporter les lignes et les arcs utilisés dans le projet au Trimble Survey Controller. Cette option n’est disponible que si vous avez des lignes et des arcs compris dans le groupe de sélection, et n’est disponible que pour le Trimble Survey Controller 10.00 ou plus récent.

Sortir sous

Sélectionnez les points sortis au fichier de périphérique.

La sortie peut être sous forme de positions de grille ou WGS-84. Selon le format de données et la version sélectionnés, ainsi que la définition du système de coordonnées de la base de données, les deux options ne sont pas toujours disponibles. Par exemple, le Trimble Survey Controller avant la version 4.00 n'accepte que les positions WGS-84; les versions Sokkia SDR20 n'acceptent que les positions de grille.

Le système de coordonnées du projet peut influencer les données sorties aux fichiers Trimble Survey Controller. Les méthodes de transformation de datum Grille de datum (par exemple, NAD83 HPGN, et NADCON) et Régression multiple ne sont pas prises en charge par le Trimble Survey Controller 6.50 ou précédente. Le Trimble Survey Controller 7.00 prend en charge les grilles de datum.

Lorsque les données sont exportées au Trimble Survey Controller que ne prend pas en charge la méthode de transformation de datum spécifiée dans le projet, l'enregistrement de datum dans le fichier .dc créé a une méthode de datum configurée sur Aucun. Les positions de grille incorrectes ne peuvent pas être affichées dans le Trimble Survey Controller. Pour plus amples informations, référez-vous à la rubrique Méthodes de transformation de datum et le Trimble Survey Controller.

De la même façon, le Trimble Survey Controller 4.00 ne peut pas travailler avec les modèles de géoïde. Si un projet utilise un Modèle de géoïde, la méthode de datum est configurée sur Aucun. Ainsi, c'est assuré que le Trimble Survey Controller 4.00 ne peut pas afficher des données incorrectes.

Pour utiliser les données de coordonnées de grille sur le terrain, vous devez effectuer une calibration sur le terrain pour lier les données de grille aux données GPS observées. Pour plus amples informations, référez-vous à la rubrique Modèles de géoïde et Survey Controllers.

Détails de fichier

Sélectionnez la case à cocher Ajouter pour inclure les données d'un fichier .dc existant dans l'exportation. Le bouton Parcourir  devient disponible. Cliquez sur Parcourir pour rechercher le fichier .dc que vous voulez inclure dans le nouveau fichier. Les données du projet courant sont alors ajoutés au nouveau fichier.

Cette option sert si vous avez un fichier .dc contenant des points de contrôle que vous voulez ajouter aux données à exporter. Ainsi le fichier .dc existant est conservé dans son état d'origine.


Unités de carnet de terrain

Lors de l'importation d'un fichier Survey Controller (.dc) ou d'un fichier au format SDR, la définition des unités dans l'en-tête du fichier est utilisée pour les besoins de conversion des unités. Trimble Geomatics Office fonctionne toujours avec des données SI (métriques) afin que les données importées d'un fichier de périphérique sont converties de façon appropriée.

Pour sélectionner comment les valeurs s'afficheront sur l'écran:

1.	Sélectionnez Fichier / Propriétés du projet.

Le dialogue Propriété du projet s'affiche.

2.	Sélectionnez la tabulation Unités et format.

Utilisez cette tabulation pour sélectionnez les unités d'affichage.

NOTE: Les fichiers SDR n'ont que les paramètres des unités de distance pieds et mètres. Pour les fichiers SDR, les valeurs pieds sont interprétées comme les valeurs de pieds américains si les unités de coordonnées sont configurées sur Pieds américains.


Option Notes d'attribut du fichier DC (Survey Controller V4)

Les fichiers de notes d'attribut sont des fichiers simples délimités par des virgules. Ils comprennent des attributs de point collectés sous forme de notes d'attribut dans un carnet de terrain exécutant le Trimble Survey Controller (version 3.00 ou 4.00). Cette option n'exporte pas des attributs collectés à l'aide d'une bibliothèque de caractéristiques et attributs (disponible avec le Trimble Survey Controller 7.00 ou ultérieure).

Utilisez cette option pour exporter seuls les attributs collectés en utilisant des enregistrement de note.

Le procédé est:

1.	Importez un fichier du Trimble Survey Controller 3.00 ou 4.00 contenant des notes d'attribut dans le projet Trimble Geomatics Office.

Un groupe de sélection est créé par Trimble Geomatics Office. Ce groupe de sélection est assigné le même nom que le nom de l'étude interne du fichier.

2.	Sélectionnez le groupe de sélection créé. Le groupe de sélection est utilisé pour la lecture des points d'un projet pour le fichier d'attributs.

Exportez le fichier .dc avec des notes d'attribut.

Un exemple du type de fichier produit est:

1000,5315.362,1056.274,82.361,ARBRE,TYPE,PIN,HAUTEUR,25,CIRCUMFERENCE,0.5

1001,5382.728,1132.692,75.472,ARBRE,TYPE,BOULEAU,HAUTEUR,20,CIRCUMFERENCE,0.4

1002,5298.048,1162.472,55.927,PE,TENSION,110KV,CABLES,3

1003,5308.682,1192.329,62.854,PE,TENSION,110KV,CABLES,3


Option Points MNT au Survey Controller

Utilisez cette option pour exporter les points MNT à un périphérique exécutant le Trimble Survey Controller.

Pour ce faire:

1.	Sélectionnez Fichier / Exporter. Le dialogue Exporter s'affiche.

2.	Sélectionnez la tabulation Levé.

3.	Sélectionnez l'option Points MNT au Survey Controller.

4.	Cliquez sur Options pour spécifier le nom du fichier et la version du Trimble Survey Controller à laquelle vous exportez.

5.	Cliquez sur OK pour accepter les paramètres et retourner au dialogue Exporter.

6.	Cliquez sur OK. Le dialogue Transfert de données Enregistrer sous s'affiche.

Les points sélectionnés pour l'exportation doivent être des point à grille réguliers. C'est à dire, ils doivent former un grille orientée Nord-Sud et Est-Ouest avec des intervalles égaux entre les points dans les deux directions. En général, les points à grille sont importés dans le système à partir d'un fichier de coordonnées ASCII. Il faut vous assurer que seuls les points à grille réguliers sont sélectionnés pour l'exportation, et qu'aucun point 'hors grille' n'a été compris dans la sélection courante.

Une fois que l'option d'exportation est sélectionnée, le système Trimble Geomatics Office trie les points à grille sélectionnés dans l'ordre approprié et crée un fichier formaté spécialement requis par le Trimble Survey Controller.


Option Points MNT au fichier (*.dtx)

Utilisez cette option pour exporter les points MNT à un fichier de modèle numérique de terrain (*.dtx).

1.	Sélectionnez Fichier / Exporter.

Le dialogue Exporter s'affiche.

2.	Sélectionnez la tabulation Levé.

3.	Sélectionnez l'option Points MNT au fichier (*.dtx).

4.	Cliquez sur Options pour spécifier le nom du fichier et la version du Survey Controller à laquelle vous exportez.

5.	Cliquez sur OK pour accepter les paramètres et retourner au dialogue Exporter.

6.	Cliquez sur OK. Le dialogue Enregistrer sous s'affiche.

7.	Saisissez le nom du fichier et sélectionnez le dossier dans lequel vous voulez l'enregistrer.

8.	Cliquez sur OK.

Les points sélectionnés pour l'exportation doivent être des point à grille réguliers. C'est à dire, ils doivent former un grille orientée Nord-Sud et Est-Ouest avec des intervalles égaux entre les points dans les deux directions. En général, les points à grille sont importés dans le système à partir d'un fichier de coordonnées ASCII. Il faut vous assurer que seuls les points à grille réguliers sont sélectionnés pour l'exportation, et qu'aucun point 'hors grille' n'a été compris dans la sélection courante.

Une fois que l'option d'exportation est sélectionnée, le système Trimble Geomatics Office trie les points à grille sélectionnés dans l'ordre approprié et crée le fichier dans le format Modèle numérique de terrain (*.dtx).


Option Grille de datum au Survey Controller (*.cdg)

Plus amples informations

 Voir aussi

Utiliser cette option pour créer les fichiers de Grille de datum combiné pour transférer au Trimble Survey Controller.

Pour ce faire:

1.	Sélectionnez Fichier / Exporter.

Le dialogue Exporter s'affiche.

2.	Sélectionnez la tabulation Levé.

3.	Sélectionnez l'option Grille de datum au Survey Controller (*.cdg).

Le dialogue Exporter grille de datum s'affiche.

Les fichiers de Grille de datum stockent les transformations de grille de datum pour une surface définie. A l'aide de l'utilitaire Coordinate System Manager™, vous pouvez définir un système de coordonnées qui utilise un fichier de Grille de datum comme la transformation de datum du système (par exemple, NADCON). Puis vous pouvez sélectionner un système de coordonnées Trimble Geomatics Office. Pour utiliser le fichier de Grille de datum sur le terrain, vous devez transférer le fichier de Grille de datum sous forme d'un fichier de Grille de datum combiné au Trimble Survey Controller 7.00 ou ultérieure. Pour plus amples informations concernant quelles méthodes de transformation de datum sont disponibles aux versions de Trimble Survey Controller différentes, référez-vous à la rubrique Transformations de datum et Survey Controllers.

Un fichier de grille de longitude et un fichier de grille de latitude existent dans le dossier Program Files\Common Files\Trimble\Geodata pour la grille de datum que vous avez sélectionné. Ces fichiers doivent être combinés avant l'exportation au Trimble Survey Controller. Lorsque vous choisissez l'option Créer un fichier de grille de datum combiné pour l'exportation vers le Survey Controller, le logiciel combine les fichiers associés à la grille de datum spécifié dans le système de coordonnées du projet.

Lorsque vous créez un fichier de Grille de datum combiné, et l'utilisez avec le Trimble Survey Controller le nom de la grille de datum d'origine est stocké dans le fichier .dc, par exemple NADCON (HPGN).

Si vous importez le fichier .dc au Trimble Geomatics Office, il utilise ce nom pour déterminer si le système de coordonnées utilisé sur le terrain correspond au système de coordonnées défini pour le projet Trimble Geomatics Office.

NOTE: Vous ne pouvez créer un fichier de Grille de datum combiné que du fichier de Grille de datum utilisé actuellement, mais vous pouvez exporter tout fichier de Grille de datum combiné.

Vous pouvez:

  transférer un fichier de Grille de datum combiné (.cdg) existant

  créer un nouveau fichier de Grille de datum combiné (.cdg) pour exporter au Trimble Survey Controller.

NOTE: Vous pouvez également utiliser l'utilitaire Grid Factory pour combiner les fichiers de Grille de datum. Pour installer l'utilitaire Grid Factory, utilisez l'option d'installation personnalisée de Trimble Geomatics Office.


Transformations de datum et Trimble Survey Controllers

Trimble Geomatics Office prend en charge l'utilisation des définitions de Grille de datum et Régression Multiple pour les transformations de datum.

Un nombre de définitions du système de coordonnées utilisant ces méthodes de transformation de datum sont fournies avec le logiciel.

Les versions précédentes la version 7.00 du Trimble Survey Controller ne prennent pas en charge ces deux types de transformation de datum. Le Trimble Survey Controller 7.00 ne prend pas en charge les grilles de datum.

Trimble Geomatics Office ne peut pas créer:

  un fichier .dc Trimble Survey Controller 6.50 ou précédente que fait référence à l'une ou l'autre de ces deux méthodes de transformation de datum.

  un fichier .dc Trimble Survey Controller 7.00 que fait référence à une transformation de datum à régression multiple.

Lorsque les points d'un projet utilisant une méthode non prise en charge sont exportés à un fichier .dc Trimble Survey Controller, Trimble Geomatics Office effectue la sortie de  l'enregistrement de datum avec la méthode configurée sur Aucun.

Ce signifie que dans le Trimble Survey Controller il n'est pas possible de changer l'affichage des points entre WGS-84 et les positions de grille. En outre il signifie que dans le travail sur le terrain il est nécessaire d'effectuer une calibration si vous avez transféré des positions de grille, avant de pouvoir effectuer tout autre travail sur le terrain utilisant ces points.

Si les positions de grille ont été transférées pour les besoins de calibration, alors ce n'est pas un problème. Si les positions WGS-84 sont transférées vous pouvez commencer l'implantation immédiatement. Cependant, il n'est pas possible d'afficher les coordonnées de grille à moins qu'une calibration sur le terrain n'est effectué.


Dialogue Exporter une grille de datum

Utilisez cette option pour exporter un fichier de Grille de datum (*.cdg).

Pour y accéder:

1.	Sélectionner Fichier / Exporter. Le dialogue Exporter s'affiche.

2.	Sélectionnez la tabulation Levé.

3.	Sélectionner l'option Grille de datum vers Survey Controller (*.cdg).

4.	Cliquez sur OK.

Lorsque vous sélectionner l'option Grille de datum vers Survey Controller (*.cdg), le dialogue Exporter une grille de datum s'affiche.

Exporter le fichier de Grille de datum combiné au Survey Controller

Sélectionnez cette option pour exporter un fichier de Grille de datum combiné au Trimble Survey Controller (version 7.00 ou ultérieure). Vous pouvez choisir d'exporter tout fichier de Grille de datum combiné existant dans le dossier GeoData. Si vous sélectionnez cette option, le dialogue Sélectionner s'affiche.

Créer une Grille de datum combiné pour l'exportation au Survey Controller

Sélectionnez cette option pour créer un nouveau fichier de Grille de datum combiné utilisant le fichier de Grille de datum utilisé par le système de coordonnées du projet courant. Le dialogue Créer un fichier de grille de datum combiné s'affiche.


Dialogue Enregistrer sous (Grille de datum/Exportation de grille de géoïde)

Utilisez ce dialogue pour saisir un nom pour le fichier Grille de datum combiné (*.cdg) ou le fichier Grille de géoïde (*.ggf).

Cliquez sur OK pour ouvrir le dialogue Transfert de données Enregistrer sous.


Dialogue Créer un fichier de grille de datum combiné

Utilisez ce dialogue pour définir la surface pour laquelle vous voulez créer un fichier de Grille de datum combiné (*.cdg). Puis vous pouvez envoyer le fichier de Grille de datum combiné au Trimble Survey Controller afin de l'utiliser sur le terrain.

Pour créer un fichier .cdg, le système de coordonnées du projet doit utiliser une grille de datum pour son système de coordonnées. Pour plus amples informations concernant l'utilisation des fichiers de Grille de datum, référez-vous à l'aide en ligne de l'utilitaire Trimble Coordinate System Manager™.

Pour y accéder:

1.	Sélectionnez Fichier / Exporter. Le dialogue Exporter s'affiche.

2.	Sélectionnez la tabulation Levé.

3.	Sélectionnez l'option Grille de datum au Survey Controller (*.cdg).

4.	Cliquez sur OK. Le dialogue Exporter une grille de datum s'affiche.

5.	Sélectionnez l'option Créer un fichier de grille de datum combiné pour l'exportation vers le Survey Controller.

6.	Cliquez sur OK.

La boîte jaune sur le globe définit la surface couverte par les fichiers de Grille de datum de latitude et de longitude.

Utilisez les outils pour faire glisser déplacer une boîte sur une surface du globe à l'intérieur de la grille de datum que vous voulez couvrir avec la grille de datum combiné.

Propriétés du fichier

Le groupe Propriétés du fichier affiche les fichiers utilisés pour créer le fichier .cdg.

Grille de datum

Affiche le fichier de Grille de datum utilisé pour créer le fichier .cdg.

Fichier de grille latitude

Affiche le fichier de grille de latitude utilisé pour créer le fichier .cdg.

Fichier de grille longitude

Affiche le fichier de grille de longitude utilisé pour créer le fichier .cdg.

Propriétés de grille

Le groupe Propriétés de grille affiche les propriété de la grille que vous définissez.

Limites de latitude

Affiche les limites de latitude de la boîte courant.

Limites de longitude

Affiche les limites de longitude de la boîte courant.

Taille du fichier

Affiche la taille du fichier .cdg créé. La valeur Taille de fichier change lorsque vous changez la forme de la boîte sur le globe.

Nom du Survey Controller

Affiche le nom interne du fichier .cdg qui sera utilisé dans le Trimble Survey Controller. Le champ Nom du Survey Controller passe par défaut au nom de la grille de datum originale. Si le fichier .cdg est utilisé dans une étude Trimble Survey Controller, le nom interne est référencé. Lors de l'importation du fichier de Carnet de terrain (*.dc), Trimble Geomatics Office utilise le nom interne pour déterminer si la grille de datum combiné utilisée dans le fichier .dc correspond à la grille de datum du projet Trimble Geomatics Office.

NOTE: Si vous éditez ce champ, assurez vous que vous n'ajouterez qu'un suffixe ou un préfixe au nom. Ainsi il sera assuré que le nom est unique et permet à la grille de datum originale d'être reconnue lors de l'importation. Vous ne pouvez pas continuer jusqu'à ce que le champ Nom du Survey Controller contient le nom de la grille de datum.

Propriété de globe

Le groupe Propriétés de globe affiche la position du curseur et fournit des options pour configurer l'apparence du globe.

Position du curseur

Affiche la position de latitude et longitude du pointeur de la souris dans le globe.

Afficher frontières entre Etats

Sélectionnez cette case à cocher pour afficher les frontières entre Etats dans le globe.

Afficher frontières nationales

Sélectionnez cette case à cocher pour afficher les frontières nationales dans le globe

Afficher fleuves

Sélectionnez cette case à cocher pour afficher les fleuves dans le globe.


Option Fichier de géoïde vers Survey Controller (*.ggf)

Plus amples informations

Comment ...	Dialogue	Voir aussi

Vue d'ensemble

Les Fichiers de grille de géoïde (*.ggf) stockent la séparation entre l'ellipsoïde WGS-84 et le géoïde pour la surface définie. Pour plus amples informations concernant les modèles de géoïde, référez-vous à la rubrique Modèles de géoïde – Vue d'ensemble.

Si un fichier .ggf est utilisé comme partie du système de coordonnées d'un projet, des élévations approximatives des points GPS sont calculées. Pour utiliser un fichier .ggf sur le terrain, utilisez Trimble Geomatics Office pour transférer le fichier .ggf au Trimble Survey Controller 6.00 ou ultérieure.

Pour ce faire:

1.	Sélectionnez Fichier / Exporter. Le dialogue Exporter s'affiche.

2.	Sélectionnez la tabulation Levé.

3.	Sélectionnez l'option Fichiers de géoïde vers Survey Controller

Le dialogue Exporter un modèle de géoïde s'affiche.

Avec cette option, vous pouvez sélectionner de:

  Transférer tout fichier .ggf dans le dossier Program files\Common files\Trimble\GeoData.

  Effectuer une sous-grille du fichier .ggf utilisé dans le projet Trimble Geomatics Office afin d'extraire un plus petit fichier de Grille de géoïde. Ce procédé réduit la taille du fichier .ggf que vous envoyez au périphérique TSC1 exécutant le Trimble Survey Controller 6.00 ou ultérieure.

Pour plus amples informations concernant la prise en charge des modèles de géoïde pour les versions du Trimble Survey Controller différentes, référez-vous à la rubrique Modèles de géoïde et Survey Controllers.

Lorsque vous créez un nouveau fichier .ggf en effectuant une sous-grille, le nom du modèle de géoïde original est stocké (par exemple, GEOID96 Conus). Ce nom interne est également compris dans les fichiers .dc utilisant le fichier .ggf à sous-grille. Lors de l'importation d'un fichier .dc, Trimble Geomatics Office utilise ce nom pour déterminer si le modèle de géoïde utilisé sur le terrain correspond au modèle de géoïde défini pour le projet Trimble Geomatics Office.

NOTE: Vous pouvez également utiliser l'utilitaire Grid Factory pour effectuer une sous-grille des fichiers de modèle de géoïde. Pour installer l'utilitaire Grid Factory, utilisez l'option d'installation personnalisée de Trimble Geomatics Office.


Option Fichier de géoïd au TDS Survey Pro CE (*.ggf)

Plus amples informations

Comment ...	Dialogue	Voir aussi
Vue d’ensemble

Les fichiers de Grille de géoïde (*.ggf) stockent la séparation entre l’ellipsoïde WGS-84 et le géoïde pour la superficie définie. Pour plus amples informations concernant les modèles de géoïde, référez-vous à la rubrique Modèles de géoïde − Vue d’ensemble.

Si un fichier .ggf est utilisé comme partie du système de coordonnées d’un projet Trimble Geomatics Office, les élévations approximatives des points GPS sont calculées. Pour utiliser un fichier .ggf sur le terrain, utilisez Trimble Geomatics Office afin de transférer le fichier .ggf au TDS Survey Pro CE (v3.5 ou ultérieur.)

Pour ce faire:

1.	Sélectionnez Fichier / Exporter. Le dialogue Exporter s'affiche.

2.	Sélectionnez la tabulation Levé.

3.	Sélectionnez l’option Fichier de géoïde vers TDS Survey Pro CE.

Le dialogue Exporter un modèle de géoïde s’affiche.

Avec cette option vous pouvez choisir de:

  Transférer tout fichier .ggf dans le dossier Program files\Common files\Trimble\GeoData.

  Faire une sous-grille du fichier .ggf utilisé dans le projet Trimble Geomatics Office courant pour extraire un fichier de Grille de géoïde plus petit. Ce procédé réduit la taille du fichier .ggf que vous envoyez au périphérique TSCE exécutant TDS Survey Pro CE (v3.5 ou ultérieur).

NOTE: Vous pouvez également utiliser l’utilitaire Grid Factory pour faire des sous-grilles des fichioers de modèle de géoïde. L’utilitaire Grid Factory fait parti de l’option Personnalisé lors de l’installation Trimble Geomatics Office.


Modèles de géoïde et Trimble Survey Controller

Les modèles de géoïde peuvent être transférés au Trimble Survey Controller 6.00 (ou ultérieure). Le Trimble Survey Controller utilise ces modèles de géoïde lors du changement de l'affichage des points entre les positions WGS-84 et les positions (NEE) de grille.

En général, les modèles de géoïde sont des fichiers très grands. Normalement il n'est pas possible de télécharger un modèle de géoïde complet au Trimble Survey Controller 6.00 ou ultérieure.

Utilisez l'option Fichier / Exporter / Fichiers de géoïde vers Survey Controller pour créer un modèle de géoïde plus petit pour votre surface. Puis vous pouvez transférer ce fichier plus petit au Trimble Survey Controller.

Versions de Trimble Survey Controller précédentes

Les points suivants sont relatifs aux versions précédentes du Trimble Survey Controller.

Version 5.00

C'est une version militaire et n'est pas disponible pour la sortie générale.

Version 4.00

Cette version ne peut pas travailler avec les modèles de géoïde. Si Trimble Geomatics Office écrivait une définition du système de coordonnées entier suivie des positions de points sous forme de positions WGS-84 ou de grille vers un fichier .dc du Trimble Survey Controller 4.00, des hauteurs/élévations incorrectes pourraient être affichées dans le Trimble Survey Controller.

Ce se produirait dans le Trimble Survey Controller lors du changement de l'affichage des données, avec des hauteurs WGS-84 affichées lorsque les positions de grille ont été transférées, et des élévations de point de grille incorrectes affichées lorsque les positions WGS-84 ont été transférées.

Afin d'éviter cette situation, Trimble Geomatics Office effectue la sortie de l'enregistrement de datum de la définition du projet, puis vous avertit que ce n'est pas possible de convertir entre les valeurs NEE et WGS-84 à cause d'un modèle de géoïde, et vous avertit qu'une transformation vide a été ajoutée. Celui-ci est suivi d'un autre enregistrement de datum avec la méthode configurée sur Aucun. Un enregistrement de datum ultérieur toujours annule tout enregistrement de datum précédent. Ce signifie que dans le Trimble Survey Controller 4.00 il n'est pas possible de changer l'affichage des points entre les positions WGS-84 et de grille. Ainsi l'affichage de toute erreur erronée est évitée. .

Pour le travail sur le terrain, ce signifie que si vous téléchargez des positions de grille, vous devez effectuez une calibration. Vous devez le faire avant de commencer tout autre travail sur le terrain utilisant ces points. (Si les positions de grille ont été transférées pour les besoins de calibration ce n'est pas un problème).

NOTE: Si les positions WGS-84 sont transférées, vous pouvez commencer l'implantation immédiatement. Cependant, il n'est pas possible d'afficher les coordonnées de grille à moins qu'un calibration sur le terrain n'est effectuée.

Versions précédentes la version 4.00

Les versions pécédentes du Trimble Survey Controller ne peut prendre en charge que le téléchargement des positions WGS-84 et de ce fait ne sont pas affectées pour l'utilisation d'un modèle de géoïde dans un projet.


Dialogue Exporter un modèle de géoïde

Lorsque vous sélectionnez l'option Fichiers de géoïde vers Survey Controller (*.ggf), ou l’option Fichiers de géoïde vers TDS Survey Pro CE (*.ggf), le dialogue Exporter un modèle de géoïde s'affiche.

Exporter un modèle de géoïde existant au Survey Controller

Sélectionnez cette option pour exporter un fichier de Grille de géoïde au Trimble Survey Controller de version 6.00 ou ultérieure, ou au TDS Survey Pro CE ou ultérieure. Vous pouvez exporter tout fichier de Grille de géoïde existant dans le dossier GeoData. Si vous sélectionnez cette option, le dialogue Sélectionner s'affiche.

Créer une sous-grille d'un modèle de géoïde pour l'exportation au Survey Controller

Sélectionnez cette option pour effectuer une sous-grille du fichier de Grille de géoïde utilisé dans le système de coordonnées du projet courant. Le dialogue Créer fichier de grille de géoïde s'affiche.


Dialogue Créer fichier de grille de géoïde

Utilisez ce dialogue pour définir la zone pour laquelle vous voulez créer un nouveau fichier de Grille de géoïde qui est basée sur le fichier de Grille de géoïde utilisé dans le système de coordonnées du projet.

Pour y accéder:

1.	Sélectionnez Fichier / Exporter. Le dialogue Exporter s'affiche.

2.	Sélectionnez la tabulation Levé.

3.	Sélectionnez l'option Fichiers de géoïde vers Survey Controller (*.ggf) ou l’option Fichiers de géoïde vers TDS Survey Pro CE (*.ggf).

4.	Cliquez sur OK. Le dialogue Exporter un modèle de géoïde s'affiche.

5.	Sélectionnez l'option Créer une sous-grille d'un modèle de géoïde pour l'option d’exportation.

6.	Cliquez sur OK.

La boîte jaune sur le globe définit la surface couverte par les fichiers de géoïde.

Utilisez les outils pour faire glisser déplacer une boîte sur une surface du globe à l'intérieur de la grille de géoïde que vous voulez couvrir avec la grille de géoïde. La taille du fichier de la boîte courante est affichée.

Propriétés du fichier

Le groupe Propriétés du fichier affiche les propriétés du fichier de Grille de géoïde utilisé pour créer le fichier .ggf.

Géoïde

Affiche le modèle de géoïde utilisé dans le système de coordonnées du projet.

Fichier de grille de géoïde

Affiche le nom du fichier de Grille de géoïde utilisé dans le modèle de géoïde.

Propriétés de grille

Le groupe Propriétés de grille affiche les propriété de la grille que vous définissez.

Limites de latitude

Affiche les limites de latitude de la boîte courant.

Limites de longitude

Affiche les limites de longitude de la boîte courant.

Taille du fichier

Affiche la taille du fichier de Grille de géoïde à créer. La valeur Taille de fichier change lorsque vous changez la forme de la boîte sur le globe.

Nom

Affiche le nom interne du fichier de sous-grille qui sera utilisé dans le logiciel Trimble Survey Controller ou TDS Survey Pro CE.  Le champ Nom du Survey Controller passe par défaut au nom du modèle de géoïde original. Si le fichier .ggf est utilisé dans une étude Trimble Survey Controller, le nom interne est référencé. Lors de l'importation du fichier de Carnet de terrain (*.dc), Trimble Geomatics Office utilise le nom interne pour déterminer si le modèle de géoïde utilisé dans le fichier .dc correspond au modèle de géoïde du projet Trimble Geomatics Office.

NOTE: Si vous éditez ce champ, assurez-vous que vous n'ajouterez qu'un suffixe ou un préfixe au nom. Ainsi il sera assuré que le nom est unique et permet le modèle de géoïde original d'être reconnu lors de l'importation. Vous ne pouvez pas continuer jusqu'à ce que le champ Nom du Survey Controller contient le nom du modèle de géoïde.

Propriétés de globe

Le groupe Propriétés de globe affiche la position du curseur et fournit des options pour configurer l'apparence du globe.

Position du curseur

Affiche la position de latitude et longitude du pointeur de la souris dans le globe.

Afficher frontières entre Etats

Sélectionnez cette case à cocher pour afficher les frontières entre Etats dans le globe.

Afficher frontières nationales

Sélectionnez cette case à cocher pour afficher les frontières nationales dans le globe

Afficher fleuves

Sélectionnez cette case à cocher pour afficher les fleuves dans le globe.

NOTE: Vous pouvez également utiliser l'utilitaire Grid Factory pour effectuer une sous-grille des fichiers de modèle de géoïde. L’utilitaire Grid Factory fait partie de l’option Personnalisé lors de l’installation de Trimble Geomatics Office.


Option Bibliothèque de caractéristiques et attributs au Survey Controller (*.fcl)

 Vue d'ensemble et plus amples informations

Pour exporter une bibliothèque de caractéristiques et attributs au Trimble Survey Controller:

1.	Sélectionnez Fichier / Exporter. Le dialogue Exporter s'affiche.

2.	Sélectionnez la tabulation Levé.

3.	Sélectionnez l'option Bibliothèque de caractéristiques et attributs au Survey Controller (*.fcl)

Le dialogue Sélectionner bibliothèque d'attributs et caractéristiques s'affiche.

Utilisez cette option pour exporter une bibliothèque d'attributs et caractéristiques au Trimble Survey Controller afin que vous pouvez collecter des codes de caractéristiques sur le terrain. Vous pouvez également utiliser la bibliothèque d'attributs et caractéristiques pour collecter des attributs pour les codes de caractéristiques avec des attributs définis.


Dialogue Sélectionner bibliothèque d'attributs et caractéristiques

Utilisez ce dialogue pour sélectionner la bibliothèque d'attributs et caractéristiques que vous voulez transférer au Périphérique de levé.

Parcourir

Cliquez sur Parcourir pour sélectionner une bibliothèque.

Fichier

Le champ Fichier par défaut est le nom du dernier fichier exporté.

Exporter attributs

Vous ne pouvez le faire que si vous exportez au Trimble Survey Controller de version 7.00 ou ultérieure. Sélectionnez cette case à cocher pour exporter les définitions d'attribut contenues dans la bibliothèque d'attributs et caractéristiques.

Utilisez cette option si vous voulez collecter des informations d'attributs sur le terrain.

Si vous effacez cette case à cocher, tout attribut défini dans le fichier de bibliothèque n'est pas exporté. Ce champ n'est pas disponible si vous exportez au logiciel TDS Survey Pro CE ou Trimble 3600/5600 TDS.

OK

Cliquez sur OK une fois que vous avez choisi la bibliothèque d'attributs et caractéristiques appropriée.

Le dialogue Transfert de données Enregistrer sous s'affiche.

NOTE: Les codes de contrôle sont exportés au Trimble Survey Controller.


Dialogue Options de bibliothèque de caractéristiques et attributs

Utilisez ce dialogue pour configurer la version du Trimble Survey Controller à laquelle vous exportez.

Pour y accéder:

1.	Sélectionnez Fichier / Exporter. Le dialogue Exporter s'affiche.

2.	Sélectionnez la tabulation Levé.

3.	Sélectionnez l'option Bibliothèque de caractéristiques et attributs au Survey Controller.

4.	Cliquez sur Options.

Version

Si vous sélectionnez une version du Trimble Survey Controller précédente la version 7.00, la case à cocher Exporter attributs dans le dialogue Sélectionner bibliothèque d'attributs et caractéristiques n'est pas disponible. C'est à cause du fait que ces versions ne prennent pas en charge les attributs.


Option Dictionnaire d'attributs au Survey Controller

 Vue d'ensemble et plus amples informations

Utilisez cette option pour exporter un fichier de dictionnaire d'attributs au Trimble Survey Controller de version 7.00 ou ultérieure. Dans le Trimble Survey Controller, utilisez le fichier de dictionnaire d'attributs pour collecter les attributs.

Pour exporter un fichier de dictionnaire d'attributs au Trimble Survey Controller:

1.	Sélectionnez Fichier / Exporter. Le dialogue Exporter s'affiche.

2.	Sélectionnez la tabulation Levé.

3.	Sélectionnez l'option Fichier de dictionnaire d'attributs au Survey Controller (*.ddf).

4.	Cliquez sur OK.

Le dialogue Sélectionner s'affiche.


Dialogue Sélectionner (Exportation des fichiers de dictionnaire d'attributs au Trimble Survey Controller)

Utilisez ce dialogue pour sélectionner le fichier de Dictionnaire d'attributs (*.ddf)  que vous voulez transférer au Trimble Survey Controller.

Lorsque vous cliquez sur Ouvrir, le dialogue Transfert de données Enregistrer sous s'affiche.


Transfert des données du Trimble Survey Controller - Vue d'ensemble

 Comment ...

Vue d'ensemble

Vous pouvez connecter à un périphérique exécutant le Trimble Survey Controller et transférer des études à Trimble Geomatics Office. Lors de l'importation les études sont converties aux  Fichiers de carnet de terrain (*.dc). Pour ce faire, vous devez avoir un périphérique Survey Controller configuré dans l'utilitaire Data Transfer.

NOTE: Lorsque toute étude a des fichiers de données brutes GPS Trimble (*.dat) associés, les fichiers .dat sont importés avec les études. Si vous importez un fichier .dat plus d'une fois au même projet, les occupations existantes dans le projet ne sont pas sélectionnées par défaut dans le dialogue Entrée DAT.

Lors de l'importation d'un fichier .dc, Trimble Geomatics Office utilise la classe assignée à chaque point dans le Trimble Survey Controller et la méthode utilisée pour déterminer la position du point afin d'assigner une qualité de point au point. Un Recalcul utilise la qualité pour déterminer la position calculée pour le point.

NOTE: Lorsque vous transférez des données vers ou depuis le Trimble Survey Controller 7.60, utilisez le format de fichier .dc de version 7.50 dc.

NOTE: Si vous avez l'intention d'effectuer un ajustement de réseau utilisant les données téléchargées d'un Trimble Survey Controller, assurez-vous de saisir une pression, une température et une correction de courbure et réfraction dans le Style de levé du Survey Controller lors de la réalisation du levé.


Enregistrements de note - Vue d'ensemble

Les notes dans le projet sont des notes entrés de l'utilisateur à partir du Trimble Survey Controller. Aucune note des informations de levé, telle que 'Initialisation obtenue' et 'Initialisation perdue' n'est mise dans le projet.

Les enregistrements de note sont utilisées dans les versions plus anciennes du Trimble Survey Controller pour les raisons suivantes:

Lorsque vous utilisez des versions plus anciennes du Trimble Survey Controller et que vous voulez utiliser les enregistrements de note, choisissez le dialogue Sélectionner / Par types pour sélectionner tous les enregistrements de note dans le projet.


Fichiers d'enregistreur de données Sokkia (*.sdr)

Trimble Geomatics Office prend en charge les données du format Fichier d'enregistreur de données Sokkia (*.sdr).

A l'aide de Trimble Geomatics Office, vous pouvez:

  importer un Fichier d'enregistreur de données Sokkia (*.sdr) à un projet Trimble Geomatics Office.

  exporter un projet Trimble Geomatics Office à un Fichier d'enregistreur de données Sokkia (*.sdr).

Pour importer un Fichier d'enregistreur de données Sokkia (*.sdr):

1.	Sélectionnez Fichier / Importer. Le dialogue Importer s'affiche.

2.	Sélectionnez la tabulation Levé.

3.	Sélectionnez l'option Fichiers d'enregistreur de données Sokkia (*.sdr). Le dialogue Ouvrir s'affiche.

4.	Sélectionnez le fichier que vous voulez importer d'un dossier et cliquez sur Ouvrir.

Pour exporter le projet Trimble Geomatics Office courant à un Fichier d'enregistreur de données Sokkia (*.sdr):

1.	Sélectionnez Fichier / Exporter. Le dialogue Exporter s'affiche.

2.	Sélectionnez la tabulation Levé.

3.	Sélectionnez l'option Fichiers SDR (*.sdr).

4.	Cliquez sur Options pour ouvrir le dialogue Options de carnet de terrain pour spécifier le nom interne du fichier et la version du Fichier d'enregistreur de données Sokkia.

5.	Cliquez sur OK pour accepter les paramètres et retourner au dialogue Exporter.

6.	Cliquez sur OK. Le dialogue Enregistrer sous!JumpID(`DataXfer.hlp',`IDH_RDM_send_Save_As_dialog') s’affiche.

7.	Sélectionnez le dossier auquel vous voulez exporter le fichier, entrez un nom du fichier et cliquez sur Enregistrer.


Transfert des fichiers de grille nationale UK au Trimble Survey Controller

 Comment ...

Utilisez l'utilitaire Data Transfer pour transférer les fichiers de grille de projection de grille nationale UK (*.pgf) directement à un Trimble Survey Controller 7.5 ou ultérieure.

Vous pouvez transférer tout fichier .pgf existant trouvé dans le dossier Program Files\Common Files\Trimble\Geodata.

NOTE: Cette option d'exportation n'est disponible que lors de l'exécution de l'utilitaire Data Transfer en mode autonome. Cette option n'est pas disponible à partir du dialogue Exporter de Trimble Geomatics Office.


Remplissage du champ - Vue d'ensemble

Plus amples informations

 Comment ...

Vue d'ensemble

Utilisez le remplissage du champ pour saisir des valeurs dans le champs. Par exemple, vous pouvez remplir un champ de couches en cliquant sur un point qui se trouve dans la couche voulue.

Dans les champs qui reconnaissent le remplissage du champ, le pointeur dans la fenêtre graphique indique:

NOTE: Si l'entité sélectionnée dans la fenêtre de graphiques ne peut pas fournir la valeur requise par le champ souligné, le champ dans le dialogue reste inchangé.

Certains champs se remplissent comme un groupe de valeurs. Un seul clique remplit tous les champs groupés. Par exemple, les champs Nord, Est et Elévation en générale forment un groupe. Pour remplir un seul champ, sélectionnez-le, enfoncez Alt et cliquez.

La sélection de Trimble Geomatics Office passe au champ vide suivant après le remplissage du champ.

Utilisations particulières du remplissage du champ

Le remplissage du champ a les utilisations particulières suivantes:

  Pour saisir la valeur de l'azimut opposant lors du remplissage d'un champ d'azimut, enfoncez Maj et cliquez. Il n'est pas possible d'utiliser le remplissage du champ lorsque vous ajoutez des points par azimut et distance.

  En enfonçant Ctrl pendant l'utilisation de remplissage du champ vous évitez de passer instantanément aux entités—seules les coordonnées sont choisies. Ainsi vous pouvez choisir les coordonnées les plus près d'une entité.

  Utilisez les commandes du menu Affichage (par exemple, Zoom et Déplacer) pour localiser des entités avec les valeurs requises pour le remplissage du champ. Une fois localisé, vous devez rendre le champ actif avant de copier la valeur.


Saisie des angles

Les chiffres à droite de la décimale sont interprétées selon la configuration de l'angle, de l'unité et du format.

Pour configurer le format, utilisez la tabulation Unités et format dans le dialogue Propriétés du projet.

Par exemple:

La saisie des chiffres 10.1234


Conversions de l'unité

Pour configurer les unités d'affichage pour les champs ayant les unités (tels que, l'affichage à l'écran, l'importation, l'exportation et les rapports) utilisez la tabulation Unités et format dans le dialogue Propriétés du projet.

Toute valeur numérique saisie dans un champ est interprétée comme une unité de la valeur configurée pour le système. Par exemple, lorsque les paramètres de l'élévation sont configurés sur mètres, toute valeur numérique saisie dans un champ d'élévation est interprétée comme une valeur métrique.

Vous pouvez annuler cette configuration en saisissant la valeur numérique suivie d'une entrée alphabétique.

Le système vous permet de saisir d'autres valeurs en spécifiant une unité. Par exemple, lorsque vous saisissez une élévation comme '25 pdsUS', Trimble Geomatics Office reconnaît l'unité (comme pieds américains) et interprète la valeur comme il faut.

La table suivante donne les valeurs qui s'appliquent à cette caractéristique:


Le déplacement des entités

Pour déplacer ou faire tourner une entité sélectionnée avec la souris, utilisez les handles d'édition.

Par exemple, vous pouvez changer la position d'une annotation afin de meilleur présenter l'information y contenue et améliorer la disposition de la fenêtre graphique principale.

Texte/Annotations

Pour déplacer les entités de texte ou des annotations, ou toutes les deux, déplacez le pointeur de la souris sur la trame de texte et localisez l'un des deux handles. Le pointeur de la souris change à une icône appropriée quand vous mettez le pointeur sur le handle. Le pointeur de déplacement vous permet de déplacer l'entité. Le pointeur de rotation vous permet de pivoter l'entité.

Par exemple:

Sélectionnez le handle d'édition requis et déplacez. La position à laquelle vous libérez le bouton de la souris est maintenue.

NOTE: A moins que le type d'ancre ne soit pas dans la position à droite, le handle de rotation sera dans la position en haut à droite. De la même façon, lorsqu'un type d'ancre à droite est sélectionné, le handle de rotation, sera dans la position en haut à gauche.

Les handles d'édition peuvent être trouvés dans les positions suivantes:

NOTE: La ligne de base indique la positions la plus basse des lettres qui ne comprennent pas de descendeurs. (C'est à dire, l'espace du fond jusqu'à la ligne de base est la zone dans laquelle les descendeurs en bas de casse sont positionnés).

Courbes

Pour changer la tangente d'entrée ou la tangente de sortie des courbes, sélectionnez une courbe (ou des courbes), déplacez le pointeur de la souris sur la courbe (ou les courbes) et trouvez les handles de rotation. Il y en a deux handles pour chaque courbe (un handle de tangente d'entrée et un handle de tangente de sortie).

Le pointeur se change à [image] lorsque vous mettez le pointeur sur le handle d'édition.

Le pointeur de rotation vous permet de pivoter la courbe ou les courbes sélectionnées.

NOTE: Avec deux courbes qui se joignent, vous pouvez pivoter les deux courbes en utilisant un seul handle d'édition. Trimble Geomatics Office automatiquement fixe et maintient une tangente commune entre les courbes qui se joignent.

Pour passer instantanément aux entités existantes, enfoncez Ctrl. Par défaut, le passage instantanément aux entités existantes est désactivé.

Points

Le déplacement d'un point (ou des points) constitue une translation simple des points de base de données. Pour déplacer un point ou des points sélectionnés, sélectionnez un point, déplacez le pointeur sur le point et trouvez le handle d'édition. Le pointeur se change à [image] lorsque vous mettez le pointeur sur le handle. Lorsque le(s) point(s) se trouve dans la position requise, libérez le bouton. Le dialogue Déplacer points et texte s'affiche. Utilisez ce dialogue pour éditer et confirmer les valeurs.

Le déplacement des points cause tous les points et le texte sélectionnés de se déplacer.


Points n'ayant aucun nom

Dans le projet, chaque entité est enregistrée avec un handle unique. Le handle est un valeur de nombre entier utilisée par Trimble Geomatics Office pour l'identification du point. Ainsi, même lorsqu'un point n'a aucun nom assigné, Trimble Geomatics Office reconnaît le point de son handle interne.

Lorsqu'un point n'ayant aucun nom assigné a besoin d'un nom (par exemple, lorsqu'il est sélectionné comme un point de fin dans le dialogue Ajouter point), le handle de base de données même s'affiche. Un signe d'égalité (=) préfixe le nom. C'est ainsi que le logiciel identifie quel point vous avez sélectionné.


Barre d'état

La barre d'état se trouve en bas de la fenêtre de graphiques. Elle affiche des informations concernant l'état courant du logiciel, y compris les informations de coordonnées.

Lorsque vous déplacez la souris sur la surface d'affichage Timeline, la barre d'état affiche des informations de l'heure. Il peut afficher l'heure locale de Windows ou les secondes/semaines GPS, selon la configuration de votre option de l'heure.

Lorsque Trimble Geomatics Office est en train d'exécuter certaines tâches, tel que le traitement des lignes de base, la barre d'état devient un indicateur de progression.

Pour plus amples informations concernant les indicateurs de la barre d'état, consultez le Manuel de l'utilisateur de Trimble Geomatics Office.


Barre d'état - Coin de la fenêtre

Le coin de la fenêtre vous permet de redimensionner la fenêtre graphique principale.


Indicateur de progression

L'affichage de la barre d'état se change à un indicateur de progression lorsque Trimble Geomatics Office est en train d'exécuter certaines tâches.

Une brève description de la tâche s'affiche, suivie d'un graphique à barre qui montre le pourcentage de la tâche complété.

Lorsqu'une tâche peut être interrompue, l'indicateur de progression affiche le message Appuyer sur ECHAP pour annuler. Lorsqu'il s'affiche, appuyez sur Echap pour terminer la tâche. Lorsqu'il est impossible d'interrompre une tâche, ce message ne s'affiche pas.


Commande Restaurer

Utilisez cette commande pour retourner la fenêtre active à sa taille et sa position d'origine.

Pour y accéder:

1.	Effectuez l'une des choses suivantes:

  Cliquez sur le bouton Trimble Geomatics Office [image] sur la barre des titres.

  Cliquez avec le bouton droit sur la barre des titres.

  Cliquez sur le bouton Système [image] sur la barre de menus.

Le menu Système s'ouvre.

2.	Sélectionnez Restaurer du menu Système.


Commande Déplacer

Utilisez cette commande pour afficher une flèche à quatre têtes pour vous permettre de déplacer la fenêtre ou la boîte de dialogue active à l'aide des touches de direction.

Pour y accéder:

1.	Effectuez l'une des choses suivantes:

 Cliquez sur le bouton Trimble Geomatics Office [image] sur la barre des titres.

 Cliquez avec le bouton droit sur la barre des titres.

 Cliquez sur le bouton Système [image] sur la barre de menus.

Le menu Système s'ouvre.

2.	Sélectionnez Déplacer du menu Système.

NOTE: Cette commande n'est plus disponible lorsque vous maximisez la fenêtre.


Commande Taille

Utilisez cette commande pour afficher une flèche à quatre têtes pour vous permettre de dimensionner la fenêtre active à l'aide des touches de direction.

Pour y accéder:

1.	Effectuez l'une des choses suivantes:

 Cliquez sur le bouton Trimble Geomatics Office [image] sur la barre des titres.

 Cliquez avec le bouton droit sur la barre des titres.

 Cliquez sur le bouton Système [image] sur la barre de menus.

Le menu Système s'ouvre.

2.	Sélectionnez Taille du menu Système.

Après que le pointeur s'est changé à une flèche à quatre têtes:

1.	Appuyez sur l'une des touches de direction (touche de direction à gauche, à droite, en haut ou en bas) pour déplacer le pointeur à la bordure que vous voulez déplacer.

2.	Appuyez sur une touche de direction pour déplacer la bordure.

3.	Appuyez sur Entrée lorsque la fenêtre est la taille souhaitée.

NOTE: Cette commande n'est plus disponible lorsque vous maximisez la fenêtre.


Commande Minimiser

Utilisez cette commande pour réduire la fenêtre de Trimble Geomatics Office à une icône.

Pour y accéder:

1.	Effectuez l'une des choses suivantes:

 Cliquez sur le bouton Trimble Geomatics Office [image] sur la barre des titres.

 Cliquez avec le bouton droit sur la barre des titres.

 Cliquez sur le bouton Système [image] sur la barre de menus.

Le menu Système s'ouvre.

2.	Sélectionnez Minimiser du menu Système.


Commande Maximiser

Utiliser cette commande pour agrandir la fenêtre active à l'espace disponible.

Pour y accéder:

1.	Effectuez l'une des choses suivantes:

 Cliquez sur le bouton Trimble Geomatics Office [image] sur la barre des titres.

 Cliquez avec le bouton droit sur la barre des titres.

 Cliquez sur le bouton Système [image] sur la barre de menus.

Le menu Système s'ouvre.

2.	Sélectionnez Maximiser du menu Système.


Commande Fermer

Utilisez cette commande pour fermer la fenêtre ou la boîte de dialogue active.

Pour y accéder:

1.	Effectuez l'une des choses suivantes:

 Cliquez sur le bouton Trimble Geomatics Office [image] sur la barre des titres.

 Cliquez avec le bouton droit sur la barre des titres.

 Cliquez sur le bouton Système [image] sur la barre de menus.

Le menu Système s'ouvre.

2.	Sélectionnez Fermer du menu Système.


Commande Suivant

Utilisez cette commande pour passer à la fenêtre de graphiques suivante selon l'ordre dont vous les avez ouvert.

Pour y accéder:

1.	Effectuez l'une des choses suivantes:

 Cliquez sur le bouton Trimble Geomatics Office [image] sur la barre des titres.

 Cliquez avec le bouton droit sur la barre des titres.

 Cliquez sur le bouton Système [image] sur la barre de menus.

Le menu Système s'ouvre.

2.	Sélectionnez Suivant du menu Système.


Commande Précédent

Utilisez cette commande pour passer à la fenêtre ouverte précédente selon l'ordre dont vous les avez ouvert.

Pour y accéder:

1.	Effectuez l'une des choses suivantes:

 Cliquez sur le bouton Trimble Geomatics Office [image] sur la barre des titres.

 Cliquez avec le bouton droit sur la barre des titres.

 Cliquez sur le bouton Système [image] sur la barre de menus.

Le menu Système s'ouvre.

2.	Sélectionnez Précédent du menu Système.


Commande Basculer vers

Utilisez cette commande pour afficher une liste de toutes les applications ouvertes.

Utilisez l'option Liste des tâches pour basculer vers ou fermer une application dans la liste.

Options du dialogue

Lorsque vous sélectionnez la commande Basculer vers, un dialogue avec les options suivantes s'affiche:

Option	Description
Liste des tâches	Sélectionne l'application que vous voulez activer ou fermer.
Basculer vers	Rend l'application sélectionnée active.
Fin de tâche	Ferme l'application sélectionnée.
Annuler	Ferme l'option Liste des tâches
Cascade	Affiche toutes les applications en les superposant afin que vous pouvez voir chaque barre de titres. Cette option n'affecte pas les applications réduites en icônes.
Mosaïque	Affiche toutes les applications sans les superposer. Cette option n'affecte pas les applications réduites en icônes.
Réorganiser les icônes	Réorganise les icônes de toutes les applications minimisées au bas de l'écran.


Glisser et déposer des fichiers au Trimble Geomatics Office

Vue d'ensemble et plus amples informations	Comment ...

Utilisez la fonctionnalité 'glisser et déposer' de l'Explorateur Windows et Microsoft Windows pour importer les fichiers de données d'un dossier sur votre ordinateur vers un projet Trimble Geomatics Office.

Vous pouvez importer tout type de fichier disponible dans les tabulations du dialogue Importer.

NOTE: Vous ne pouvez pas glisser et déposer les fichiers .job ou .raw d'une carte PC vers un projet Trimble Geomatics Office. Pour transférer ces fichiers vers un projet Trimble Geomatics Office, utilisez Trimble Geomatics Office pour les transférer. Ce logiciel convertit les fichiers au format correct.

Formats uniques

Vous pouvez utiliser n'importe lequel des deux boutons de la souris pour glisser et déposer les fichiers.

Il est toujours meilleur d'utiliser le bouton droit de la souris parce que lorsque vous libérez:

  le bouton droit de la souris après avoir glisser, une liste des formats d'importation disponibles s'affiche. Vous pouvez sélectionner le format correct de la liste.

  le bouton gauche de la souris après avoir glisser, Trimble Geomatics Office utilise l'extension du fichier et détermine le format du fichier importé. Il n'y a aucune  garantie que le format de fichier correct sera sélectionné par le logiciel.

Formats multiples

Vous pouvez utiliser n'importe lequel des deux boutons de la souris pour glisser et déposer les fichiers multiples.

  Avec le bouton droit de la souris, vous pouvez glisser et déposer des fichiers multiples du même type. Une liste des formats disponibles s'affiche de laquelle vous pouvez sélectionner le format correct à importer. Si vous glissez et déposez une sélection de types de fichiers différents (par exemple un fichier .dc et un fichier .dat), les options d'importation individuelles ne s'affichent pas dans la liste. Sélectionnez Défaut de la liste afin d'utiliser les options d'importation pour chaque type de fichier. Trimble Geomatics Office utilise l'extension des fichiers et détermine les formats de fichier que vous importez. Cependant, il n'y a aucune  garantie que le format de fichier correct sera sélectionné par le logiciel.

  Avec le bouton gauche de la souris, vous pouvez glisser et déposer des fichiers multiples de types différents. Trimble Geomatics Office utilise l'extension des fichiers et détermine les formats de fichier que vous importez. Cependant, il n'y a aucune  garantie que le format de fichier correct sera sélectionné par le logiciel.

NOTE: Lorsque Trimble Geomatics Office ne reconnaît aucun des formats importés avec glisser et déposer, un menu de raccourcis s'affiche fournissant l'option de l'importer le fichier sous forme d'un fichier de carnet de terrain tiers ou un fichier NMEA. Si vous choisissez l'option Défaut, Trimble Geomatics Office essayera de l'importer comme un fichier NMEA.

Formats personnalisés

Tout format d'importation personnalisé que vous créez s'affichera dans le menu de raccourci lorsque vous utilisez le bouton droit de la souris pour glisser déposer un fichier vers un projet Trimble Geomatics Office.

ASTUCE: Lorsque vous n'avez pas un projet ouvert, vous pouvez glisser le fichier Options.ini sur Trimble Geomatics Office pour ouvrir le projet.


Menus de raccourcis

Les menus de raccourcis sont des menus contextuels ayant des commandes qui varient selon la position du pointeur.

Vous pouvez utilisez les menus de raccourcis pour les tâches les plus utilisées telles que:

  l'accès aux commandes de menu et aux dialogues les plus utilisés

  basculant entre les paramètres

  l'insertion des codes de champ

Pour accéder à un menu de raccourcis:

1.	Mettez le pointeur sur la zone appropriée.

2.	Cliquez sur le bouton droit de la souris.


Touches d'accès rapide

Les touches d'accès rapide fournissent une méthode facile d'utiliser le clavier pour la sélection d'une option d'un menu ou d'un dialogue.

Appuyez sur la touche du clavier correspondant à la lettre soulignée dans l'option que vous voulez sélectionner.


Systèmes de coordonnées - Vue d'ensemble

Plus amples informations

Comment ...	Dialogue	Voir aussi
Vue d'ensemble

Trimble Geomatics Office utilise un système de coordonnées pour transformer les mesures sur une surface en courbe (la terre) à une surface plate (une carte ou un plan). Par exemple, un système de coordonnées est utilisé pour calculer les coordonnées de grille d'un point mesuré à l'aide de GPS (les mesures GPS sont effectuées sur l'ellipsoïde WGS-84).

Un système de coordonnées peut consister en les éléments suivants:

  une transformation de datum (entre l'ellipsoïde WGS-84 et l'ellipsoïde local)

  une projection

  un modèle de géoïde

  une calibration du site GPS (consistant en un ajustement horizontal et un ajustement vertical)

Vous devez sélectionner un système de coordonnées pour chaque projet. Sinon, lorsque vous n'avez pas un système (ou que vous ne savez pas quel système à sélectionner) il faut utiliser la projection par défaut.

NOTE: Vous devez vous assurer que les points dans un projet se trouvent dans une distance raisonnable de l'origine de la projection selon les propriétés de la projection utilisée.

Base de données du système de coordonnées

La base de données du système de coordonnées est stockée sous forme d'un fichier appelé Current.csd. Il contient toutes les informations du système de coordonnées. Trimble fournit un groupe répandu de systèmes de coordonnées publiés d'autour du monde. Pour définir ou éditer les systèmes de coordonnées, les zones, les sites, les transformations de datum, les ellipsoïdes et les modèles de géoïde, utilisez l’Utilitaire Coordinate System Manager.

Sélection d'un système de  coordonnées

Pour sélectionner un système de coordonnées pour un projet, utilisez l'un des suivants:

  un système de coordonnées et une zone de la base de données du système de coordonnées

  un site de la base de données du système de coordonnées

  une projection par défaut non définie

  une projection par défaut définie

  le système de coordonnées spécifié dans un fichier Survey Controller (*.dc) lors de l'importation

Lorsque vous sélectionnez un système de coordonnées dans le logiciel Trimble Geomatics Office, les paramètres sont obtenus de la base de données du système de coordonnées.

Changement du système de coordonnées d'un projet

Pour changer le système de coordonnées d'un projet, effectuez l'une des choses suivantes:

  Spécifiez un nouveau système de coordonnées à l'aide du dialogue Propriétés du projet.

  Importez un fichier Survey Controller (*.dc) avec un système de coordonnées différent dans un projet Trimble Geomatics Office. C'est à dire, lorsque le projet a un système de coordonnées et le fichier .dc a un système de coordonnées différent, pendant l'importation le logiciel détecte la différence et vous demande de sélectionner quel système il faut utiliser. C'est une méthode pour changer le système de coordonnées d'un projet si vous décidez d'utiliser le système du fichier .dc.

Lorsque vous changez le système de coordonnés d'un projet, Trimble Geomatics Office calcule les coordonnées de tous les points. Les coordonnées sont transformées du système de coordonnées ancien au nouveau système de coordonnées. Le procédé général pour ce faire est le suivant:

1.	Toutes les coordonnées sont réduites à l'ellipsoïde. (En général, l'ellipsoïde est l'ellipsoïde WGS-84 mais lorsque seulement la projection a changé, c'est l'ellipsoïde local).

2.	Des nouvelles coordonnées sont calculées utilisant les nouveaux paramètres.

Ces étapes sont illustrées dans la figure suivante:

Si un modèle de géoïde ou un ajustement vertical fait partie du système de coordonnées ancien, il est utilisé pour calculer les hauteurs WGS-84 de tous les points qui ne sont pas dérivés des observations sur le terrain. De la même façon, si un modèle de géoïde ou un ajustement vertical est appliqué dans le nouveau système de coordonnées, il est utilisé pour calculer la nouvelle élévation des coordonnées de grille des hauteurs WGS-84.

L'effet du changement du système de coordonnées sur le logiciel Trimble RoadLink et Trimble DTMLink

Lorsque vous exportez des points du projet aux utilitaires RoadLink ou DTMLink de Trimble, les points exportés ne sont pas mis à jour par les changements au projet Trimble Geomatics Office.

Trimble vous recommande de ne pas exporter des points jusqu'à ce que vous êtes sûr que tous les changements au projet Trimble Geomatics Office sont complets.

Si vous avez déjà exporté des points du projet Trimble Geomatics Office à Roadlink ou à DTMLink, et vous changez le système de coordonnées ou calibrez vos données dans le projet Trimble Geomatics Office, l'avertissement suivant s'affiche:

Les coordonnées des points de DTMLink et RoadLink ne seront pas mises à jour quand vous changez le système de coordonnées.

Effectuez l'une des choses suivantes:

  Cliquez sur OK pour reconnaître que les coordonnées du point ne changeront pas dans RoadLink ou DTMLink.

  Cliquez sur Annuler pour annuler le changement.

Afin d'assurer que les coordonnées les plus récentes sont utilisées dans les utilitaires RoadLink et DTMLink:

1.	Supprimez les points non requis dans RoadLink et/ou DTMLink.

2.	Exportez les points à nouveau du projet.


Systèmes de coordonnées au sol

Si vous avez besoin des coordonnées au niveau du sol plutôt qu'au niveau de la projection, utilisez un système de coordonnées au sol. Lorsque vous sélectionnez un système de coordonnées au sol, les distances de grille sont égales aux distances au sol.

Pour utiliser vos coordonnées au sol dans Trimble Geomatics Office, vous devez:

1.	Définir un système de coordonnées pour le projet.

2.	Saisir les coordonnées pour l'emplacement du projet. L'emplacement du projet est enregistré sous forme de latitude et longitude.

3.	Saisir le facteur d'échelle au sol, ou le calculer utilisant les coordonnées de l'emplacement du projet.

Définition d'un système de coordonnées d'un projet

Pour des informations à propos de comment sélectionner et changer un système de coordonnées dans un projet Trimble Geomatics Office, référez-vous à la rubrique Système de coordonnées - Vue d'ensemble.

Saisie des coordonnées pour l'emplacement du projet

Pour ce faire:

1.	Sélectionnez Fichier / Propriétés du projet. Le dialogue Propriétés du projet s'affiche.

2.	Sélectionnez la tabulation Système de coordonnées.

3.	Dans le groupe Paramètres du site local, cliquez sur Changer. Le dialogue Paramètres du site local s'affiche.

4.	Dans le groupe Emplacement du projet, saisissez les coordonnées appropriées. Si le système de coordonnées n'est pas défini, vous serez demandé de mettre à jour le système de coordonnées.

La latitude et la longitude doivent être approximatives, et l'élévation doit être une estimation raisonnable de l'élévation, donc saisissez une élévation moyenne pour le projet.

Facteur d'échelle au sol

Pour ce faire:

1.	Dans le dialogue Paramètres du site local, sélectionnez la case à cocher Utiliser coordonnées au sol et effectuez l'une des choses suivantes:

  Saisissez le Facteur d'échelle au sol.

  Sélectionnez la case à cocher Calculer le facteur d'échelle de la position du projet. Le logiciel calcule le facteur d'échelle au sol des valeurs dans le groupe Emplacement du projet.

  Spécifiez comment les coordonnées au sol sont affichées. Vous pouvez spécifier un déport nord et/ou est, ou des coordonnées à quatre ou cinq chiffres. Utilisez cette option pour différencier facilement entre les coordonnées.

NOTE: Le système de coordonnées au sol suppose que vous travaillez près de l'emplacement du projet et qu'il n'y a aucune variation importante dans la hauteur du projet.

Le facteur d'échelle au sol est calculé par:

1.	La détermination de l'élévation de référence (h)

2.	Le calcul du facteur d'élévation

Où:

R est le rayon ellipsoïdal

h est l'élévation de référence

a est le demi-grand axe

e2 est l'excentricité carrée

φ est la latitude d'origine de la projection

3. 	Le calcul du Facteur d'échelle au sol au centroïde à l'emplacement du projet.

Où:

s est le facteur d'échelle de grille à l'emplacement du projet.

Dans le Trimble Survey Controller de version 10, les Systèmes de coordonnées au sol sont pris en charge complètement. Les notes suivantes se réfèrent aux Systèmes de coordonnées au sol dans le Trimble Survey Controller de version 7.7 et antérieures.

NOTES

Importation des projets avec des coordonnées au sol de Trimble Survey Controller de version 7.5 – 7.7

Lorsque vous importez une étude Trimble Survey Controller (version 7.5 ou 7.7) avec une projection Mercator transversal dans laquelle un facteur d'échelle au sol a été défini (par exemple, une étude sans projection/sans datum  avec une calibration) laissez la case à cocher Utiliser coordonnées au sol dans le dialogue Paramètres du site local vide. Autrement le facteur d'échelle au sol sera appliqué deux fois.

Lorsque vous importez une étude Trimble Survey Controller avec un facteur d'échelle au sol défini dans un projet Trimble Geomatics Office que n'a pas un système de coordonnées au sol défini, un message d'avertissement vous informe que les systèmes de coordonnées dans l'étude et dans le projet sont différents. Utilisez le système de coordonnées que se trouve dans le projet Trimble Geomatics Office et puis sélectionnez la case à cocher Utiliser coordonnées au sol dans le dialogue Paramètres du site local.

Exportation des projets avec des coordonnées au sol à Trimble Survey Controller de version 7.5-7.7

Lorsque vous exportez un projet Trimble Geomatics Office avec un système de coordonnées au sol au Trimble Survey Controller (version 7.70 ou précédente), le système de coordonnées sera différent lorsque vous l'importez à nouveau dans le logiciel Trimble Geomatics Office. Il faut vérifier que le système de coordonnées soit accepté.

Trimble Survey Controller (version 7.7 ou précédente) ne prend pas en charge les coordonnées au sol avec la projection Conforme conique parallèle 2 de Lambert. Le système de coordonnées est exporté de Trimble Geomatics Office au Trimble Survey Controller avec une projection Conforme conique parallèle 1 de Lambert.

Trimble Survey Controller (version 7.7 ou précédente) ne prend pas en charge les coordonnées au sol avec la projection Conique égal de surface d'Albers. Si vous voulez exporter un projet avec une projection Conique égal de surface d'Albers définie au logiciel Trimble Survey Controller, supprimez les coordonnées au sol ou changez les projections.


Transformations de datum - Vue d'ensemble

Plus amples informations

 Voir aussi

Vue d'ensemble

Une transformation de datum est un groupe de paramètres utilisé pour la conversion des coordonnées d'un ellipsoïde (par exemple, l'ellipsoïde WGS-84) aux coordonnées sur l'ellipsoïde local. (L'ellipsoïde local est l'ellipsoïde utilisé par la projection.) Une transformation de datum fait partie d'une définition du système de coordonnées.

Les méthodes de transformation de datum suivantes sont disponibles:

  Molodensky (trois paramètres)

  sept paramètres

  grille de datum

  régression multiple

Toutes les méthodes sont stockées dans la base de données du système de coordonnées.

Pour définir des nouvelles transformations de datum, éditez la base de données du système de coordonnées à l'aide de l'utilitaire Coordinate System Manager™.


Transformation de datum - Molodensky (trois paramètres)

 Vue d'ensemble et plus amples informations

La transformation de datum Molodensky définit une translation dans les directions X, Y, et Z entre l'origine de l'ellipsoïde WGS-84 et l'origine de l'ellipsoïde local (ou vice versa).

La transformation de datum Molodensky est également appelée une transformation de datum à trois paramètres.

Affichage des détails

Lorsque vous avez une transformation de datum Molodensky définie comme partie du système de coordonnées de votre projet, vous pouvez afficher les valeurs de transformation de datum.

Pour ce faire, utilisez le dialogue Détails système de coordonnées.

Ces champs définissent une transformation cartésienne (X,Y,Z) en trois dimensions de l'origine de l'ellipsoïde WGS-84 à l'origine de l'ellipsoïde local.

Les transformations de datum sont stockées dans la base de données du système de coordonnées (Current.csd). Pour les éditer, utilisez l'utilitaire Coordinate System Manager™.


Transformation de datum - sept paramètres

 Vue d'ensemble et plus amples informations

La transformation de datum à sept paramètres définit une transformation de datum de l'ellipsoïde WGS-84 à un ellipsoïde local, ou d'un ellipsoïde local à l'ellipsoïde WGS-84.

Translation

Ces champs définissent une transformation cartésienne (X,Y,Z) en trois dimensions de l'origine de l'ellipsoïde WGS-84 à l'origine de l'ellipsoïde local.

Rotation

Les valeurs de rotation X définissent les angles de rotation autour des axes  X, Y, et Z respectivement. Ceux sont les angles à travers desquels les axes cartésiens de l'ellipsoïde WGS-84 doivent être tournée afin de les aligner aux axes de l'ellipsoïde local.

L'angle de rotation  est positif pour une rotation dans le sens inverse d'horloge autour de chaque axe, observé de l'extrémité positive de l'axe.

Facteur d'échelle

Toutes les coordonnées sont multipliées par cette valeur. Elle est définie en parties par million. Si le facteur d'échelle est 6,02, toutes les coordonnées sont multipliées par 1,00000602.

Affichage des détails

Lorsque vous avez une transformation de datum à sept paramètres définie comme partie du système de coordonnées de votre projet, vous pouvez afficher les valeurs de transformation de datum.

Pour ce faire, utilisez le dialogue Détails système de coordonnées – tabulation Transformation de datum.

Les transformations de datum sont stockées dans la base de données du système de coordonnées (Current.csd). Pour les éditer, utilisez l'utilitaire Coordinate System Manager™.


Transformation de datum - Grille de datum

 Vue d'ensemble et plus amples informations

Une transformation de datum Grille de datum est un fichier publié qui contient les valeurs des paramètres de transformation de datum de la zone. Le fichier est utilisé pour interpoler les valeurs de transformation de datum à votre emplacement.

Pour utiliser une grille de datum comme partie de la définition du système de coordonnées du projet, vous devez sélectionner une transformation de datum définie par un fichier de grille de datum de latitude et longitude (*.dgf).

Pour afficher ou créer une transformation de datum utilisant des Fichiers de grille de datum, utilisez l'utilitaire Coordinate System Manager™.

Pour utiliser le fichier de grille de datum sur le terrain, vous devez transférer le fichier de grille de datum sous forme d'un fichier de grille de datum combiné au Trimble Survey Controller (version 7.00 ou ultérieure).


Transformation de datum - Régression multiple

 Vue d'ensemble et plus amples informations

Une transformation de datum à régression multiple définit une transformation entre l'ellipsoïde WGS-84 et un ellipsoïde local par l'adaptation d'une équation polynôme aux déplacements prévus dans la zone locale. Pour ajouter une transformation de datum à régression multiple, utilisez l'utilitaire Coordinate System Manager™.

Affichage des détails

Lorsque vous avez une transformation de datum à régression multiple définie comme partie du système de coordonnées de votre projet, vous pouvez afficher les valeurs de transformation de datum.

Pour ce faire, utilisez le dialogue Détails système de coordonnées – tabulation Transformation de datum.

Les transformations de datum sont stockées dans la base de données du système de coordonnées (Current.csd). Pour les éditer, utilisez l'utilitaire Trimble Coordinate System Manager™.


Projections - Vue d'ensemble

Plus amples informations

 Voir aussi

Vue d'ensemble

Une projection est une expression mathématique pour la surface en courbe d'un ellipsoïde sur une grille de coordonnées rectangulaire.

Utilisez une projection pour transformer des coordonnées géodésiques locales aux coordonnées de grille locales.

Les coordonnées de grille sont tracées afin de créer des cartes représentant la surface de la terre ou des parties de cette surface.

Les pays et les régions différents utilisent des projections différentes afin d'obtenir une distorsion minimale dans les coordonnées de grille résultantes. L'utilitaire Coordinate System Manager™ contient de nombreuses projections (par exemple, Mercator transversal, Parallèle 1 de Lambert). Pour plus amples informations concernant les projections spécifiques, référez-vous à l'aide de Coordinate System Manager.

NOTE: Vérifiez que les points dans un projet se trouvent dans une distance raisonnable de l'origine de la projection selon les propriétés de la projection utilisée.

Pour définir des paramètres de projection comme partie d'une définition du système de coordonnées, utilisez l'utilitaire Coordinate System Manager™.


Modèles de géoïde - Vue d'ensemble

Plus amples informations

 Voir aussi

Vue d'ensemble

Un géoïde est une surface de potentiel de gravitation constante que se rapproche au niveau de la mer.

Les points observés avec GPS ont des hauteurs basées sur l'ellipsoïde WGS-84. Ces hauteurs sont connues comme des hauteurs d'ellipsoïde.

Un modèle de géoïde nous permet d'estimer des élévations basées sur les hauteurs d'ellipsoïde.

Les modèles de géoïde sont stockés sous forme de fichiers de Grille de géoïde (*.ggf). Un fichier .ggf contient les séparations de géoïde d'un nombre de positions sur une grille. A l'aide de l'interpolation, il est possible de calculer ces séparations de géoïde pour tout point qui se trouve dans la grille.

Lorsqu'un modèle de géoïde est sélectionné pour chaque point observé à l'aide de GPS, Trimble Geomatics Office utilise le fichier .ggf pour interpoler la séparation de géoïde (N) à cet emplacement. Ensuite, le logiciel soustrait ou ajout la valeur de la hauteur d'ellipsoïde (h) observée. Ainsi une élévation approximative au-dessus du niveau de la mer (é) est obtenue pour le point GPS. Le rapport entre le géoïde et l'ellipsoïde local est illustré dans la figure suivante:

Pour définir un nouveau modèle de géoïde utilisant un fichier de grille de géoïde existant, utilisez l'utilitaire Coordinate System Manager. Tous les modèles de géoïde dans la base de données du système de coordonnées sont disponibles lorsque vous sélectionnez une système de coordonnées pour un projet.

Vous pouvez utiliser Trimble Geomatics Office pour exporter les fichiers de grille de géoïde au Trimble Survey Controller ou le logiciel TDS Survey Pro CE (version 3.5 ou ultérieure),et créer des sous-grilles des fichiers de grille de géoïde existants.

NOTE: Lorsque vous n'utilisez pas un modèle de géoïde ou n'effectuez pas une calibration du site GPS, l'élévation d'un point sera la même que la hauteur d'ellipsoïde et ne sera pas une élévation vraie. Si un modèle de géoïde est utilisé dans le système de coordonnées du projet, il est utilisé pour la conversion entre les hauteurs d'ellipsoïde locales et les élévations pour tous les types de points, non seulement GPS.


Changer le système de coordonnées - dialogue Sélectionner système de coordonnées

Utilisez ce dialogue pour sélectionner un système de coordonnées existant, un site, ou pour définir une projection par défaut pour le projet.

Ce dialogue est la première étape d'un assistant que vous guide à travers les paramètres de sélection.

Cliquez sur Suivant pour avancer et Précédent pour retourner à travers les dialogues de sélection.

Trimble Geomatics Office utilise les systèmes de coordonnées dans le fichier de base de données du système de coordonnées (Current.csd). Vous pouvez éditer les paramètres d'un système de coordonnées à l'aide de l'utilitaire Coordinate System Manager™.

Nouveau système

Pour sélectionner un nouveau système de coordonnées et une nouvelle zone, un site calibré local enregistré dans la base de données du système de coordonnées, ou une projection par défaut:

1.	Sélectionnez Nouveau système.

2.	Cliquez sur Suivant pour passer au dialogue Sélectionner le Type de système de coordonnées .

3.	Utilisez l'assistant pour sélectionner un système de coordonnées pour le projet.

Système utilisé récemment

Pour sélectionner un système de coordonnées d'une liste des dix systèmes de coordonnées sélectionnés les plus récemment:

1.	Cliquez sur Système utilisé récemment.

2.	Cliquez sur les boutons directionnels pour défiler à travers les résumés de vos systèmes.

3.	Sélectionnez le système de coordonnées que vous voulez utiliser pour votre projet.

4.	Cliquez sur Finir.

NOTE: Si c'est la première fois que vous avez utilisé la base de données du système de coordonnées, le dialogue Sélectionner type de système de coordonnées s'affiche (et pas le dialogue Changer le système de coordonnées – Sélectionner système de coordonnées). C'est à cause du fait qu'il n'y a aucun système utilisé récemment à sélectionner.

NOTE: Trimble Geomatics Office ne prend pas en charge tous les systèmes de coordonnées disponibles dans l'utilitaire Coordinate System Manager. Lorsque vous sélectionnez un système de coordonnées que n'est pas pris en charge, le message suivant s'affiche:

Type de système de coordonnées inconnu


Dialogue Sélectionner le Type de système de coordonnées

Utilisez ce dialogue pour sélectionner le type de système de coordonnées que vous voulez utiliser.

Système de coordonnées et zone

Sélectionnez cette option pour choisir un système de coordonnées de la base de données de votre système de coordonnées.

Site calibré

Sélectionnez cette option pour choisir un site calibré que vous avez enregistré précédemment dans la base de données du système de coordonnées. Les sites sont des détails de système de coordonnées stockés pour les sites locaux que peuvent comprendre des paramètres de calibration GPS.

Projection par défaut (Mercator transversal)

Sélectionnez cette option pour spécifier une projection par défaut.

Précédent

Cliquez sur Précédent pour retourner au dialogue Sélectionner système de coordonnées.

NOTE: Si c'est la première fois que vous avez utilisé la base de données du système de coordonnées, Précédent n'est pas disponible à cause du fait qu'il n'y a aucun système utilisé récemment à sélectionner.

Suivant

Cliquez sur Suivant. Selon l'option que vous avez sélectionné, l'un des dialogues suivants s'affiche:

  dialogue Sélectionner la zone du système de coordonnées.

  dialogue Sélectionner site calibré.

  dialogue Projection par défaut (Mercator transversal).


Dialogue Sélectionner la zone du système de coordonnées

Utilisez ce dialogue pour sélectionner le système de coordonnées pour le projet de la liste des systèmes de coordonnées disponibles dans la base de données du système de coordonnées.

Pour afficher les propriétés d'un groupe ou une zone de système de coordonnées:

1.	Soulignez le groupe ou la zone de système de coordonnées.

2.	Cliquez avec le bouton droit de la souris pour accéder au menu de raccourcis et sélectionnez Propriétés.

Groupe de système de coordonnées

Cette liste contient les systèmes de coordonnées dans la base de données du système de coordonnées. Un système contient une ou plusieurs zones définies pour des zones diverses.

Zone

Cette liste contient les zones disponibles pour le système de coordonnées sélectionné. Une zone contient une transformation de datum (y compris un ellipsoïde et un groupe de paramètres de transformation de datum pour la transformation des données de l'ellipsoïde WGS-84 à l'ellipsoïde local), un modèle de géoïde et les paramètres de la projection.

Précédent

Cliquez sur Précédent pour retourner au dialogue Type du système de coordonnées.

Suivant

Cliquez sur Suivant. Le dialogue Sélectionner modèle de géoïde s'affiche.


Dialogue Sélectionner site calibré

Utilisez ce dialogue pour sélectionner un site calibré de la liste de sites calibrés disponibles dans la base de données du système de coordonnées. Les sites sont stockés dans la base de données du système de coordonnées. Vous créez les sites dans l'utilitaire Coordinate System Manager™, ou dans Trimble Geomatics Office (dans la tabulation Fichier / Propriétés du projet / Système de coordonnées ou Levé / Calibration du site GPS.)

Pour afficher les propriétés d'une site:

1.	Soulignez le nom du site.

2.	Cliquez avec le bouton droit de la souris pour accéder au menu de raccourcis et sélectionnez Propriétés.

Détails du site

Ce groupe affiche la transformation de datum et le modèle de géoïde du site souligné.

Précédent

Cliquez sur Précédent pour retourner au dialogue Sélectionner le type du système de coordonnées.

Finir

Cliquez sur Finir pour compléter la définition du système de coordonnées et retourner au dialogue Propriétés du projet. La définition du site sélectionnée est le système de coordonnées du projet.

NOTE: Vous ne pouvez pas changer le modèle de géoïde d'un site dans le logiciel Trimble Geomatics Office. Le modèle de géoïde spécifié pour le site est utilisé. Pour changer le modèle de géoïde d'un site, utilisez l'utilitaire Coordinate System Manager. L'utilitaire vous permet d'éditer le site.


Dialogue Projection par défaut (Mercator transversal)

Utilisez ce dialogue pour spécifier les paramètres de la projection par défaut. Trimble Geomatics Office peut calculer les valeurs ou vous pouvez les entrer.

Calculer les paramètres de projection automatiquement

Sélectionnez cette case à cocher pour configurer Trimble Geomatics Office pour qu'il utilise le première position WGS-84 saisie dans le projet comme l'origine, et la première coordonnée de grille (nord et est) pour calculer la constante nord et la constante est.

NOTE: Vous devez également spécifier le système d'azimut sud (si requis) et la direction de coordonnée positive. Trimble Geomatics Office ne met pas à jour ces champs automatiquement.

Effacez cette case à cocher pour saisir des paramètres de la projection. Les champs suivants dans le dialogue deviennent disponibles. Saisissez les valeurs dans ces champs.

Latitude central

Saisissez la latitude de l'origine de la projection.

Longitude centrale

Saisissez la longitude de l'origine de la projection.

Constante nord et Constante est

Saisissez la constante est et la constante nord de la projection par défaut. La constante est et la constante nord  sont les coordonnées de grille de l'origine de la projection par défaut.

Facteur d'échelle

Saisissez le facteur d'échelle de l'origine. Il devrait être basé sur un facteur d'échelle connu pour la projection. Celui-ci représente le ratio de distance sur la projection à la distance sur la surface de la terre. Toutes les coordonnées au sol sont mises à l'échelle par cette valeur pour fournir des distances de grille.

Système d'azimut sud

Sélectionnez ce champ pour spécifier un système d'azimut sud. Un système d'azimut sud utilise le sud comme l'azimut de zéro degrés. Il est utilisée en Hawaii et en Afrique du Sud (d'autres systèmes utilisent le nord comme l'azimut de zéro degrés).

Direction de coordonnée positive

Sélectionnez les options pour spécifier la direction dans laquelle vous voulez que les valeurs de coordonnées s'augmentent.

Précédent

Cliquez sur Précédent pour retourner au dialogue Sélectionner  le Type de système de coordonnées.

Suivant

Lorsque vous avez terminé, cliquez sur Suivant. Le dialogue Sélectionner modèle de géoïde s'affiche.


Dialogue Sélectionner modèle de géoïde

Utilisez ce dialogue pour sélectionner un modèle de géoïde pour le projet. Lorsqu'un modèle de géoïde est défini pour le système de coordonnées dans la base de données du système de coordonnées, le modèle de géoïde est sélectionné automatiquement.

Un modèle de géoïde est une surface mathématique composée des séparations de géoïde/ellipsoïde. C'est une estimation de la séparation entre l'ellipsoïde WGS-84 et le niveau moyen de la mer. Un modèle de géoïde sert à dériver des hauteurs orthométriques approximatives des points GPS.

Aucun modèle de géoïde

Sélectionnez cette option pour ne spécifier aucun modèle de géoïde pour le système de coordonnées. Lorsque vous ne spécifiez pas un modèle de géoïde, l'élévation du point est la hauteur au-dessus de l'ellipsoïde local.

Si vous sélectionnez cette option, Trimble vous recommande d'effecteur une calibration du site GPS afin d'obtenir des élévations plus précises pour vos observations GPS.

Modèle de géoïde prédéfini

Pour spécifier un modèle de géoïde pour le projet, sélectionnez-le de la liste. La liste des modèles de géoïde est de la base de données du système de coordonnées (Current.csd).

Selon la précision du modèle de géoïde, l'élévation dérivée de la hauteur WGS-84 et de la séparation de géoïde est assignée une qualité. Pour spécifier la qualité du modèle de géoïde, utilisez le champ Qualité du modèle de géoïde dans le dialogue Propriétés du projet / tabulation Recalculer. La qualité est utilisée pour spécifier la qualité du champ d'élévation des points.

Précédent

Cliquez sur Précédent pour retourner au dialogue Sélectionner la zone du système de coordonnées ou au dialogue Projection par défaut (Mercator transversal).

Finir

Cliquez sur Finir pour compléter la définition du système de coordonnées et retourner au dialogue Propriétés du projet.


Sites - Vue d'ensemble

En général un site est utilisé pour le stockage des détails de calibration du site GPS local sur un système de coordonnées et une zone existants.

Lorsque vous avez effectué une calibration du site GPS vous pouvez l'enregistrer ainsi que le système de coordonnées dans la base de données du système de coordonnées sous forme d'une site afin que vous pouvez l'utiliser facilement à nouveau.

Pour enregistrer un site, effectuez l'une des choses suivantes:

  Sélectionnez Levé / Calibration du site GPS.

  Sélectionnez Fichier / Propriétés du projet / tabulation Système de coordonnées.

Sinon, vous pouvez créer un nouveau site à l'aide de l'utilitaire Coordinate System Manager™. Quand vous utilisez cet utilitaire, les sites enregistrés s'affichent dans la tabulation Sites.

Pour plus amples informations, référez-vous à l'aide de l'utilitaire Coordinate System Manager.

NOTE: Vous ne pouvez pas enregistrer les paramètres de transformation d'ajustement sous forme d'un site.


Utilisation d'une projection par défaut - Vue d'ensemble / dialogue Définition de projection par défaut

Vue d'ensemble

Une projection par défaut non définie est une projection par défaut sans aucune valeur de latitude d'origine et de longitude d'origine. Lorsque vous saisissez des données de levé dans un projet pour la première fois utilisant une projection par défaut non définie, le dialogue Définition de projection par défaut s'affiche. Le dialogue Définition de projection par défaut suggère l'utilisation des paramètres de projection appropriés aux données que vous saisissez. Des champs divers dans le dialogue sont disponibles selon le type de données ajouté au projet. Les scénarios suivants sont possibles:

  L'ajout des points GPS à un projet (projection par défaut non définie) contenant des points de grille.

  Un fichier avec une projection définie à une projet avec une projection par défaut non définie

  L'ajout d'un point WGS-84 à un projet (projection par défaut non définie).

  L'utilisation d'un fichier à facteur d'échelle seul (projection par défaut non définie).

  L'ajout des points de grille à un projet vide.

Pour utiliser une projection par défaut non définie, utilisez un modèle de projet que n'a pas un système coordonnées défini (par exemple, les modèles métriques et de pieds américains de Trimble).


L'ajout des points de grille (projection par défaut non définie)

 Vue d'ensemble et plus amples informations

Le dialogue Définition de projection par défaut s'affiche lorsque la première position ajoutée au projet Trimble Geomatics Office ayant une projection par défaut non définie (aucun système de coordonnées) est une position de grille.

Le dialogue Définition de projection par défaut contient les champs suivants:

Constante nord

Le nord à l'origine de la projection. La valeur du nord est dérivée du premier point dans le fichier. Vous pouvez modifier cette valeur.

Constante est

L'est à l'origine de la projection. La valeur de l'est est dérivée du premier point dans le fichier. Vous pouvez modifier cette valeur.

OK

Cliquez sur OK pour accepter les paramètres de projection par défaut. Ces valeurs sont utilisées pour définir le système de coordonnées du projet. Pour afficher ces paramètres, sélectionnez Fichier / Propriétés du projet / tabulation Système de coordonnées et cliquez sur Détails.

NOTE: Du fait que votre projection par défaut n'est pas encore définie complètement (la latitude d'origine et la longitude d'origine ne sont pas définies),  vous pouvez mettre à jour la projection par défaut par l'importation d'un fichier .dc contenant un ou plusieurs points GPS.


L'ajout des points GPS à un projet (projection par défaut non définie) contenant des points de grille

 Vue d'ensemble et plus amples informations

Le dialogue Définition de projection par défaut s'affiche lorsque le projet Trimble Geomatics Office a une projection par défaut non définie (aucune latitude d'origine et aucune longitude d'origine), contient une ou plusieurs positions de grille, et:

  un fichier contenant au moins un point GPS est importé;

ou

  un point WGS-84 est entré.

Le dialogue Définition de projection par défaut contient les champs suivants:

Nord

Ce champ est nul. Saisissez la meilleure coordonnée nord connue pour le point GPS utilisé pour définir la projection par défaut.

Est

Ce champ est nul. Saisissez la meilleure coordonnée est connue pour le point GPS utilisé pour définir la projection par défaut.

Latitude

La latitude WGS-84 du premier point GPS dans le fichier s'affiche ici. Ce champ n'est pas disponible.

Longitude

La longitude WGS-84 du premier point GPS dans le fichier s'affiche ici. Ce champ n'est pas disponible.

OK

Cliquez sur OK pour accepter les coordonnées de grille pour le point GPS. Les valeurs de latitude WGS-84 et de longitude WGS-84 pour l'origine sont déterminées sur la base des coordonnées constantes de l'origine (déjà définies), et les valeurs pour le point GPS spécifié dans le dialogue. Pour afficher tous les paramètres, sélectionnez Fichier / Propriétés du projet / tabulation Système de coordonnées et cliquez sur Détails.

NOTE: Dès qu'un point GPS a été ajouté à la base de données, la projection par défaut est complètement définie, à partir de maintenant lorsque vous importez un fichier, ou que vous ajoutez un point au projet Trimble Geomatics Office, les paramètres de projection par défaut, ne seront pas mis à jour. Cependant, vous pouvez mettre à jour l'origine de la projection en effectuant une calibration GPS (sélectionnez Levé / Calibration du site GPS).


L'ajout d'un point WGS-84 à un projet (projection par défaut non définie)

Le dialogue Définition de projection par défaut s'affiche lorsque le projet Trimble Geomatics Office a une projection par défaut non définie (aucune latitude d'origine et aucune longitude d'origine) et

  un point WGS-84 est entré

  un point WGS-84 est importé.

Le dialogue Définition de projection par défaut contient les champs suivants:

Nord

Ce champ passe par défaut à 0. Saisissez la meilleure coordonnée nord connue pour le point utilisé à définir la projection par défaut. Cette valeur est utilisée comme la constante nord de la projection.

Est

Ce champ passe par défaut à 0. Saisissez la meilleure coordonnée est connue pour le point utilisé à définir la projection par défaut. Cette valeur est utilisée comme la constante est de la projection.

Latitude origine

La latitude WGS-84 du point s'affiche ici. Ce champ est la latitude d'origine pour la projection. Ce champ n'est pas disponible.

Longitude origine

La longitude WGS-84 du point s'affiche ici. Ce champ est la longitude d'origine pour la projection. Ce champ n'est pas disponible.

OK

Cliquez sur OK pour accepter les paramètres de projection par défaut. Ces valeurs sont utilisées pour définir les système de coordonnées par défaut du projet. Pour afficher ces paramètres, sélectionnez Fichier / Propriétés du projet / tabulation Système de coordonnées et cliquez sur Détails.

NOTE: Dès que la projection par défaut est complètement définie, à partir de maintenant lorsque vous importez un fichier .dc, ou que vous ajoutez un point au projet Trimble Geomatics Office, les paramètres de projection par défaut, ne seront pas mis à jour. Cependant, vous pouvez mettre à jour l'origine de la projection en effectuant une calibration GPS (sélectionnez Levé / Calibration du site GPS).


L'importation d'un fichier (avec un projection définie) à un projet avec une projection par défaut non définie

 Vue d'ensemble et plus amples informations

Le système de coordonnées du projet est changé automatiquement au système de coordonnées spécifié dans le fichier.

Un message est écrit dans le rapport d'importation.  Il dit:

« Les systèmes de coordonnées du projet et importé sont différents. La définition importée a été utilisée de préférence à la projection (Mercator transversal) par défaut non définie. «


L'utilisation du système de coordonnées d'un fichier

 Vue d'ensemble et plus amples informations

Lorsqu'un fichier est importé avec un système de coordonnées défini, la définition du système de coordonnées enregistrée dans le fichier est comparée à la définition du système de coordonnées du projet. Lorsqu'ils  diffèrent, le dialogue Système de coordonnées du projet s'affiche.

Pour utiliser le système de coordonnées spécifié dans le fichier, sélectionnez l'option Convertir à la définition du carnet de terrain dans le dialogue Système de coordonnées du projet.

Lorsque le système de coordonnées dans le fichier est tout autre qu'un système à facteur d'échelle seul, la définition du système de coordonnées du projet est changée à la définition spécifiée dans le fichier.

Lorsque le système de coordonnées dans le fichier est de Facteur d'échelle seul, ce que se passe dépend du système de coordonnées défini pour le projet.

NOTE: Lorsque le système de coordonnées du projet est une projection par défaut non définie et n'a aucun point, le système de coordonnées dans le fichier est utilisé automatiquement et le système de coordonnées du projet est changé au système de coordonnées dans le fichier.


Fichiers de station totale (facteur d'échelle seul) sans aucun système de coordonnées défini - Vue d'ensemble

Vue d'ensemble

Un système de coordonnées à facteur d'échelle seul peut être utilisé pour des levés dont toutes les observations ont été effectuées à l'aide d'un instrument conventionnel (c'est à dire, aucune observation GPS).

La façon dont les fichiers contenant seulement les observations terrestres sont traités dans Trimble Geomatics Office dépend du système de coordonnées défini dans le projet Trimble Geomatics Office.

Projection par défaut non définie

Si le projet Trimble Geomatics Office a une projection par défaut non définie (aucune latitude d'origine ou aucune longitude d'origine), le dialogue Projection par défaut s'affiche. Vous pouvez entrer les valeurs de constante nord, de constante est et d’échelle.

AVERTISSEMENT: Lorsqu'un point existe déjà dans la base de données, l'échelle dans le fichier n'est pas utilisée pour mettre à jour la projection par défaut. Vérifiez que l'échelle dans le fichier importé est la même que l'échelle dans le projection par défaut du projet.

Système de coordonnées défini

Lorsque le projet Trimble Geomatics Office a un système de coordonnées défini (y compris la projection par défaut dans laquelle les valeurs de latitude et de longitude d'origine ont été définies), le système de coordonnées du projet est utilisé, et on suppose que l'échelle dans le fichier est la même que l'échelle dans le système de coordonnées du projet.


L'utilisation des fichiers à facteur d'échelle seul (projection par défaut non définie)

 Vue d'ensemble et plus amples informations

Le dialogue Définition de projection par défaut s'affiche lorsque:

  le fichier que vous importez contient une projection à facteur d'échelle seul

  le projet Trimble Geomatics Office a une projection par défaut non définie (c'est à dire, la latitude d'origine et la longitude d'origine ne sont pas définies)

  il n'y a aucun point dans la base de données.

Le dialogue contient les champs suivants:

Constante nord

La valeur nord est dérivée du premier point dans le fichier. Vous pouvez modifier les valeurs de l'origine constante.

Constante est

La valeur est dérivée du premier point dans le fichier. Vous pouvez modifier les valeurs de l'origine constante.

Facteur d'échelle

Le facteur d'échelle est dérivé du dernier enregistrement de facteur d'échelle dans le fichier. Vous pouvez modifier les valeurs du facteur d'échelle.

NOTE: Lors de la recherche du premier point dans le fichier, les points supprimés ne sont pas considérés.

OK

Cliquez sur OK pour accepter les paramètres de la projection par défaut. Ces valeurs sont utilisées pour définir le système de coordonnées du projet. Pour afficher ces paramètres, sélectionnez la tabulation Fichier / Propriétés du projet / Système de coordonnées et cliquez sur Détails.

NOTE: Du fait que votre projection par défaut n'est pas encore complètement définie (c'est à dire, la latitude d'origine et la longitude d'origine et la longitude d'origine ne sont pas encore définies), la Projection par défaut sera mise à jour lorsqu'un fichier contenant une ou plusieurs positions GPS est importé.

La correction au niveau de la mer est désactivée à moins que vous n'avez la case à cocher Utiliser coordonnées au sol sélectionné dans le dialogue Paramètres du site local.


L'utilisation des fichiers terrestres avec le système de coordonnées du projet défini

Lorsqu'un système de coordonnées est défini pour le projet Trimble Geomatics Office, et vous importez un fichier, un message s'affiche indiquant que le système de coordonnées du projet sera utilisé. Puis les points dans le fichier sont importés.

Trimble Geomatics Office suppose que l'échelle du projet dans le système de coordonnées existant, et le facteur d’échelle au sol, est la même que l'échelle dans le fichier.

Si l'option Utiliser coordonnées au sol n'est pas sélectionnée dans le dialogue Paramètres du site local, il faut vérifier que les échelles sont identiques avant l'importation, et que le facteur d’échelle dans le fichier est égal au facteur d’échelle combiné.

Si la case à cocher Utiliser coordonnées au sol est sélectionnée dans le dialogue Paramètres du site local, vérifiez que l'élévation du projet dans Trimble Geomatic Office est correcte pour les données importées.


Dialogue Système de coordonnées du projet

Ce dialogue s'affiche lorsqu'un fichier est importé et la définition du système de coordonnées enregistrée dans le fichier .dc diffère de la définition du système de coordonnées du projet, et  le système de coordonnées du projet est défini (c'est à dire, le système de coordonnées du projet n'est pas une projection par défaut non définie).

Système de coordonnées

Pour utiliser le système de coordonnées dans le fichier, sélectionnez l'option Convertir à la définition du carnet de terrain. Toutes les données existantes dans la base de données sont mises à jour utilisant la nouvelle définition du système de coordonnées.

Pour conserver la définition du système de coordonnées du projet sélectionnez l'option Conserver la définition existante du projet. Les données dans le fichier du carnet de terrain sont interprétées utilisant la définition du système de coordonnées par défaut du projet.

Pour afficher les détails de l'un ou l'autre système de coordonnées, cliquez sur le bouton Détails approprié.

Résumé

Cliquez sur Résumé pour afficher un rapport de comparaison des deux systèmes de coordonnées. Utilisez le rapport pour examiner les paramètres complets des systèmes de coordonnées et trouver les différences entre les systèmes.


L'utilisation du dialogue Système de coordonnées du projet lors de l'importation du fichier

Lors de l'importation d'un fichier, la définition du système de coordonnées enregistrée dans le fichier est comparée à la définition du système de coordonnées par défaut du projet. Pour utiliser le système de coordonnées spécifié dans le projet, sélectionnez l'option Conserver la définition existante du projet  dans le dialogue Système de coordonnées du projet.

Projection par défaut non définie

Si le système de coordonnées du projet est une projection par défaut non définie, le dialogue Mettre à jour projection par défaut s'affiche. Le premier point dans le fichier est utilisé pour mettre à jour la Projection par défaut.

Autres systèmes de coordonnées

Le fichier est importé utilisant le système de coordonnées du projet.

NOTE: Si le système de coordonnées dans le fichier est de Facteur d'échelle seul, ce que se passe dépend du système de coordonnées défini pour le projet.


L'utilisation d'une projection de plan à partir de Trimble Survey Office ou Survey Controller

Trimble Geomatics Office ne mettra pas à jour une projection de plan non définie.

Lorsque vous effectuez l'une des choses suivantes:

  exportez un fichier Trimble Graphics (*.tgf) de Trimble Survey Office dans lequel le projet Trimble Survey Office  a une projection de plan

  exportez un fichier .dc d'un projet Trimble Survey Office avec une projection de plan

et puis vous importez le fichier à Trimble Geomatics Office, la projection de plan peut toujours être utilisée pour le projet Trimble Geomatics Office. (Vous devez sélectionner l'option Utiliser le système de coordonnées du fichier .dc lorsque le dialogue Système de coordonnées du projet s'affiche.)

La projection de plan devient le système de coordonnées sélectionné pour le projet. Utiliser le dialogue Propriétés du projet / tabulation Système de coordonnées pour afficher les détails du système de coordonnées.


Réalisation d'un Ajustement de réseau avec données d'un fichier à facteur d'échelle seul (projection par défaut non définie)

Un fichier de données terrestres (c’est à dire, un fichier sans des observations GPS) peut avoir un système de coordonnées n'ayant aucun paramètre défini autre qu'un échelle; c'est à dire, il n'a pas une transformation de datum, un ajustement horizontal ou un ajustement vertical. Les systèmes de coordonnées à facteur d'échelle seul sont créés dans le Trimble Survey Controller (version 7.00 ou ultérieure).

Vous devez définir un système de coordonnées avant de pouvoir effectuer un ajustement de réseau. Donc, si vous avez importé une étude à facteur d'échelle seul du Trimble Survey Controller dans un système de coordonnées du projet par défaut non défini dans Trimble Geomatics Office, le logiciel vous demande de mettre à jour la projection par défaut non définie.

Dans le dialogue que s'affiche, vous devez saisir une latitude et une longitude pour l'origine de projection, et les coordonnées de constante nord et de constante est.


Paramètres de rotation de la terre

Lorsque vous traitez les lignes de base d’une longueur de plus de 100 km, le processeur de lignes de base utilise les paramètres de rotation de la terre pour modeler les marées de pôle.

Ces paramètres de rotation de la terre, surtout les coordonnées du pôle, sont disponibles chaque semaine sous forme de (fichiers .erp) des organisations telle que le Service GPS International (SGI). Si un fichier .erp n’est pas disponible pour le temps de la ligne de base, les coordonnées du pôle prévues sont utilisées.

Utilisez l’Explorateur de Windows pour copier tous les fichiers .erp relatifs dans le dossier Data Files\Precise Ephemeris (Fichiers de données\Ephéméride précise) sous le dossier de projet. Le processeur de lignes de base recherche automatiquement dans ce dossier pour ces fichiers s’ils sont requis. S’il ne les trouve pas, il utilise les valeurs prévues.

NOTE: Si vous utilisez un fichier .erp dans plus d’un projet, vous devez copier ce fichier dans chacun des projets.


Réseau géodésique à haute précision

Les fichiers de grille de datum HPGN (High Precision Geodetic Network) pour les systèmes US State Plane sont fournis comme partie de l’installation du système Trimble Geomatics Office. Les fichiers de Grille de datum sont stockés dans le dossier GeoData sous le dossier Trimble Common Files (Fichiers communs Trimble) (en général Program Files\Common Files\Trimble\GeoData). Les fichiers de grille de datum HPGN ne sont pas disponibles pour tous les états des Etats-Unis.

La convention du nom de fichier de grille est:

  L’identificateur d’état à deux caractères + hpgn + la (latitude) ou lo (longitude), avec l’extension de fichier .dgf. Par exemple, les fichiers de grille California HPGN sont Cahpgnla.dgf et Cahpgnlo.dgf.

Du fait que la gestion du système de coordonnées dans Trimble Geomatics Office est basé sur l’ellipsoïde WGS-84, les grilles de datum HPGN (conçues pour la conversion bidirectionnelle entre NAD83 (1986) et NAD83 (HPGN)) ne s’adaptent pas nettement dans ce modèle de système de coordonnées. Donc, et afin d’éviter toute confusion, le logiciel ne fournit aucune définition de système de coordonnées prédéfinie pour référencer ces grilles de datum.

Vous pouvez configurer vos propres définitions de système de coordonnées utilisant les fichiers de grille de datum HPGN appropriés dans l’utilitaire Coordinate System Manager.

Pour ce faire:

1.	Créez une nouvelle définition de datum qui fait référence aux fichiers de grille de datum HPGN appropriés pour l’état.

2.	Créez une nouvelle définition de système de coordonnées utilisant les paramètres de projection State Plane appropriés et faisant référence à la définition de datum nouvellement créée.

Utilisez la nouvelle définition de système de coordonnées dans un projet Trimble Geomatics Office afin d’obtenir des valeurs NAD-83 (1986) de latitude et longitude des coordonnées de projection HPGN connues; ou des valeurs de latitude, longitude et hauteur.

Effectuez les choses suivantes:

1.	Entrez les valeurs HPGN comme des coordonnées de grille; ou des valeurs de latitude, longitude et hauteur locales comme approprié.

2.	Lisez les valeurs NAD-83 (1986) de latitude et longitude des champs WGS-84 affichés.

Obtenez des valeurs HPGN des valeurs NAD-83 (1986) de latitude, longitude, et de hauteur  connues comme suit:

1.	Entrez les valeurs NAD-83 (1986) de latitude, longitude, et de hauteur comme des valeurs WGS-84.

2.	Lisez les coordonnées de grille HPGN équivalentes; ou les valeurs de latitude, longitude, et de hauteur des champs de grille ou local respectivement.


Système de coordonnées New Brunswick (Canada)

Maintenant la base de données du système de coordonnées a la projection Stéréographique en double et l’ellipsoïde Average Terrestrial System 1977 (ATS77). Vous devez acheter la grille de datum et le modèle de géoïde du Geodetic Survey of Canada avant d’installer le système de coordonnées New Brunswick.

Le système de coordonnées New Brunswick (Canada) utilise:

  Ellipsoïde: ATS77

  Datum: NTv2 Datum Grid

  Geoïde: HT101

  Projection: Stéréographique double:

– Latitude centrale: 46° 30' 00" N

– Longitude centrale: 66° 30' 00" W

– Constante nord: 800 000 m

– Constante est: 300 000 m

– Facteur d’échelle: 0.999912

Pour créer le système de coordonnée de New Brunswick (Canada):

1.	Utilisez l’utilitaire Grid Factory pour créer des fichiers de Grille de géoïde (*.ggf) du modèle de géoïde HT101 acheté du Geodetic Survey of Canada.

2.	Utilisez l’utilitaire Grid Factory pour créer des fichiers de Grille de datum(*lat.dgf, *lon.dgf) de la grille de datum NTv2 achetée du Geodetic Survey of Canada. Pour plus amples informations, référez-vous à l’aide de l’utilitaire Grid Factory.

3.	Utilisez l’utilitaire Coordinate System Manager pour ajouter un nouveau système de coordonnées. Pour plus amples informations, référez-vous à la rubrique Commande Ajouter système de coordonnées dans l’Aide de l’utilitaire Coordinate System Manager.


 Glossaire

This glossary defines the Ce glossaire définit les termes spécifiques à GPS et à Trimble Geomatics Office. Il ne comprend pas les termes de Microsoft Windows ou les termes généraux concernant les ordinateurs.

Symboles

1-sigma

A

à deux variables

AASHTO

Accès sélectif (SA)

affichage actuel

ajustement de réseau

ajustement libre

ajustement vertical

ajustement

algorithme

almanach

ambiguïté entière

ambiguïté

angle cartographique

annotation

Anti-déception électronique

APC

aplatissement

azimut géodésique

azimut

azimut de section normale

B

baud

Bifréquence

C

calibration du site

Centré à la terre - Fixé à la terre (ECEF)

centre de phase

cinématique en temps réel

CMR

Code d'acquisition approximative (C/A)

code P

Code Y

code

codes de caractéristiques

codes de champ

composant

constellation

contraint au minimum

contraint complètement

contrainte

contraintes internes

contraint

convergence d'ajustement

conversion de grille

Coordonnées cartésiennes

coordonnées de calibration

coordonnées fixes

coordonnées géodésiques locales

coordonnées géographiques

coordonnées rectangulaires

correction de phase d'antenne

correction tropo

corrélée

couches

courbe de distribution normale

covariance

critère tau

D

datum d'ajustement

datum du projet

datum géodésique

datum

Décalage de l'horloge

décorrélér

défaut du datum

degrés de liberté

délai de zénith

delta élévation

delta hauteur

delta N, delta E, delta U

delta X, delta Y, delta Z

dérapage de cycle

déviation du vertical

différentiation double

distance au sol

distance de grille

distance d'ellipsoïde

distance horizontale

distance inclinée

distribution de fréquence

données brutes GPS

DOP

E

échelle

effet Doppler

élévation

ellipse d'erreur

ellipsoïde local

ellipsoïde

enregistrement des données

enregistrements QC

entités

éphéméride précise

éphéméride

époque

erreur d'installation

erreur propagée

erreur standard d'unité de poids

erreur standard

erreur

erreurs a posteriori

erreurs a priori

erreurs de centrage de tribraque

erreurs relatives

erreurs systématiques

est

estimation de composant de variance

état de réseau

état

étiquette

événements

exactitude

F

facteur de référence

fermeture de boucle

fermeture

fichier de bibliothèque d'attributs et caractéristiques (*.fcl)

fixe

G

GDOP

géoïde

géomatique

géométrie de satellites

GPS

grand axe

grande voie

grille

groupe de composants de variance

groupe de paramètres de transformation

H

hauteur d'antenne

hauteur d'ellipsoïde

hauteur orthométrique

HDOP

heure GPS

HI

histogramme

horizon géodésique local

I

inconnus

indépendant

initialisation de point connu

intervalle d'époque

inverse d'aplatissement

iono libre

ionosphère

itération

J

jalon de mesure de hauteur

K

L

L1

L2

levé cinématique

ligne de base GPS

ligne de base

lignes de base redondantes

M

masque d'élévation

masque PDOP

matrice de covariance

matrice variance-covariance

méridien tourné

message de données

méthode de recherche OTF (en vol)

MNT

mobile

modèle de géoïde

modèle stochastique

modèle tropo

modèles de centre de phase

modélisation

modélisation ionosphérique

moindres carrés

mono-fréquence

multitrajet

N

NAVDATA

niveau de confiance

niveau d'importance

niveau moyen de la mer

NMEA

nombre de redondance

nord

normal

O

observation conventionnelle

observation du géoïde

observation redondante

observation terrestre

observations de levé

observations GPS

observations

onde

origine

P

paramètre

paramètres de transformation

parité

PDOP

petit axe

pied américain

plan du ciel

plomb

poids

poids

point de contrôle horizontal

point de contrôle vertical

point de contrôle

pondération scalaire

position horizontale

positionnement autonome

positionnement différentiel

positions de points

post traiter

poursuite

PPM

précision

prise de vue isolée

PRN

probabilité

processeur de lignes de base

profil de colonne réduit

projection cartographique

projection

projet

Q

R

ratio

RDOP

recherche de nombre entier

redondance

réseau statique

réseau

résiduel d'observation

résiduel

RINEX

RMS

rotation

RTCM

RTK

S

scalaire

SDMS

semi grand axe

semi petit axe

séparation du géoïde

session

seuil PDOP

sigma

signe algébrique

site calibré

SNR

solution finale

solution fixe

solution flottante

sous-réseau

State Plane Coordinates

station de base

station de référence

statique (levé)

Statique rapide

stratégie de pondération

styles CAD

styles d'ajustement

Super-trak

surdéterminé

SV

symboles et types de lignes

système de coordonnées

système de référence

T

tau (valeur)

TDOP

temps d'occupation

test d'Acceptance de qualité

test du chi carré

TOW

transformation de datum

transformation

tribraque

types de solution

U

univariable

URA

US National

UTC

V

valeur aberrante

valeur probable

valeurs ajustées

valeurs de covariance

variance de référence

variance

VDOP

vecteur

vertical

voie étroite

W

WGS-84

X

X, Y, et Z

Y

Z


1-sigma

Une erreur standard de la moyenne.


erreurs a posteriori

Des erreurs a priori multipliées par l'erreur standard du poids unitaire (facteur de référence) résultant d'un ajustement de réseau.


erreur a priori

Des erreurs estimées pour les observations avant un ajustement de réseau.


AASHTO

American Association of State Highway and Transportation Officials


exactitude

La proximité d'une mesure à la valeur (vraie) actuelle de la quantité pour laquelle la mesure est en cours.


valeurs ajustées

Une valeur dérivée des données observées (de mesure) par l'application d'un procédé d'élimination des erreurs dans ces données dans un ajustement de réseau.


ajustement

Le procédé de détermination et d'application des corrections aux observations afin de réduire les erreurs dans un ajustement de réseau.


convergence d'ajustement

Lorsque l'ajustement de réseau se conforme à la tolérance résiduelle définie ou à la tolérance résiduelle ultime dans un nombre défini d'itérations.


datum d'ajustement

Le datum utilisé dans une itération d'ajustement de réseau courante. Trimble Geomatics Office vous permet de sélectionner soit le datum du projet soit WGS-84.


styles d'ajustement

Les paramètres par défaut et définis de l'utilisateur de Trimble utilisés pour un ajustement de réseau.


signe algébrique

Le signe (+ ou −) associé avec une valeur qui la désigne comme un nombre positif ou négatif.


algorithme

Un groupe de règles qui sert à résoudre un problème dans un nombre fini d'étapes.


almanach

Les données, transmises d'un satellite GPS, qui comprennent des informations d'orbite de tous les satellites, la correction d'horloge, et les paramètres de retard atmosphérique.

L'almanach permet une acquisition rapide des SV. Les informations d'orbite sont un sous-groupe des données d'éphéméride avec une précision réduite.


ambiguïté

Le nombre entier de cycles inconnu de la phase porteuse reconstruite contenu dans un groupe de mesures ininterrompu. Le récepteur compte les ondes radio (depuis le satellite lorsqu'elles passent l'antenne) à un haut degré d'exactitude. Cependant il n'a aucune information du nombre des ondes au satellite au temps de commencer le compte. Ce nombre inconnu de longueurs d'onde entre le satellite et l'antenne est donc l'ambiguïté. Egalement connue comme ambiguïté entière  ou biais entier.


annotation

Une pièce de texte décrivant un autre enregistrement dans la base de données. Pour sélectionner et éditer les annotations, il faut utiliser la fenêtre Propriétés.

Une annotation est vive—tout champ est développé à nouveau chaque fois que l'entité mère change.


hauteur d'antenne

La hauteur d'un centre de phase d'antenne GPS au-dessus du point observé.

La hauteur d'antenne non corrigée est mesurée à partir du point observé à un point désigné sur l'antenne, puis corrigée au vertical vrai soit manuellement soit automatiquement dans le logiciel.


correction de phase d'antenne

Le centre de phase d'une antenne GPS n'est pas un point physique ni stable. Le centre de phase d'une antenne GPS change par rapport à la direction variable du signal d'un satellite. La plupart de la variation du centre de phase dépend de l'élévation du satellite. Effectuant un modèle de cette variation au site du centre de phase d'antenne permet l'utilisation d'une variété de types d'antenne dans un seul levé. Les corrections du centre de phase d'antenne ne sont pas aussi critiques lorsque deux antennes identiques sont utilisées parce que les erreurs communes s'annulent.


Anti-déception électronique (AS)

Une caractéristique permettant au Département de Défense des Etats-Unis de transmettre un code Y crypté à la place d'un code P. Le code Y est destiné à servir seuls des utilisateurs autorisés (essentiellement militaires). AS (anti-déception électronique) est utilisé avec Selective Availability (accès sélectif) afin de refuser la précision entière de GPS aux utilisateurs civils.


Antenna Phase Center (Centre de phase d'antenne)

Le centre électronique de l'antenne. Souvent il ne correspond pas au centre physique de l'antenne. Le signal radio est mesuré à l'APC.

Dans la fenêtre Propriétés, la hauteur d'un point peut être l'élévation de l'APC. Lorsque la hauteur est spécifiée comme APC, c'est la hauteur de l'APC—pas la hauteur au sol.


positionnement autonome

Un mode d'opération dans lequel un récepteur GPS calcule les positions en temps réel uniquement des données de satellite, sans tenir compte des données fournies d'une station de base. Le positionnement autonome est la procédure de positionnement la moindre précise pouvant être effectuée d'un récepteur GPS, produisant des positions d'une précision de ±100 mètres RMS horizontal lorsqu'il y a d'accès sélectif, et d'une précision de ±10--20 mètres lorsqu'il n'y en a pas. Egalement connu comme le positionnement absolu et le positionnement de point.


azimut

Une observation de levé utilisée pour mesurer l'angle formé par une ligne de base horizontale et le nord géodésique. Lorsqu'appliqué aux observations GPS, il fait référence à un azimut de section normal.


station de base

Une antenne et un récepteur installés sur un emplacement connu. Elle est utilisée pour les levés différentiels ou cinématiques en temps réel (RTK). Les données peuvent être enregistrées à la station de base pour un post traitement plus tard. Une station de référence Trimble, tel que le logiciel Trimble Reference Station (TRS), ou le logiciel Universal Reference Station (URS) de Trimble et un récepteur en mode de station de base sont des exemples des stations de base.

Pendant la topographie GPS, vous observez et calculez des lignes de base (c'est à dire, la position d'un récepteur par rapport à un autre). La station de base sert comme la position à partir de laquelle toute autre position inconnue est dérivée.


ligne de base

La position d'un point par rapport à un autre point. Dans la topographie GPS, c'est la position d'un récepteur par rapport à un autre. Lorsque les données de ces deux récepteurs sont combinées, le résultat est une ligne de base consistant en un vecteur en trois dimensions entre les deux stations.


Processeur de lignes de base

Un programme informatique qui calcule les solutions de lignes de base à partir des mesures de satellite. Il peut s'exécuter comme un postprocesseur sur un ordinateur personnel, ou comme un processeur en temps réel dans un récepteur. WAVE (Weighted Ambiguity Vector Estimator – Estimateur de vecteur d'ambiguïté pondéré) est le processeur de lignes de base de Trimble.


baud

Une unité de la vitesse du transfert de données (à partir d'un dispositif numérique binaire vers un autre) utilisée lors de la description des communications en série.


à deux variables

Une fonction mathématique décrivant le comportement des erreurs aléatoires en deux dimensions dans les ellipses d'erreur de:

  nord/est

  latitude/longitude

  X/Y


styles CAD

Les styles CAD définissent l'apparence des points, des lignes, des arcs, des courbes, du texte et des annotations dans un projet. Par exemple, un style peut consister en un symbole, un type de ligne, une couleur ou une police. Les définitions de style sont stockées dans un projet.

Afin d'avoir des styles disponibles pour plusieurs projets, il faut définir les styles dans un modèle de projet.


site calibré

Une définition du site utilise une définition du système de coordonnées existant plus une transformation de corrections. Ainsi, la meilleure adaptation pour les données GPS est obtenue dans une surface (ou site) spécifique.

Les transformations de corrections supplémentaires sont nécessaires parce qu'un système de coordonnées est conçu pour une application sur une surface très grande. Il ne tient pas en compte les variations dans les coordonnées locales.

Il faut que tout nouveau travail s'adapte au contrôle existant, donc les transformations de corrections supplémentaires corrigeront pour ces variations locales. Les corrections supplémentaires ne sont valides que sur une surface limitée. Ainsi, la terminologie 'site' est expliquée.

Trimble Geomatics Office peut calculer les transformations supplémentaires requises pour adapter au contrôle local et enregistrer ces définitions dans la base de données du système de coordonnées.


coordonnées de calibration

Les coordonnées WGS-84 (latitude/longitude/hauteur d'ellipsoïde) générées à partir d'un ajustement de réseau contraint imperceptiblement de votre observation GPS et enregistrées pour une utilisation ultérieure dans une calibration du site GPS.

Les coordonnées de calibration sont utilisées comme les coordonnées GPS observées associées aux coordonnées de grille d'un point spécifique lors de la réalisation d'une calibration.


Coordonnées cartésiennes

Référez-vous à Coordonnées cartésiennes centrées à la terre − fixées à la terre.


test du chi carré

Un test statistique global de l'ajustement de réseau. C'est un test du total des poids carrés des résiduels, le nombre de degrés de liberté et une probabilité critique de 95 pour-cent ou supérieur.

L'objet du test est de rejeter ou d'accepter l'hypothèse que les erreurs prévues ont été estimées précisément.


décalage de l'horloge

La différence constante de la lecture de l'heure entre deux horloges. Avec GPS, en général il fait référence au décalage entre les horloges SV et l'horloge dans le récepteur de l'utilisateur.


fermeture

Un accord entre les parties mesurées et connues d'un réseau.


Compact Measurement Record (Enregistrement de mesure compact)

Un message de mesure du satellite diffusé par un récepteur de base et utilisé dans les levés cinématiques en temps réel (RTK) pour calculer un vecteur de ligne de base précis de la base au mobile.


code d'Acquisition approximative (C/A)

Un code de bruit pseudo-aléatoire (PRN) modulé sur un signal L1. Ce code aide le récepteur de calculer la distance du satellite.


code

Le code GPS  est un code de bruit pseudo-aléatoire (PRN) modulé sur les signaux porteurs GPS.

Le code C/A est non classifié et est disponible aux applications civiles.

Le code P est également connu et non classifié, mais il peut être crypté en ce qui concerne la défense nationale.

Les mesures de code sont la base du positionnement et de la navigation GPS. En plus le code est utilisé conjointement avec les mesures de la phase porteuse pour obtenir des solutions de lignes de base de qualité de levé plus précises.


Composant

L'une des trois observations de levé utilisée pour définir une ligne de base en trois dimensions entre deux points de contrôle. La même ligne de base peut être définie avec un azimut, delta hauteur et distance (dans les coordonnées d'ellipsoïde), avec delta X, delta Y, et delta Z (dans les coordonnées cartésiennes centrées sur la terre); et avec delta nord, delta est, et delta hauteur (dans les coordonnées de plan local).


constellation

Un groupe spécifique de satellites utilisé pour le calcul des positions: trois satellites pour les positions 2D, quatre satellites pour les positions 3D.

Tous les satellites visibles à un récepteur GPS au même temps. La constellation optimale est la constellation avec la PDOP la plus basse. Voir aussi PDOP.


contraint

Tenir (fixer) une quantité (observation et coordonnée) comme vraie dans un ajustement de réseau.


contrainte

Les limitations externes imposées sur les quantités ajustables (observations et coordonnées) dans un ajustement de réseau.


point de contrôle

Un point de référence auquel des coordonnées ont été, ou sont en cours d'être, assignées en utilisant les observations de levé.


observation conventionnelle

Une observation sur le terrain obtenue utilisant une station totale ou un théodolite.


système de coordonnées

Un groupe de transformations permettant la transformation des positions GPS (dans l'ellipsoïde WGS-84) aux coordonnées de projection ayant des élévations au-dessus du géoïde.

Il consiste en une transformation de datum, une allocation de modèle du géoïde et une définition de projection de coordonnées.

La transformation de datum est définie dans la base de données du système de coordonnées. Elle comprend la définition du datum sur lequel la projection de coordonnées est basé.

Un modèle du géoïde existant peut être assigné au système de coordonnées, mais il est également possible de spécifier une séparation géoïdale constante au lieu d'utiliser le modèle du géoïde.  Utilisez l'utilitaire Trimble Coordinate System Manager™ (Gestionnaire du système de coordonnées de Trimble) pour définir les modèles du géoïde dans la base de données du système de coordonnées.

Vous pouvez assigner certains types de projection de coordonnées à un système de coordonnées (par exemple, Mercator transversal ou Lambert 1 parallèle). Les types de projection divers sont utilisés dans des pays et des régions diverses afin d'obtenir des résultats optimums (c'est à dire, une distorsion minimale) dans les coordonnées de projection. Les méthodes de projection de coordonnées projettent des valeurs de latitude et longitude sur le datum approprié aux valeurs de coordonnées cartésiennes. Les élévations pour les coordonnées de projection sont obtenues utilisant le modèle du géoïde assigné au système de coordonnées.

L'utilitaire Coordinate System Manager de Trimble vous permet d'afficher, éditer et ajouter aux définitions du système de coordonnées fournit avec Trimble Geomatics Office.


correlée
corrélée

Le terme pour deux ou plusieurs observations (ou quantités dérivées) ayant au moins une source commune d'erreur.


covariance

Une mesure de la corrélation d'erreurs entre deux observations ou quantités dérivées. Elle fait référence également à un terme de diagonale excentrée (c'est à dire, pas une variance) dans une matrice de covariance-variance.


matrice de covariance

Une matrice qui définit la variance et la covariance d'une observation. Les éléments de la diagonale sont la variance et les éléments de chaque côté de la diagonale sont la covariance.


valeurs de covariance

Comme utilisées par Trimble Geomatics Office, c'est la publication des erreurs a posteriori propagées (calculées) dans l'azimut, la distance et la hauteur entre les paires de points de contrôle résultant d'un ajustement de réseau. Le terme covariance indique que ce calcul nécessite l'utilisation des termes de covariance dans la matrice de variance-covariance des points de contrôle ajustés.


affichage actuel

Vous pouvez ouvrir plus d'un affichage sur la base de données en utilisant la commande Fenêtre / Nouvelle fenêtre. Chacun des affichages peut avoir des paramètres d'affichage différents. L'affichage actuel est l'affichage actif et c'est identifié par l'utilisation de la barre de titres active.


dérapage de cycle

Une interruption à la capture par un récepteur des signaux radio d'un satellite. Un dérapage de cycle nécessite la réestimation des termes de l'ambiguïté de nombre entier lors du traitement des lignes de base.


enregistrement des données

Le procédé d'enregistrement des données de satellites dans un fichier stocké dans le récepteur, un carnet de terrain exécutant Trimble Survey Controller, ou sur une carte PC.


message de données

Un message, compris dans le signal GPS, qui fournit un rapport sur l'emplacement et la santé des satellites ainsi que toute correction d'horloge. Il comprend des informations concernant d'autres satellites ainsi que leur position approximative.


datum

Un modèle mathématique de la terre conçu afin de s'adapter à une partie du géoïde ou tout le géoïde. Il est défini par le rapport entre un ellipsoïde et un point sur la surface topographique établi comme l'origine du datum. En général, il est connu comme un datum géodésique.

En général, les dimensions et la forme d'un ellipsoïde, et l'emplacement du centre de l'ellipsoïde par rapport au centre de la terre, définissent les datums géodésiques mondiaux.


défaut du datum

Les désaccords inconnus entre deux groupes de coordonnées qui ne peuvent être corrigés qu'en utilisant une transformation de datum comme partie d'un ajustement de réseau.


transformation de datum

Définit la transformation utilisée pour transformer les coordonnées d'un point défini dans un datum aux coordonnées dans un datum différent.

Il y a de nombreuses méthodes de transformation de datum différentes prises en charge par Trimble Geomatics Office:

  Sept paramètres

  Trois paramètres (également connu comme Molodensky)

  Grille de datum

  Régression multiple

En général les transformations de datum effectuent la transformation des données collectées dans le datum WGS-84, par des méthodes GPS, sur les datums utilisés pour les usages de cartographie et topographie dans les régions et les pays individuels.


décorréler

Supprimer les covariances entre les observations. Il est possible de le faire au moyen des transformations orthogonales complexes, ou par le calcul des ajustements horizontal et vertical séparés.


déviation du vertical

La différence angulaire, entre le direction vers le haut du fil à plomb (vertical) et la perpendiculaire (normal) à l'ellipsoïde.


degrés de liberté

Une mesure de la redondance dans un réseau.


delta élévation

La différence de l'élévation entre deux points.


delta hauteur

Le composant vertical dans l'expression de Trimble Geomatics Office des lignes de base GPS. C'est la différence de hauteur ou le changement de hauteur.


delta N, delta E, delta U
delta N, delta E, delta H

Les différences de coordonnées exprimées dans un système de coordonnées de l'horizon géodésique local.


delta X, delta Y, delta Z

Les différences de coordonnées exprimées dans un système de coordonnées cartésien.


positionnement différentiel

La mesure précise d'une position relative de deux récepteurs qui sont en  train de poursuivre les mêmes satellites simultanément.


DOP - Dilution of Precision (Dilution de précision)

Un indicateur de la qualité d'une position GPS. Il tient compte de l'emplacement de chaque satellite par rapport aux autres satellites dans la constellation, et leur géométrie par rapport au récepteur GPS. Une valeur DOP basse indique une plus haute probabilité de précision. Les DOP standard pour les applications GPS sont:

  Position PDOP (trois coordonnées)

  HDOP horizontale (deux coordonnées horizontales)

  RDOP

  VDOP verticale (hauteur seulement)

  TDOP Heure (décalage d'horloge seulement)


Effet Doppler

La modification apparente de la fréquence d'un signal due au mouvement relatif des satellites et du récepteur.


différentiation double

Une méthode mathématique pour la différentiation des phases porteuses mesurées simultanément par deux récepteurs poursuivant les mêmes satellites.  Cette méthode supprime les erreurs d'horloge du récepteur et du satellite.


DTM - Digital Terrain Model (Modèle numérique de terrain)

Une représentation électronique du terrain en trois dimensions.


bifréquence

Un type de récepteur utilisant les signaux L1 et L2 des satellites GPS. Un récepteur bifréquence peut calculer des positions plus précises sur les distances plus longues et sous les conditions plus défavorables parce qu'il effectue une compensation des délais ionosphériques.


Earth-Centered-Earth-Fixed (ECEF) (centré à la terre - fixé à la terre)

Un système de coordonnées cartésien utilisé par le système de référence WGS-84. Dans ce système de coordonnées, le centre du système est au centre de masse de la terre. L'axe z est identique à l'axe de rotation moyen de la terre et l'axe x passe à travers de 0°N et 0°E. L'axe y est perpendiculaire au plan des axes x et z.


est

La lecture dans la direction est des valeurs de grille. De gauche à droite sur une grille (axe X).


élévation

La hauteur au-dessus du niveau moyenne de la mer ou la distance verticale au-dessus du géoïde. Quelques fois l'élévation est connue comme la hauteur orthométrique.


masque d'élévation

Un angle qui est en général réglé sur 13 degrés. Lorsque vous suivez les satellites au-dessus de cet angle, vous évitez d'interférence produite par des constructions, des arbres et des erreurs de multitrajet.

Trimble recommande de ne pas suivre des satellites au-dessous de 13 degrés.


ellipsoïde

Un modèle mathématique de la terre formé par la rotation d'une ellipse autour de son axe mineur. Pour les ellipsoïdes qui modèlent la terre, l'axe mineur est l'axe polaire et l'axe majeur est l'axe équatorial.

Un ellipsoïde est  défini en spécifiant les longueurs de tous les deux axes, ou en spécifiant la longueur de l'axe majeur et l'aplatissement.

Deux quantités définissent un ellipsoïde; en général elles sont exprimées comme

la longueur du demi grand axe, a, et

l'aplatissement, [image], où b est la longueur du demi petit axe.


distance d'ellipsoïde

Comme utilisée dans Trimble Geomatics Office, c'est la longueur de la section normale entre deux points.

ATTENTION: La distance d'ellipsoïde n'est pas la même que la distance géodésique.


hauteur d'ellipsoïde

La distance mesurée le long du normal, de la surface de l'ellipsoïde à un point.


entités

Les éléments graphiques primaires que vous pouvez afficher et sélectionner de la fenêtre de graphiques.

Les entités disponibles dans Trimble Geomatics Office sont des points, des lignes, des arcs, des courbes, du texte et des annotations.


éphéméride

Un groupe de données décrivant la position d'un objet céleste sous forme d'une fonction de temps. Chaque satellite GPS transmet périodiquement une éphéméride diffusée décrivant ses positions prévues dans un proche avenir, téléchargée par le Segment de contrôle. Les programmes de post traitement peuvent également utiliser une éphéméride précise qui décrit les positions exactes d'un satellite dans le passé.


époque

L'intervalle de mesure d'un récepteur GPS. L'époque varie selon le type du levé:

  pour des levés en temps réel elle est réglé sur une seconde.

  pour des levés post traités elle peut être réglé à un taux d'entre une seconde et une minute.


intervalle de l'époque

L'intervalle de mesure utilisé par un récepteur GPS, également appelé un cycle.


erreur

La différence entre la valeur mesurée d'un quantité et sa valeur réelle. Les erreurs de levé en général sont divisées dans trois catégories: les fautes, les erreurs systématiques et les erreurs aléatoires. Une analyse par les moindres carrés est utilisée pour détecter et éliminer les fautes et les erreurs systématiques, et un ajustement par les moindres carrés est utilisé pour mesurer et distribuer correctement toute erreur aléatoire.


ellipse d'erreur

Une ellipse d'erreur de coordonnée est une représentation graphique de l'ampleur et la direction de l'erreur des points de réseau ajusté.


événements

Un enregistrement de l'événement qui s'est produit, telle que la fermeture de l'obturateur d'un appareil photographique photogrammétrique. Un récepteur GPS peut enregistrer une marque d'événement contenant l'heure de l'événement et un commentaire alphanumérique saisis à l'aide du clavier pour décrire l'événement. Il est possible de lancer un événement au moyen du clavier ou une entrée de signal électrique sur l'un des ports du récepteur.


Statique rapide

Une méthode de levé GPS utilisant des occupations de jusqu'à 20 minutes pour la collecte des données GPS brutes, puis un post traitement pour obtenir des positions sous-centimétriques. En général, les temps d'occupation varient selon le nombre de satellites (SVs) visibles:

  avec 4 SVs il faut 20 minutes*

  avec 5 SVs il faut 15 minutes*

  avec 6 ou plus SVs il faut 8 minutes*

(*collectées à un taux d'époque de 15 secondes)


fichier de bibliothèque d'attributs et caractéristiques (*.fcl
fichier de bibliothèque d'attributs et caractéristiques (*.fcl)

Un fichier texte contenant les définitions des codes de caractéristiques, des attributs, des styles CAD et des codes de contrôle.


codes de caractéristiques

Les mots ou les abréviations descriptifs décrivant les caractéristiques que vous voyez.


codes de champ

Des instructions spéciales qui indiquent à Trimble Geomatics Office d'insérer des informations dans les rapports, les fichiers et les annotations.

Les codes de champ sont utilisés pour s'occuper du transfert des données de base de données dans les opérations de rapport/importation/exportation ASCII.


solution finale

Lorsque le post traitement est utilisé pour générer des vecteurs GPS, notamment pour les solutions statiques, le processeur de lignes de base passe à travers les solutions différentes utilisant une variété de techniques de traitement et de combinaisons de mesures GPS. En général, chaque solution subséquente est meilleure que la solution précédente. La solution finale fournit la meilleure estimation du vecteur GPS entre deux points.


fixe

Voir contraint.


coordonnées fixes

Les coordonnées de point que ne se déplacent pas lors de la réalisation d'un ajustement de réseau.


solution fixe

Une solution obtenue lorsque le processeur de lignes de base peut résoudre la recherche d'ambiguïté de nombres entiers avec une suffisance de confiance pour sélectionner un groupe de nombres entiers au lieu d'un autre. C'est connue comme une solution fixe parce que les ambiguïtés sont toutes fixes de leurs valeurs flottantes estimées à leurs valeurs de nombres entiers correctes.


aplatissement

Une expression mathématiques des longueurs relatives des grands axes et des petits axes d'un ellipsoïde.


inverse d'aplatissement

Une expression de l'aplatissement qui est plus facile à lire et éditer.


solution flottante

Une solution obtenue lorsque le processeur de lignes de base ne peut pas résoudre la recherche d'ambiguïté de nombres entiers avec une suffisance de confiance pour sélectionner un groupe de nombre entier au lieu d'un autre. C'est connue comme une solution flottante parce que l'ambiguïté contient une fraction et n'a pas de nombre entier.


ajustement libre

Effectuant un ajustement de réseau dont aucun point (coordonnée)  n'est contraint. L'ajustement de réseau utilise des contraintes internes.


distribution de fréquence

La taille et l'étendue des résiduels dans un groupe de données. Illustré de façon graphique dans les histogrammes.


entièrement contraint

Un ajustement de réseau dans lequel tous les points dans le réseau qui font partie d'un réseau de contrôle plus grand sont maintenus fixes à leurs valeurs de coordonnées publiées.  Utilisé pour fusionner des réseaux plus petits avec des réseaux de contrôle plus grands et des réseaux anciens avec des nouveaux réseaux.


GDOP - Geometric Dilution of Precision (Dilution de précision géométrique)

Le rapport entre les erreurs de l'heure et la position de l'utilisateur, et les erreurs dans l'étendue de satellites. Voir aussi DOP.


azimut géodésique

L'angle entre le méridien géodésique et la tangente à la ligne géodésique de l'observateur, mesuré dans le plan perpendiculaire à l'ellipsoïde normal de l'observateur. Dans le sens des aiguilles d'une montre à partir du nord.


datum géodésique

Un modèle mathématique conçu à s'adapter à une partie ou tout le géoïde (la surface physique de la terre).  Il est défini par le rapport entre un ellipsoïde et un point sur la surface topographique établi comme l'origine d'un datum.

La taille et la forme d'un ellipsoïde et la position du centre de l'ellipsoïde par rapport au centre de la terre définissent les datum géodésiques mondiaux.

Des datum divers ont été établis pour convenir aux régions spécifiques. Par exemple, les cartes européennes sont souvent basées sur le datum européen de 1950 (ED-50). Les cartes des Etats-Unis sont souvent basées sur le datum d'Amérique du Nord de 1927 ou de 1983 (NAD-27, NAD-83). Toutes les coordonnées GPS sont basées sur la surface de datum WGS-84.


coordonnées géographiques

La latitude, la longitude, et la hauteur d'ellipsoïde.


géoïde

La surface d'équipotentiel gravitationnel que se rapproche étroitement au niveau moyen de la mer. Elle n'est pas une forme mathématique uniforme, mais une figure irrégulière avec une forme totale semblable à un ellipsoïde.

En général, les élévations des points sont mesurées faisant référence au géoïde. Cependant, les points fixés par des méthodes GPS ont des hauteurs établies dans le datum WGS-84 (une figure mathématique).

Il faut déterminer le rapport entre le datum WGS-84 et le géoïde par observation, car il n'y a aucune définition mathématique simple pouvant décrire le rapport. Vous devez utiliser des méthodes de levé conventionnelles pour observer l'élévation au-dessus du géoïde, puis faire une comparaison des résultats avec la hauteur au-dessus de l'ellipsoïde WGS-84 au même point.

Par la collecte d'un grand nombre d'observations de la séparation entre le géoïde et le datum WGS-84 (la séparation géoïdale), il est possible d'établir les fichiers de grille des valeurs de séparation. Ce permet l'interpolation de la séparation géoïdale aux positions intermédiaires. Les fichiers contenant ces grilles de séparations géoïdales sont connus comme les modèles de géoïde. Avec une position WGS-84 qui se trouve dans les limites d'un modèle de géoïde, le modèle peut retourner la séparation géoïdale interpolée à cette position.


modèle de géoïde

Une représentation mathématique du géoïde d'un zone spécifique, ou de la terre entière. Le logiciel utilise le modèle de géoïde pour générer des séparations de géoïde de vos points dans le réseau.


observation du géoïde

Une séparation de géoïde, avec son erreur associée, extraite d'un modèle de géoïde. L'ajustement de réseau du de Geomatics Office les traite de la même façon que toute observation avec une erreur associée. Pendant que l'ajustement progresse les observations deviendront des observations de géoïde ajustées.


séparation de géoïde

La distance entre l'ellipsoïde et le géoïde à un point spécifique.


géomatique

La conception, la collecte, le stockage, l'analyse, l'affichage et la récupération des informations spatiales. La collecte des informations spatiales peut être d'une variété de sources, y compris les méthodes terrestres et GPS. La géomatique effectue l'intégration du levé traditionnel avec des nouvelles techniques technologiques, avec le résultat que la géomatique sert à un nombre vaste d'applications.


GPS - Global Positioning System (Système de positionnement global)

Le GPS est basé sur une constellation de vingt-quatre (24) satellites qui orbitent la terre à une altitude très élevée.


ligne de base GPS

Une mesure à trois dimensions entre une paire de stations pour lesquelles des données GPS simultanées ont été collectées et traitées avec des techniques de différentiation.

Représentée comme delta X, delta Y, et delta Z; ou azimut, distance, et hauteur delta.


observation GPS

Une ligne de base GPS avec ses erreurs associées. Pendant qu'un ajustement progresse les observations deviennent des observations GPS ajustées.


données brutes GPS

Les données collectées par un récepteur GPS pour un traitement plus tard. Elles peuvent être sous la forme d'un fichier .dat (format de fichier de données brutes Trimble) ou un fichier RINEX.


Heure GPS

Une mesure de l'heure utilisée par le système NAVSTAR GPS. L'heure GPS se base sur le Temps universel coordonné (UTC) mais n'ajoute pas des 'secondes intercalaires' afin de compenser pour les changements dans la période de rotation de la terre.


grille

Un système de coordonnées rectangulaire horizontal à deux dimensions, telle qu'une projection de carte.


conversion de grille

La conversion entre les coordonnées de projection de carte et géographiques.


distance de grille

La distance entre deux points exprimée sous forme de coordonnées de projection de carte.


distance au sol

La distance (distance horizontale avec une courbure appliquée) entre deux points au sol.


HDOP

Horizontal Dilution of Precision

Dilution de précision horizontale


jalon de mesure de hauteur

Un outil de mesure fourni avec une antenne GPS externe et utilisé pour la mesure de la hauteur de l'antenne au-dessus d'un point.


HI

Height of instrument.

Hauteur de l'instrument

Synonyme des hauteurs d'antenne pour GPS.


histogramme

Un affichage graphique de l'ampleur et la distribution des résiduels dans un ajustement de réseau.


point de contrôle horizontal

Un point avec seulement une précision de coordonnées horizontale. L'élévation ou la hauteur d'ellipsoïde est d'un ordre de précision inférieur ou elle est inconnue.


distance horizontale

La distance horizontale entre deux points, calculée horizontalement à partir de l'élévation de l'un ou l'autre point.


position horizontale

Un point avec seulement des coordonnées horizontales.


indépendant

Tout sous-réseau, observation, et point de contrôle qui n'est pas connecté par la géométrie ou les erreurs. Ce terme est l'opposé de corrélé.


contrainte interne

Un ajustement de réseau calculé sans fixer aucune coordonnée de point. Trimble Geomatics Office utilise le centroïde du réseau comme une contrainte interne.


ambiguïté de nombre entier

Le nombre entier de cycles dans une pseudodistance de phase porteuse entre le satellite GPS et le récepteur GPS.


recherche de nombre entier

Le traitement de lignes de base GPS, soit en temps réel soit post traité, nécessite des solutions de nombre entier fixes pour obtenir les meilleurs résultats possibles. Le logiciel traitant les mesures GPS utilisées pour dériver une ligne de base effectue une recherche de nombre entier afin d'obtenir une solution de nombre entier fixe. La recherche entraîne la mise à l'essai de diverses combinaisons de valeurs de nombre entier et la sélection des meilleurs résultats.


iono free

ionospheric free solution (IonoFree)

iono libre

solution iono libre (Iono libre)

Une solution utilisant une combinaison de mesures GPS pour modeler et supprimer les effets de l'ionosphère sur les signaux GPS. Cette solution est utilisée souvent pour des levés de contrôle d'ordre supérieur, en particulière lors de l'observation des lignes de base longues.


ionosphère

La bande de particules chargées que se trouve de 80 à 120 milles au-dessus de la surface de la terre. Elle a d'influence sur la précision des mesures GPS lors de la mesure des lignes de base longues avec des récepteurs à une fréquence.


modélisation ionosphérique

Le délai à cause de l'ionosphère varie relativement à la fréquence des signaux GPS et a d'influence différente sur les signaux L1 et L2. Lorsque les récepteurs à bi-fréquence sont utilisés les observation de phase porteuse pour les deux fréquences peuvent être utilisées pour modeler et éliminer la plupart des effets ionosphériques. Lorsque les mesures à bi-fréquence ne sont pas disponibles un modèle ionosphérique émis par les satellites GPS peut être utilisé pour réduire les effets ionosphériques.  Cependant, l'utilisation du modèle émis n'est pas aussi efficace que l'utilisation des mesures à bi-fréquence.


itération

Un groupe complet de calculs d'ajustement y compris la formation des équations d'observation, les ajustements de coordonnées et le calcul des résiduels.


levé cinématique

Une méthode de levé GPS utilisant des occupations d'arrêt et début courtes, pendant la capture en continu d'au moins 4 satellites. Il est possible de le faire en temps réel ou post traité aux précisions centimétriques.


initialisation de point connu

L'initialisation de point connu est utilisée conjointement avec l'initialisation cinématique. Lorsque deux points connus sont disponibles, le processeur de lignes de base peut calculer un inverse entre les deux points et dériver un vecteur d'initialisation. Ce vecteur d'initialisation, avec des composants de ligne de base connus, est utilisé dans la solution de l'ambiguïté de nombre entier. Si le processeur peut résoudre cette ambiguïté avec succès une solution de nombre entier fixe sera possible, fournissant des meilleures solutions pour un levé cinématique.


L1

La porteuse de bande L primaire utilisée par les satellites GPS pour transmettre les données de satellites. Sa fréquence est 1575.42 MHz. Elle est modulée par le code C/A, le code P et un message de navigation.


L2

La porteuse de bande L secondaire utilisée par les satellites GPS pour transmettre les données de satellites. Sa fréquence est 1227.6 MHz. Elle est modulée par le code P et un message de navigation.


étiquette

Des informations que vous pouvez assigner aux points dans le projet. Ils s'affichent à côté des points ainsi vous assistant à les trouver facilement. Les étiquettes sont visibles dans Vue de levé et Vue en plan.

Vous pouvez utiliser la définition d’étiquette prédéfinie de Trimble Geomatics Office pour étiqueter les points (par exemple, avec leurs noms, leurs codes de caractéristiques et leurs élévations).

Les étiquettes utilisent la même police et la même taille que les Info-bulles dans vos paramètres d'Apparence Microsoft Windows.

Pour assigner les étiquettes aux points dans la base de données, utilisez Affichage / Etiquettes de point.


couches

Un endroit dans lequel les données groupées ensemble peuvent être stockées.

L'organisation des données dans les couches de cette façon rend leur gestion plus facile. Vous pouvez avoir n'importe combien de couches dans un projet. Pour assigner ou réassigner une entité à une couche, utilisez la fenêtre Propriétés. Un nom de couche peut consister en jusqu'à 100 caractères alphanumériques.


moindres carrés

Une méthode mathématique pour l'ajustement des observations, basée sur la théorie de probabilité. Dans cette méthode d'ajustement, le total des carrés de tous les résiduels pondérés est minimisé.


niveau de confiance

Une mesure de la confiance dans nos résultats, exprimée sous forme d'un pourcentage ou un sigma.


niveau d'importance

Une expression de probabilité. On dit qu'une erreur avec un sigma (standard) a un niveau d'importance de 68 pour-cent. Pour les erreurs à une dimension, un niveau d'importance de 95 pour-cent est exprimé par 1.96 sigma, et un niveau d'importance de pour-cent est exprimé par 2.576 sigma.


ellipsoïde local

L'ellipsoïde spécifié par un système de coordonnées. En premier les coordonnées WGS-84 sont transformées sur cet ellipsoïde, puis converties en coordonnées de grille.


coordonnées géodésiques locales

La latitude, la longitude et la hauteur d'un point. Les coordonnées sont exprimées en fonction de l'ellipsoïde local.


horizon géodésique local

A tout point, un plan à la hauteur d'ellipsoïde d'un point donné qui est parallèle au plan tangente à l'ellipsoïde à ce point-là. Les valeurs de coordonnées pour l'horizon géodésique local sont exprimées sous forme de Est, Nord et Haut. Le HGL est utilisé afin de faire tourner les différences de coordonnées cartésiennes EC (centré sur la terre), avant la modélisation d'une ligne de base sur l'ellipsoïde. Comme une partie de la modélisation sur l'ellipsoïde, les valeurs d'azimut calculées à partir des composants HGL doivent être corrigées pour tout normal oblique.


fermeture de boucle

Les fermetures de boucle fournissent une indication du degré d'erreur dans une groupe d'observations dans un réseau.

Une fermeture de boucle est calculée par la sélection d'un point à partir duquel une ou plusieurs observations ont été prises, l'ajout d'une de ces observations aux coordonnées du point, et le calcul des coordonnées du deuxième point basées sur cette observation. Le procédé est répété une ou plusieurs fois autour d'une boucle, jusqu'à ce qu'on arrive au point de début d'origine. Lorsqu'il n'y avait aucune erreur dans les observations, la coordonnée finale calculée serait exactement la même que la coordonnée de début d'origine. Par la soustraction de la coordonnée calculée de la coordonnée d'origine une fermeture est déterminée. La division de cette erreur par la longueur de la ligne permet l'expression de l'erreur sous forme de parties par million.

Il est également possible d'utiliser cette technique entre deux points différents lorsque les deux points sont connus avec un haut degré de précision. C'est également connu comme une fermeture transversale.


grand axe

Voir ellipsoïde.


angle cartographique

L'angle entre la grille nord sur une projection cartographique et le méridien de longitude à un point déterminé. Egalement connu comme une convergence.


projection cartographique

Une expression mathématique rigoureuse de la surface courbée de l'ellipsoïde sur une grille de coordonnées rectangulaire.


niveau moyen de la mer

La hauteur moyenne de la surface de l'océan pour toutes les étapes de la marée. Utilisé comme un référence pour les élévations.


contraint au minimum

Un ajustement de réseau dans lequel seulement des contraintes suffisantes pour déterminer le système de coordonnées sont utilisées. Utilisé pour la mesure de l'uniformité interne des observations.


petit axe

Voir ellipsoïde.


modélisation

L'expression d'une observation et ses erreurs relatives mathématiquement et géométriquement sur un système de coordonnées défini, tel qu'un ellipsoïde.


multitrajet

L'interférence (semblable aux 'images secondaires' sur la télévision) que se produit lorsque les signaux GPS arrivent à une antenne après avoir parcouru des trajets divers. Le signal qui parcourt le trajet plus long fournit une estimation de pseudodistance plus grande et augmente l'erreur. Les multitrajets peuvent se produire des réflexions des structures près de l'antenne.


voie étroite

Une combinaison linéaire des observation de phase porteuse L1 et L2 (L1 + L2) qui sert à neutraliser les effets ionosphériques dans les données de lignes de base collectées. La longueur d'onde réelle de la voie étroite est de 10,7 centimètres.


NAVDATA

NAVDATA est le message de navigation à 1500 bit émit par chaque satellite. Ce message contient l'heure du système, les paramètres de correction d'horloge, les paramètres du modèle de retard ionosphérique, et les détails de la santé et l'éphéméride du satellite. Les informations sont utilisées pour le traitement des signaux GPS afin d'obtenir la vitesse et la position de l'utilisateur.


réseau

Un groupe de lignes de base. Voir aussi sous-réseau.


ajustement de réseau

La solution des équations simultanées conçue afin d'obtenir une fermeture dans un réseau de levé en minimisant la somme des carrés pondérés des résiduels des observations.

La technique d'ajustement utilisé par Trimble Geomatics Office parfois s'appelle la variation des coordonnées, et parfois la méthode d'observation indirectes.


état de réseau

Une indication qu'une observation spécifique sera inclue dans l'ajustement

 Réseau signifie qu'elle est inclue dans l'ajustement

 Non-réseau signifie qu'elle est exclue de l'ajustement


NMEA

National Marine Electronics Association

Le standard NMEA 0183 définit l'interface pour les périphériques de navigation électroniques maritimes. Ce standard  définit un nombre de 'chaînes'  connues comme les chaînes NMEA contenant des détails de navigation telle que les positions.

La plupart des récepteurs GPS de Trimble peuvent sortir les positions sous forme des chaînes NMEA.


normal

Dans la géodésie, la ligne droite perpendiculaire à la surface de l'ellipsoïde.


courbe de distribution normale

Une illustration graphique des variables aléatoires autour d'une valeur attendue selon la théorie de probabilité. Utilisée avec les histogrammes.


azimut de section normale

L'angle mesuré du méridien géodésique de l'observateur au point observé sur l'ellipsoïde


nord

La lecture vers le nord d'une valeur de grille.


résiduel d'observation

La correction appliquée à une observation, comme déterminée par l'ajustement.


observations

Voir observations de levé.


temps d'occupation

Le temps requis sur une station, ou un point, pour réaliser le traitement d'une ligne de base GPS avec succès. Le temps variera selon la technique de levé, le type de récepteur GPS utilisé et la précision requise pour les résultats définitifs. Les temps d'occupation peuvent varier entre deux secondes (levés cinématiques) et plusieurs heures (levés de contrôle ou de déformation nécessitant les plus hauts niveaux de précision et de répétition).


origine

L'intersection des axes dans un système de coordonnées. Le point de commencement.


hauteur orthométrique

La distance entre un point et la surface du géoïde. En général appelée l'élévation.


méthode de recherche OTF (en vol)

Le traitement des lignes de base GPS, soit en temps réel soit post traité, requiert des solutions de nombre entière pour les meilleurs résultats possibles. (Voir recherche de nombre entier.)

Dans le passé, on a effectué cette recherche avec des mesures collectées lorsque deux ou plusieurs récepteurs étaient immobiles sur leurs points respectifs. Les récepteurs et les logiciels modernes peuvent utiliser les mesures collectées lorsque le récepteur mobile est en mouvement. Du fait que le récepteur est un mouvement, les données sont décrites comme collectées 'en vol' (OTF) et la recherche de nombre entier utilisant ces données est une 'recherche en vol' (OTF).


valeur aberrante

Une observation identifiée par l'analyse statistique comme ayant un résiduel trop grand pour son erreur estimée. Le terme se dérive de la position graphique d'une observation dans un histogramme.


sur-déterminé

Un réseau pour lequel plus de mesures que nécessaires ont été effectuées afin de calculer les coordonnées du réseau. Lié à la redondance.


Code P

Le code 'précis' transmis par les satellites GPS. Chaque satellite possède un code unique qui est modulé sur les ondes porteuses L1 et L2. Le code P est remplacé d'un code Y lorsque l'anti-déception électronique est active.


paramètre

Une variable indépendante dans laquelle les coordonnées des points sur une ligne ou une surface sont exprimées. Voir inconnus.


parité

Un type de vérification d'erreur utilisé dans le stockage et le transfert des données numériques binaires. Les options pour la vérification de parité comprennent Pair, Impair ou Aucun.


PDOP

Position Dilution of Precision

Dilution de précision de position

Un facteur de mérite sans unité qui exprime le rapport entre l'erreur dans la position de l'utilisateur et l'erreur dans la position du satellite. Géométriquement, PDOP est proportionnelle à 1 divisé par le volume de la pyramide constituée des lignes passant du récepteur à quatre satellites observés. Les valeurs considérées comme "bonnes" pour le positionnement sont petites, par exemple 3. Les valeurs supérieures à 7 sont considérées médiocres. Ainsi, une petite PDOP est associée aux satellites largement séparés.

PDOP est liée à la DOP horizontale et verticale par:


seuil de PDOP

Un paramètre du récepteur spécifiant une valeur PDOP maximum pour le positionnement. Lorsque l'orientation géométrique des satellites fournit une PDOP supérieure à la valeur du masque, le récepteur cesse de calculer les relevés de position.


masque PDOP

La valeur PDOP la plus élevée à laquelle un récepteur effectuera le calcul des positions.


centre de phase

Voir correction de phase d'antenne.


modèles de centre de phase

Un modèle utilisé pour appliquer une correction à un signal GPS basé sur un type d'antenne spécifique. La correction se base sur l'élévation du satellite au-dessus de l'horizon et  modèle les variations électriques dans l'emplacement du centre de phase d'antenne. Ces modèles servent à éliminer les erreurs introduites lors de l'utilisation des antennes non-identiques aux points de base et mobile. Voir aussi correction de phase d'antenne.


plomb

L'action d'aligner l'antenne ou l'instrument le long d'une ligne verticale (verticale du centre de perspective) perpendiculaire à la surface équipotentielle du champ de gravité de la terre.


positions de point

Voir positionnement autonome.


post traiter

Traiter les données de satellite sur un ordinateur après les avoir collecté.


PPM

Parts per million

Parties par million

Une représentation standardisée d'une erreur d'échelle dans les mesures de distances. Une erreur  de 1 PPM produirait 1 millimètre d'erreur de mesure pour chaque 1000 mètres de distance parcourus.


éphéméride précise

Voir éphéméride.


précision

Une mesure indiquant avec quel degré de proximité les variables aléatoires tendent de se grouper autour d'une valeur calculée. Une précision haute signifie des petits résidus. En général exprimée comme une partie dans, ou alternativement, comme parties par million.


PRN

Pseudorandom number

nombre pseudo-aléatoire

Une séquence de 1's et 0's numériques qui semblent d'être répartis d'une manière aléatoire comme le bruit, mais qui peuvent être reproduits exactement. Les codes PRN ont une valeur d'autocorrélation basse pour tout délai ou retard sauf quand ils coïncident exactement.

Chaque satellite NAVSTAR peut être identifié par ses codes de bruit pseudo-aléatoire P et C/A uniques, donc quelques fois, le terme 'PRN'  est utilisé comme un autre nom pour un satellite ou SV GPS.


probabilité

Un pourcentage statistique qui exprime quelle partie d'un nombre d'observations hypothétique se trouvera dans les limites définies. Connue quelques fois comme le niveau d'importance.


valeur probable

La valeur ajustée des observations et d'autres quantités, supposant que l'ajustement a été effectué correctement. L'approximation la plus près possible à la valeur réelle.


projet

Trimble Geomatics Office fonctionne sur les données contenues dans des projets. Vous pouvez considérer un projet comme l'espace de travail dans lequel vous travaillez. Les nouveaux projets sont toujours créés à partir des modèles existants et ils héritent tous les éléments du modèle. A partir de ce point, tout nouveau travail n'est enregistré que dans le nouveau projet.

Un projet contient toutes les observations brutes, les points calculés, la définition du système de coordonnées, les lignes, le texte et les styles CAD.


datum du projet

Le datum associé à un projet dans Trimble Geomatics Office. Toutes les coordonnées locales sont affichées utilisant le datum du projet.


projection

Utilisée pour créer des cartes plates représentant la surface de la terre ou des parties de cette surface.


erreur propagée

Les erreurs calculées dérivées des erreurs d'observation estimées et exprimées en fonction des positions de coordonnées. Les erreurs de coordonnées propagées peuvent, à leur tour, être propagées dans des erreurs relatives d'azimut, de distance, et de hauteur delta entre des points.


test d'Acceptance de qualité

Un ou plusieurs tests d'évaluation effectués à l'aide du logiciel sur les données de mesures GPS brutes pour déterminer si les données sont acceptées ou rejetées par un groupe de valeurs de tolérance définies de l'utilisateur. Ces tests suppriment les données de tout traitement ultérieur ou marquent les données ayant besoin des améliorations de qualité.


enregistrements QC

Quality Control records

Enregistrements de contrôle de qualité

Les enregistrements QC contiennent des informations concernant la qualité de la position GPS mesurée. Ils sont stockés avec l'enregistrement de point.


ratio

Lors de l'initialisation, le récepteur détermine le nombre entier des longueurs d'onde pour chaque satellite. Pour un groupe spécifique de nombres entiers, il calcule la probabilité que ce soit le groupe correct.

Ratio est le ratio de la probabilité de l'exactitude du meilleur groupe de nombres entiers courant à la probabilité de l'exactitude du meilleur groupe après. Ainsi, un ratio élevé indique que le meilleur groupe de nombres entiers est beaucoup meilleur que tout autre groupe. Donc, nous avons la confiance que c'est correct. Le ratio doit être supérieur à 5 pour des initialisations en vol (OTF) et de nouveau point.


RDOP

Relative Dilution of Precision

Dilution de précision relative


cinématique en temps réel

Une méthode de levé GPS en temps réel utilisant une occupation (arrêt et marche) courte, lors du maintien de la capture d'au moins 4 satellites.  Cette méthode nécessite une liaison de données sans fil entre les récepteurs de base et mobile.


coordonnées rectangulaires

Les coordonnées de tout système dont les axes de référence s'intersectent à angle droit.


profil à colonne réduite

Une version abrégée des équations normales dans laquelle les équations sont réorganisées afin de minimiser la mémoire ordinateur requise pour le stockage de tous les éléments non zéro.


redondance

Le montant par lequel un réseau de contrôle est surdéterminé, ou possède plus d'observations que nécessaires pour calculer ses parties exactement.


numéro de redondance

Une mesure des degrés de liberté dans une partie, plutôt que la totalité, d'un réseau de contrôle.


lignes de base redondantes

Une ligne de base observée à un point déjà connecté au réseau par d'autres observations. Une ligne de base redondante peut être une réobservation indépendante d'une mesure précédente, ou une observation à un point à partir d'une autre base. Elle est redondante parce qu'elle fournit plus d'informations que nécessaires pour déterminer uniquement un point. Néanmoins, les observations redondantes sont très utiles car elles fournissent une vérification de la qualité des mesures précédentes.


observation redondante

Une observation répétée, ou une observation qui contribue à la surdétermination  d'un réseau.


facteur de référence

Voir erreur standard de poids unitaire.


système de référence

Le système de coordonnées d'un datum.


station de référence

C'est une station de base.


variance de référence

Le carré du facteur de référence.


erreurs relatives

Les erreurs et les précisions exprimées pour et entre les paires de points de contrôle ajustés de réseau.


résiduel

La correction, ou l'ajustement d'une observation afin d'obtenir une fermeture totale d'un réseau de contrôle. En outre, toute différence entre une quantité observée et une valeur calculée pour cette quantité.


RINEX

Receiver INdependent EXchange format

Format  indépendant de récepteur

Un format de fichier de données brutes GPS utilisé pour échanger des fichiers multiples de fabricants tiers.


RMS

Root Mean Square

Valeur moyenne quadratique

RMS exprime la précision de la mesure du point. C'est le rayon du cercle d'erreur dans lequel se trouvent approximativement 70% des fixes de positon. Il peut être exprimé sous forme d'unités de distance ou de cycles de longueur d'onde.


méridien tourné

Une zone constante pour la projection de carte Mercator oblique.


rotation

Dans les transformations, un angle à travers duquel un axe de coordonnée est déplacé autour de l'origine du système de coordonnées.


mobile

Tout récepteur GPS mobile et tout ordinateur de terrain effectuant la collecte de données sur le terrain. La position d'un récepteur mobile peut être corrigée différentiellement par rapport au récepteur GPS de base immobile.


RTCM

Radio Technical Commission for Maritime Services

Une Commission établie pour définir une liaison de données différentielle pour la correction différentielle en temps réel des récepteurs GPS mobiles. Il y a deux types de messages de correction différentielle RTCM, mais tous les récepteurs GPS de Trimble utilisent le protocole RTCM Type 2 le plus récent.


RTK

Real-time kinematic

Cinématique en temps réel

Un type de levé GPS.


géométrie des satellites

La position et le mouvement des satellites GPS lors d'un levé GPS.


scalaire

Dans les moindres carrés, une valeur appliquée aux variances (erreurs) basée sur le niveau de confiance requis.


pondération scalaire

Un procédé d'application d'un scalaire aux erreurs estimées pour obtenir une pondération correcte de l'observation. Les trois types de scalaire disponibles dans un ajustement de réseau dans Trimble Geomatics Office sont:

 Par défaut signifie que le scalaire est fixé sur 1.00, l'erreur initiale estimée reste la même.

 Alternatif signifie que le scalaire est fixé sur le Facteur de référence de l'ajustement précédent.

 Défini de l'utilisateur signifie que vous pouvez saisir une valeur pour le scalaire

Le scalaire s'applique aux erreurs d'observation utilisant l'une des méthodes suivantes:

 Toutes les observations

 Chaque observation

 Groupes de composants de variance


échelle

Un multiplicateur utilisé sur des variables de coordonnée et d'autres variables linéaires, telles que les projections de carte et les transformations.


SDMS

Survey Data Management System

Système de gestion de données de levé

Un groupe de définitions de format pour le stockage des données de levé.  AASHTO maintient ce système.


Accès sélectif (SA)

La dégradation artificielle du signal de satellite GPS par le Département de la Défense des Etats-Unis. L'erreur dans la position à cause de SA peut être jusqu'à 100 mètres.


demi-grand axe

Une moitié du grand axe.


demi-petit axe

Une moitié du petit axe.


session

Une période pendant laquelle un ou plusieurs récepteurs GPS enregistrent les données de mesure du satellite.


erreurs d'installation

Les erreurs du centrage de tribraque ou de la hauteur d'instrument à un point de contrôle.


prise de vue isolée

Une ligne de base observée sans aucune redondance.


sigma

Un symbole ou un terme mathématique pour l'erreur standard.


mono-fréquence

Un type de récepteur qui n'utilise que le signal GPS L1. Il n'y a aucune compensation des effets ionosphériques.


calibration du site

Un procédé pour le calcul des paramètres qui établissent le rapport entre les positions WGS-84 (latitude, longitude et hauteur d'ellipsoïde) déterminées par les observations GPS et  les coordonnées connues locales définies par une projection de carte et les élévations au-dessus du niveau moyen de la mer. Les paramètres sont utilisés pour générer des coordonnées de grille locale à partir de WGS-84 (et vice versa) en temps réel sur le terrain lors de l'utilisation des méthodes de levé RTK.


Plan du ciel

Un plan polaire affichant les trajets des satellites visibles pendant l'intervalle de temps sélectionné pour le graphique. L'élévation du satellite est représentée dans la dimension radiale et l'azimut est montré dans la dimension angulaire. Le résultat représente le trajet du satellite comme il apparaît à un observateur regardant du haut depuis une position directement au-dessus du point de levé.


types de solution

Une description des données et des techniques utilisées pour obtenir les solutions de lignes de base à partir des mesures GPS. Les types de solution typiques comprennent des descriptions tels que code, flottant et fixe. Ceux-ci décrivent les techniques utilisées par le processeur de lignes de base afin d'obtenir une solution de lignes de base. Les types de solution peuvent également comprendre des descriptions tels que L1, L2, grande voie, voie étroite, ou libre d'iono. Elles décrivent la façon dans laquelle les mesures GPS sont combinées pour arriver aux résultats spécifiques. Pour plus amples informations, référez-vous aux références au sujet du traitement GPS pour une discussion plus profonde concernant ces termes et techniques.


distance inclinée

La distance dans le plan parallèle à la différence verticale (inclinaison) entre les points.


SNR

Signal-to-Noise Ratio

Rapport signal à bruit

Une mesure de l'intensité d'un signal de satellite. SNR s'étend de 0 (aucun signal) jusqu'à environ 35.


erreur standard

Une estimation statistique d'erreur, selon laquelle 68 pour-cent d'un nombre infini d'observations auront théoriquement des erreurs absolues inférieures ou égales à cette valeur.


erreur standard de poids unitaire

Une mesure de l'ampleur des résiduels d'observation dans un réseau ajusté par rapport  aux erreurs d'observation estimées avant l'ajustement.


State Plane Coordinates

Les définitions spéciales des projections cartographiques  Transverse Mercator et Lambert conforme adoptées par statut dans les Etats-Unis. Il y a un groupe de telles zones pour NAD-27 et un autre pour NAD-83.


statique (levé)

Une méthode de levé GPS utilisant des occupations longues (quelques fois de plusieurs heures) pour la collecte des données brutes GPS, puis le post traitement afin d'obtenir des précisions inférieures à un centimètre.


réseau statique

Le réseau statique décrive la géométrie et l'ordre dont les lignes de base GPS collectées au moyen des techniques statique et statique rapide sont organisées et traitées. D'abord le processeur de lignes de base examine le projet afin de trouver les points ayant les coordonnées de la plus haute qualité, et puis construit le réseau de traitement de ces points. Le résultat est un groupe de lignes de base statiques qui sont dérivées en utilisant des coordonnées initiales précises.


état

Chaque observation et chaque groupe de coordonnées entré pour un point a un champ d'état (disponible dans la page Résumé de la fenêtre Propriétés).

L'état peut être Activé, Activé comme vérification, ou Désactivé:

  Les observations et les coordonnées activées sont toujours utilisées par le procédé de recalcul dans la détermination de la position calculée pour le point.

  Les observations et les coordonnées  activées comme vérification ne sont utilisées que lorsqu'il n'y a aucune Activée.

  Les observations et les coordonnées désactivées ne sont jamais utilisées


modèle stochastique

Une référence générale aux techniques utilisées pour l'estimation des erreurs dans un ajustement de réseau.


sous-réseau

Un groupe de lignes de base connectées par des points de contrôle communs, et indépendantes (séparées) de toute autre ligne de base. Comme utilisé dans Trimble Geomatics Office, un réseau peut consister en un ou plusieurs sous-réseaux, et tout réseau peut consister en seulement une ligne de base avec deux points de contrôle.


Super-trak

Une méthode propriétaire de Trimble pour le traitement du signal L2 lorsque le code P est encrypté.


observations de levé

Les mesures effectuées à ou entre des points de contrôle en utilisant des appareils de levé, y compris les récepteurs GPS et l'équipement conventionnel.


SV

Satellite Vehicle (or Space Vehicle)

Véhicule de satellite  (ou Véhicule spatial)


types de ligne et symboles

Les types de ligne et symboles sont maintenus en utilisant les utilitaires Symbol Editor (Editeur de symboles) et Line Type Editor (Editeur de types de ligne) de Trimble. Utilisez ces éditeurs pour la création des nouveaux symboles et types de ligne ainsi que l'édition de ceux existants. Les symboles sont stockés dans des bibliothèques de symboles et les types de ligne dans des bibliothèques de types de ligne.

Trimble Geomatics Office utilise les bibliothèques de symboles et types de ligne courantes. Lors de l'enregistrement d'une bibliothèque éditée vous pouvez la spécifier comme la bibliothèque du système (si elle n'est pas déjà la bibliothèque du système courante).


erreurs systématiques

Une erreur que se produit avec le même signe, et souvent la même ampleur, dans un nombre d'observations liées.


tau (valeur)

Une valeur calculée d'une distribution de fréquence interne basée sur le nombre d'observations, les degrés de liberté et un pourcentage (95%) donné de probabilité. Cette valeur est utilisée pour déterminer si une observation ne s'adapte pas aux autres dans l'ajustement. Lorsqu'un résidu d'observations dépasse le tau, il est indiqué comme une observation aberrante. Connues comme les lignes tau dans le histogramme des résidus standardisés, les lignes verticales à droite et à gauche de la ligne verticale centrale.


critère tau

La technique statistique d'Allen Pope pour la détection des observations aberrantes.  Pour plus amples informations, référez-vous à Pope (1976).


TDOP

Time Dilution of Precision

Dilution de précision du temps


observation terrestre

Une observation terrestre est une observation sur le terrain utilisant un télémètre à laser ou un instrument conventionnel.


TOW

Time of Week

Heure de la semaine

TOW en secondes, depuis minuit du soir samedi/matin dimanche selon l'heure GPS.


poursuite

Le procédé de recevoir et de reconnaître les signaux d'un satellite.


transformation

La rotation, le déplacement et la mise à l'échelle d'un réseau afin de le déplacer d'un système de cordonnées à un autre.


groupe de paramètres de transformation

Un groupe d'observations sélectionné utilisé pour calculer des paramètres de transformation uniques à ce groupe d'observations. En général, les observations dans le groupe sont du même type ayant des erreurs semblables et mesurées utilisant une méthode commune.


paramètres de transformation

Un groupe de paramètres dérivés pour un ajustement de réseau ou définis de l'utilisateur, qui transforment un datum à un autre. En général avec le GPS, les paramètres sont générés pour transformer WGS-84 au datum local.


tribraque

Un dispositif de centrage utilisé pour le montage des antennes GPS et d'autres instruments de levé sur les trépieds de levé.


erreurs de centrage de tribraque

Les erreurs associées au centrage (plomb) du tribraque sur un point observé. Ces erreurs sont estimées. L'estimation est basée sur l'évaluation de la qualité des méthodes de levé et devrait être modeste.


correction tropo

tropospheric correction

correction troposphérique

La correction appliquée à une mesure de satellite afin d'effectuer une correction pour le délai troposphérique.


modèle tropo

tropospheric model

modèle troposphèrique

Les signaux GPS sont retardés par le troposphère.  L'amplitude du délai variera selon la température, l'humidité, la pression, la hauteur de la station au-dessus du niveau de la mer et l'élévation des satellites GPS au-dessus de l'horizon. Les corrections aux mesures de code et de phase peuvent être effectuées à l'aide d'un modèle tropo pour représenter des délais.


univariable

Une fonction mathématique décrivant le comportement des erreurs aléatoires dans une dimension, dans:

  l'angle

  la distance

  la différence de hauteur

  l'élévation

  la hauteur d'ellipsoïde


URA

User Range Accuracy

précision en distance de l'utilisateur

Une mesure des erreurs que peuvent s'introduire à cause des problèmes de satellite et l'accès sélectif (SA) lorsqu'un véhicule de satellite (SV) spécifique est utilisé. Une URA de 32 mètres indique que SA est activé. La valeur URA est réglée par le Segment de contrôle et émise par les satellites.


inconnus

Les ajustements calculés aux coordonnées et aux paramètres de transformation. Egalement utilisés pour calculer les résidus d'observations.


US National

L'agence du gouvernement des Etats-Unis qui maintient le datum géodésique national et tous les réseaux de contrôle de levé géodésique dans les Etats-Unis et ses territoires.


Pied américain

1200/3937 mètre. L'unité officielle de mesure linéaire pour NAD-27.


UTC

Universal Time Coordinated

temps universel coordonné;

Un standard de temps basé sur le temps moyen local au méridien de Greenwich. Voir aussi Heure GPS.


variance

Le carré de l'erreur standard.


estimation de composant de variance

Une technique des moindres carrés pour l'estimation de l'erreur relative des parties différentes d'un réseau.


groupe de composants de variance

L'un des groupes d'observations pour lequel l'estimation de composant de variance est utilisée dans un ajustement de réseau.


matrice de variance-covariance

Un groupe de chiffres exprimant les variances et les covariances dans un groupe d'observations.


VDOP

Vertical Dilution of Precision

Dilution de précision verticale


vecteur

Une ligne en trois dimensions entre deux points.


vertical

Semblable au normal, à l'exception qu'il est calculé à partir du plan tangente au géoïde au lieu de l'ellipsoïde.


ajustement vertical

Un ajustement de réseau des observations et coordonnées  verticales seulement.


point de contrôle vertical

Un point n'ayant qu'une précision de coordonnée verticale. La position horizontale est d'un ordre de précision inférieur ou est inconnue.


WAVE

Weighted Ambiguity Vector Estimator

Estimateur de vecteur d'ambiguïté pondérée

WAVE est le processeur de lignes de base de Trimble. Il calcule les vecteurs GPS à partir de observations sur le terrain effectuées utilisant les procédures de collecte de données cinématique, statique rapide ou statique.


poids

L'inverse de la variance d'une observation.


stratégie de pondération

La collection des valeurs utilisées pour augmenter les matrices de variance-covariance dans Trimble Geomatics Office.


poids

Le groupe de poids, ou l'inverse de la matrice de variance-covariance des observation corrélées.


WGS-84

World Geodetic System (1984)

Système géodésique mondial (1984)

L'ellipsoïde mathématique utilisé par GPS depuis janvier 1987.


grande voie

Une combinaison linéaire des observations de phase porteuse L1 et L2 (L1 − L2). Ce sert à cause de sa longueur d'onde effective (86,2 cm) et à la recherche des ambiguïtés de nombre entier sur les lignes de base longues.


X, Y et Z

Dans le système cartésien centré à la terre, X fait référence à la direction de l'axe de coordonnée passant de l'origine du système au méridien Greenwich, Y à l'axe passant de l'origine à travers du méridien de longitude est 90°, et Z à la calotte glaciaire polaire. Dans le systèmes de coordonnés rectangulaires, X fait référence à l'axe est-ouest, Y à l'axe nord-sud, et Z à l'axe de hauteur.


code Y

Le code Y est une forme encryptée des informations contenues dans le code P. Les satellites transmettent le code Y au lieu du code P lorsque l'effet de l'Anti-déception est en cours.


délai du zénith

Le délai, à cause du troposphère, d'un signal GPS observé depuis un satellite directement au-dessus. Pendant que le satellite approche vers l'horizon, le trajet du signal à travers du troposphère devient plus long et le délai augmente.


Fichiers d'éphéméride précise GPS

Un fichier d'éphéméride précise contient les données de position précises ainsi que les corrections d'horloge de chaque satellite pour une période de temps spécifiée, en général un jour. Ces informations sont utilisées par le processeur de lignes de base pour améliorer la précision de lignes de base en réduisant les effets des erreurs d'éphéméride émises. En plus, l'utilisation d'une éphéméride précise aide le processeur de lignes de base  à obtenir des solutions fixes sur les lignes de base longues.

Trimble Geomatics Office prend en charge l'utilisation des éphémérides précises en deux formats de fichier:

  .sp3 − Un format ASCII défini par le U.S. National Geodetic Survey (Service de Géodésie national des Etats-Unis). C'est le format le plus disponible.

  .ef18 − Un fichier binaire avec des informations semblables à celles contenues dans les fichiers .sp3. Les .ef18 requièrent moins d'espace disque et peuvent être utilisés de façon plus efficace par le processeur de lignes de base.

L'utilisation des éphémérides précises est contrôlé par l'option Ephéméride dans le dialogue Style de traitement de lignes de base GPS actif. Les fichiers d'éphéméride précise doivent être importés dans chaque projet dans lequel ils seront utilisés. Utilisez Fichier / Importer pour charger un ou plusieurs fichiers d'éphéméride précise dans un projet.

A l'origine, les fichiers d'éphéméride précise ont fourni des positions de satellite et les décalages d'horloge pour une semaine. Cependant, les fichiers courants, ne fournissent ces informations que pour un seul jour. Ce signifie que les fichiers peuvent devenir  disponibles dans une période de temps plus courte.

Pour le processeur de lignes de base, il faut que la durée de l'éphéméride précise dépasse la durée des mesures GPS dont le traitement est en cours par un minimum de huit époques SV, avant et après le début des données. Ces information d'éphéméride précise supplémentaires sont requises afin d'assurer l'interpolation correcte au début et à la fin du levé.

L'utilitaire d'importation d'éphéméride précise effectue le suivant:

  Vérifie le format du fichier. Si le fichier est dans le format .sp3 il sera converti au format .ef18 automatiquement lors de l'importation. Si le fichier est dans le format .ef18, il est utilisé directement.

  Détermine s'il y a des fichiers d'éphéméride précise existants déjà chargés dans le projet. Si c'est le cas, il vérifie la couverture des fichiers afin de déterminer si le nouveau fichier couvre la même durée que la couverture existante. Si c'est le cas, le nouveau fichier écrasera le fichier ancien et un message d'avertissement est fourni dans le rapport d'importation pour indiquer cet événement.

  Copie le fichier au dossier Ephéméride précise du projet.

Les fichiers d'éphéméride précise sont disponible à partir d'une variété de sources, y compris:

  U.S. National Geodetic Survey (Service de Géodésie national des Etats-Unis)

  Center for Orbit Determination in Europe (CODE) (Centre pour le détermination d'orbite en Europe)

  European Space Agency (Agence spatiale européenne)

  Scripps Orbit and Permanent Array Center (SOPAC)

  Canadian Department of Energy, Mines and Resources (EMR) (Service d'énergie, des mines et des ressources canadien)

  NASA Jet Propulsion Laboratory (JPL) (Laboratoire de propulsion à réaction NASA)


Pour importer un fichier d'éphéméride précise à votre ordinateur (.sp3, .e18)

1.	Sélectionnez Fichier / Importer. Le dialogue Importer apparaît.

2.	Dans la tabulation Levé, sélectionnez l'option Fichiers d'éphéméride précise (*.sp3, *.e18).

3.	Cliquez sur OK. Le dialogue Ouvrir apparaît.

4.	Trouvez les fichiers .sp3 ou .e18 que vous voulez importer.

5.	Cliquez sur Ouvrir. Les fichiers sont chargés dans le dossier d'Ephéméride précise du projet.

NOTE: Si les fichiers d'éphéméride précise importés couvrent la même période de temps que les fichiers d'éphéméride précise existants dans le projet déjà, un message s'affiche. Ce message indique qu'un ou plusieurs messages ont été générés lors de l'importation. Pour afficher les détails de ces fichiers, cliquez sur Oui dans ce message.

NOTE: L'utilisation de l'éphéméride précise est contrôlée par l'option Ephéméride dans Styles de traitement de lignes de base GPS actifs.


Vue d'ensemble d'enregistrement

Si vous enregistrez votre logiciel Trimble Geomatics Office et fournissez vos détails de contact, Trimble peut vous envoyer toute nouvelle information concernant le logiciel dès que c'est disponible—y compris les informations concernant tout nouveau développement dans le produit. Votre inscription assiste la compagnie Trimble à améliorer continuellement son service après vente et de développement.

Vous pouvez enregistrer votre logiciel Trimble Geomatics Office lorsque vous êtes demandé à la fin du procédé d'installation, ou bien à tout moment dans Trimble Geomatics Office.

Pour ce faire:

1.	Dans Trimble Geomatics Office, sélectionnez Aide / Enregistrement.

2.	Sélectionnez une méthode d'enregistrement. Vous pouvez faire l'une des choses suivantes:

  Complétez et soumettez le formulaire d'enregistrement fournit sur l'Internet.

  Imprimez la formulaire d'enregistrement texte fourni, complétez-le, et puis envoyez-le par télécopie ou courrier.

Les demandes dans les formulaires d'enregistrement par télécopie/courrier et sur l'Internet sont identiques.

Les informations fournies ne sont ni utilisées ni distribuées en dehors de Trimble. Les formulaires d'enregistrement également comprennent des demandes à propos de comment et pourquoi vous utilisez Trimble Geomatics Office. Ces demandes ne sont pas obligatoires, mais vos réponses assisteront la compagnie Trimble d'améliorer son service après vente et de développement.


Enregistrement - commande Via Internet
Enregistrement − commande Via Internet

 Vue d'ensemble et plus amples informations

Pour enregistrer votre logiciel Trimble Geomatics via Internet:

1.	Sélectionnez Aide / Enregistrement / Via Internet. La page d'enregistrement en ligne s'affichera dans votre navigateur Web.

2.	Complétez le formulaire et assurez-vous de fournir les détails pour tous les champs marqués *.

3.	Cliquez sur Envoyer enregistrement maintenant.


Enregistrement - Via télécopie ou courrier
Enregistrement − Via télécopie ou courrier

 Vue d'ensemble et plus amples informations

Pour enregistrer votre logiciel Trimble Geomatics Office via télécopie ou courrier:

1.	Sélectionnez Aide / Enregistrement / Via télécopie ou courrier.

Le formulaire d'enregistrement Trimble Geomatics Office s'affichera dans votre éditeur texte par défaut.

2.	Complétez le formulaire et assurez-vous de fournir les détails pour tous les champs marqués *.

3.	Enregistrez le formulaire

4.	Envoyez le formulaire à:

Land Survey Marketing

Trimble Navigation

7403 Church Ranch Boulevard, Suite 100

Westminster, CO 80021

USA

Attention: Technical Publications Group

Télécopie: +1 720 887-6101


    Rapport de fermetures de boucle GPS - Vue d'ensemble

Plus amples informations

Comment ...	Rapport	Voir aussi

Vue d'ensemble

Le Rapport de fermetures de boucle GPS peut être utilisé pour assister à l'estimation de la qualité d'un groupe d'observations dans un réseau. Vous pouvez l'utiliser afin de vérifier des erreurs dans vos résultats de traitement de lignes de base (si vous avez le module Traitement de lignes de base installée). Cependant, pour générer des fermetures de boucle, il faut que vous aviez conçu votre réseau pour des figures géométriques fermées. Si c'est le cas,  Trimble Geomatics Office générera toutes les boucles possibles automatiquement et produira un Rapport de fermetures de boucle GPS.

Le dialogue Paramètres de fermeture de boucle est utilisé pour contrôler la façon dont les boucles sont construites et analysées. En outre il est utilisé pour contrôler la quantité des informations dans le Rapport de fermetures de boucle GPS. Les paramètres de fermeture de boucle sont accédés à partir de Rapports / Configuration / Rapport de fermetures de boucle GPS.

La Rapport de fermetures de boucle GPS par défaut comprend les sections suivantes:

  En-tête

  Résumé

  Boucles échouées

  Observations dans boucles échouées

  Occupations dans boucles échouées

Ces sections supplémentaires du rapport peuvent être comprises dans le rapport en les sélectionnant dans le dialogue Paramètres de fermeture de boucle:

  Détails des boucles échouées

  Boucles réussies

  Détails des boucles réussies

Considérez ces lignes directrices:

  Des petites erreurs de fermeture sont requises. Une grande erreur de fermeture indique qu'il y a un problème avec une ou plusieurs observations dans la boucle.

  Souvent les grandes erreurs de fermeture sont à cause des fautes effectuées sur le terrain, telle que la saisie des informations d'antenne incorrectes ou la configuration incorrecte d'une station. Vérifiez vos notes de terrain afin de vérifier les informations associées aux observations marquées comme faisant partie de boucles échouées multiples.

  Lors de la conception de votre réseau afin de consister en figures géométriques fermées, assurez-vous que le réseau soit observé afin de comprendre les lignes de base des sessions différentes dans chaque figure. Lorsqu'une boucle est générée utilisant des lignes de base qui ont été toutes observées dans la même session, il est impossible de détecter les erreurs de configuration de station communes.

NOTE: A partir du Rapport de fermeture de boucle, vous pouvez vous lier aux Boucles GPS dans la Vue de levé.


    Rapport de fermetures de boucle - En-tête

Plus amples informations

Comment ...	Rapport	Voir aussi
L'en-tête du rapport est comprise dans chaque rapport. Elle comprend des informations générales du projet extraites et une section Contenu. La section Contenu a des liens hypertextes pouvant être utilisés pour sauter aux sections différentes du rapport.

Il affiche les champs suivants, la plupart desquels contiennent les informations que vous avez spécifié dans les propriétés du projet:

Projet

Ce champ affiche le nom du projet.

Nom de l'utilisateur

Ce champ affiche le nom.

Date et heure

Ce champ affiche la date et l'heure de la génération du rapport.

Système de coordonnées

Ce champ affiche le système de coordonnées.

Zone

Ce champ affiche la zone du système de coordonnées. Si un site de calibration GPS est sélectionné, le nom du site est affiché dans le champ Zone.

Tous les systèmes de coordonnées publiés ont une zone. La seule exception est la projection par défaut qui affiche 'défaut' dans le champ Zone. Le défaut est utilisé lorsque vous supposez des coordonnées d'un site, par exemple N = 10000, E = 10000, et élév = 100.

Datum du projet

Ce champ affiche le datum du projet. Il ne représente pas le datum sélectionné pour l'ajustement.

Datum vertical

Vous pouvez ajouter ce champ aux informations d'en-tête du Rapport Ajustement de réseau. Pour ce faire, ajoutez les détails nécessaires au champ Datum vertical dans le dialogue Propriétés du projet – tabulation Détails du projet.

Modèle de géoïde

Ce champ affiche le modèle de géoïde si un modèle a été assigné dans les propriétés du projet. Lorsqu'aucun modèle de géoïde n'est assigné au projet, 'Pas sélectionné' s'affiche dans ce champ.

Unités de coordonnées

Ce champ affiche les unités du système de coordonnées .

Unités de distance

Ce champ affiche les unités de distance.

Unités de hauteur

Ce champ affiche les unités de hauteur.

Contenu

Le contenu du Rapport de fermetures de boucle GPS est contrôlé par les paramètres dans le dialogue Paramètres de fermeture de boucle . Le Rapport de fermetures de boucle GPS par défaut comprend les sections suivantes:

  En-tête

  Résumé

  Boucles échouées

  Observations dans boucles échouées

  Occupations dans boucles échouées

Ces sections du rapport supplémentaires peuvent être comprises dans le rapport en les sélectionnant dans le dialogue Paramètres de fermeture de boucle:

  Détails des boucles échouées

  Boucles réussies

  Détails des boucles réussies


    Rapport de fermetures de boucle GPS - Résumé

Plus amples informations

Comment ...	Rapport	Voir aussi
La section de résumé fournit une vue d'ensemble rapide des résultats de l'analyse de fermeture de boucle. Elle est comprise dans le rapport par défaut et comprend les informations suivantes:

Etapes dans la boucle

Ce champ affiche le nombre d'observations utilisées pour former les côtés de la boucle. Le nombre par défaut est 3. Si plus de 3 étapes sont spécifiées, toutes les boucles ayant ce nombre d'étapes, ou moins, sont générées. Lorsque le nombre 5 est spécifié, for exemple, toutes les boucles ayant 5, 4 ou 3 côtés sont considérées dans le rapport.

Nombre de boucles

Ce champ affiche le nombre total de boucles générées.

Nombre réussi

Ce champ affiche le nombre de boucles qui ont réussi les critères d'acceptance.

Nombre échoué

Ce champ affiche le nombre de boucles qui ont échoué les critères d'acceptance.

Critères de réussite / échec

Ces champs affichent la tolérance horizontale et verticale absolue.

Meilleur

Ces champs affichent les résultats de la meilleure boucle générée.

Plus mauvais

Ces champs affichent les résultats de la boucles la plus mauvaise. .

Boucle moyenne

Ces champs affichent les résultats de la moyenne de toutes les boucles générées pour ce rapport.

Déviation standard

Ces champs affichent la déviation standard de toutes les boucles générées pour ce rapport.


    Rapport de fermetures de boucle GPS - Boucles échouées

Plus amples informations

Comment ...	Rapport	Voir aussi
La section de boucles échouées fournit des informations concernant chaque boucle qui échoue le test de tolérance d'erreur de fermeture de boucle. Une boucle échoue lorsque les erreurs de fermeture sont plus grandes que les tolérances spécifiées dans le dialogue Paramètres de fermeture de boucle. Cette section est comprise dans le rapport par défaut.

Un exemple d'une section de rapport présentant une boucle échouée est indiquée ci-dessous:

Boucle GPS 1:
LUNE 2:ONDE:N 245:LUNE 2

Observations:

		De	Vers	Type de solution	Heure de début
LB3	S1	LUNE 2	ONDE	L1 fixe	13:45:03 24 Fév 1999
LB47	S5	LUNE 2	ONDE	L1 fixe	13:11:03 24 Fév 1999
LB2	S8	ONDE	N 245	L1 fixe	13:45:03 24 Fév 1999
LB46	S4	LUNE 2	N 245	L1 fixe	13:43:48 24 Fév 1999
LB49	S7	LUNE 2	N 245	L1 fixe	12:28:33 24 Fév 1999

Combinaisons échouées pour la boucle 1:

	Longueur	dHoriz	dVert	Ppm
S5 – S8 – S7	3295,499m	0,026m	0,034m	12,915

Table des observations

Les premières deux colonnes dans la table représentent les ID de solution et de ligne de base respectivement, de l'observation GPS utilisée dans la boucle. L'ID de ligne de base représente la combinaison de deux groupes de données GPS qui se chevauchent pouvant être traités pour former une ligne de base. L'ID de solution représente une résolution actuelle de la ligne de base par le processeur de lignes de base. On peut avoir des ID de solution multiples pour une seule ID de ligne de base. Ce se produit lorsque la case à cocher Ecraser les solutions de lignes de base en double est vide lors de l'Enregistrement des résultats du traitement GPS.

Les autres colonnes sont les suivantes:

De

Ce champ affiche le point de début de l'observation.

Vers

Ce champ affiche le point de fin de l'observation.

Type de solution

Ce champ affiche le type de solution pour l'observation.

Heure de début

Ce champ affiche l'heure de début de l'observation.

Table Combinaisons échouées pour la boucle #

La table Combinaisons échouées pour la boucle identifie la combinaison des observations GPS qui ont généré une erreur de fermeture supérieure aux critères d'acceptance. Les combinaisons multiples sont listées dans la table lorsque plusieurs échecs se produisent. La première colonne de la table affiche les solutions de ligne de base dans la combinaison échouée.

ASTUCE: Lorsque vous cliquez sur la combinaison échouée, les observations sont soulignées dans la Vue de levé.

Les autres colonnes sont les suivantes:

Longueur

Ce champ affiche la distance totale traversée autour de la boucle.

dHoriz

Ce champ affiche l'erreur de fermeture horizontale générée par cette boucle.

dVert

Ce champ affiche l'erreur de fermeture verticale générée par cette boucle

PPM

Ce champ affiche l'erreur de fermeture horizontale et verticale totale exprimée sous forme de parties par million.


    Rapport de fermetures de boucle GPS - Observations dans boucles échouées

Plus amples informations

Comment ...	Rapport	Voir aussi
Cette section aide à l'identification des lignes de base GPS qui ne s'adaptent pas au réseau. Ça se fait par l'étude de toutes les boucles échouées et la réalisation d'un rapport des lignes de base qui se trouvent dans des boucles échouées multiples. La liste est triée afin que toutes les lignes de base que se produisent dans le plus grand nombre de boucles échouées sont listées en premier. Cette liste contient aussi des statistiques de résumé de lignes de base que pourraient aider à l'identification des solutions insuffisantes.

Cette section est comprise dans le rapport par défaut.

 Un exemple d'une section de rapport est indiqué ci-dessous:

		De	Vers	Type de solution	Ratio	Var réf	RMS	# d'occurrences
LB2	S8	ONDE	N 245	L1 fixe	21,8	2,489	0,006	3
LB47	S5	LUNE 2	WAVE	L1 fixe	2,9	1,693	0,005	2
LB5	S19	COW-BOY	KTOM	L1 fixe	8,5	1,117	0,009	1

Retour en haut

Les premières deux colonnes dans la table représentent les ID de solution et de ligne de base respectivement, de l'observation GPS utilisée dans la boucle. L'ID de ligne de base représente la combinaison de deux groupes de données GPS qui se chevauchent pouvant être traités pour former une ligne de base. L'ID de solution représente une résolution actuelle de la ligne de base par le processeur de lignes de base. On peut avoir des ID de solution multiples pour une seule ID de ligne de base. Ce se produit lorsque la case à cocher Ecraser les solutions de lignes de base en double est vide lors de l'enregistrement des résultats du traitement GPS.

Les autres colonnes sont les suivantes:

De

Ce champ affiche le point de début de l'observation.

Vers

Ce champ affiche le point de fin de l'observation.

Heure de début

Ce champ affiche l'heure et la date de l'observation.

Ratio

Ce champ affiche le ratio de l'observation GPS comme déterminée par le processeur de lignes de base.

Var réf

Ce champ affiche la variance de référence de l'observation GPS comme déterminée par le processeur de lignes de base.

RMS

Ce champ affiche le RMS de l'observation GPS comme déterminée par le processeur de lignes de base.

# d'occurrences

Ce champ affiche le nombre de boucles échouées dont l'observation GPS a fait partie. Les observations avec un nombre plus grand d'occurrences devraient être revues pour estimer comment elles s'adaptent à votre réseau.


    Rapport de fermetures de boucle GPS - Occupations dans boucles échouées

Plus amples informations

Comment ...	Rapport	Voir aussi
Cette section aide à l'identification des occupations de station GPS pour les lignes de base que ne s'adaptent pas au réseau. Ces occupations de station peuvent avoir des erreurs de configuration de station. Ça se fait par l'étude de toutes les boucles échouées, la détermination des lignes de base trouvées dans des boucles échouées multiples, et la réalisation d'un rapport des occupations qui sont communes à ces lignes de base. La liste est triée afin que toutes les occupations que se produisent dans le plus grand nombre de boucles échouées sont listées en premier.

Cette section est comprise dans le rapport par défaut.

Un exemple d'une section de rapport est indiqué ci-dessous:

Occupations dans boucles échouées

Point	heure de début	# d'occurrences
KTOM	10:46:18 24 Fév 1999	6
ONDE	13:11:03 24 Fév 1999	5
LUNE2	12:04:48 24 Fév 1999	4
COW-BOY	14:05:03 24 Fév 1999	3
N 245	12:28:33 24 Fév 1999	2
COW-BOY	12:45:48 24 Fév 1999	1
Retour en haut

Les colonnes sont les suivantes:

Point

Ce champ affiche le nom du point auquel l'occupation s'est produite.

Heure de début

Ce champ affiche l'heure de l'occupation. Utilisez ce temps pour vérifier vos notes de terrain contre les informations contenues dans le projet.

# d'occurrences

Ce champ affiche le nombre de fois que cette occupation a été utilisée pour une observation faisant partie d'une boucle échouée. Les occupations avec un nombre plus grand d'occurrences devraient être examinées pour des erreurs de configuration de station.


    Rapport de fermetures de boucle GPS - Détails des boucles échouées

Plus amples informations

Comment ...	Rapport	Voir aussi
Cette section fournit plus amples informations concernant chaque boucle qui échoue le test de tolérance d'erreur de fermeture. Cette section n'est pas comprise dans le rapport par défaut. Vous pouvez activer cette section du dialogue Paramètres de fermeture de boucle .

Une exemple d'une section de rapport est indiqué ci-dessous:

Détails des boucles échouées

Boucle S1 - S86 - S3:

De:	LUNE 2
Vers:	ONDE		L1 fixe	13:45:02 24 Fév 1999
		Pente:	962,407m	606819,055m
		Total:	962,407m	1828548,125m
				29,499m
Vers:	N 245		L1 flottant	13:45:02 24 Fév 1999
		Pente:	852,542m	606662,061m
		Total:	1814,949m	1829386,030m
				33,251m
Vers:	LUNE 2		L1 fixe	13:43:47 24 Fév 1999
		Pente:	1480,504m	605983,905m
		Total:	3295,453m	1828070,086m
				26,770m
Précision:		32,407ppm
Erreurs	Horiz:	0,096m	Vert:	–0,046m

En haut de la table est une liste des solutions de lignes de base faisant partie de la boucle. Celle-ci est suivie de la station De, étant le point à partir duquel la boucle provient.

Ces informations initiales sont suivies de trois ou plus sous-sections, chacune composée de quatre lignes. Ces sous-sections contiennent des informations de station, et elles ont fournies dans l'ordre dans lequel le programme traverse autour de la boucle. La première ligne de chaque sous-section commence avec l'identificateur Vers:. Les informations suivantes sont fournies dans ces sous-sections de station:

Ligne 1

L'identificateur Vers: est la première ligne dans une sous-section à quatre lignes. La ligne fournit des informations concernant l'observation GPS. L'identificateur Vers: est suivi du nom du point, listé dans l'ordre dans lequel la boucle est traversée. Celui-ci est suivi du type d'observation GPS et la date de la collecte des mesures GPS.

Les informations fournies dans cette ligne correspondent à l'une des ID de solution fournies en haut de la table. Plus amples informations concernant une solution spécifique de ligne de base peuvent être obtenues en étudiant ses statistiques en utilisant la fenêtre Propriétés dans Vue de levé, ou en créant un Rapport du traitement de lignes de base GPS pour cette ligne de base.

Ligne 2

L'identificateur Pente indique la deuxième ligne d'une sous-section à quatre lignes. L'identificateur est suivi de la distance inclinée de l'observation GPS. Celle-ci est suivie du composant nord des coordonnées du point Vers courant.

Ligne 3

L'identificateur Total indique la troisième ligne d'une sous-section à quatre lignes. L'identificateur est suivi de la distance totale traversée jusqu'à ce moment dans la construction de la boucle. Celle-ci est suivie du composant est des coordonnées du point Vers courant.

Ligne 4

La dernière ligne de la sous-section n'a pas un identificateur. Elle contient la hauteur des coordonnées du point Vers courant.

NOTE: Les coordonnées de nord, est et hauteur rapportées ici ne seront pas les mêmes coordonnées que celles trouvées dans le projet. Les valeurs affichées ici sont dérivées en ajoutant les observations GPS aux coordonnées initiales de la station De et en continuant cet ajout de vecteur autour de la boucle. A la fin de la boucle, les coordonnées dérivées de cette façon sont soustraites des coordonnées initiales du point De pour générer l'erreur de fermeture.

La dernière sous-section de station dans cette table est pour la station De, indiquant que la boucle a été fermée. Les deux dernières lignes de la table fournissent des informations concernant l'erreur de fermeture.

Précision

Ce champ affiche l'erreur de fermeture horizontale et verticale exprimée en parties par million.

Erreurs

Ce champ affiche les erreurs de fermeture verticales et horizontales générées par cette boucle.


    Rapport de fermetures de boucle - Boucles réussies

Plus amples informations

Comment ...	Rapport	Voir aussi
Cette section fournit des informations concernant chaque boucle qui réussit au test de tolérance. Une boucle réussit lorsque les erreurs de fermeture de la boucle sont moins de ou égales aux tolérances spécifiées. Cette section n'est pas comprise dans le rapport par défaut. Vous pouvez activer cette section du dialogue Paramètres de fermeture de boucle .

Une exemple de section de rapport est indiqué ci-dessous:

Boucles réussies

Boucle GPS 1:
LUNE 2: ONDE: COW-BOY: LUNE 2

Observations:

ID de ligne de base	ID de solution	De	Vers	Type de solution	Heure de début
LB3	S1	LUNE 2	ONDE	L1 fixe	13:45:02 24 Fév 1999
LB47	S4	LUNE 2	ONDE	L1 fixe	13:11:02 24 Fév 1999
LB1	S9	ONDE	COW-BOY	L1 fixe	14:05:02 24 Fév 1999
LB45	S2	LUNE 2	COW-BOY	L1 fixe	14:05:02 24 Fév 1999
LB48	S5	LUNE 2	COW-BOY	L1 fixe	12:45:47 24 Fév 1999
Combinaisons réussies pour la Boucle 1:

	Longueur	dHoriz	dVert	Ppm
S1 - S9 - S2	3795,569m	0,001m	-0,002m	0,508
S1 - S9 - S5	3795,577m	0,009m	0,001m	2,296
S4 - S9 - S2	3795,568m	0,018m	-0,012m	5,683
S4 - S9 - S5	3795,577m	0,020m	-0,009m	5,872

Table des observations

Les premières deux colonnes dans la table représentent les ID de solution et de ligne de base respectivement, de l'observation GPS utilisée dans la boucle. L'ID de ligne de base représente la combinaison de deux groupes de données GPS qui se chevauchent pouvant être traités pour former une ligne de base. L'ID de solution représente une résolution actuelle de la ligne de base par le processeur de lignes de base. On peut avoir des ID de solution multiples pour une seule ID de ligne de base. Ce se produit lorsque la case à cocher Ecraser les solutions de lignes de base en double est vide lors de l'Enregistrement des résultats du traitement GPS.

Les autres colonnes sont les suivantes:

De

Ce champ affiche le point de début de l'observation.

Vers

Ce champ affiche le point de fin de l'observation.

Type de solution

Ce champ affiche le type de solution pour l'observation.

Heure de début

Ce champ affiche l'heure de début de l'observation.

Table Combinaisons réussies pour la boucle #

La table Combinaisons échouées pour la boucle # identifie la combinaison des Observations GPS qui ont généré une erreur de fermeture moins de ou égale aux critères d'acceptance. La première colonne de la table affiche les solutions de ligne de base dans la combinaison réussie.

ASTUCE: Lorsque vous cliquez sur la combinaison réussie, les observations sont soulignées dans la Vue de levé.

Le deuxième groupe de colonnes sont les suivantes:

Longueur

Ce champ affiche la distance totale traversée autour de la boucle.

dHoriz

Ce champ affiche l'erreur de fermeture horizontale générée par cette boucle.

dVert

Ce champ affiche l'erreur de fermeture verticale générée par cette boucle

Ppm

Ce champ affiche l'erreur de fermeture horizontale et verticale totale exprimée sous forme de parties par million.


    Rapport de fermetures de boucle GPS - Détails de boucles réussies

Plus amples informations

Comment ...	Rapport	Voir aussi
Cette section fournit plus amples informations concernant chaque boucle que réussit au test de tolérance d'erreur de fermeture. Elle n'est pas comprise dans le rapport par défaut. Vous pouvez activer cette section en utilisant le dialogue dialogue Paramètres de fermeture de boucle GPS.Un exemple de section de rapport est indiqué ci-dessous:

Détails des boucles réussies

Boucle S1 - S9 - S2:

De:	LUNE 2
Vers:	ONDE		L1 fixe	13:45:02 24 Fév 1999
		Pente:	962,407m	606819,055m
		Total:	962,407m	1828548,125m
				29,499m
Vers:	COW-BOY		L1 fixe	14:05:02 24 Fév 1999
		Pente:	1076,569m	607738,388m
		Total:	2038,976m	1827988,084m
				22,331m
Vers:	LUNE 2		L1 fixe	14:05:02 24 Fév 1999
		Pente:	1756,593m	605983,815m
		Total:	3795,569m	1828070,120m
				26,815m
Précision:	0,50858508651573ppm
Erreurs	Horiz:	0,001m	Vert:	–0,002m

En haut de la table est une liste des solutions de lignes de base faisant partie de la boucle. Celle-ci est suivie de la station De, étant le point à partir duquel la boucle provient.

Ces informations initiales sont suivies de trois ou plus sous-sections, chacune composée de quatre lignes. Ces sous-sections contiennent des informations de station, et elles ont fournies dans l'ordre dans lequel le programme traverse autour de la boucle. La première ligne de chaque sous-section commence avec l'identificateur Vers:. Les informations suivantes sont fournies dans ces sous-sections de station:

Ligne 1

L'identificateur Vers: est la première ligne dans une sous-section à quatre lignes. La ligne fournit des informations concernant l'observation GPS. L'identificateur Vers: est suivi du nom du point, listé dans l'ordre dans lequel la boucle est traversée. Celui-ci est suivi du type d'observation GPS et la date de la collecte des mesures GPS.

Les informations fournies dans cette ligne correspondent à l'une des ID de solution fournies en haut de la table. Plus amples informations concernant une solution spécifique de ligne de base peuvent être obtenues en examinant ses statistiques en utilisant la fenêtre Propriétés dans Vue de levé, ou en créant un Rapport du traitement de lignes de base GPS pour cette ligne de base.

Ligne 2

L'identificateur Pente indique la deuxième ligne d'une sous-section à quatre lignes. L'identificateur est suivi de la distance inclinée de l'observation GPS. Celle-ci suivie du composant nord des coordonnées du point Vers courant.

Ligne 3

L'identificateur Total indique la troisième ligne d'une sous-section à 4 lignes. L'identificateur est suivi de la distance totale traversée jusqu'à ce moment dans la construction de la boucle. C'est suivi du composant est des coordonnées du point Vers courant.

Ligne 4

La dernière ligne de la sous-section n'a pas un identificateur. Elle contient la hauteur des coordonnées du point Vers courant.

NOTE: Les coordonnées nord, est et hauteur rapportées ici ne seront pas les mêmes coordonnées que celles trouvées dans le projet. Les valeurs affichées ici sont dérivées en ajoutant les observations GPS aux coordonnées initiales de la station De et en continuant cet ajout de vecteur autour de la boucle. A la fin de la boucle, les coordonnées dérivées de cette façon sont soustraites des coordonnées initiales du point De pour générer l'erreur de fermeture.

La dernière sous-section de station dans cette table est pour la station De, indiquant que la boucle a été fermée. Les deux dernières lignes de la table fournissent des informations concernant l'erreur de fermeture.

Précision

Ce champ affiche l'erreur de fermeture horizontale et verticale exprimée en parties par million.

Erreurs

Ce champ affiche les erreurs de fermeture verticales et horizontales générées par cette boucle.


    Dialogue Paramètres de fermeture de boucle GPS

Plus amples informations

Comment ...
Utilisez ce dialogue pour configurer les options de Fermeture de boucle et pour sélectionner les sections inclues dans le Rapport de fermetures de boucle.

Pour afficher le dialogue Paramètres de fermeture de boucle, effectuez l'une des choses suivantes:

  Sélectionnez Rapports / Configuration / Rapport de fermetures de boucle.

  Appuyez sur Alt+RSL.

Le dialogue Paramètres de fermeture de boucle s'affiche. Les contrôles ou groupes suivants sont compris:

Etapes dans la boucle

Editez ce champ pour spécifier le nombre d'observations utilisées pour former les côtés de la boucle. Le nombre par défaut est 3. Si plus de 3 étapes sont spécifiées, toutes les boucles ayant ce nombre d'étapes, ou moins, sont générées. Lorsque le nombre 5 est spécifié, for exemple, toutes les boucles ayant 5, 4 ou 3 côtés sont considérées dans le rapport.

Tolérance

Editez ces champs pour définir l'erreur de fermeture maximum acceptable que conforme aux exigences de précision requises pour votre projet. Les erreurs de fermeture qui dépassent ces tolérances seront rapportées comme des boucles échouées dans le Rapport de fermetures de boucle GPS . La tolérance peut être basée sur une distance absolue ou sous forme d'un ratio exprimé en parties par million.

Lorsque l'option Absolu est sélectionnée les champs Horizontal et Vertical sont disponibles. Saisissez des tolérances de distance verticales et horizontales dans ces champs pour spécifier l'erreur de fermeture maximum permise.

Lorsque Ratio est sélectionné le champ Ppm est disponible. Saisissez l'erreur de fermeture maximum permise en ppm. Lorsque Ration est sélectionné les erreurs de fermeture verticales et horizontales sont combinées avant d'effectuer le test de tolérance d'erreur de fermeture.

Sections du rapport

Utilisez les cases à cocher dans ce groupe pour sélectionner les sections s'affichant dans le Rapport de fermetures de boucles GPS. Les sections suivantes peuvent être comprises:

  Résumé général

  Boucles échouées

  Détails des boucles échouées

  Observations dans boucles échouées

  Occupations dans boucles échouées

  Boucles réussies

  Détails des boucles réussies

Par défaut, les sections suivantes sont comprises:

  Résumé général

  Boucles échouées

  Observations dans boucles échouées

  Occupations dans boucles échouées

Il est possible d'enlever toute section du rapport par l'effacement de sa case à cocher. Toute autre section peut être comprise dans le rapport par la sélection de ses cases à cocher.

N'utiliser que les solutions actives

Sélectionnez cette case à cocher si vous ne voulez utiliser que des solutions actives dans le Rapport de fermetures de boucle.


    Commande Rapport de fermetures de boucle GPS

Comment ...	Rapport
Utilisez cette commande pour effectuer un analyse de fermeture de boucle et afficher les résultats sous forme de format de rapport.

Pour afficher le Rapport de fermetures de boucle GPS:

1.	Sélectionnez Rapport / Rapport de fermetures de boucle GPS. Si vous avez sélectionné des observations dans la fenêtre graphique, le dialogue Rapport de fermetures de boucle s'affiche.

2.	Sélectionnez l'une des options suivantes:

  Sélection – Effectue une fermeture de boucle sur les lignes de base sélectionnées.

  BD entière – Effectue une fermeture de boucle sur la base de données entière.

L'afficheur HTML par défaut de votre ordinateur se démarre après l'achèvement de l'analyse de fermeture de boucle et charge le  Rapport de fermetures de boucle GPS.


    Configuration du Rapport de fermetures de boucle GPS

Plus amples informations

Comment ...	Dialogue

Sélectionnez cette commande pour choisir les sections comprises dans le Rapport de fermetures de boucle GPS.

Pour afficher le dialogue Configuration du Rapport de fermetures de boucle GPS, effectuez l'une des choses suivantes:

  Sélectionnez Rapport / Configuration / Rapport de fermetures de boucle GPS.

  Appuyez sur Alt+RSL.

L'afficheur HTML par défaut de votre ordinateur se démarre et le dialogue Configuration du rapport de fermetures de boucle GPS s'affiche.

NOTE: Plusieurs sections s'affichent par défaut et des sections supplémentaires peuvent être ajoutées. Vous pouvez désactiver ou activer toute combinaison de cases à cocher pour personnaliser le contenu du rapport.


Dialogue Rapport de fermeture de boucle

Utilisez ce dialogue pour sélectionner s'il faut rapporter sur la base de données entière ou sur des boucles sélectionnées.

Pour afficher le Rapport de fermetures de boucle GPS:

1.	Sélectionnez Rapport / Rapport de fermetures de boucle GPS. Si vous avez sélectionné des observations dans la fenêtre graphique, le dialogue Rapport de fermetures de boucle s'affiche.

2.	Sélectionnez l'une des options suivantes:

  Sélection – Effectue une fermeture de boucle sur les lignes de base sélectionnées.

  BD entière – Effectue une fermeture de boucle sur la base de données entière.

L'afficheur HTML par défaut de votre ordinateur se démarre après l'achèvement de l'analyse de fermeture de boucle et charge le  Rapport de fermetures de boucle GPS.


Pour vérifier la fermeture de boucle

Après le traitement des lignes de base, il se peut qu'un indicateur rouge d'avertissement s'affiche dans la fenêtre Vue de levé pour marquer chaque ligne de base que n'a pas réussie les critères d'acceptance de qualité. Les lignes de base avec des indicateurs d'avertissement peuvent également causer des grandes erreurs de fermeture lors de la réalisation de l'analyse de fermeture de boucle.

Des causes d'erreur contribuant aux erreurs de fermeture insuffisantes peuvent comprendre:

  L'installation sur un point de levé incorrecte.

  La saisie des informations d'antenne incorrectes.

  Les résultats de traitement insuffisants.

Pour vérifier une fermeture de boucle:

1.	Sélectionnez Rapport / Rapport de fermetures de boucle GPS. Si vous avez sélectionné des observations dans la fenêtre graphique, le dialogue Rapport de fermetures de boucle s'affiche.

2.	Sélectionnez l'une des options suivantes:

  Sélection − Effectue une fermeture de boucle sur les lignes de base sélectionnées.

  BD entière − Effectue une fermeture de boucle sur la base de données entière.

L'afficheur HTML par défaut de votre ordinateur se démarre après l'achèvement de l'analyse de Fermeture de boucle et le rapport s'affiche avec l'en-tête en haut du rapport.

3.	Cliquez sur Résumé.

La section Résumé du Rapport de fermeture de boucle s'affiche.


Pour configurer la Rapport de fermetures de boucle

 Vue d'ensemble et plus amples informations

Les sections du Rapport de fermetures de boucle GPS peuvent être activées ou désactivées pour contrôler le contenu du rapport. Par défaut, ces sections du rapport sont activées:

  Résumé

  Boucles échouées

  Observations dans boucles échouées

  Occupations dans boucles échouées

Lorsqu'activées, ces sections du rapport sont également disponibles:

  Détails des boucles échouées

  Boucles réussies

  Détails des boucles réussies

Pour activer ou désactiver les sections du rapport:

1.	Pour afficher le Rapport de fermetures de boucle GPS, effectuez l'une des choses suivantes:

  Sélectionnez Rapport / Configuration / Rapport de fermetures de boucle GPS.

  Appuyez sur Alt+RSL.

Le dialogue Paramètres de fermeture de boucle s'affiche.

2.	Sélectionnez ou effacez les cases à cocher applicables dans le groupe Sections du rapport.


Pour afficher un résume du Rapport de fermetures de boucle GPS

 Vue d'ensemble et plus amples informations

Pour afficher le résumé de rapport:

1.	Sélectionnez Rapport / Rapport de fermetures de boucle GPS. Si vous avez sélectionné des observations dans la fenêtre graphique, le dialogue Rapport de fermetures de boucle s'affiche.

2.	Sélectionnez l'une des options suivantes:

  Sélection – Effectue une fermeture de boucle sur les lignes de base sélectionnées.

  BD entière – Effectue une fermeture de boucle sur la base de données entière.

L'afficheur HTML par défaut de votre ordinateur se démarre après l'achèvement de l'analyse de Fermeture de boucle et le rapport s'affiche avec l'en-tête en haut du rapport.

3.	Cliquez sur Résumé.

La section Résumé s'affiche.


Pour afficher le Rapport de fermetures de boucle GPS

Vue d'ensemble et plus amples informations
Pour afficher le rapport:

1.	Sélectionnez Rapport / Rapport de fermetures de boucle GPS. Si vous avez sélectionné des observations dans la fenêtre graphique, le dialogue Rapport de fermetures de boucle s'affiche.

2.	Sélectionnez l'une des options suivantes:

  Sélection – Effectue une fermeture de boucle sur les lignes de base sélectionnées.

  BD entière – Effectue une fermeture de boucle sur la base de données entière.

L'afficheur HTML par défaut de votre ordinateur se démarre après l'achèvement de l'analyse de fermeture de boucle, et le rapport s'affiche.

3.	Utilisez les liens hypertextes pour sauter aux sections du rapport et étudier les données.


Pour afficher l'En-tête du rapport de fermetures de boucle GPS

Vue d'ensemble et plus amples informations
Pour afficher l'en-tête du rapport:

1.	Sélectionnez Rapport / Rapport de fermetures de boucle GPS. Si vous avez sélectionné des observations dans la fenêtre graphique, le dialogue Rapport de fermetures de boucle s'affiche.

2.	Sélectionnez l'une des options suivantes:

  Sélection − Effectue une fermeture de boucle sur les lignes de base sélectionnées.

  BD entière − Effectue une fermeture de boucle sur la base de données entière.

L'afficheur HTML par défaut de votre ordinateur se démarre après l'achèvement de l'analyse de Fermeture de boucle et le rapport s'affiche avec l'en-tête en haut du rapport.

3.	Utilisez les barres de défilement pour étudier les données.


Pour afficher les statistiques de boucles réussies de fermeture de boucle GPS

Vue d'ensemble et plus amples informations
La section Boucles réussies identifie quelles boucles  ont réussies le test de tolérance d'erreur de fermeture. Cette section du rapport ne s'affiche pas dans le rapport par défaut. Vous pouvez inclure cette section dans le rapport en la sélectionnant dans le dialogue Paramètres de fermeture de boucle .

Pour afficher les statistiques de boucles réussies:

1.	Sélectionnez Rapport / Rapport de fermetures de boucle GPS. Si vous avez sélectionné des observations dans la fenêtre graphique, le dialogue Rapport de fermetures de boucle s'affiche.

2.	Sélectionnez l'une des options suivantes:

  Sélection – Effectue une fermeture de boucle sur les lignes de base sélectionnées.

  BD entière – Effectue une fermeture de boucle sur la base de données entière.

L'afficheur HTML par défaut de votre ordinateur se démarre après l'achèvement de l'analyse de Fermeture de boucle et le rapport s'affiche avec l'en-tête en haut du rapport.

3.	Cliquez sur Boucles réussies.

La section Détails de boucles réussies s'affiche.


Pour afficher les Détails de boucles réussies de fermeture de boucle GPS

Vue d'ensemble et plus amples informations
La section Détails de boucles réussies fournit plus amples informations concernant les boucles qui ont réussi le test de tolérance d'erreur de fermeture. Cette section du rapport ne s'affiche pas dans le rapport par défaut. Vous pouvez inclure cette section dans le rapport en la sélectionnant dans le dialogue Paramètres de fermeture de boucle.

Pour afficher les statistiques de boucles réussies:

1.	Sélectionnez Rapport / Rapport de fermetures de boucle GPS. Si vous avez sélectionné des observations dans la fenêtre graphique, le dialogue Rapport de fermetures de boucle s'affiche.

2.	Sélectionnez l'une des options suivantes:

  Sélection – Effectue une fermeture de boucle sur les lignes de base sélectionnées.

  BD entière – Effectue une fermeture de boucle sur la base de données entière.

L'afficheur HTML par défaut de votre ordinateur se démarre après l'achèvement de l'analyse de Fermeture de boucle et le rapport s'affiche avec l'en-tête en haut du rapport.

3.	Cliquez sur Détails de boucles réussies.

La section Détails de boucles réussies s'affiche.


Pour afficher les Détails de boucles échouées de fermeture de boucle GPS

Vue d'ensemble et plus amples informations
La section Détails de boucles échouées fournit plus amples informations concernant les boucles que n'ont pas réussi le test de tolérance d'erreur de fermeture. Cette section du rapport ne s'affiche pas dans le rapport par défaut. Vous pouvez inclure cette section dans le rapport en la sélectionnant dans le dialogue Paramètres de fermeture de boucle.

Pour afficher les statistiques de boucles échouées:

1.	Sélectionnez Rapport / Rapport de fermetures de boucle GPS. Si vous avez sélectionné des observations dans la fenêtre graphique, le dialogue Rapport de fermetures de boucle s'affiche.

2.	Sélectionnez l'une des options suivantes:

  Sélection – Effectue une fermeture de boucle sur les lignes de base sélectionnées.

  BD entière – Effectue une fermeture de boucle sur la base de données entière.

L'afficheur HTML par défaut de votre ordinateur se démarre après l'achèvement de l'analyse de Fermeture de boucle et le rapport s'affiche avec l'en-tête en haut du rapport

3.	Cliquez sur Détails de boucles échouées.

La section Détails de boucles échouées s'affiche.


Pour afficher les statistiques des Observations dans boucles échouées

Vue d'ensemble et plus amples informations
La section Observations dans boucles échouées aide à l'identification des lignes de base GPS qui ne s'adaptent pas au réseau. Ça se fait par l'étude de toutes les boucles échouées et la réalisation d'un rapport des lignes de base qui se trouvent dans des boucles échouées multiples. La liste est triée afin que toutes les lignes de base que se produisent dans le plus grand nombre de boucles échouées sont listées en premier. Cette également contient des statistiques de résumée de lignes de base que pourraient aider à l'identification des solutions.

Pour afficher les observations dans boucles échouées:

1.	Pour afficher le Rapport de fermetures de boucle GPS, effectuez l'une des choses suivantes:

  Sélectionnez Rapport / Configuration / Rapport de fermetures de boucle GPS.

  Appuyez sur Alt+RSL..

L'afficheur HTML par défaut de votre ordinateur se démarre après l'achèvement de l'analyse de Fermeture de boucle et le rapport s'affiche avec l'en-tête en haut du rapport.

2.	Cliquez sur Observations dans boucles échouées.

La section Observations dans boucles échouées s'affiche.


Pour afficher les statistiques des Occupations dans boucles échouées

Vue d'ensemble et plus amples informations
La section Observations dans boucles échouées aide à l'identification des lignes de base GPS qui ne s'adaptent pas au réseau. Ça se fait par l'étude de toutes les boucles échouées et la réalisation d'un rapport des lignes de base qui se trouvent dans des boucles échouées multiples. La liste est triée afin que toutes les lignes de base que se produisent dans le plus grand nombre de boucles échouées sont listées en premier. Cette également contient des statistiques de résumée de lignes de base que pourraient aider à l'identification des solutions.

Pour afficher les observations dans boucles échouées:

1.	Sélectionnez Rapport / Rapport de fermetures de boucle GPS. Si vous avez sélectionné des observations dans la fenêtre graphique, le dialogue Rapport de fermetures de boucle s'affiche.

2. 	Sélectionnez l'une des options suivantes:

  Sélection – Effectue une fermeture de boucle sur les lignes de base sélectionnées.

  BD entière – Effectue une fermeture de boucle sur la base de données entière.

L'afficheur HTML par défaut de votre ordinateur se démarre après l'achèvement de l'analyse de Fermeture de boucle et le rapport s'affiche avec l'en-tête en haut du rapport.

3.	Cliquez sur Occupations dans boucles échouées.

La section Occupations dans boucles échouées s'affiche.


Pour afficher les statistiques de boucles échouées

 Vue d'ensemble et plus amples informations

Les statistiques de Boucles échouées identifient pourquoi les étapes dans une boucle de ligne de base ont échoué de fermer..

Pour afficher les statistiques de boucles échouées:

1.	Sélectionnez Rapport / Rapport de fermetures de boucle GPS. Si vous avez sélectionné des observations dans la fenêtre graphique, le dialogue Rapport de fermetures de boucle s'affiche.

2. 	Sélectionnez l'une des options suivantes:

  Sélection – Effectue une fermeture de boucle sur les lignes de base sélectionnées.

  BD entière – Effectue une fermeture de boucle sur la base de données entière.

L'afficheur HTML par défaut de votre ordinateur se démarre après l'achèvement de l'analyse de Fermeture de boucle et le rapport s'affiche avec l'en-tête en haut du rapport.

3.	Cliquez sur Boucles échouées.

La section Boucles échouéess'affiche.


Pour afficher les propriété de l'indicateur d'avertissement de ligne de base

Vue d'ensemble et plus amples informations
Après les traitement de lignes de base, un indicateur d'avertissement rouge s'affiche dans la Fenêtre graphique pour indiquer chaque ligne de base que n'a pas réussi les critères d'acceptance de qualité.

Pour afficher les détails de l'indicateur d'avertissement:

1.	Sélectionnez une ligne de base.

2.	Ouvrez la fenêtre Propriétés (Cliquez avec le bouton droit de la souris pour accéder au menu de raccourci et sélectionnez Propriétés).

La fenêtre Propriétés s'affiche.

3.	Cliquez sur l'icône plus (+) à côté de la ligne de base dans la liste de sélection de la fenêtre Propriétés.

L'arborescence de la Liste de sélection se développe et affiche les points 'de' et 'vers' de la ligne de base. Un indicateur rouge indique un problème de traitement de lignes de base.

4.	Sélectionnez tout point avec un indicateur.

5.	Effectuez l'une des choses suivantes:

  Cliquez sur la bouton Indicateurs d'avertissement.

  Dans le menu Page détaillée, sélectionnez Avertissements.

Un message affiche pourquoi la ligne de base a un indicateur. Utilisez ces informations pour résoudre le problème empêchant la ligne de base de réussir les critères d'acceptance. Pour plus amples informations, référez-vous à Analyse de données de traitement GPS.


Recalcul - Vue d'ensemble

Le recalcul est le processus d’établissement de la position calculée d’un point.

Trimble Geomatics Office effectue un recalcul sur toutes les données, c.-à-d. sur toutes les observations GPS, conventionnelles et par télémètre à laser, ainsi que sur toutes les coordonnées saisies au clavier ou importées pour un point. Il utilise ces observations et les coordonnées saisies au clavier pour déterminer une position d’une qualité aussi élevée que possible pour un point.

Le recalcul :

  calcule la position pour les points mesurés sur le terrain

  donne une qualité appropriée au point

  détecte et fait un rapport sur les erreurs de fermeture des données

  crée un Rapport de recalcul

Si un point comporte des données de levé redondantes, le Rapport de dérivation du point indique quelle(s) observation(s) ou coordonnées saisies au clavier ont été utilisées pour établir la position calculée. Le Rapport de recalcul contient toutes les observations redondantes qui sont hors tolérance. Les tolérances sont définies dans la tabulation Recalculer du dialogue Propriétés du projet.

Un recalcul est requis lorsque des données sont ajoutées au projet, ou des données existantes éditées, par exemple si vous modifiez une hauteur d’antenne pour une observation GPS.

Cependant, Trimble Geomatics Office n’effectue pas de recalcul automatiquement. Lorsqu’un recalcul est requis, l’icône Recalculer [image] s’affiche sur la barre d'état de Trimble Geomatics Office.

Pour recalculer, effectuez l’une des opérations suivantes :

  Sélectionnez Levé / Recalculer.

  Appuyez sur F4.

  Dans la barre d'état, double-cliquez sur l’icône Recalculer (si elle est affichée).

  Cliquez avec le bouton droit de la souris pour accéder au menu de raccourcis et sélectionnez Recalculer.

NOTE : Vous pouvez effectuer un recalcul à tout moment ; cependant, si l’icône Recalculer n’est pas d’abord affichée sur la barre d'état, rien ne change.

Le recalcul :

  ne répartit pas les erreurs

  utilise les coordonnées saisies au clavier de qualité Contrôle ou Levé plutôt que les observations.

NOTE : Le terme de coordonnées saisies au clavier fait référence aux coordonnées d’un point qui sont saisies au clavier à l’aide de Trimble Geomatics Office ou importées d’un fichier de données dans le projet Trimble Geomatics Office.


Recalcul - Comment la position calculée des points observés est déterminée

 Vue d'ensemble et plus amples informations

L’organigramme montre comment un recalcul détermine la position calculée pour des points observés sur le terrain.

NOTE : Lors de la détermination de la position calculée pour un point observé, le recalcul n’utilise pas les observations désactivées ni les coordonnées saisies au clavier désactivées. Il utilise les observations ou les coordonnées saisies au clavier activées comme vérifications uniquement lorsque les positions des points du projet ne peuvent pas être déterminées par d’autres observations ou coordonnées. Pour activer et désactiver des observations, utilisez la fenêtre Propriétés.

Les rubriques suivantes décrivent les cadres de l’organigramme :

  Comment un recalcul détermine les points de début potentiels

  Comment un recalcul détermine la position de tous les points de début potentiels

  Recalcul − Comment les qualités de composant déterminent le point de début courant

  Recalcul − Comment les observations sont appliquées à partir du point de début courant

  Recalcul − Comment la qualité des points observés est déterminée


Recalcul - Comment les points de début potentiels sont déterminés

 Vue d'ensemble et plus amples informations

Un recalcul peut débuter avec tout point présentant les caractéristiques suivantes :

  Coordonnée saisie au clavier, et

  Une ou plusieurs observations à partir du point

Par exemple, la figure suivante représente une base cinématique en temps réel (RTK). Il s’agit d’un point de début potentiel.


Recalcul - Comment la position de tous les points de début potentiels est déterminée

 Vue d'ensemble et plus amples informations

La position du point de début est déterminée à partir de la coordonnée saisie au clavier. Un point de début potentiel ne comporte habituellement qu’un seul groupe de coordonnées saisies au clavier. La qualité de chaque composant de la position (NE,é,h) est héritée de la qualité de la coordonnée saisie au clavier. Si le composant d’élévation ou de hauteur n’a pas été saisi au clavier, alors le modèle de géoïde est utilisé pour déduire le composant manquant.

Un point peut comporter plus d’un groupe de coordonnées saisies au clavier si vous :

  saisissez des coordonnées au clavier pour un point et effectuez un ajustement (si le module Ajustement de réseau est installé sur votre ordinateur) sans garder le point fixé (cela entraîne la création d’un groupe de coordonnées ajusté pour le point)

  importez des coordonnées WGS-84 et des coordonnées de contrôle de grille pour un point

  importez une coordonnée WGS-84 et une élévation. Une élévation est enregistrée comme partie d’une coordonnée de grille.

Dans ce cas, le recalcul utilise certains critères pour déterminer quelle coordonnée utiliser comme position calculée. S’il ne trouve aucun groupe de coordonnées à l’aide du premier critère, il applique le critère suivant. Les critères, dans l’ordre d’utilisation, sont les suivants :

1.	L’état (activé/désactivé) des coordonnées

2.	Le composant de la plus haute qualité.

3.	Les coordonnées WGS-84, plutôt que les coordonnées de grille, sont utilisées.


Recalcul - Comment la position de tous les points de début potentiels est déterminée : (1) L'état des coordonnées

 Voir aussi

L’état (activé/désactivé) des coordonnées

Seules les coordonnées activées sont prises en compte lors de la détermination de la position des points de début. Les coordonnées de vérification sont utilisées pour la détermination des points de début uniquement si les observations ne sont pas utilisées pour déterminer les coordonnées du point.

NOTE : Les coordonnées de vérification sont rarement utilisées pour déterminer les coordonnées des points de début.

Les coordonnées désactivées ne sont jamais prises en compte. Pour activer ou désactiver des coordonnées, utilisez la fenêtre Propriétés ou le dialogue Edition multiple.

Le composant de la plus haute qualité

Si plus d’un groupe de coordonnées est activé, le recalcul choisit les composants de la plus haute qualité parmi toutes les coordonnées saisies au clavier disponibles pour le point. Les trois composants sont les suivants :

  horizontal

  élévation

  hauteur

NOTE : Il n’est pas nécessaire que ces composants proviennent du même groupe de coordonnées.

La table suivante contient les qualités des trois composants et classe ces qualités de la plus élevée à la plus basse. L’ordre des qualités est appelé la « hiérarchie de qualités ».

Dans la fenêtre Propriétés, le symbole de la qualité est affiché à côté de chaque composant.

Type de coordonnées

Si les deux coordonnées ont la même qualité, le recalcul fait une sélection selon le type de la coordonnée. Les coordonnées WGS-84, plutôt que les coordonnées de grille/locales, sont utilisées.


Recalcul - Comment les qualités de composant déterminent le point de début courant

 Vue d’ensemble et plus amples informations

S’il y a plus d’un point de début potentiel, le recalcul choisit le point de la plus haute qualité comme point de début courant. Il utilise la hiérarchie suivante :

1.	Contrôle

2.	Ajusté

3.	Levé

4.	Mappage

5.	Inconnu

Chaque qualité de composant est considérée. Si la qualité de la position du point est différente pour chaque composant, la qualité horizontale est prise en compte d’abord, puis la hauteur et finalement l’élévation.

La table suivante contient une partie de la hiérarchie de qualités, laquelle détermine quel point de début potentiel devient le point de début courant (la hiérarchie de qualités s’étend au-delà de ce qui figure dans la table). Les cases hachurées indiquent les qualités préférées par le recalcul :

Par exemple, un point peut avoir les qualités de composant suivantes :

  Qualité horizontale − contrôle

  Qualité de hauteur − inconnu

  Qualité d’élévation − inconnu

Il est alors utilisé comme point de début avant un point ayant les qualités de composant suivantes :

  Qualité horizontale − levé

  Qualité de hauteur − levé

  Qualité d’élévation − mappage

Vérifiez que le point que vous souhaitez utiliser comme point de début ou coordonnée de sélection du meilleur point de début comporte des composants de qualité de contrôle.

NOTE : S’il y a plus d’un point de début potentiel avec les mêmes qualités pour tous les composants, le recalcul utilise le premier point du projet.


Recalcul - Comment les observations sont appliquées à partir du point de début courant

 Vue d'ensemble et plus amples informations

Le recalcul applique une observation à partir du point de début de manière différente selon le type d’observation. Il applique :

  les observations GPS  à l’ellipsoïde WGS-84.

  les observations conventionnelles et par télémètre à laser à l’ellipsoïde local — sauf si l’option Réduire les observations terrestres à l’ellipsoïde (correction au niveau de la mer) n’est pas sélectionnée.

Cette option se trouve dans la tabulation Recalculer du dialogue Propriétés du projet. Dans ce cas, le recalcul applique des observations terrestres au sol. Pour de plus amples informations, référez-vous à la rubrique Correction du niveau d’ellipsoïde. Si un ajustement est effectué, toutes les distances sont réduites à l’ellipsoïde.

  Delta élévations − sont appliquées à des élévations existantes dans un recalcul initial.


Recalcul - Comment les observations GPS sont appliquées

 Vue d’ensemble et plus amples informations

Les observations GPS sont enregistrées dans le projet Trimble Geomatics Office comme vecteurs sol-sol. Cependant, les vecteurs GPS sont observés sur le terrain du centre de phase d’antenne au centre de phase d’antenne (APC-APC).

Durant l’importation, Trimble Geomatics Office convertit les vecteurs GPS en vecteurs sol-sol à l’aide des hauteurs d’antenne des antennes GPS à chaque extrémité de la ligne de base. Si les hauteurs d’antenne sont éditées, le vecteur sol-sol est automatiquement mis à jour.

NOTE : Le processeur de lignes de base génère les vecteurs sol-sol.

Le recalcul effectue les opérations suivantes pour déterminer la position pour les points observés à l’aide du système GPS :

1.	Convertit les coordonnées du point de début courant en latitude, longitude et hauteur WGS-84 (LLh).

2.	Convertit ces coordonnées en coordonnées cartésiennes WGS-84 (X, Y, Z).

3.	Applique le vecteur GPS sol-sol (delta X, delta Y et delta Z) aux coordonnées cartésiennes WGS-84 du point de début. Cela donne une position cartésienne WGS-84 pour le point observé.

NOTE : Le vecteur GPS appliqué est un vecteur ajusté si des paramètres de réseau ajustés sont utilisés dans le projet, même si la ligne de base n’a pas été utilisée dans l’ajustement de réseau.

4.	Convertit le point observé en une latitude, longitude et hauteur WGS-84.

5.	Utilise le système de coordonnées sélectionné pour le projet pour déduire les coordonnées de projection pour chaque point de sa position WGS-84. Il utilise également toute Calibration de site GPS appliquée au projet.

6.	La coordonnée qui en résulte devient la position courante du point.


Recalcul - Direction des lignes de base GPS

 Vue d’ensemble et plus amples informations

Par défaut, le recalcul applique toutes les observations GPS cinématiques en temps réel (RTK) dans la direction dans laquelle le vecteur a été observé (base à mobile).

La direction des lignes de base statiques et faststatic post-traitées est établie par la détermination du meilleur point de début durant le traitement des lignes de base. La direction est enregistrée dans le projet et indiquée par une flèche sur la ligne de base dans la fenêtre graphique. L’algorithme de détermination du meilleur point de début garantit que les vecteurs sont traités du point ayant la position de la plus haute qualité au point ayant la position de la plus basse qualité. Le recalcul applique le vecteur dans la même direction que celle enregistrée dans le projet.

Vous pouvez modifier la direction d’un vecteur GPS de manière à ce que le recalcul applique le vecteur dans la direction opposée.

Pour cela, sélectionnez l’observation puis Editer / Inverser flux d’observation. Dans ce cas, vérifiez que la position pour le point duquel vous souhaitez que le vecteur parte est connue, par exemple coordonnées de latitude, longitude et hauteur WGS-84 saisies au clavier. Si la position du point n’est pas connue, l’observation n’est pas appliquée.


Recalcul - Comment les observations conventionnelles et par télémètre à laser sont appliquées

 Vue d’ensemble et plus amples informations

Lorsque des observations conventionnelles sont importées :

  les angles verticaux sont corrigés pour éliminer la courbure et la réfraction

  les distances inclinées sont corrigées pour éliminer les erreurs atmosphériques (correction ppm)

  les observations horizontales sont converties en angles horizontaux à partir de la visée arrière.

Le recalcul effectue les opérations suivantes pour déterminer la position des points observés conventionnellement ou avec un télémètre à laser :

1.	Il effectue l'une des opérations suivantes :

  Si l’observation est une observation par télémètre à laser, le recalcul applique la déclinaison magnétique à l’azimut observé.

  Si l’observation est une observation conventionnelle, le recalcul applique la constante prisme à la distance inclinée.

2.	Il réduit les distances inclinées observées à des distances au sol horizontales.

3.	Si l’option Réduire les observations terrestres à l’ellipsoïde (correction au niveau de la mer) est sélectionnée dans la tabulation Recalculer du dialogue Propriétés du projet, il réduit les distances au sol à des distances d’ellipsoïde.

4.	Il ajoute la hauteur d’instrument à la hauteur locale.

5.	Il utilise le système de coordonnées défini pour le projet pour convertir le point de début courant à la latitude, longitude et hauteur locales (l’observation est appliquée le long de l’ellipsoïde).

6.	Il utilise les règles suivantes pour déterminer l’azimut de visée arrière (pour les observations conventionnelles uniquement) :

  Lorsque seules les positions du point d’instrument et des points de visée arrière sont connues (c.-à-d. qu’aucun azimut n’a été saisi au clavier), le recalcul utilise l’azimut de visée arrière déterminé à partir de l’inverse entre les deux points.

  Lorsque seul un azimut saisi au clavier est disponible et que le point « vers » n’a pas de coordonnée, le recalcul utilise celui-ci comme azimut de visée arrière.

  Si les positions du point d’instrument et du point de visée arrière sont connues, et qu’un azimut saisi au clavier est disponible, le recalcul détermine la qualité de chaque azimut et utilise la hiérarchie de la table ci-dessous pour déterminer quel azimut utiliser comme azimut de visée arrière.

Par exemple, si un instrument de qualité Levé et une paire de points de visée arrière existent, ainsi qu’un azimut saisi au clavier de qualité Contrôle, l’azimut saisi au clavier est utilisé pour l’azimut de visée arrière.

NOTE : Pour calculer la qualité de l’azimut de visée arrière, on examine la qualité horizontale du point d’instrument et du point de visée arrière et l’on choisit la qualité la plus basse. Par exemple, si la qualité du point d’instrument est Contrôle et que celle du point de visée arrière est Mappage, l’azimut de visée arrière aura la qualité Mappage.

Type	Qualité
Dérivé des coordonnées	Contrôle
Azimut saisi au clavier	Contrôle
Dérivé des coordonnées	Levé
Azimut saisi au clavier	Levé
Dérivé des coordonnées	Mappage
Azimut saisi au clavier	Mappage
NOTE : Si vous voulez qu’un recalcul utilise l’azimut saisi au clavier plutôt que l’azimut déterminé entre l’instrument et le point de visée arrière, augmentez la qualité de la position de l’azimut saisi au clavier de manière à ce qu’elle soit supérieure dans la hiérarchie à l’azimut déterminé entre l’instrument et le point de visée arrière.

ASTUCE : La qualité d’un azimut saisi au clavier peut être éditée :

7.	Il applique l’angle horizontal à l’azimut de visée arrière pour déterminer l’azimut de grille (pour les observations conventionnelles uniquement).

8.	Il applique la déclinaison magnétique à l’azimut magnétique pour déterminer l’azimut de grille (pour le laser uniquement).

9.	Il applique l’observation le long de l’ellipsoïde pour déterminer la latitude et la longitude locales du point observé.

10.	Pour déterminer l’élévation du point observé, il soustrait la hauteur de destination de la hauteur d’instrument pour calculer la delta hauteur. Il ajoute ensuite la valeur obtenue à l’élévation du point d’installation de l’instrument.

Il utilise le système de coordonnées sélectionné pour le projet afin de convertir les coordonnées en coordonnées de grille locales.


Recalcul - Comment les delta élévations sont appliquées

Des delta élévations sont appliquées lors du recalcul. Le recalcul effectue les opérations suivantes pour générer des delta élévations :

  L’élévation du point de début est déterminée.

  La delta élévation est appliquée au point de début pour déterminer l’élévation du nouveau point.

Dans un recalcul, le point de début doit comprendre une élévation saisie au clavier pour que des delta élévations puissent être générées à partir de celui-ci.

Vous devez entrer une élévation au début de la série de niveaux afin d’assurer que les delta élévations peuvent être générées à partir du début de la série de niveaux.

Par exemple, dans le diagramme suivant, une élévation doit être saisie au clavier pour le Point A afin que les points B, C et D puissent être déterminés à l’aide des observations de delta élévation. Si une observation vers A contenait des informations suffisantes pour calculer une élévation au point A, les delta élévations ne seraient toujours pas calculées (à moins qu’elle comporte une élévation saisie au clavier).


Recalcul - Exemple

 Vue d’ensemble et plus amples informations

La figure suivante correspond à un exemple de la méthode utilisée par un recalcul pour déterminer les coordonnées des points. Les points A et I sont tous les deux des points de qualité Contrôle, le reste des points étant de qualité Levé.

Le recalcul établit les points A et I comme points de début potentiels puisque ces deux points ont des coordonnées saisies au clavier et des observations partant d’eux. Les Points A et I ont la même qualité (contrôle L,L,h), le recalcul choisit donc le point A comme point de début courant parce que c’est le premier à avoir été entré dans le projet.

Le recalcul utilise toutes les observations du point de début pour déterminer les coordonnées des points observés jusqu’à ce qu’il ne puisse plus continuer. Par conséquent, les points suivants obtiennent leurs coordonnées du point A : B, C, D, E, F, G et H.

Il n’existe plus aucune observation à partir du point de début courant (point A), le recalcul choisit donc le point I comme nouveau point de début courant. Le recalcul détermine des coordonnées pour les points J, K et L à partir du point I.

Il existe des observations du point I aux points D et H. Le recalcul ayant déjà déterminé des coordonnées pour ces points, une fermeture est indiquée pour ces points. Si l’erreur de fermeture est supérieure aux paramètres de tolérance de la tabulation Recalculer du dialogue Propriétés du projet, un indicateur d’avertissement s’affiche sur le point observé dans la fenêtre graphique. La fermeture est rapportée dans le Rapport de recalcul.

AVERTISSEMENT : Le Point D peut être dérivé d’observations de B et I. L’observation de B est utilisée quelle que soit la qualité de ces observations. Cela est dû au fait que D a d’abord obtenu ses coordonnées de B. Si vous souhaitez vérifier que l’observation de I est utilisée, modifiez l’état de l’observation de B à D à « activée comme vérification ».


Recalcul - Observations et coordonnées multiples pour un point

 Vue d'ensemble et plus amples informations

Des observations multiples peuvent être effectuées vers un point et ce point peut s’accompagner de coordonnées saisies au clavier multiples. Si l’option Utiliser meilleure observation de la tabulation Recalculer du dialogue Propriétés du projet est sélectionnée, une seule observation ou coordonnée peut être utilisée pour déduire un composant de position (NE,é,h) pour le point. Par exemple, deux observations peuvent être effectuées vers un point et ce point peut s’accompagner d’une coordonnée saisie au clavier. La coordonnée saisie au clavier est utilisée, plutôt que l’observation, pour déterminer les coordonnées du point. Les autres observations ou coordonnées saisies au clavier qui ne sont pas utilisées par le recalcul sont rapportées dans le Rapport de recalcul.

NOTE : Si l’option Calculer la moyenne à partir du même type d’observation (dans la tabulation Recalculer du dialogue Propriétés du projet) est sélectionnée, toutes les observations peuvent contribuer au calcul des composants de coordonnées du point de prise de vue isolée. La moyenne est calculée uniquement pour les points de prise de vue isolée.

Observations multiples vers un point

Pour déterminer les coordonnées calculées pour un point comportant de multiples observations (lorsque l’option Utiliser meilleure observation est sélectionnée), le recalcul utilise la hiérarchie d’observations suivante :

1.	Etat d’observation (activée, activée comme vérification, désactivée)

2.	Qualité d’observation

3.	Type d’observation (GPS, conventionnelle, laser, réduite)

4.	Précision

5.	Première observation du projet


Recalcul - Observations multiples vers les points de prise de vue isolée

 Vue d'ensemble et plus amples informations

Lorsque des observations (GPS ou conventionnelles) sont transférées à Trimble Geomatics Office, le logiciel détermine quelle observation est une prise de vue isolée. Une prise de vue isolée est une observation vers un point :

  à partir duquel aucune observation n’a été effectuée

  vers lequel aucune observation statique ou FastStatic n’a été effectuée

  qui n’est pas un point de contrôle

La configuration du groupe Observations de prise de vue isolée multiples (dans la tabulation Recalculer du dialogue Propriétés du projet) détermine quelles observations sont utilisées pour calculer la position du point de prise de vue isolée.

NOTE : Le recalcul ne calcule pas de moyenne pour les élévations de prises de vue isolées multiples.


Recalcul - Observations multiples et calcul de la moyenne de coordonnées

 Vue d’ensemble et plus amples informations

Par défaut, l’option Calculer la moyenne du même type d’observations de la tabulation Recalculer du dialogue Propriétés du projet n’est pas sélectionnée. Par conséquent, la moyenne des observations multiples n’est pas calculée.

Si vous sélectionnez cette option, Trimble Geomatics Office calcule la moyenne des coordonnées calculées à partir de chaque observation vers un point de prise de vue isolée.

NOTE : Si le point est un point de cheminement, le recalcul n’utilise qu’une seule observation ou coordonnée pour déterminer la position du point.

Les règles du calcul de la moyenne de la position pour les points de prise de vue isolée sont les suivantes :

  Le recalcul fait la moyenne du même type d’observation uniquement. Par exemple, la moyenne d’une observation GPS et d’une observation conventionnelle ne peut pas être calculée. Dans ce cas, le recalcul utilise l’observation GPS plutôt que l’observation conventionnelle.

  Le recalcul fait la moyenne des coordonnées dérivées d’observations uniquement. Par exemple, si un point comporte deux observations et des coordonnées saisies au clavier, le recalcul fait la moyenne des observations et détermine les coordonnées ainsi que les qualités. Il compare ensuite les coordonnées saisies au clavier à celles dont la moyenne a été calculée, et utilise la coordonnée de la plus haute qualité comme position pour le point.

NOTE : Si une ou plusieurs observations sont activées, le recalcul n’utilise pas les observations activées comme vérifications pour déterminer une position moyenne.

Pour déterminer la position calculée d’un point, le recalcul calcule les positions de chaque observation effectuée vers le point. Chaque observation est pondérée en fonction de sa valeur de précision. Plus la précision est élevée, plus l’observation a de l’influence sur les valeurs de coordonnées moyennes.

Pour les observations GPS, le recalcul utilise le champ Précision. Pour afficher ce champ :

1.	Sélectionnez une observation GPS.

2.	Ouvrez la fenêtre Propriétés.

3.	Cliquez sur le bouton du menu Statistiques d’observation

Pour les observations conventionnelles une pondération égale est utilisée.

Le recalcul utilise la position moyenne comme centre d’un cercle dont le rayon est défini par la distance de tolérance. Cette distance est spécifiée dans la tabulation Recalculer du dialogue Propriétés du projet. Si le composant de coordonnées est horizontal (Nord, Est), le recalcul utilise la Tolérance horizontale. Si le composant de coordonnées est vertical (Elévation ou Hauteur), il utilise la Tolérance verticale.

Si une quelconque position se trouve en dehors du cercle défini par la tolérance, la position la plus éloignée de la position moyenne est rejetée et le processus est répété. Dans la figure suivante, trois observations conventionnelles ont été effectuées vers un point de prise de vue isolée.

L’observation 3 est en dehors du cercle de tolérance et donc omise pour l’itération suivante.

Le processus est répété avec les observations restantes jusqu’à ce que toutes les positions calculées se trouvent dans le cercle de tolérance. La figure suivante montre comment les coordonnées de grille moyennes calculées deviennent les coordonnées calculées finales pour le point.

Toutes les observations utilisées pour déterminer les coordonnées moyennes sont maintenant dans la tolérance spécifiée, le recalcul utilise donc ces coordonnées dont la moyenne a été calculée afin de calculer la position du point.

NOTE : Si vous désélectionnez la case à cocher Utiliser tolérance sous la tabulation Recalculer du dialogue Propriétés du projet, la moyenne de toutes les observations du même type est calculée sans vérification. Les valeurs aberrantes ne sont donc pas détectées.


Recalcul - Observations multiples vers des points de cheminement

 Vue d’ensemble et plus amples informations

Dans un cheminement de levé, il est possible que plus d’une observation ait été effectuée vers un point de cheminement. Il s’agit d’une observation multiple.

La moyenne des observations multiples vers les points de cheminement n’est pas calculée. Cela est illustré dans la figure suivante. Deux observations sont réalisées vers le Point 1 et une observation est réalisée à partir du Point 1. L’observation utilisée pour déduire la position du Point 1 est la première à positionner le point.


Recalcul - Points avec des coordonnées saisies au clavier et des observations

 Vue d'ensemble et plus amples informations

La position affichée pour un point peut être dérivée de :

  observation(s)

  coordonnée(s) saisie(s) au clavier, par exemple coordonnées de grille importées

Si des observations ont été effectuées vers un point et qu’il s’accompagne de coordonnées saisies au clavier, le recalcul utilise la qualité de(s) coordonnée(s) saisie(s) au clavier et d’observation(s) pour déterminer la position du point. Si les qualités sont les mêmes, il utilise les coordonnées saisies au clavier plutôt que les coordonnées d’observation(s). Par conséquent, si vous souhaitez utiliser des observations pour la position d’un point, vérifiez que vous sélectionnez pour l’observation des qualités plus élevées que celles des coordonnées saisies au clavier pour le point.

Le recalcul considère séparément la qualité du composant horizontal et celle du composant vertical. Par exemple, il est possible que les coordonnées de grille importées pour un point comportent un composant horizontal de qualité Levé (Nord, Est) et un composant d’élévation de qualité Mappage. Une observation vers le point a également été effectuée. Les coordonnées horizontales et verticales dérivées de l’observation sont de qualité Levé.

Cette situation est résumée dans la table suivante.

Les cases hachurées indiquent comment la position du point est déterminée. Les qualités des composants horizontaux sont les mêmes, le recalcul utilise donc les coordonnées plutôt que l’observation. La qualité du composant vertical de la coordonnée importée est inférieure à celle de l’observation, le recalcul déduit donc l’élévation de l’observation.

La meilleure élévation ou hauteur pour les points est déduite à l’aide de toute coordonnée ou observation vers le point, celle-ci étant à la fois disponible et saisie au clavier. Si le recalcul utilise une observation pour calculer une élévation ou hauteur pour le point, il calcule alors les composants verticaux restants à partir de l’un des éléments suivants :

  Le modèle de géoïde et/ou l’ajustement vertical

  Coordonnées saisies du clavier

Si un modèle de géoïde est utilisé pour le projet, la qualité du modèle de géoïde figurant sous la tabulation Recalculer du dialogue Propriétés du projet est utilisée pour déterminer la qualité de l’élévation ou de la hauteur dérivée du modèle de géoïde. Cette qualité est comparée à celle de toutes coordonnées saisies au clavier et le composant d’élévation ou de hauteur de la meilleure qualité est utilisé.

NOTE : S’il existe un ajustement vertical d’une calibration dans la définition du système de coordonnées du projet, alors la qualité de l’élévation ou hauteur dérivée est Levé.


Recalcul - Comment la qualité des points observés est déterminée

 Vue d’ensemble et plus amples informations

Le recalcul détermine la qualité d’un point observé sur la base de la qualité du point à partir duquel il a été observé et de la qualité de l’observation utilisée pour déduire la position du point. Il prend séparément en compte les composants horizontaux et verticaux.

La hiérarchie de qualités est la suivante :

1.	Contrôle

2.	Levé

3.	Mappage

4.	Inconnu

La table suivante indique comment les qualités d’observation sont déterminées à partir du type de solution ou du type de mesure.

Type d'observation	Qualité assignée à l’observation
GPS fixe	Levé
GPS flottante	Mappage
GPS autonome	Inconnu
Conventionnelle	Levé
Télémètre à laser	Mappage
Delta élévation	Levé

La qualité assignée à un point pour chaque composant (horizontal, d’élévation et de hauteur) est la plus basse parmi les qualités suivantes :

  la qualité du point à partir duquel l’observation a été effectuée

  la qualité de l’observation


Position de base autonome pour les levés cinématiques en temps réel

 Vue d’ensemble et plus amples informations

Si vous utilisez la touche [image] du Trimble Survey Controller™ pour démarrer une base cinématique en temps réel (RTK) et transférer le fichier Survey Controller (.dc) à Trimble Geomatics Office, la position de base (et par conséquent tous les points mobiles de la base) sont d’une qualité inconnue (pour tous les composants : horizontaux, de hauteur et d’élévation).

Si vous utilisez Trimble Geomatics Office pour effectuer une calibration de site GPS, les qualités des composants horizontaux et de hauteur ne changent pas.

Si vous êtes sûr que la calibration du site GPS a amélioré la qualité des points, utilisez la fenêtre Propriétés pour modifier la qualité du point de base. L’augmentation de la qualité de la base entraîne l’amélioration de la qualité des points mobiles.

NOTE : La qualité du composant d’élévation est Levé s’il existe un ajustement vertical dans la définition du système de coordonnées du projet.


Recalcul - Points ayant été déplacés ou ajustés à l'aide de Trimble Geomatics Office

 Vue d’ensemble et plus amples informations

Le recalcul ne déplace pas les points qui ont été décalés par :

  le déplacement d’une coordonnée.

  une transformation de coordonnées.

  un ajustement d’élévation.

  un ajustement de réseau.

Lorsque l’une de ces opérations est appliquée à un point, les positions dérivées des observations vers le point ne sont pas utilisées.


Recalcul - Hauteur du projet

 Vue d’ensemble et plus amples informations

La hauteur du projet est un paramètre pour tout le projet utilisé comme hauteur approximative pour tout calcul relatif à des points sans hauteur. Tous les points sans hauteur ne sont pas mis à jour à l’aide de la valeur de hauteur du projet. Cette valeur est utilisée uniquement pour convertir des coordonnées lorsque :

  le système de coordonnées du projet est modifié et que des hauteurs sont nulles

  les corrections au niveau de la mer sont appliquées parce que les points n’ont pas de hauteurs

Le champ Hauteur du projet se trouve dans la tabulation Système de coordonnées du dialogue Propriétés du projet.


Recalcul - Observations de vérification

 Vue d’ensemble et plus amples informations

Trimble Survey Controller™ vous permet d’enregistrer des observations de vérification vers des points. Cette section décrit comment ces points sont traités lorsqu’un fichier .dc est importé dans Trimble Geomatics Office.

Vérifier les points vers des points autres que les points de contrôle saisis au clavier

Lorsqu’un fichier .dc est importé, la position calculée pour le point ajouté est déduite à l’aide de l’observation/des observations vers le point activée(s). L’observation de vérification est importée mais n’est pas utilisée pour déterminer la position calculée du point à moins qu’il n’y ait aucune observation d’état Activé vers le point (le recalcul utilise toujours les observations d’état Activé avant d’utiliser les Observations activées comme vérifications). L’observation se voit octroyer l’état Activée comme vérification et s’affiche sous forme d’une ligne pointillée dans la fenêtre graphique. Pour afficher et éditer l’état d’une observation, utilisez la page Résumé de la fenêtre Propriétés. Une erreur de fermeture est calculée entre la position calculée pour le point et les coordonnées déterminées par l’observation de vérification. Cela est rapporté dans le Rapport de recalcul généré pendant l’importation et enregistré dans le dossier Recalcul du dossier Rapports du projet.

NOTE :

Si vous enregistrez une observation de vérification vers un point déjà enregistré et supprimez le point collecté précédemment, l’observation de vérification est importée et un nouveau point est ajouté.

L’Etat de l’observation demeure Activée comme vérification, et la position calculée pour le point est dérivée de l’observation.


Trimble 3600/5600 GDM - Vue d'ensemble

Vous pouvez transférer les données entre Trimble Geomatics Office et une unité de contrôle Trimble 3600/5600 GDM sur un instrument Trimble 3600/5600. Trimble recommande que l’instrument ait le firmware 632.11.00 ou ultérieur.

Vous pouvez importer les données des fichier 3600/5600 GDM Job (*.job) et Area (*.are), et exporter les données de votre projet Trimble Geomatics Office courant dans un fichier Area (*.are). Sinon, importez ou exporter les données directement utilisant l’option d’importation Périphériques de levé.

La table suivante affiche les types de fichier pouvant être transférés entre Trimble Geomatics Office et une unité de contrôle Trimble 3600/5600 GDM:

Importation	Exportation
.job − données collectées d’une unité de contrôle Trimble 3600/5600 GDM	Aucune
.are − fichier de coordonnées	.are
Aucune	.pco − dans l’unité de contrôle Trimble 3600/5600 GDM
Aucune	.uds − une séquence définie par l'utilisateur
Afin d’assurer que Trimble Geomatics Office puisse interpréter correctement des fichiers 3600/5600 GDM importés, effectuez les choses suivantes:

  	Vérifiez que l’unité de contrôle Trimble 3600/5600 GDM a le numéro de programme activé.

  	Configurez le Program 32 afin de stocker la distance inclinée plutôt qu’une distance horizontale et verticale. Pour plus amples informations, référez-vous à la documentation de l’unité de contrôle Trimble 3600/5600 GDM.

  	Pour afficher les observations de Program 23 et 24 dans Trimble Geomatics Office, les tables de l’utilisateur de programme Trimble 3600/5600 GDM doivent comprendre des enregistrements de l’angle horizontal, l’angle vertical et la distance inclinée après les enregistrements standards, Pour plus amples informations concernant comment modifier les tables de l’utilisateur, référez-vous à la documentation de l’unité de contrôle Trimble 3600/5600 GDM.


Séquences définies par l'utilisateur (UDS) - Vue d'ensemble

Une séquence définie par l’utilisateur détermine la façon dont les données de levé sont collectées et stockées sur le terrain utilisant une unité de contrôle Trimble 3600/5600 GDM sur un instrument Trimble 3600/5600. Chaque séquence est stockée dans un fichier Séquence définie par l’utilisateur (*.uds).

Les fichiers UDS devraient comprendre les enregistrements suivants:

2	Station d’instrument
3	Hauteur d’instrument	(Facultatif)
62	Point de visée arrière
21	Azimut de visée arrière
5	Nom du point
6	Hauteur cible	(Facultatif)
4	Code	(Facultatif)
7	Angle horizontal
8	Angle vertical
9	Distance inclinée

Pour utiliser une séquence définie par l’utilisateur sur le terrain, transférez un fichier UDS à l’unité de contrôle Trimble 3600/5600 GDM utilisant l’utilitaire Data Transfer autonome. (Vous ne pouvez pas utiliser le dialogue Exporter de Trimble Geomatics Office.) Pour plus amples informations, référez-vous à l’aide de Transfert de données.

Dès que vous avez collecté et stocké vos données de levé, transférez-les dans Trimble Geomatics Office.

NOTE: Les fichiers UDS prédéfinis sont disponibles sur le CD Trimble Geomatics Office.

Voir aussi


Fichiers Pcode (*.pco)

Les fichiers Pcode comprennent des codes numériques et des descriptions alphanumériques qui définissent les caractéristiques et les attributs. Ils sont utilisés dans une unité de contrôle Trimble 3600/5600 GDM sur un instrument Trimble 3600/5600. Vous pouvez entrer les codes pendant la collecte de données sur le terrain afin de définir les points observés rapidement. Par exemple, si un fichier Pcode comprend un code de “201” et une description de “Arbre”, sur le terrain vous pouvez entrer “201” et “Arbre” s’affichera sur l’unité de contrôle.

Un fichier Pcode exporté dans une unité de contrôle Trimble 3600/5600 GDM doit avoir les choses suivantes:

  Des codes numériques avec un numéro inférieur à 256

  Des descriptions lesquelles ensemble sont composées de moins de 800 caractères

Pour utiliser un fichier Pcode, créez un fichier d’une bibliothèque de caractéristiques et attributs dans l’utilitaire Feature and Attribute Editor de Trimble et puis transférez-le dans l’unité de contrôle.

Voir aussi


Fichiers Area (*.are)

Trimble Geomatics Office prend en charge les données Area d’une unité de contrôle Trimble 3600/5600 GDM (fichiers *.are).

Pour plus amples informations, référez-vous à:

  Importation des fichiers Trimble 3600/5600 GDM CU, comment

  Exportation vers un format Trimble 3600/5600 GDM CU, comment


Fichiers Job (*.job)

Trimble Geomatics Office prend en charge les données Job d’une unité de contrôle Trimble 3600/5600 GDM (fichiers *.job).

Pour plus amples informations, référez-vous à Importation des fichiers Trimble 3600/5600 GDM CU, comment